(369,'2024-10-10','07:52:10','90.00000','105.00000','pH Buffer','NA ','NA','E4INS24005316','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,1,101,'pH','12.46 pH',1,'84378','Shivanand. W','2024-10-10 08:06:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.49 pH\r\nmV                   -321.5 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(370,'2024-10-10','08:50:13','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005318','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-10 09:14:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   310.9 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\n\r\n\r\n'),
(371,'2024-10-10','08:52:22','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005318','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-10 09:14:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   172.4 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\n\r\n\r\n'),
(372,'2024-10-10','08:54:23','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005318','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-10 09:15:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -2.9 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\n\r\n\r\n'),
(373,'2024-10-10','08:56:20','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005318','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-10 09:15:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.00 pH\r\nmV                   -177.5 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\n\r\n\r\n'),
(374,'2024-10-10','08:58:12','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005318','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-10 09:16:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.45 pH\r\nmV                   -321.5 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\n\r\n\r\n'),
(375,'2024-10-10','09:44:04','90.00000','105.00000','Penicillin G Benzathine','dissolution','PG0124001A','E4STFP24000554','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Penicillin G',1,'90018','Ch. Satyam','2024-10-10 16:54:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.50 pH\r\nmV                   -31.0 mV\r\nTemperature          32.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(376,'2024-10-10','10:06:45','90.00000','105.00000','Penicillin G','NA','PG0124003A','E4STFP24000566','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-15 13:04:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.47 pH\r\nmV                   29.0 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\n\r\n\r\n'),
(377,'2024-10-10','10:08:54','90.00000','105.00000','Penicillin G','NA','PG0124002A','E4STFP24000558','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-15 13:09:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.50 pH\r\nmV                   27.9 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\nLog #                132\r\n\r\n\r\n'),
(378,'2024-10-10','10:12:29','90.00000','105.00000','Penicillin G','NA','PG0124001A','E4STFP24000551','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-15 13:09:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.50 pH\r\nmV                   27.5 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\n\r\n\r\n'),
(379,'2024-10-10','10:22:21','90.00000','105.00000','Piperacillin and Tazobactam for Injection','Assay','PN0324059A','E4STFP24000628','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-10 10:33:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   85.9 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\nLog #                137\r\n\r\n\r\n'),
(380,'2024-10-10','10:30:18','90.00000','105.00000','AMPICILLIN FOR INJECTION','RS','AP0224021','E4FP24000679','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205043','T.V. Subbaiah',2,1,101,'pH','Ampicillin pH',1,'84378','Shivanand. W','2024-10-10 10:39:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.04 pH\r\nmV                   115.1 mV\r\nTemperature          29.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\nLog #                138\r\n\r\n\r\n'),
(381,'2024-10-10','10:43:46','90.00000','105.00000','PIPERACILLIN','NA','TZ0024094','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-10 13:26:04','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.86 pH\r\nmV                   50.2 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             10-10-24,05:17\r\n\r\n\r\n'),
(382,'2024-10-10','11:18:08','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC','TR0424002','E4FP24000680','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205058','C. Srinivasulu',2,1,101,'pH','Pipertazoinj',1,'84378','Shivanand. W','2024-10-10 11:28:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.03 pH\r\nmV                   56.2 mV\r\nTemperature          29.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\nLog #                140\r\n\r\n\r\n'),
(383,'2024-10-10','11:26:07','90.00000','105.00000','PIPERACILLIN','NA','TZ0024094','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-10 13:29:54','ph value not considered because temperature high','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.88 pH\r\nmV                   49.8 mV\r\nTemperature          28.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             10-10-24,05:17\r\nLog #                114\r\n\r\n\r\n'),
(384,'2024-10-10','11:34:58','90.00000','105.00000','PIPERACILLIN','NA','TZ0024094','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-10 13:28:17','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.97 pH\r\nmV                   43.5 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             10-10-24,05:17\r\nLog #                115\r\n\r\n\r\n'),
(385,'2024-10-10','11:40:04','90.00000','105.00000','Soyabean casein digest agar','NA','RS022/24006','E4SO24000652','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SCDAIH',1,'205143','Ch. Prasad','2024-10-15 13:09:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.43 pH\r\nmV                   -27.2 mV\r\nTemperature          26.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\nLog #                134\r\n\r\n\r\n'),
(386,'2024-10-10','11:50:32','90.00000','105.00000','Sabourarud dextrose agar','NA','RS021/24006','E4SO24000654','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SDAIH',1,'205143','Ch. Prasad','2024-10-15 13:09:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   74.6 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\n\r\n\r\n'),
(387,'2024-10-10','12:20:36','90.00000','105.00000','PIPERACILLIN','NA','TZ0024094','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-10 13:28:29','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.26 pH\r\nmV                   27.0 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             10-10-24,05:17\r\n\r\n\r\n'),
(388,'2024-10-10','13:03:06','90.00000','105.00000','PIPERACILLIN','NA','TZ0024094','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-10 13:28:43','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.42 pH\r\nmV                   18.0 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             10-10-24,05:17\r\n\r\n\r\n'),
(389,'2024-10-10','13:46:31','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005441','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-10 16:52:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.79 pH\r\nmV                   69.3 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(390,'2024-10-10','13:48:03','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005442','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-10 16:52:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.65 pH\r\nmV                   77.5 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(391,'2024-10-10','13:49:42','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005443','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90018','Ch. Satyam','2024-10-10 16:53:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.87 pH\r\nmV                   64.6 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(392,'2024-10-10','13:52:08','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005444','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90018','Ch. Satyam','2024-10-10 16:53:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   74.0 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(393,'2024-10-10','13:54:26','90.00000','105.00000','Water For Injection','pH','WFI/SP33','E4RM24005445','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90018','Ch. Satyam','2024-10-10 16:54:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.73 pH\r\nmV                   72.5 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(394,'2024-10-10','13:56:34','90.00000','105.00000','Pure steam','pH','PSC/SP21','E4RM24005446','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'90018','Ch. Satyam','2024-10-10 16:54:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   75.2 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(395,'2024-10-10','13:59:01','90.00000','105.00000','Purified water','pH','PW/SP24','E4RM24005447','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-10 16:54:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.72 pH\r\nmV                   73.3 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(396,'2024-10-10','14:01:12','90.00000','105.00000','Water For Injection','pH','WFI/SP25','E4RM24005448','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90018','Ch. Satyam','2024-10-10 16:53:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   71.5 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(397,'2024-10-10','14:33:43','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024094','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','3',1,'84472','G. Ashok','2024-10-10 18:32:08','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   25.9 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             10-10-24,05:17\r\nLog #                118\r\n\r\n\r\n'),
(398,'2024-10-10','14:56:54','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024094','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','3',1,'84472','G. Ashok','2024-10-10 18:32:39','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.30 pH\r\nmV                   25.1 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             10-10-24,05:17\r\n\r\n\r\n'),
(399,'2024-10-10','14:56:51','90.00000','105.00000','FLUID-A','NA','RS012/24007','E4SO24000658','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Fluid-A',1,'205143','Ch. Prasad','2024-10-15 13:10:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.25 pH\r\nmV                   42.6 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\nLog #                136\r\n\r\n\r\n'),
(400,'2024-10-10','15:01:40','90.00000','105.00000','Fluid-D','NA','RS010/24008','E4SO24000659','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Fluid-D',1,'205143','Ch. Prasad','2024-10-15 13:10:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.43 pH\r\nmV                   31.9 mV\r\nTemperature          26.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\n\r\n\r\n'),
(401,'2024-10-10','15:39:39','90.00000','105.00000','Soyabean casein digest agar','NA','RS022/24006','E4SO24000652','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDAIH',1,'205143','Ch. Prasad','2024-10-15 13:11:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.42 pH\r\nmV                   -27.1 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\n\r\n\r\n'),
(402,'2024-10-10','15:43:56','90.00000','105.00000','Sabourarud dextrose agar','NA','RS021/24006','E4SO24000654','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SDAIH',1,'205143','Ch. Prasad','2024-10-15 13:11:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.58 pH\r\nmV                   81.8 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\n\r\n\r\n'),
(403,'2024-10-10','16:48:27','90.00000','105.00000','Glass Vials for pH','pH','NA','E4EQ24000360','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'90018','Ch. Satyam','2024-10-10 16:53:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.73 pH\r\nmV                   248.3 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(404,'2024-10-10','16:55:41','90.00000','105.00000','Glass Vials for pH','pH (vial-2)','NA','E4EQ24000360','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'90018','Ch. Satyam','2024-10-10 17:04:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.26 pH\r\nmV                   275.4 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(405,'2024-10-10','17:59:34','90.00000','105.00000','Piperacillin and Tazobactam for Injection','ph','TZ0024094','E4IP24001240','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','pH (Bulk)',1,'90018','Ch. Satyam','2024-10-10 18:05:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.23 pH\r\nmV                   43.2 mV\r\nTemperature          22.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(406,'2024-10-10','20:59:46','90.00000','105.00000','FLUID-A','NA','RS012/24007','E4SO24000658','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','Fluid-A',1,'205143','Ch. Prasad','2024-10-15 13:12:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.67 pH\r\nmV                   18.0 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\n\r\n\r\n'),
(407,'2024-10-10','21:01:34','90.00000','105.00000','Fluid-D','NA','RS010/24008','E4SO24000659','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','Fluid-D',1,'205143','Ch. Prasad','2024-10-15 13:13:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.80 pH\r\nmV                   9.9 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             10-10-24,08:46\r\n\r\n\r\n'),
(408,'2024-10-11','00:37:35','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005327','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','1.68 pH',1,'90018','Ch. Satyam','2024-10-11 01:27:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.64 pH\r\nmV                   313.0 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(409,'2024-10-11','00:40:18','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005327','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','4.01pH',1,'90018','Ch. Satyam','2024-10-11 01:28:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.04 pH\r\nmV                   173.0 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(410,'2024-10-11','00:42:58','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005327','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','7.00pH',1,'90018','Ch. Satyam','2024-10-11 01:28:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -1.9 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(411,'2024-10-11','00:52:09','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005327','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','10.01 pH',1,'90018','Ch. Satyam','2024-10-11 01:28:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.02 pH\r\nmV                   -176.6 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(412,'2024-10-11','00:54:26','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005327','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','12.46 pH',1,'90018','Ch. Satyam','2024-10-11 01:28:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.46 pH\r\nmV                   -318.4 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(413,'2024-10-11','07:16:31','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005337','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-11 09:51:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   312.4 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-10-24,07:12\r\n\r\n\r\n'),
(414,'2024-10-11','07:18:40','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005337','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-11 09:51:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   173.1 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-10-24,07:12\r\n\r\n\r\n'),
(415,'2024-10-11','07:20:36','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005337','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-11 09:52:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -2.9 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-10-24,07:12\r\n\r\n\r\n'),
(416,'2024-10-11','07:23:06','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005337','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-11 09:52:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.00 pH\r\nmV                   -178.4 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-10-24,07:12\r\n\r\n\r\n'),
(417,'2024-10-11','07:52:37','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005337','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-11 09:52:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.43 pH\r\nmV                   -318.6 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-10-24,07:12\r\nLog #                146\r\n\r\n\r\n'),
(418,'2024-10-11','07:54:19','90.00000','105.00000','Penicillin G Benzathine','dissolution','PG0124002A','E4STFP24000559','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Penicillin G',1,'90018','Ch. Satyam','2024-10-11 11:15:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.31 pH\r\nmV                   219.4 mV\r\nTemperature          30.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\nLog #                157\r\n\r\n\r\n'),
(419,'2024-10-11','08:02:00','90.00000','105.00000','Penicillin G Benzathine','dissolution','PG0124002A','E4STFP24000559','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Penicillin G',1,'90018','Ch. Satyam','2024-10-11 11:16:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.50 pH\r\nmV                   -29.6 mV\r\nTemperature          31.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(420,'2024-10-11','09:10:20','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH Test','TR0424002','E4FP24000680','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-11 12:01:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.81 pH\r\nmV                   11.3 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(421,'2024-10-11','09:13:01','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH Test','TR0424003','E4FP24000684','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-11 12:01:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.50 pH\r\nmV                   29.4 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(422,'2024-10-11','09:52:19','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH Test','TZ0424091','E4FP24000685','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-11 12:02:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.08 pH\r\nmV                   54.0 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(423,'2024-10-11','09:55:56','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH Test','TZ0424092','E4FP24000689','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-11 12:03:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.06 pH\r\nmV                   55.1 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(424,'2024-10-11','10:18:42','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH Test','PT0424109','E4FP24000687','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-11 12:03:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.90 pH\r\nmV                   64.4 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(425,'2024-10-11','11:31:03','90.00000','105.00000','Peptone','NA','RS007/24035','E4SO24000665','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-15 13:14:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.42 pH\r\nmV                   32.2 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-10-24,07:12\r\n\r\n\r\n'),
(426,'2024-10-11','11:33:35','90.00000','105.00000','Soyabean casein digest broth for sterilty','NA','RS033/24008','E4SO24000663','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SCDBSIH',1,'205143','Ch. Prasad','2024-10-15 13:14:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.33 pH\r\nmV                   -21.6 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-10-24,07:12\r\n\r\n\r\n'),
(427,'2024-10-11','11:36:22','90.00000','105.00000','Fluid thioglycollate meddiuam','NA','RS026/24009','E4SO2400664','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','FTGMIH',1,'205143','Ch. Prasad','2024-10-15 13:14:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.23 pH\r\nmV                   -15.4 mV\r\nTemperature          26.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-10-24,07:12\r\nLog #                149\r\n\r\n\r\n'),
(428,'2024-10-11','11:36:48','90.00000','105.00000','AMPICILLIN FOR INJECTION','pH','AP0224023','E4FP24000686','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Ampicillin pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-11 12:04:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.24 pH\r\nmV                   -131.1 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\nLog #                165\r\n\r\n\r\n'),
(429,'2024-10-11','11:43:24','90.00000','105.00000','AMPICILLIN FOR INJECTION','pH','AP0224024','E4FP24000688','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Ampicillin pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-11 12:04:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.14 pH\r\nmV                   -125.8 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(430,'2024-10-11','11:50:54','90.00000','105.00000','Penicillin G','NA','PG0124001A','E4STFP24000551','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-15 13:15:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.66 pH\r\nmV                   17.7 mV\r\nTemperature          26.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-10-24,07:12\r\nLog #                150\r\n\r\n\r\n'),
(431,'2024-10-11','11:58:58','90.00000','105.00000','Penicillin G','NA','PG0124001A','E4STFP24000551','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-15 13:15:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -4.0 mV\r\nTemperature          26.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-10-24,07:12\r\n\r\n\r\n'),
(432,'2024-10-11','12:01:00','90.00000','105.00000','Penicillin G','NA','PG0124002A','E4STFP24000558','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-15 13:15:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.87 pH\r\nmV                   5.2 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-10-24,07:12\r\n\r\n\r\n'),
(433,'2024-10-11','12:07:57','90.00000','105.00000','Penicillin G','NA','PG0124003A','E4STFP24000566','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-15 13:16:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.81 pH\r\nmV                   9.1 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-10-24,07:12\r\n\r\n\r\n'),
(434,'2024-10-11','13:31:58','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005455','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-10-11 17:29:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.89 pH\r\nmV                   64.7 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\nLog #                167\r\n\r\n\r\n'),
(435,'2024-10-11','13:33:53','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005456','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-10-11 17:30:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.85 pH\r\nmV                   67.0 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(436,'2024-10-11','13:35:41','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005457','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-10-11 17:30:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   68.3 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(437,'2024-10-11','13:37:58','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005458','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-10-11 17:31:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   68.2 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(438,'2024-10-11','13:40:12','90.00000','105.00000','Purified water','pH','PW/SP34','E4RM24005459','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-10-11 17:31:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.73 pH\r\nmV                   74.2 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(439,'2024-10-11','13:42:03','90.00000','105.00000','Purified water','pH','PW/SP36','E4RM24005460','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-10-11 17:32:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.85 pH\r\nmV                   67.4 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(440,'2024-10-11','13:43:55','90.00000','105.00000','Purified water','pH','PW/SP25','E4RM24005461','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-10-11 17:33:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.86 pH\r\nmV                   66.6 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(441,'2024-10-11','13:45:53','90.00000','105.00000','Water For Injection','pH','WFI/SP26','E4RM24005462','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-10-11 17:33:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.80 pH\r\nmV                   69.9 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(442,'2024-10-11','13:47:56','90.00000','105.00000','Water For Injection','pH','WFI/SP34','E4RM24005463','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-10-11 17:34:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.42 pH\r\nmV                   33.8 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(443,'2024-10-11','13:49:54','90.00000','105.00000','Purified water','pH','PW/SP26','E4RM24005464','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-10-11 17:34:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   73.0 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(444,'2024-10-11','13:52:02','90.00000','105.00000','Water For Injection','pH','WFI/SP27','E4RM24005465','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-10-11 17:35:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.81 pH\r\nmV                   69.3 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-10-24,00:27\r\n\r\n\r\n'),
(445,'2024-10-11','14:32:33','90.00000','105.00000','Peptone','NA','RS007/24035','E4SO24000665','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-15 13:16:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.94 pH\r\nmV                   1.2 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-10-24,07:12\r\n\r\n\r\n'),
(446,'2024-10-11','14:37:19','90.00000','105.00000','Soyabean casein digest broth for sterilty','NA','RS033/24008','E4SO24000665','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SCDBSIH',1,'205143','Ch. Prasad','2024-10-15 13:16:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.32 pH\r\nmV                   -21.0 mV\r\nTemperature          26.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-10-24,07:12\r\nLog #                155\r\n\r\n\r\n'),
(447,'2024-10-11','14:41:23','90.00000','105.00000','Fluid thioglycollate meddiuam','NA','RS026/24009','E4SO24000664','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','FTGMIH',1,'205143','Ch. Prasad','2024-10-15 13:16:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.98 pH\r\nmV                   -0.9 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-10-24,07:12\r\n\r\n\r\n'),
(448,'2024-10-11','15:29:38','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005345','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','87624','N. Nani babu',2,1,102,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-10-11 16:14:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   1.71 pH\r\nmV                   312.1 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             11-10-24,15:23\r\nLog #                126\r\n\r\n\r\n'),
(449,'2024-10-11','15:35:25','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005345','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','87624','N. Nani babu',2,1,102,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-10-11 16:15:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   4.00 pH\r\nmV                   174.4 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             11-10-24,15:23\r\n\r\n\r\n'),
(450,'2024-10-11','15:38:41','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005345','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','87624','N. Nani babu',2,1,102,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-10-11 16:15:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.99 pH\r\nmV                   -1.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             11-10-24,15:23\r\n\r\n\r\n'),
(451,'2024-10-11','15:40:55','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005345','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','87624','N. Nani babu',2,1,102,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-10-11 16:15:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -171.1 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             11-10-24,15:23\r\n\r\n\r\n'),
(452,'2024-10-11','15:43:42','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005345','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','87624','N. Nani babu',2,1,102,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-10-11 16:15:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -318.4 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             11-10-24,15:23\r\n\r\n\r\n'),
(453,'2024-10-11','15:54:57','90.00000','105.00000','TAZOBACTAM','NA','TZ0024095','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','1',1,'95681','T. Sivanarayana','2024-10-11 20:57:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.54 pH\r\nmV                   127.0 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             11-10-24,14:18\r\n\r\n\r\n'),
(454,'2024-10-11','18:40:34','90.00000','105.00000','PIPERACILLIN','NA','TZ0024095','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','88332','P.V.A.K. Raju',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-11 20:58:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   51.9 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             11-10-24,14:18\r\n\r\n\r\n'),
(455,'2024-10-11','19:24:28','90.00000','105.00000','PIPERACILLIN','NA','TZ0024095','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','88332','P.V.A.K. Raju',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-11 20:58:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.85 pH\r\nmV                   51.0 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             11-10-24,14:18\r\n\r\n\r\n'),
(456,'2024-10-11','20:09:57','90.00000','105.00000','PIPERACILLIN','NA','TZ0024095','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','88332','P.V.A.K. Raju',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-11 20:59:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.20 pH\r\nmV                   31.0 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             11-10-24,14:18\r\n\r\n\r\n'),
(457,'2024-10-11','21:01:38','90.00000','105.00000','PIPERACILLIN','NA','TZ0024095','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','88332','P.V.A.K. Raju',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-11 22:03:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.46 pH\r\nmV                   15.4 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             11-10-24,14:18\r\n\r\n\r\n'),
(458,'2024-10-11','22:27:56','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024095','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','3',1,'88515','Prashanth B','2024-10-11 23:20:06','Before carbon filtration-1','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.25 pH\r\nmV                   27.4 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             11-10-24,14:18\r\n\r\n\r\n'),
(459,'2024-10-11','22:44:38','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024095','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','3',1,'88515','Prashanth B','2024-10-11 23:23:40','Before carbon filtration-2','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.21 pH\r\nmV                   29.6 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             11-10-24,14:18\r\n\r\n\r\n'),
(460,'2024-10-12','20:55:51','90.00000','105.00000','TAZOBACTAM','NA','TZ0024096','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','1',1,'95681','T. Sivanarayana','2024-10-12 22:00:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.53 pH\r\nmV                   128.1 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             12-10-24,15:30\r\n\r\n\r\n'),
(461,'2024-10-13','00:42:17','90.00000','105.00000','PIPERACILLIN','NA','TZ0024096','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'88515','Prashanth B','2024-10-13 02:37:02','Reviewed(Initial Piperacillin)','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.91 pH\r\nmV                   48.7 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             13-10-24,00:25\r\n\r\n\r\n'),
(462,'2024-10-13','01:32:56','90.00000','105.00000','PIPERACILLIN','NA','TZ0024096','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'88515','Prashanth B','2024-10-13 02:38:31','Reviewed(Piperacillin-First adjustment)','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.93 pH\r\nmV                   47.4 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             13-10-24,00:25\r\n\r\n\r\n'),
(463,'2024-10-13','02:14:02','90.00000','105.00000','PIPERACILLIN','NA','TZ0024096','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'88515','Prashanth B','2024-10-13 02:39:22','Reviewed(Piperacillin-Second adjustment)','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.20 pH\r\nmV                   32.0 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             13-10-24,00:25\r\n\r\n\r\n'),
(464,'2024-10-13','03:03:38','90.00000','105.00000','PIPERACILLIN','NA','TZ0024096','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'88515','Prashanth B','2024-10-13 05:20:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.45 pH\r\nmV                   17.3 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             13-10-24,00:25\r\n\r\n\r\n'),
(465,'2024-10-13','04:51:19','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024096','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','3',1,'88515','Prashanth B','2024-10-13 05:20:59','Reviewed(Before Carbon filtration-1)','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.31 pH\r\nmV                   25.3 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             13-10-24,00:25\r\n\r\n\r\n'),
(466,'2024-10-13','05:20:39','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024096','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','3',1,'88515','Prashanth B','2024-10-13 05:25:04','Reviewed(Before Carbon filtration-2)','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.29 pH\r\nmV                   26.5 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             13-10-24,00:25\r\n\r\n\r\n'),
(467,'2024-10-13','10:49:53','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005359','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-10-13 11:30:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.67 pH\r\nmV                   310.4 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             13-10-24,10:43\r\nLog #                179\r\n\r\n\r\n'),
(468,'2024-10-13','10:53:13','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005359','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-10-13 11:31:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   173.1 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             13-10-24,10:43\r\nLog #                180\r\n\r\n\r\n'),
(469,'2024-10-13','10:56:00','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005359','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-10-13 11:31:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -1.8 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             13-10-24,10:43\r\nLog #                181\r\n\r\n\r\n'),
(470,'2024-10-13','11:04:47','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005359','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-10-13 11:31:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -176.9 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             13-10-24,10:43\r\nLog #                182\r\n\r\n\r\n'),
(471,'2024-10-13','11:08:35','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005359','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-10-13 11:31:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.49 pH\r\nmV                   -322.1 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             13-10-24,10:43\r\nLog #                183\r\n\r\n\r\n'),
(472,'2024-10-13','12:01:28','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','pH','TZ0024095','E4IP24001245','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','3',1,'82735','M.D. Babar Qhadri','2024-10-13 12:18:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.85 pH\r\nmV                   66.5 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             13-10-24,10:43\r\n\r\n\r\n'),
(473,'2024-10-13','12:09:19','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','pH','TZ0024096','E4IP24001250','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','3',1,'82735','M.D. Babar Qhadri','2024-10-13 12:19:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.05 pH\r\nmV                   54.7 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             13-10-24,10:43\r\n\r\n\r\n'),
(474,'2024-10-14','07:09:20','90.00000','105.00000','TAZOBACTAM','NA','TZ0024097','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','1',1,'95681','T. Sivanarayana','2024-10-14 11:13:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.62 pH\r\nmV                   123.1 mV\r\nTemperature          23.1 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             14-10-24,06:33\r\n\r\n\r\n'),
(475,'2024-10-14','07:33:46','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005360','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','1.68 pH',1,'90018','Ch. Satyam','2024-10-14 07:56:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.65 pH\r\nmV                   310.0 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(476,'2024-10-14','07:36:29','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005360','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','4.01pH',1,'90018','Ch. Satyam','2024-10-14 07:56:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   171.7 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(477,'2024-10-14','07:38:46','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005360','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','7.00pH',1,'90018','Ch. Satyam','2024-10-14 07:56:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -2.3 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(478,'2024-10-14','07:43:52','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005360','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','10.01 pH',1,'90018','Ch. Satyam','2024-10-14 07:56:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.02 pH\r\nmV                   -178.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(479,'2024-10-14','07:46:59','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005360','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','12.46 pH',1,'90018','Ch. Satyam','2024-10-14 07:57:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -322.9 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(480,'2024-10-14','10:06:50','90.00000','105.00000','Piperacillin and Tazobactam For injection','Mobile Phase A','TZ0424091','E4FP24000685','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,1,101,'pH','Pipertazoinj',1,'90018','Ch. Satyam','2024-10-14 10:35:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   84.3 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(481,'2024-10-14','10:08:07','90.00000','105.00000','PIPERACILLIN','NA','TZ0024097','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-14 11:14:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.93 pH\r\nmV                   46.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             14-10-24,06:33\r\nLog #                139\r\n\r\n\r\n'),
(482,'2024-10-14','10:16:16','90.00000','105.00000','Piperacillin and Tazobactam For injection','Mobile phase B','TZ0424091','E4FP24000685','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,1,101,'pH','Pipertazoinj',1,'90018','Ch. Satyam','2024-10-14 10:36:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   84.7 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\nLog #                192\r\n\r\n\r\n'),
(483,'2024-10-14','10:28:51','90.00000','105.00000','Penicillin G Benzathine','dissolution','PG0124002A','E4STFP24000563','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Penicillin G',1,'90018','Ch. Satyam','2024-10-14 11:34:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.32 pH\r\nmV                   215.0 mV\r\nTemperature          28.7 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(484,'2024-10-14','10:34:27','90.00000','105.00000','Penicillin G Benzathine','dissolution','PG0124002A','E4STFP24000563','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Penicillin G',1,'90018','Ch. Satyam','2024-10-14 11:34:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.50 pH\r\nmV                   -32.0 mV\r\nTemperature          30.8 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(485,'2024-10-14','10:50:42','90.00000','105.00000','PIPERACILLIN','NA','TZ0024097','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-14 11:14:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.95 pH\r\nmV                   45.7 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             14-10-24,06:33\r\n\r\n\r\n'),
(486,'2024-10-14','10:50:03','90.00000','105.00000','Piperacillin and Tazobactam for Injection','Mobile phase RS test','PT0424074A','E4STFP24000614','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205189','P. Ravi Kumar',2,0,101,'pH','PIPERTAZO',1,'90018','Ch. Satyam','2024-10-14 11:35:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.80 pH\r\nmV                   185.5 mV\r\nTemperature          26.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\nLog #                196\r\n\r\n\r\n'),
(487,'2024-10-14','11:36:12','90.00000','105.00000','PIPERACILLIN','NA','TZ0024097','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-14 13:58:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.24 pH\r\nmV                   29.3 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             14-10-24,06:33\r\n\r\n\r\n'),
(488,'2024-10-14','11:54:05','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005373','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-14 13:42:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   310.8 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             14-10-24,11:49\r\n\r\n\r\n'),
(489,'2024-10-14','12:22:30','90.00000','105.00000','PIPERACILLIN','NA','TZ0024097','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-14 13:59:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.40 pH\r\nmV                   19.5 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             14-10-24,06:33\r\n\r\n\r\n'),
(490,'2024-10-14','12:29:59','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005373','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-14 13:42:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.05 pH\r\nmV                   170.8 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             14-10-24,11:49\r\nLog #                158\r\n\r\n\r\n'),
(491,'2024-10-14','12:33:58','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005373','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-14 13:43:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -3.9 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             14-10-24,11:49\r\n\r\n\r\n'),
(492,'2024-10-14','12:35:54','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005373','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-14 13:43:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -177.9 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             14-10-24,11:49\r\n\r\n\r\n'),
(493,'2024-10-14','12:37:59','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005373','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-14 13:44:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.45 pH\r\nmV                   -321.5 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             14-10-24,11:49\r\n\r\n\r\n'),
(494,'2024-10-14','13:52:31','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005495','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-14 15:10:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   73.3 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(495,'2024-10-14','13:54:27','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005496','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-14 15:11:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.73 pH\r\nmV                   71.8 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(496,'2024-10-14','14:00:00','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005497','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-14 15:12:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   71.2 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(497,'2024-10-14','14:02:00','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005498','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-14 15:13:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   73.9 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(498,'2024-10-14','14:04:37','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024097','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','3',1,'95681','T. Sivanarayana','2024-10-14 16:33:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   26.6 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             14-10-24,06:33\r\nLog #                143\r\n\r\n\r\n'),
(499,'2024-10-14','14:03:49','90.00000','105.00000','Purified water','pH','PW/SP03','E4RM24005499','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-14 15:14:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   72.7 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\nLog #                201\r\n\r\n\r\n'),
(500,'2024-10-14','14:07:57','90.00000','105.00000','Pure steam','pH','PSC/SP09','E4RM24005500','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'205208','S. Siddhartha Kumar Reddy','2024-10-14 15:14:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.79 pH\r\nmV                   68.4 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(501,'2024-10-14','14:09:46','90.00000','105.00000','Purified water','pH','PW/SP21','E4RM24005501','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-14 15:16:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   71.0 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(502,'2024-10-14','14:12:03','90.00000','105.00000','Water For Injection','pH','WFI/SP22','E4RM24005502','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-14 15:17:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.77 pH\r\nmV                   69.4 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(503,'2024-10-14','14:14:08','90.00000','105.00000','Pure steam','pH','PSC/SP14','E4RM24005503','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'205208','S. Siddhartha Kumar Reddy','2024-10-14 15:18:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.06 pH\r\nmV                   52.3 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(504,'2024-10-14','14:21:28','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ002097','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','3',1,'95681','T. Sivanarayana','2024-10-14 16:33:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.25 pH\r\nmV                   30.0 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             14-10-24,06:33\r\n\r\n\r\n'),
(505,'2024-10-14','15:38:22','90.00000','105.00000','Macckankey broth','NA','RS013/24007','E4SO24000673','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','NA',1,'205143','Ch. Prasad','2024-10-14 17:46:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.43 pH\r\nmV                   -26.6 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             14-10-24,11:49\r\n\r\n\r\n'),
(506,'2024-10-14','17:53:18','90.00000','105.00000','Epuipment Qualification samples pH','pH','NA','E4EQ24000368','Equipment Qualification Samples','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Epuipment Qualification',1,'205208','S. Siddhartha Kumar Reddy','2024-10-14 18:03:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.58 pH\r\nmV                   80.5 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(507,'2024-10-14','18:17:36','90.00000','105.00000','Peptone','NA','RS007/24036','E4SO24000674','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-15 13:17:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.61 pH\r\nmV                   21.8 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             14-10-24,11:49\r\n\r\n\r\n'),
(508,'2024-10-14','18:23:00','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0024097','E4IP24001255','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH (Bulk)',1,'205208','S. Siddhartha Kumar Reddy','2024-10-14 18:35:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.29 pH\r\nmV                   38.9 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(509,'2024-10-14','18:28:29','90.00000','105.00000','Macckankey broth','NA','RS013/24007','E4SO24000673','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','NA',1,'205143','Ch. Prasad','2024-10-15 13:17:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.45 pH\r\nmV                   -27.8 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             14-10-24,11:49\r\n\r\n\r\n'),
(510,'2024-10-14','19:11:16','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC','TZ0424091','E4FP24000685','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205058','C. Srinivasulu',2,0,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-14 19:41:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   84.6 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(511,'2024-10-14','19:24:18','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC','TZ0424091','E4FP24000685','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205058','C. Srinivasulu',2,0,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-14 19:42:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   84.0 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             14-10-24,07:27\r\n\r\n\r\n'),
(512,'2024-10-14','21:04:01','90.00000','105.00000','Peptone','NA','RS007/24036','E4SO24000674','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-15 13:18:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.58 pH\r\nmV                   23.4 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             14-10-24,11:49\r\n\r\n\r\n'),
(513,'2024-10-15','07:12:00','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005377','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-10-15 07:51:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   311.8 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(514,'2024-10-15','07:16:21','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005377','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-10-15 07:51:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.04 pH\r\nmV                   173.6 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(515,'2024-10-15','07:20:51','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005377','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-10-15 07:51:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -1.6 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(516,'2024-10-15','07:36:15','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005377','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-10-15 07:52:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.05 pH\r\nmV                   -178.4 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(517,'2024-10-15','07:40:34','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005377','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-10-15 07:52:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -321.7 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(518,'2024-10-15','11:03:16','90.00000','105.00000','Sodium Hydrogen Carbonate','Identification','2449200363','E4RM24005427','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,1,101,'pH','pH   Test',1,'86253','D. Jayapal Reddy','2024-10-15 17:22:00','Reviewed (Refer PNC No : PNC-EP4-QC-24-0138)','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   91.1 mV\r\nTemperature          35.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(519,'2024-10-15','11:20:14','90.00000','105.00000','Piperacillin and Tazobactam for Injection','ASSAY','PT0424074A','E4STFP24000614','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,0,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-15 11:43:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   90.5 mV\r\nTemperature          34.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(520,'2024-10-15','11:31:12','90.00000','105.00000','Tazobactam USP','Assay','A2403A020','E4AQ24000054','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205266','K. Vara prasad',2,1,101,'pH','Assay',1,'82735','M.D. Babar Qhadri','2024-10-15 11:44:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   88.8 mV\r\nTemperature          31.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(521,'2024-10-15','12:02:29','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005390','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','88599','MD Mahamad',2,1,102,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-10-15 12:29:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   1.67 pH\r\nmV                   311.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             15-10-24,11:58\r\n\r\n\r\n'),
(522,'2024-10-15','12:04:25','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005390','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','88599','MD Mahamad',2,1,102,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-10-15 12:30:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   175.4 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             15-10-24,11:58\r\n\r\n\r\n'),
(523,'2024-10-15','12:06:13','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005390','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','88599','MD Mahamad',2,1,102,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-10-15 12:30:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   0.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             15-10-24,11:58\r\n\r\n\r\n'),
(524,'2024-10-15','12:17:56','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005390','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','88599','MD Mahamad',2,1,102,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-10-15 12:30:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   10.05 pH\r\nmV                   -177.4 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             15-10-24,11:58\r\n\r\n\r\n'),
(525,'2024-10-15','12:20:04','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005390','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','88599','MD Mahamad',2,1,102,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-10-15 12:30:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   12.49 pH\r\nmV                   -318.8 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             15-10-24,11:58\r\n\r\n\r\n'),
(526,'2024-10-15','14:44:38','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424111','E4FP24000696','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Pipertazoinj',1,'84378','Shivanand. W','2024-10-15 15:13:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.87 pH\r\nmV                   66.7 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(527,'2024-10-15','14:49:17','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PT0424001A','E4STFP24000636','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-15 15:14:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.40 pH\r\nmV                   94.3 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(528,'2024-10-15','15:02:39','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PT0424026A','E4STFP24000637','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-15 15:15:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.20 pH\r\nmV                   105.7 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(529,'2024-10-15','15:05:07','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PT0424026A','E4STFP24000638','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-15 15:16:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   87.4 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(530,'2024-10-15','15:28:13','90.00000','105.00000','Piperacillin and Tazobactam for Injection','Assay','TI0323002A','E4STFP24000596','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','90017','G. Srishailam',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-15 16:05:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   90.2 mV\r\nTemperature          33.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(531,'2024-10-15','15:39:01','90.00000','105.00000','TAZOBACTAM','NA','TZ0024098','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','1',1,'88515','Prashanth B','2024-10-15 22:20:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.57 pH\r\nmV                   125.4 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             15-10-24,10:28\r\n\r\n\r\n'),
(532,'2024-10-15','15:40:02','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005516','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-10-15 16:07:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.72 pH\r\nmV                   74.7 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(533,'2024-10-15','15:41:38','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005517','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-10-15 16:08:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   77.3 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(534,'2024-10-15','15:44:39','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005518','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-10-15 16:09:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.73 pH\r\nmV                   74.6 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(535,'2024-10-15','15:47:01','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005519','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-10-15 16:09:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.76 pH\r\nmV                   73.1 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(536,'2024-10-15','15:49:14','90.00000','105.00000','Water For Injection','pH','WFI/SP13','E4RM24005520','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-10-15 16:10:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.67 pH\r\nmV                   78.3 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(537,'2024-10-15','15:51:12','90.00000','105.00000','Pure steam','pH','PSC/SP10','E4RM24005521','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'84378','Shivanand. W','2024-10-15 16:10:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.67 pH\r\nmV                   78.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(538,'2024-10-15','15:53:16','90.00000','105.00000','Purified water','pH','PW/SP30','E4RM24005522','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-10-15 16:11:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.66 pH\r\nmV                   78.7 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(539,'2024-10-15','15:55:03','90.00000','105.00000','Purified water','pH','PW/SP50','E4RM24005523','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-10-15 16:11:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   73.3 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(540,'2024-10-15','15:57:00','90.00000','105.00000','Water For Injection','pH','WFI/SP36','E4RM24005524','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-10-15 16:12:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   76.6 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(541,'2024-10-15','15:59:14','90.00000','105.00000','Pure steam','pH','PSC/SP25','E4RM24005525','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'84378','Shivanand. W','2024-10-15 16:13:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.72 pH\r\nmV                   75.3 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(542,'2024-10-15','17:04:33','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0423011A','E4STFP24000634','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-10-15 17:16:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.04 pH\r\nmV                   56.9 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(543,'2024-10-15','17:08:51','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0423011A','E4STFP24000635','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-10-15 17:16:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.90 pH\r\nmV                   64.6 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(544,'2024-10-15','17:19:35','90.00000','105.00000','PIPERACILLIN','NA','TZ0024098','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'88515','Prashanth B','2024-10-15 22:20:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.89 pH\r\nmV                   48.9 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             15-10-24,10:28\r\n\r\n\r\n'),
(545,'2024-10-15','18:02:37','90.00000','105.00000','PIPERACILLIN','NA','TZ0024098','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'88515','Prashanth B','2024-10-15 22:21:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.91 pH\r\nmV                   47.6 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             15-10-24,10:28\r\n\r\n\r\n'),
(546,'2024-10-15','18:38:33','90.00000','105.00000','Glass Vials for pH','pH (Vial-1)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:34:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.28 pH\r\nmV                   277.3 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(547,'2024-10-15','18:40:39','90.00000','105.00000','Glass Vials for pH','pH(Vial-2)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:35:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.27 pH\r\nmV                   277.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(548,'2024-10-15','18:43:07','90.00000','105.00000','Glass Vials for pH','pH(Vial-3)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:35:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.32 pH\r\nmV                   274.2 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(549,'2024-10-15','18:46:06','90.00000','105.00000','Glass Vials for pH','pH(Vial-4)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:35:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.39 pH\r\nmV                   270.4 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(550,'2024-10-15','18:52:08','90.00000','105.00000','PIPERACILLIN','NA','TZ0024098','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'88515','Prashanth B','2024-10-15 22:21:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.29 pH\r\nmV                   25.9 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             15-10-24,10:28\r\n\r\n\r\n'),
(551,'2024-10-15','18:52:21','90.00000','105.00000','Glass Vials for pH','pH(Vial-5)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:36:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.31 pH\r\nmV                   275.1 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(552,'2024-10-15','18:54:24','90.00000','105.00000','Glass Vials for pH','pH(Vial-6)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:37:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.61 pH\r\nmV                   257.6 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(553,'2024-10-15','18:57:22','90.00000','105.00000','Glass Vials for pH','pH(Vial-7)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:39:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.60 pH\r\nmV                   257.5 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(554,'2024-10-15','18:59:36','90.00000','105.00000','Glass Vials for pH','pH(Vial-8)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:39:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.79 pH\r\nmV                   246.7 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(555,'2024-10-15','19:02:03','90.00000','105.00000','Glass Vials for pH','pH(Vial-9)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:39:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.54 pH\r\nmV                   261.8 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(556,'2024-10-15','19:04:23','90.00000','105.00000','Glass Vials for pH','pH(Vial-10)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:39:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.27 pH\r\nmV                   277.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(557,'2024-10-15','19:07:15','90.00000','105.00000','Glass Vials for pH','pH(Vial-11)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:40:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.42 pH\r\nmV                   268.3 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(558,'2024-10-15','19:10:10','90.00000','105.00000','Glass Vials for pH','pH(Vial-12)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:41:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.56 pH\r\nmV                   260.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(559,'2024-10-15','19:12:36','90.00000','105.00000','Glass Vials for pH','pH(Vial-13)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:41:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.71 pH\r\nmV                   252.1 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(560,'2024-10-15','19:15:05','90.00000','105.00000','Glass Vials for pH','pH(Vial-14)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:40:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.46 pH\r\nmV                   266.6 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(561,'2024-10-15','19:17:01','90.00000','105.00000','Glass Vials for pH','pH(Vial-15)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:40:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.55 pH\r\nmV                   261.0 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(562,'2024-10-15','19:18:59','90.00000','105.00000','Glass Vials for pH','pH(Vial-16)','NA','E4EQ24000364','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:42:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.76 pH\r\nmV                   248.4 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(563,'2024-10-15','19:42:46','90.00000','105.00000','PIPERACILLIN','NA','TZ0024098','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'88515','Prashanth B','2024-10-15 22:22:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.48 pH\r\nmV                   14.3 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             15-10-24,10:28\r\n\r\n\r\n'),
(564,'2024-10-15','20:20:34','90.00000','105.00000','Glass Vials for pH','pH(Vial-1)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:42:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.28 pH\r\nmV                   277.2 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(565,'2024-10-15','20:24:50','90.00000','105.00000','Glass Vials for pH','pH(Vial-2)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:42:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.38 pH\r\nmV                   271.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(566,'2024-10-15','20:27:06','90.00000','105.00000','Glass Vials for pH','pH(Vial-3)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:43:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.13 pH\r\nmV                   285.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(567,'2024-10-15','20:29:07','90.00000','105.00000','Glass Vials for pH','pH(Vial-4)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:43:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.34 pH\r\nmV                   273.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(568,'2024-10-15','20:30:51','90.00000','105.00000','Glass Vials for pH','pH(Vial-5)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:43:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.36 pH\r\nmV                   271.9 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(569,'2024-10-15','20:32:38','90.00000','105.00000','Glass Vials for pH','pH(Vial-6)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:44:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.48 pH\r\nmV                   265.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(570,'2024-10-15','20:35:15','90.00000','105.00000','Glass Vials for pH','pH(Vial-7)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:44:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.63 pH\r\nmV                   256.5 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(571,'2024-10-15','20:37:31','90.00000','105.00000','Glass Vials for pH','pH(Vial-8)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:44:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.78 pH\r\nmV                   247.0 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(572,'2024-10-15','20:39:46','90.00000','105.00000','Glass Vials for pH','pH(Vial-9)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:43:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.52 pH\r\nmV                   262.4 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(573,'2024-10-15','20:41:57','90.00000','105.00000','Glass Vials for pH','pH(Vial-10)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:45:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.58 pH\r\nmV                   259.6 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(574,'2024-10-15','20:44:18','90.00000','105.00000','Glass Vials for pH','pH(Vial-11)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:45:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.65 pH\r\nmV                   255.0 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(575,'2024-10-15','20:46:38','90.00000','105.00000','Glass Vials for pH','pH(Vial-12)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:45:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.40 pH\r\nmV                   269.2 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(576,'2024-10-15','20:48:46','90.00000','105.00000','Glass Vials for pH','pH(Vial-13)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:45:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.56 pH\r\nmV                   260.0 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(577,'2024-10-15','20:51:35','90.00000','105.00000','Glass Vials for pH','pH(Vial-14)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:46:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.71 pH\r\nmV                   251.5 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(578,'2024-10-15','20:54:14','90.00000','105.00000','Glass Vials for pH','pH(Vial-15)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:46:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.51 pH\r\nmV                   263.0 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(579,'2024-10-15','20:57:04','90.00000','105.00000','Glass Vials for pH','pH(Vial-16)','NA','E4EQ24000365','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'86253','D. Jayapal Reddy','2024-10-16 09:46:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.76 pH\r\nmV                   248.5 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             15-10-24,07:06\r\n\r\n\r\n'),
(580,'2024-10-15','22:35:54','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024098','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','3',1,'88629','Raghuram Reddy','2024-10-15 23:16:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.39 pH\r\nmV                   20.2 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             15-10-24,10:28\r\n\r\n\r\n'),
(581,'2024-10-15','22:50:02','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024098','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','3',1,'88629','Raghuram Reddy','2024-10-15 23:17:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.36 pH\r\nmV                   21.3 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             15-10-24,10:28\r\n\r\n\r\n'),
(582,'2024-10-16','07:06:26','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005397','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','1.68 pH',1,'90778','B. Bhaskar','2024-10-16 07:32:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.67 pH\r\nmV                   312.5 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(583,'2024-10-16','07:09:25','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005397','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','4.01pH',1,'90778','B. Bhaskar','2024-10-16 07:33:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   173.0 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(584,'2024-10-16','07:11:33','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005397','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','7.00pH',1,'90778','B. Bhaskar','2024-10-16 07:33:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -1.9 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(585,'2024-10-16','07:13:54','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005397','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','10.01 pH',1,'90778','B. Bhaskar','2024-10-16 07:34:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.03 pH\r\nmV                   -179.1 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(586,'2024-10-16','07:16:42','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005397','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','12.46 pH',1,'90778','B. Bhaskar','2024-10-16 07:34:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.46 pH\r\nmV                   -321.2 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(587,'2024-10-16','08:12:37','90.00000','105.00000','Penicillin G Benzathine','dissolution','PG0124003A','E4STFP24000571','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Penicillin G',1,'205208','S. Siddhartha Kumar Reddy','2024-10-16 11:52:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.32 pH\r\nmV                   218.9 mV\r\nTemperature          30.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(588,'2024-10-16','08:18:30','90.00000','105.00000','Penicillin G Benzathine','dissolution','PG0124003A','E4STFP24000571','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Penicillin G',1,'205208','S. Siddhartha Kumar Reddy','2024-10-16 11:55:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.49 pH\r\nmV                   -31.1 mV\r\nTemperature          32.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(589,'2024-10-16','10:31:38','90.00000','105.00000','AMPICILLIN AND SULBACTAM','UOD','AS0124009','E4FP24000690','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-10-16 12:14:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.98 pH\r\nmV                   120.3 mV\r\nTemperature          32.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(590,'2024-10-16','10:49:51','90.00000','105.00000','Nafcillin For Injection','RS','NF0122001A','E4STFP24000621','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205043','T.V. Subbaiah',2,0,101,'pH','pH_Test',1,'90778','B. Bhaskar','2024-10-16 11:10:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.51 pH\r\nmV                   208.7 mV\r\nTemperature          32.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(591,'2024-10-16','11:05:53','90.00000','105.00000','Piperacillin and Tazobactam For injection','Assay','TZ0424092','E4FP24000689','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','89853','J.Ravi',2,1,101,'pH','Pipertazoinj',1,'90778','B. Bhaskar','2024-10-16 11:45:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   87.2 mV\r\nTemperature          31.9 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(592,'2024-10-16','11:23:44','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0024098','E4IP24001260','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','pH (Bulk)',1,'90778','B. Bhaskar','2024-10-16 11:34:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.16 pH\r\nmV                   47.5 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\nLog #                277\r\n\r\n\r\n'),
(593,'2024-10-16','11:59:50','90.00000','105.00000','Sodium bicarbonate','Carbonate','SL/214/166','E4RM24005526','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','Carbonate test',1,'90778','B. Bhaskar','2024-10-16 12:25:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.09 pH\r\nmV                   -65.0 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(594,'2024-10-16','12:04:43','90.00000','105.00000','Sodium bicarbonate','Carbonate','SL/214/167','E4RM24005527','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','Carbonate test',1,'90778','B. Bhaskar','2024-10-16 12:26:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.11 pH\r\nmV                   -66.6 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\nLog #                279\r\n\r\n\r\n'),
(595,'2024-10-16','12:11:10','90.00000','105.00000','Sodium bicarbonate','Carbonate','SL/214/169','E4RM24005528','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','Carbonate test',1,'90778','B. Bhaskar','2024-10-16 12:27:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.12 pH\r\nmV                   -67.1 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(596,'2024-10-16','12:22:42','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005406','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-16 13:39:27','verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   311.4 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             16-10-24,11:57\r\n\r\n\r\n'),
(597,'2024-10-16','12:20:53','90.00000','105.00000','AMPICILLIN AND SULBACTAM','pH','AS0124009','E4FP24000690','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','AS',1,'90778','B. Bhaskar','2024-10-16 12:41:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.18 pH\r\nmV                   -129.8 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\nLog #                282\r\n\r\n\r\n'),
(598,'2024-10-16','12:29:21','90.00000','105.00000','AMPICILLIN AND SULBACTAM','pH','AS0324073','E4FP24000693','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','AS',1,'90778','B. Bhaskar','2024-10-16 12:42:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.10 pH\r\nmV                   -125.1 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(599,'2024-10-16','12:33:41','90.00000','105.00000','pH Buffer','NA','NA','E4IS24005406','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-16 13:40:22','verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.06 pH\r\nmV                   170.4 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             16-10-24,11:57\r\nLog #                169\r\n\r\n\r\n'),
(600,'2024-10-16','12:38:00','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005406','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-16 13:40:40','verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.05 pH\r\nmV                   -4.1 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             16-10-24,11:57\r\n\r\n\r\n'),
(601,'2024-10-16','12:41:51','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005406','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-16 13:41:29','verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   9.99 pH\r\nmV                   -176.5 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             16-10-24,11:57\r\nLog #                171\r\n\r\n\r\n'),
(602,'2024-10-16','12:44:35','90.00000','105.00000','pH Buffer','NA','NA','	E4INS24005406','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-16 13:43:09','verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.45 pH\r\nmV                   -320.9 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             16-10-24,11:57\r\n\r\n\r\n'),
(603,'2024-10-16','12:46:34','90.00000','105.00000','AMPICILLIN AND SULBACTAM','SST pH','AS0124009','E4FP24000690','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','AS',1,'84378','Shivanand. W','2024-10-17 07:32:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.03 pH\r\nmV                   114.0 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(604,'2024-10-16','13:59:32','90.00000','105.00000','Ampicillin and sulbactam for Injection','pH','AS1524001A','E4STFP24000639','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','Ampisulbactam',1,'90778','B. Bhaskar','2024-10-16 14:12:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.26 pH\r\nmV                   -134.0 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(605,'2024-10-16','14:03:30','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TR0424001A','E4STFP24000640','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-10-16 14:13:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.31 pH\r\nmV                   39.0 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(606,'2024-10-16','15:14:31','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005567','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-16 16:11:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   72.1 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(607,'2024-10-16','15:16:11','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005568','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-16 16:11:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.05 pH\r\nmV                   53.9 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(608,'2024-10-16','15:18:14','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005569','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-10-16 16:12:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.77 pH\r\nmV                   70.2 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(609,'2024-10-16','15:20:18','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005570','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-10-16 16:49:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   74.1 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(610,'2024-10-16','15:22:06','90.00000','105.00000','Purified water','pH','PW/SP05','E4RM24005571','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-16 16:12:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   73.9 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(611,'2024-10-16','15:25:53','90.00000','105.00000','Peptone','	NA','RS007/24037','E4SO24000682','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-18 14:26:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.40 pH\r\nmV                   33.3 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             16-10-24,11:57\r\n\r\n\r\n'),
(612,'2024-10-16','15:24:05','90.00000','105.00000','Pure steam','pH','PSC/SP20','E4RM24005572','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'86253','D. Jayapal Reddy','2024-10-16 16:12:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.81 pH\r\nmV                   68.2 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\nLog #                292\r\n\r\n\r\n'),
(613,'2024-10-16','15:28:00','90.00000','105.00000','Purified water','pH','PW/SP31','E4RM24005573','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-16 16:13:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   71.9 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(614,'2024-10-16','15:29:56','90.00000','105.00000','Water For Injection','pH','WFI/SP37','E4RM24005574','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-10-16 16:13:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   74.0 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(615,'2024-10-16','15:31:35','90.00000','105.00000','Water For Injection','pH','WFI/SP42','E4RM24005575','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-10-16 16:13:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.73 pH\r\nmV                   72.8 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(616,'2024-10-16','15:33:33','90.00000','105.00000','Pure steam','pH','PSC/SP27','E4RM24005576','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'86253','D. Jayapal Reddy','2024-10-16 16:14:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   72.1 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(617,'2024-10-16','16:00:26','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424110','E4FP24000691','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-10-16 16:24:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.13 pH\r\nmV                   49.9 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(618,'2024-10-16','16:09:30','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0424093','E4FP24000692','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-10-16 16:25:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.02 pH\r\nmV                   55.7 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(619,'2024-10-16','16:13:16','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0424094','E4FP24000694','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-10-16 16:25:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.07 pH\r\nmV                   53.0 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(620,'2024-10-16','16:16:13','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0424095','E4FP24000695','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-10-16 16:25:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.09 pH\r\nmV                   51.6 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(621,'2024-10-16','16:50:37','90.00000','105.00000','AMPICILLIN AND SULBACTAM','pH','AS0124010','E4FP24000697','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-10-16 17:01:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.23 pH\r\nmV                   -132.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(622,'2024-10-16','16:53:48','90.00000','105.00000','AMPICILLIN AND SULBACTAM','pH','AS0324074','E4FP24000698','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-10-16 17:01:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.24 pH\r\nmV                   -132.4 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\n\r\n\r\n'),
(623,'2024-10-16','18:08:28','90.00000','105.00000','Piperacillin and Tazobactam For injection','UOD','PT0424110','E4FP24000691','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,1,101,'pH','Pipertazoinj',1,'84378','Shivanand. W','2024-10-17 10:21:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   88.7 mV\r\nTemperature          36.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_118\r\nLast Cal             16-10-24,07:03\r\nLog #                303\r\n\r\n\r\n'),
(624,'2024-10-16','19:21:51','90.00000','105.00000','Peptone','NA','RS007/24037','E4SO24000682','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-18 14:26:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.59 pH\r\nmV                   22.2 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             16-10-24,11:57\r\n\r\n\r\n'),
(625,'2024-10-16','22:28:33','90.00000','105.00000','Penicillin G','NA','PG0124001A','E4STFP24000551','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-18 14:27:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.53 pH\r\nmV                   25.7 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             16-10-24,11:57\r\n\r\n\r\n'),
(626,'2024-10-16','22:30:55','90.00000','105.00000','Penicillin G','NA','PG0124002A','E4STFP24000558','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-18 14:27:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.43 pH\r\nmV                   31.5 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             16-10-24,11:57\r\n\r\n\r\n'),
(627,'2024-10-16','22:34:42','90.00000','105.00000','Penicillin G','NA','PG0124003A','E4STFP24000566','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-18 14:27:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.54 pH\r\nmV                   25.2 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             16-10-24,11:57\r\n\r\n\r\n'),
(628,'2024-10-17','07:15:07','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005419','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','1.68 pH',1,'84378','Shivanand. W','2024-10-17 07:47:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   311.0 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\nLog #                304\r\n\r\n\r\n'),
(629,'2024-10-17','07:17:20','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005419','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','4.01pH',1,'84378','Shivanand. W','2024-10-17 07:47:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.04 pH\r\nmV                   172.2 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\nLog #                305\r\n\r\n\r\n'),
(630,'2024-10-17','07:19:19','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005419','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','7.00pH',1,'84378','Shivanand. W','2024-10-17 07:48:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -2.6 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\nLog #                306\r\n\r\n\r\n'),
(631,'2024-10-17','07:21:45','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005419','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','10.01 pH',1,'84378','Shivanand. W','2024-10-17 07:48:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.03 pH\r\nmV                   -179.5 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\nLog #                307\r\n\r\n\r\n'),
(632,'2024-10-17','07:23:52','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005419','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','12.46 pH',1,'84378','Shivanand. W','2024-10-17 07:48:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.43 pH\r\nmV                   -320.2 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\nLog #                308\r\n\r\n\r\n'),
(633,'2024-10-17','08:39:43','90.00000','105.00000','AMPISULBACTAM','Assay pH','2449102395','E4RM24005530','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','Assay buffer pH',1,'84378','Shivanand. W','2024-10-17 11:34:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.03 pH\r\nmV                   117.0 mV\r\nTemperature          31.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\n\r\n\r\n'),
(634,'2024-10-17','08:47:12','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005423','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-17 10:29:13','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.65 pH\r\nmV                   307.9 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             17-10-24,08:45\r\n\r\n\r\n'),
(635,'2024-10-17','08:50:18','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005423','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-17 10:29:24','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.04 pH\r\nmV                   168.4 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             17-10-24,08:45\r\n\r\n\r\n'),
(636,'2024-10-17','08:52:01','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005423','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-17 10:29:41','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -6.6 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             17-10-24,08:45\r\n\r\n\r\n'),
(637,'2024-10-17','08:53:25','90.00000','105.00000','pH Buffer','NA','NA','	E4INS24005423','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-17 10:29:49','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.00 pH\r\nmV                   -179.0 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             17-10-24,08:45\r\n\r\n\r\n'),
(638,'2024-10-17','08:56:20','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005423','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-17 10:29:56','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.42 pH\r\nmV                   -319.6 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             17-10-24,08:45\r\n\r\n\r\n'),
(639,'2024-10-17','09:21:13','90.00000','105.00000','AMPISULBACTAM','Identification','2449101352','E4RM24005472','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','89853','J.Ravi',2,0,101,'pH','Assay buffer pH',1,'84378','Shivanand. W','2024-10-17 11:39:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.01 pH\r\nmV                   118.4 mV\r\nTemperature          32.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\n\r\n\r\n'),
(640,'2024-10-17','09:44:37','90.00000','105.00000','Aseptic media fill','pH','DP24008','E4AMV24000147','Media fill validation','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','Aseptic media fill',1,'90018','Ch. Satyam','2024-10-17 09:51:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.34 pH\r\nmV                   -21.5 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\n\r\n\r\n'),
(641,'2024-10-17','11:13:01','90.00000','105.00000','Ampicillin and sulbactam for Injection','RS BY HPLC','AS0123001A','E4STFP24000625','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,1,101,'pH','Ampisulbactam',1,'84378','Shivanand. W','2024-10-17 11:37:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.97 pH\r\nmV                   119.4 mV\r\nTemperature          28.3 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\n\r\n\r\n'),
(642,'2024-10-17','11:49:45','90.00000','105.00000','0.9% SALINE','NA','RS008/24011','E4SO24000688','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','SALIH',1,'89384','M.V.V. Satish','2024-10-17 15:39:12','verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -6.4 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             17-10-24,08:45\r\n\r\n\r\n'),
(643,'2024-10-17','13:39:34','90.00000','105.00000','R2A','NA','MPT0519','E4SO24000684','PH for Ready to use media','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','R2A',1,'89384','M.V.V. Satish','2024-10-17 15:39:47','verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.25 pH\r\nmV                   -18.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             17-10-24,08:45\r\n\r\n\r\n'),
(644,'2024-10-17','13:42:24','90.00000','105.00000','MSA','NA','MPT0545','E4SO24000685','PH for Ready to use media','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','MSA',1,'89384','M.V.V. Satish','2024-10-17 15:40:39','verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.36 pH\r\nmV                   -25.4 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             17-10-24,08:45\r\n\r\n\r\n'),
(645,'2024-10-17','14:40:21','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005581','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-17 16:30:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.85 pH\r\nmV                   66.2 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\n\r\n\r\n'),
(646,'2024-10-17','14:43:19','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005582','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-17 16:30:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.82 pH\r\nmV                   67.7 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\n\r\n\r\n'),
(647,'2024-10-17','14:57:44','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005583','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90018','Ch. Satyam','2024-10-17 16:32:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.97 pH\r\nmV                   59.0 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\n\r\n\r\n'),
(648,'2024-10-17','15:00:06','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005584','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90018','Ch. Satyam','2024-10-17 16:32:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.82 pH\r\nmV                   67.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\n\r\n\r\n'),
(649,'2024-10-17','15:02:11','90.00000','105.00000','Water For Injection','pH','WFI/SP33','E4RM24005585','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90018','Ch. Satyam','2024-10-17 16:33:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   71.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\n\r\n\r\n'),
(650,'2024-10-17','15:04:28','90.00000','105.00000','Pure steam','pH','PSC/SP21','E4RM24005586','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'90018','Ch. Satyam','2024-10-17 16:33:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.81 pH\r\nmV                   68.6 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\n\r\n\r\n'),
(651,'2024-10-17','15:07:13','90.00000','105.00000','Purified water','pH','PW/SP32','E4RM24005587','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-17 16:33:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.78 pH\r\nmV                   70.4 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\n\r\n\r\n'),
(652,'2024-10-17','15:10:08','90.00000','105.00000','Water For Injection','pH','WFI/SP39','E4RM24005588','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90018','Ch. Satyam','2024-10-17 16:34:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.84 pH\r\nmV                   66.6 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\n\r\n\r\n'),
(653,'2024-10-17','15:44:44','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24016','E4SO24000687','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDBIH',1,'205143','Ch. Prasad','2024-10-18 14:29:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.41 pH\r\nmV                   -27.8 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             17-10-24,08:45\r\n\r\n\r\n'),
(654,'2024-10-17','15:50:36','90.00000','105.00000','0.9% SALINE','NA','RS008/24011','E4SO24000688','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SALIH',1,'205143','Ch. Prasad','2024-10-18 14:30:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.13 pH\r\nmV                   46.9 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             17-10-24,08:45\r\n\r\n\r\n'),
(655,'2024-10-17','16:20:52','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TE0423001A','E4STFP24000641','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'90018','Ch. Satyam','2024-10-17 16:35:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.95 pH\r\nmV                   60.1 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\n\r\n\r\n'),
(656,'2024-10-17','16:24:07','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TE0423001A','E4STFP24000642','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'90018','Ch. Satyam','2024-10-17 16:36:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.72 pH\r\nmV                   73.8 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\n\r\n\r\n'),
(657,'2024-10-17','18:18:10','90.00000','105.00000','Epuipment Qualification samples pH','pH','NA','E4EQ24000374','Equipment Qualification Samples','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Epuipment Qualification',1,'205208','S. Siddhartha Kumar Reddy','2024-10-18 11:28:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.88 pH\r\nmV                   64.4 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             17-10-24,07:11\r\n\r\n\r\n'),
(658,'2024-10-17','18:23:28','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24016','E4SO24000687','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDBIH',1,'205143','Ch. Prasad','2024-10-18 14:32:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.39 pH\r\nmV                   -26.4 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             17-10-24,08:45\r\n\r\n\r\n'),
(659,'2024-10-18','07:34:50','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005446','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','1.68 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-18 07:59:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.64 pH\r\nmV                   305.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(660,'2024-10-18','07:39:12','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005446','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','4.01pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-18 07:59:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   167.0 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(661,'2024-10-18','07:42:41','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005446','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','7.00pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-18 08:00:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -9.2 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(662,'2024-10-18','07:46:12','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005446','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','10.01 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-18 08:00:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -183.8 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(663,'2024-10-18','07:49:46','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005446','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','12.46 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-18 08:00:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.43 pH\r\nmV                   -324.9 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(664,'2024-10-18','08:24:24','90.00000','105.00000','Ampicillin and Sulbactam','UOD','AS0324074','E4FP24000698','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','90017','G. Srishailam',2,1,101,'pH','AS',1,'205208','S. Siddhartha Kumar Reddy','2024-10-18 08:53:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.98 pH\r\nmV                   112.9 mV\r\nTemperature          31.2 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(665,'2024-10-18','08:44:11','90.00000','105.00000','Ampicillin and Sulbactam','RS BY HPLC','AS0124009','E4FP24000690','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,1,101,'pH','AS',1,'205208','S. Siddhartha Kumar Reddy','2024-10-18 08:54:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.04 pH\r\nmV                   107.4 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(666,'2024-10-18','08:58:46','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005451','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-18 10:09:45','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   307.4 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             18-10-24,08:56\r\n\r\n\r\n'),
(667,'2024-10-18','08:55:08','90.00000','105.00000','Ampicillin and Sulbactam','RS BY HPLC','2449102395','E4RM24005530','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205189','P. Ravi Kumar',2,0,101,'pH','RS Mobile phase A',1,'205208','S. Siddhartha Kumar Reddy','2024-10-18 09:34:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.03 pH\r\nmV                   107.5 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\nLog #                331\r\n\r\n\r\n'),
(668,'2024-10-18','09:04:42','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005451','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-18 10:10:03','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.04 pH\r\nmV                   169.1 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             18-10-24,08:56\r\nLog #                191\r\n\r\n\r\n'),
(669,'2024-10-18','09:09:20','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005451','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-18 10:10:21','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -6.2 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             18-10-24,08:56\r\n\r\n\r\n'),
(670,'2024-10-18','09:10:15','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005451','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-18 10:10:42','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.00 pH\r\nmV                   -179.4 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             18-10-24,08:56\r\nLog #                193\r\n\r\n\r\n'),
(671,'2024-10-18','09:12:27','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005451','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-18 10:10:52','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.44 pH\r\nmV                   -322.5 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             18-10-24,08:56\r\nLog #                194\r\n\r\n\r\n'),
(672,'2024-10-18','10:16:25','90.00000','105.00000','Piperacillin and Tazobactam For injection','UOD','PT0424111','E4FP24000696','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,1,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-18 11:30:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   82.0 mV\r\nTemperature          35.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(673,'2024-10-18','10:44:05','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS BY HPLC MP pH','PT0424011A','E4STFP24000607','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,1,101,'pH','PIPERTAZO',1,'205208','S. Siddhartha Kumar Reddy','2024-10-18 11:32:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.98 pH\r\nmV                   52.8 mV\r\nTemperature          29.4 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\nLog #                333\r\n\r\n\r\n'),
(674,'2024-10-18','11:27:48','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0424096','E4FP24000700','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-18 11:41:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.26 pH\r\nmV                   35.9 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(675,'2024-10-18','11:31:09','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0424097','E4FP24000701','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-18 12:41:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.24 pH\r\nmV                   36.5 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(676,'2024-10-18','12:55:07','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324075','E4FP24000702','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-10-18 15:16:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.19 pH\r\nmV                   -135.4 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(677,'2024-10-18','13:00:59','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324076','E4FP24000704','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-10-18 15:11:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.06 pH\r\nmV                   -128.1 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(678,'2024-10-18','14:26:51','90.00000','105.00000','Aseptic media fill','pH','DP24008','E4AMV24000151','Media fill validation','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','Aseptic media fill',1,'86253','D. Jayapal Reddy','2024-10-18 15:10:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.30 pH\r\nmV                   -24.9 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(679,'2024-10-18','14:37:41','90.00000','105.00000','Epuipment Qualification samples pH','pH','NA','E4EQ24000375','Equipment Qualification Samples','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Epuipment Qualification',1,'86253','D. Jayapal Reddy','2024-10-18 15:12:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.95 pH\r\nmV                   54.0 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(680,'2024-10-18','14:39:40','90.00000','105.00000','Epuipment Qualification samples pH','pH','NA','E4EQ24000377','Equipment Qualification Samples','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Epuipment Qualification',1,'86253','D. Jayapal Reddy','2024-10-18 15:12:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.81 pH\r\nmV                   62.1 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(681,'2024-10-18','14:41:58','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005613','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-18 15:13:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   65.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(682,'2024-10-18','14:43:47','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005614','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-18 15:13:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   61.3 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(683,'2024-10-18','14:46:00','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005615','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-10-18 15:13:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.81 pH\r\nmV                   61.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(684,'2024-10-18','14:48:18','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005616','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-10-18 15:14:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.78 pH\r\nmV                   64.1 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(685,'2024-10-18','14:50:48','90.00000','105.00000','Purified water','pH','PW/SP34','E4RM24005617','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-18 15:14:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   64.3 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(686,'2024-10-18','14:54:10','90.00000','105.00000','Purified water','pH','PW/SP36','E4RM24005618','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-18 15:14:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   60.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(687,'2024-10-18','14:56:24','90.00000','105.00000','Purified water','pH','PW/SP40','E4RM24005619','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-18 15:14:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.78 pH\r\nmV                   63.9 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(688,'2024-10-18','14:58:09','90.00000','105.00000','Pure steam','pH','PSC/SP04','E4RM24005620','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'86253','D. Jayapal Reddy','2024-10-18 15:15:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.76 pH\r\nmV                   65.1 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(689,'2024-10-18','15:10:30','90.00000','105.00000','Ampicillin and Sulbactam','UOD','AS0324073','E4FP24000693','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-10-18 15:15:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.01 pH\r\nmV                   112.0 mV\r\nTemperature          32.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_120\r\nLast Cal             18-10-24,07:27\r\n\r\n\r\n'),
(690,'2024-10-19','01:19:33','90.00000','105.00000','TAZOBACTAM','NA','TZ0024099','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','1',1,'88629','Raghuram Reddy','2024-10-19 03:36:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.62 pH\r\nmV                   126.7 mV\r\nTemperature          26.9 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-10-24,00:33\r\n\r\n\r\n'),
(691,'2024-10-19','02:40:12','90.00000','105.00000','PIPERACILLIN','NA','TZ0024099','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-19 03:36:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.94 pH\r\nmV                   49.3 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-10-24,00:33\r\n\r\n\r\n'),
(692,'2024-10-19','03:16:54','90.00000','105.00000','PIPERACILLIN','NA','TZ0024099','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-19 03:37:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.02 pH\r\nmV                   45.2 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-10-24,00:33\r\n\r\n\r\n'),
(693,'2024-10-19','04:02:27','90.00000','105.00000','PIPERACILLIN','NA','TZ0024099','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-19 05:41:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.23 pH\r\nmV                   32.8 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-10-24,00:33\r\n\r\n\r\n'),
(694,'2024-10-19','04:49:20','90.00000','105.00000','PIPERACILLIN','NA','TZ0024099','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-19 05:41:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.38 pH\r\nmV                   23.8 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-10-24,00:33\r\n\r\n\r\n'),
(695,'2024-10-19','06:17:47','90.00000','105.00000','Piperacillin and Tazobactam for Injection','NA','TZ0024099','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','pH (Bulk)',1,'88629','Raghuram Reddy','2024-10-19 06:35:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.31 pH\r\nmV                   27.9 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-10-24,00:33\r\n\r\n\r\n'),
(696,'2024-10-19','06:32:15','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024099','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','3',1,'88629','Raghuram Reddy','2024-10-19 06:36:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.32 pH\r\nmV                   27.2 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             19-10-24,00:33\r\n\r\n\r\n'),
(697,'2024-10-19','07:20:32','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005470','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-10-19 08:32:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.63 pH\r\nmV                   304.9 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(698,'2024-10-19','07:23:15','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005470','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-10-19 08:32:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   165.0 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(699,'2024-10-19','07:26:13','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005470','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-10-19 08:33:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -8.7 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(700,'2024-10-19','07:28:45','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005470','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-10-19 08:33:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.99 pH\r\nmV                   -182.4 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(701,'2024-10-19','07:54:10','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005470','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-10-19 08:34:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.48 pH\r\nmV                   -328.0 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\nLog #                354\r\n\r\n\r\n'),
(702,'2024-10-19','09:11:12','90.00000','105.00000','Piperacillin And Tazobactam','Identification','2449201492','E4RM24005491','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,1,101,'pH','PT',1,'82735','M.D. Babar Qhadri','2024-10-19 09:27:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.46 pH\r\nmV                   84.3 mV\r\nTemperature          34.6 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\nLog #                355\r\n\r\n\r\n'),
(703,'2024-10-19','09:20:57','90.00000','105.00000','Piperacillin USP','Assay pH','2449102410','E4WS24000012','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','Assay pH',1,'82735','M.D. Babar Qhadri','2024-10-19 09:31:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.02 pH\r\nmV                   111.4 mV\r\nTemperature          35.7 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(704,'2024-10-19','09:27:12','90.00000','105.00000','Piperacillin USP','Assay PH','2449102410','E4WS24000012','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205048','P. Nagesh',2,1,101,'pH','Assay pH',1,'82735','M.D. Babar Qhadri','2024-10-19 10:26:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.00 pH\r\nmV                   112.0 mV\r\nTemperature          35.0 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(705,'2024-10-19','09:41:07','90.00000','105.00000','Ampicillin and Sulbactam','UOD','AS0324075','E4FP24000702','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','90017','G. Srishailam',2,1,101,'pH','AS',1,'205208','S. Siddhartha Kumar Reddy','2024-10-19 15:33:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.00 pH\r\nmV                   111.3 mV\r\nTemperature          32.6 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(706,'2024-10-19','09:56:29','90.00000','105.00000','pH Buffer','NA','NA','E4INS240005478','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','1.68 pH',1,'89384','M.V.V. Satish','2024-10-19 14:36:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   308.5 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\nLog #                195\r\n\r\n\r\n'),
(707,'2024-10-19','10:00:31','90.00000','105.00000','pH Buffer','NA','NA','E4INS240005478','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','4.01pH',1,'89384','M.V.V. Satish','2024-10-19 14:36:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.05 pH\r\nmV                   169.6 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\nLog #                196\r\n\r\n\r\n'),
(708,'2024-10-19','10:04:59','90.00000','105.00000','pH Buffer','NA','NA','E4INS240005478','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','7.00pH',1,'89384','M.V.V. Satish','2024-10-19 14:38:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.05 pH\r\nmV                   -5.8 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\n\r\n\r\n'),
(709,'2024-10-19','10:06:26','90.00000','105.00000','pH Buffer','NA','NA','E4INS240005478','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'89384','M.V.V. Satish','2024-10-19 14:39:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   9.99 pH\r\nmV                   -178.2 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\n\r\n\r\n'),
(710,'2024-10-19','10:07:54','90.00000','105.00000','pH Buffer','NA','NA','E4INS240005478','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','12.46 pH',1,'89384','M.V.V. Satish','2024-10-19 14:39:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.44 pH\r\nmV                   -321.7 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\n\r\n\r\n'),
(711,'2024-10-19','10:05:19','90.00000','105.00000','Ampicillin and Sulbactam for injection','pH','2449102395','E4RM24005530','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','Ampi&Sulbactam',1,'82735','M.D. Babar Qhadri','2024-10-19 10:27:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.00 pH\r\nmV                   -123.9 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\nLog #                359\r\n\r\n\r\n'),
(712,'2024-10-19','10:11:18','90.00000','105.00000','Ampicillin and Sulbactam for injection','pH','2449102396','E4RM24005531','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','Ampi&Sulbactam',1,'82735','M.D. Babar Qhadri','2024-10-19 11:06:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.91 pH\r\nmV                   -119.1 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(713,'2024-10-19','10:25:37','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0024099','E4IP24001265','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','pH (Bulk)',1,'82735','M.D. Babar Qhadri','2024-10-19 11:07:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.10 pH\r\nmV                   44.2 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(714,'2024-10-19','10:40:11','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424112','E4FP24000703','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-10-19 11:08:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.93 pH\r\nmV                   54.1 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(715,'2024-10-19','11:06:52','90.00000','105.00000','Piperacillin and Tazobactam for Injection','ASSAY BY HPLC','PT0424011A','E4STFP24000607','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205232','K. Narender Reddy',2,0,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-19 14:00:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   80.3 mV\r\nTemperature          32.7 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(716,'2024-10-19','11:58:47','90.00000','105.00000','Ampicillin and Sulbactam for injection','pH','2449102397','E4RM24005532','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','Ampi&Sulbactam',1,'82735','M.D. Babar Qhadri','2024-10-19 14:01:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.08 pH\r\nmV                   -129.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(717,'2024-10-19','12:03:12','90.00000','105.00000','Ampicillin and Sulbactam for injection','pH','2449102398','E4RM24005533','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','Ampi&Sulbactam',1,'82735','M.D. Babar Qhadri','2024-10-19 14:02:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.08 pH\r\nmV                   -128.6 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(718,'2024-10-19','13:53:28','90.00000','105.00000','Ampicillin and Sulbactam for injection','pH','2449102399','E4RM24005534','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','Ampi&Sulbactam',1,'82735','M.D. Babar Qhadri','2024-10-19 14:02:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.05 pH\r\nmV                   -127.5 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(719,'2024-10-19','14:08:28','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005623','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-19 15:35:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.48 pH\r\nmV                   79.9 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(720,'2024-10-19','14:11:50','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005624','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-19 15:36:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.47 pH\r\nmV                   80.9 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(721,'2024-10-19','14:14:34','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005625','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-19 15:37:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.44 pH\r\nmV                   82.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(722,'2024-10-19','14:18:00','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005626','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-19 15:38:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.40 pH\r\nmV                   84.6 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(723,'2024-10-19','14:24:28','90.00000','105.00000','Water For Injection','pH','WFI/SP34','E4RM24005627','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-19 15:39:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   78.1 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(724,'2024-10-19','14:27:00','90.00000','105.00000','Purified water','pH','PW/SP45','E4RM24005628','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-19 15:40:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.41 pH\r\nmV                   83.6 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(725,'2024-10-19','14:30:31','90.00000','105.00000','Purified water','pH','PW/SP09','E4RM24005629','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-19 16:11:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.46 pH\r\nmV                   81.2 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(726,'2024-10-19','14:48:49','90.00000','105.00000','Water For Injection','pH','PG0224004','E4RM24005631','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-19 16:12:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.41 pH\r\nmV                   84.3 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(727,'2024-10-19','15:16:38','90.00000','105.00000','Ampicillin and Sulbactam for injection','pH','2449102400','E4RM24005535','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','Ampi&Sulbactam',1,'205208','S. Siddhartha Kumar Reddy','2024-10-19 16:13:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.03 pH\r\nmV                   -126.6 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(728,'2024-10-19','15:20:30','90.00000','105.00000','Ampicillin and Sulbactam for injection','pH','2449102401','E4RM24005536','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','Ampi&Sulbactam',1,'205208','S. Siddhartha Kumar Reddy','2024-10-19 16:14:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.99 pH\r\nmV                   -123.7 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_121\r\nLast Cal             19-10-24,07:13\r\n\r\n\r\n'),
(729,'2024-10-19','15:37:42','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24017','E4SO24000700','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SCDBIH',1,'89384','M.V.V. Satish','2024-10-19 17:10:37','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.31 pH\r\nmV                   -21.1 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\n\r\n\r\n'),
(730,'2024-10-19','15:40:57','90.00000','105.00000','SCDM WITH 4%Tween80 &0.5 Soya lecthin','NA','RS011/24008','E4SO24000701','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','NSCDBIH',1,'89384','M.V.V. Satish','2024-10-19 17:11:10','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.35 pH\r\nmV                   -23.4 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\n\r\n\r\n'),
(731,'2024-10-19','15:42:51','90.00000','105.00000','SCDM with 1% Tween 80','NA','RS034/24007','E4SO24000702','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','NSSCDBSIH',1,'89384','M.V.V. Satish','2024-10-19 17:11:39','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.29 pH\r\nmV                   -20.1 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\n\r\n\r\n'),
(732,'2024-10-19','17:38:46','90.00000','105.00000','Soyabean casein digest agar','NA','RS022/24007','E4SO24000704','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDAIH',1,'89384','M.V.V. Satish','2024-10-21 09:19:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.48 pH\r\nmV                   -31.2 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\n\r\n\r\n'),
(733,'2024-10-19','17:43:07','90.00000','105.00000','Sabourarud dextrose agar','NA','RS021/24007','E4SO24000703','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SDAIH',1,'89384','M.V.V. Satish','2024-10-21 09:19:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.76 pH\r\nmV                   69.2 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\n\r\n\r\n'),
(734,'2024-10-19','17:45:43','90.00000','105.00000','R2AIH AGAR','NA','RS020/24005','E4SO24000705','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','R2AIH',1,'205143','Ch. Prasad','2024-10-25 09:47:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.25 pH\r\nmV                   -17.3 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\n\r\n\r\n'),
(735,'2024-10-19','20:46:25','90.00000','105.00000','SCDM with 1% Tween 80','NA','RS034/24007','E4SO24000702','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','NSSCDBSIH',1,'205143','Ch. Prasad','2024-10-25 09:47:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.32 pH\r\nmV                   -21.6 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\n\r\n\r\n'),
(736,'2024-10-19','20:49:05','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24017','E4SO24000700','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDBIH',1,'205143','Ch. Prasad','2024-10-25 09:47:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.37 pH\r\nmV                   -24.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\n\r\n\r\n'),
(737,'2024-10-19','20:51:30','90.00000','105.00000','SCDM WITH 4%Tween80 &0.5 Soya lecthin','NA','RS011/24008','E4SO24000701','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','NSCDBIH',1,'205143','Ch. Prasad','2024-10-25 09:48:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.42 pH\r\nmV                   -27.3 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\n\r\n\r\n'),
(738,'2024-10-19','20:55:17','90.00000','105.00000','R2AIH AGAR','NA','RS020/24005','E4SO24000705','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','R2AIH',1,'205143','Ch. Prasad','2024-10-25 09:48:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.17 pH\r\nmV                   -13.0 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\n\r\n\r\n'),
(739,'2024-10-19','20:57:29','90.00000','105.00000','Sabourarud dextrose agar','NA','RS021/24007','E4SO24000703','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SDAIH',1,'205143','Ch. Prasad','2024-10-25 09:48:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.72 pH\r\nmV                   71.8 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\n\r\n\r\n'),
(740,'2024-10-19','21:01:01','90.00000','105.00000','Soyabean casein digest agar','NA','RS022/24007','E4SO24000704','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDAIH',1,'205143','Ch. Prasad','2024-10-25 09:49:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.41 pH\r\nmV                   -26.9 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             19-10-24,09:49\r\n\r\n\r\n'),
(741,'2024-10-20','04:07:31','90.00000','105.00000','TAZOBACTAM','NA','TZ0024100','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','1',1,'84472','G. Ashok','2024-10-20 06:12:54','Reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.63 pH\r\nmV                   126.5 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             20-10-24,02:53\r\n\r\n\r\n'),
(742,'2024-10-20','08:41:55','90.00000','105.00000','PIPERACILLIN','NA','TZ0024100','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-20 14:24:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.95 pH\r\nmV                   49.9 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             20-10-24,02:53\r\n\r\n\r\n'),
(743,'2024-10-20','09:37:18','90.00000','105.00000','PIPERACILLIN','NA','TZ0024100','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-20 14:24:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.02 pH\r\nmV                   45.2 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             20-10-24,02:53\r\n\r\n\r\n'),
(744,'2024-10-20','10:27:39','90.00000','105.00000','PIPERACILLIN','NA','TZ0024100','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-20 14:24:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   29.4 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             20-10-24,02:53\r\n\r\n\r\n'),
(745,'2024-10-20','11:25:30','90.00000','105.00000','PIPERACILLIN','NA','TZ0024100','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-20 14:24:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.45 pH\r\nmV                   20.4 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             20-10-24,02:53\r\n\r\n\r\n'),
(746,'2024-10-20','13:46:03','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024100','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','3',1,'95681','T. Sivanarayana','2024-10-20 14:24:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   30.6 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             20-10-24,02:53\r\n\r\n\r\n'),
(747,'2024-10-20','13:58:02','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024100','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','3',1,'95681','T. Sivanarayana','2024-10-20 14:25:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.26 pH\r\nmV                   31.5 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             20-10-24,02:53\r\n\r\n\r\n'),
(748,'2024-10-21','07:04:18','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005499','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','1.68 pH',1,'90018','Ch. Satyam','2024-10-21 07:37:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   307.3 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(749,'2024-10-21','07:08:04','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005499','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','4.01pH',1,'90018','Ch. Satyam','2024-10-21 07:37:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.05 pH\r\nmV                   168.5 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(750,'2024-10-21','07:19:16','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005499','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','7.00pH',1,'90018','Ch. Satyam','2024-10-21 07:38:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -6.3 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(751,'2024-10-21','07:22:06','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005499','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','10.01 pH',1,'90018','Ch. Satyam','2024-10-21 07:38:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.03 pH\r\nmV                   -181.3 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(752,'2024-10-21','07:25:11','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005499','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','12.46 pH',1,'90018','Ch. Satyam','2024-10-21 07:38:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -325.5 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(753,'2024-10-21','08:20:42','90.00000','105.00000','Ampicillin and Sulbactam for injection','IDENTIFICATION BUFFER pH','2449101356','E4RM24005580','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','Ampi&Sulbactam',1,'90018','Ch. Satyam','2024-10-21 10:05:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.99 pH\r\nmV                   116.4 mV\r\nTemperature          31.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(754,'2024-10-21','09:43:05','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC','TZ0424097','E4FP24000701','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,1,101,'pH','Pipertazoinj',1,'90018','Ch. Satyam','2024-10-21 11:42:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   82.3 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\nLog #                383\r\n\r\n\r\n'),
(755,'2024-10-21','09:57:10','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005504','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-21 12:18:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.65 pH\r\nmV                   306.8 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             21-10-24,09:54\r\n\r\n\r\n'),
(756,'2024-10-21','09:59:18','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC','TZ0424097','E4FP24000701','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,1,101,'pH','Pipertazoinj',1,'90018','Ch. Satyam','2024-10-21 11:42:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   82.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(757,'2024-10-21','10:03:38','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005504','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-21 10:19:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.06 pH\r\nmV                   167.0 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             21-10-24,09:54\r\nLog #                213\r\n\r\n\r\n'),
(758,'2024-10-21','10:05:07','90.00000','105.00000','Piperacillin and Tazobactam for injection','RS BY HPLC','TZ0424025A','E4CS24000410','Diluent compatibility','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,1,101,'pH','Piper Tazo',1,'90018','Ch. Satyam','2024-10-21 11:41:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   82.4 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\nLog #                385\r\n\r\n\r\n'),
(759,'2024-10-21','10:09:30','90.00000','105.00000','pH Buffer','NA','pHverification','E4INS24005504','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-21 10:19:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -5.3 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             21-10-24,09:54\r\nLog #                214\r\n\r\n\r\n'),
(760,'2024-10-21','10:13:06','90.00000','105.00000','pH Buffer','NA','pH verfication','E4INS24005504','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-21 10:19:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   9.97 pH\r\nmV                   -177.5 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             21-10-24,09:54\r\n\r\n\r\n'),
(761,'2024-10-21','10:48:42','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0024100','E4IP24001269','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','pH (Bulk)',1,'90018','Ch. Satyam','2024-10-21 11:43:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   69.7 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(762,'2024-10-21','11:05:21','90.00000','105.00000','Penicillin G Benzathine','RS1','2449104473','E4RM24005658','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205189','P. Ravi Kumar',2,1,101,'pH','3.3 pH Buffer',1,'82735','M.D. Babar Qhadri','2024-10-21 15:28:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.31 pH\r\nmV                   215.3 mV\r\nTemperature          29.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(763,'2024-10-21','11:03:11','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005504','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-21 11:19:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.44 pH\r\nmV                   -322.4 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             21-10-24,09:54\r\nLog #                216\r\n\r\n\r\n'),
(764,'2024-10-21','11:16:08','90.00000','105.00000','Piperacillin and Tazobactam for Injection','ASSAY','PA0424001A','E4STFP24000622','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205232','K. Narender Reddy',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-21 15:29:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   84.4 mV\r\nTemperature          30.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(765,'2024-10-21','11:27:37','90.00000','105.00000','Ampicillin and Sulbactam for injection','IDENTIFICATION SST pH','2449101356','E4RM24005580','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','Ampi&Sulbactam',1,'82735','M.D. Babar Qhadri','2024-10-21 15:30:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.01 pH\r\nmV                   112.7 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(766,'2024-10-21','13:44:42','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005661','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-21 15:31:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.87 pH\r\nmV                   62.4 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(767,'2024-10-21','13:49:45','90.00000','105.00000','Peptone','NA','RS007/24038','E4SO24000707','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-30 10:06:19','Attach photocopy',''),
(768,'2024-10-21','13:48:25','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005662','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-21 15:31:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.84 pH\r\nmV                   63.8 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\nLog #                391\r\n\r\n\r\n'),
(769,'2024-10-21','14:02:16','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005663','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-10-21 15:32:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.81 pH\r\nmV                   66.3 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(770,'2024-10-21','14:04:14','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005664','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-10-21 15:50:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.78 pH\r\nmV                   67.2 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(771,'2024-10-21','14:06:29','90.00000','105.00000','Purified water','pH','PW/SP03','E4RM24005665','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-21 15:41:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.66 pH\r\nmV                   73.7 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(772,'2024-10-21','14:08:31','90.00000','105.00000','Pure steam','pH','PSC/SP09','E4RM24005666','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'82735','M.D. Babar Qhadri','2024-10-21 15:41:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   68.8 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(773,'2024-10-21','14:10:43','90.00000','105.00000','Purified water','pH','PW/SP29','E4RM24005667','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-21 15:42:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.65 pH\r\nmV                   74.5 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(774,'2024-10-21','14:13:00','90.00000','105.00000','Water For Injection','pH','WFI/SP28','E4RM24005668','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-10-21 15:43:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.73 pH\r\nmV                   68.4 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(775,'2024-10-21','14:37:17','90.00000','105.00000','Ampicillin and Sulbactam','RS BY HPLC MP pH','AS0324002A','MCL24000106','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,0,101,'pH','AS',1,'82735','M.D. Babar Qhadri','2024-10-22 09:23:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.04 pH\r\nmV                   111.2 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\nLog #                398\r\n\r\n\r\n'),
(776,'2024-10-21','15:05:51','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TE0423002A','E4STFP24000643','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-21 15:48:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.11 pH\r\nmV                   48.0 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(777,'2024-10-21','15:09:46','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TE0423002A','E4STFP24000644','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-21 15:48:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.94 pH\r\nmV                   58.4 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(778,'2024-10-21','15:14:45','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TE0423003A','E4STFP24000645','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-21 15:49:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.02 pH\r\nmV                   53.6 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(779,'2024-10-21','15:23:29','90.00000','105.00000','Ampicillin and sulbactam for Injection','ASSAY','AS0123001A','E4STFP24000625','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Ampisulbactam',1,'82735','M.D. Babar Qhadri','2024-10-22 09:25:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.03 pH\r\nmV                   114.1 mV\r\nTemperature          32.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(780,'2024-10-21','15:29:16','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TE0423003A','E4STFP24000646','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-21 15:49:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.80 pH\r\nmV                   66.4 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(781,'2024-10-21','15:32:30','90.00000','105.00000','Ampcillin for injection','pH','AP0224001A','E4STFP24000647','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','Ampi',1,'82735','M.D. Babar Qhadri','2024-10-21 15:49:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.09 pH\r\nmV                   -126.5 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(782,'2024-10-21','17:02:52','90.00000','105.00000','Ampicillin and Sulbactam','UOD','AS0324077','E4FP24000706','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,0,101,'pH','AS',1,'82735','M.D. Babar Qhadri','2024-10-22 09:27:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.01 pH\r\nmV                   115.7 mV\r\nTemperature          32.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_122\r\nLast Cal             21-10-24,06:59\r\n\r\n\r\n'),
(783,'2024-10-21','18:19:28','90.00000','105.00000','Nacl with newttralizers','NA','RS008/24012','E4SO24000710','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','NSALIH',1,'205143','Ch. Prasad','2024-10-30 10:06:40','Attach photocopy',''),
(784,'2024-10-21','19:25:12','90.00000','105.00000','TAZOBACTAM','NA','TZ0024101','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','1',1,'88629','Raghuram Reddy','2024-10-21 21:13:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.55 pH\r\nmV                   126.6 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        97.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             21-10-24,17:48\r\n\r\n\r\n'),
(785,'2024-10-21','21:31:00','90.00000','105.00000','Peptone','NA','RS007/24039','E4SO24000711','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-25 13:58:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.85 pH\r\nmV                   3.9 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             21-10-24,09:54\r\n\r\n\r\n'),
(786,'2024-10-21','21:33:31','90.00000','105.00000','Nacl with newttralizers','NA','RS008/24012','E4SO24000710','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','NSALIH',1,'205143','Ch. Prasad','2024-10-25 13:59:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.67 pH\r\nmV                   14.5 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             21-10-24,09:54\r\nLog #                222\r\n\r\n\r\n'),
(787,'2024-10-21','22:50:54','90.00000','105.00000','PIPERACILLIN','na','TZ0024101','na','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'84472','G. Ashok','2024-10-22 00:38:37','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.99 pH\r\nmV                   43.7 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        97.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             21-10-24,17:48\r\n\r\n\r\n'),
(788,'2024-10-22','00:50:28','90.00000','105.00000','PIPERACILLIN','NA','TZ0024101','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'84472','G. Ashok','2024-10-22 04:47:37','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.01 pH\r\nmV                   44.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        97.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             22-10-24,00:18\r\n\r\n\r\n'),
(789,'2024-10-22','01:29:17','90.00000','105.00000','PIPERACILLIN','NA','TZ0024101','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'84472','G. Ashok','2024-10-22 04:47:57','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.32 pH\r\nmV                   26.8 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        97.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             22-10-24,00:18\r\n\r\n\r\n'),
(790,'2024-10-22','02:15:13','90.00000','105.00000','PIPERACILLIN','na','TZ0024101','na','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'84472','G. Ashok','2024-10-22 04:48:15','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.41 pH\r\nmV                   21.6 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        97.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             22-10-24,00:18\r\n\r\n\r\n'),
(791,'2024-10-22','03:39:16','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','na','TZ0024101','na','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','3',1,'84472','G. Ashok','2024-10-22 04:48:53','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.36 pH\r\nmV                   24.8 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        97.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             22-10-24,00:18\r\n\r\n\r\n'),
(792,'2024-10-22','03:55:57','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','na','TZ0024101','na','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','3',1,'84472','G. Ashok','2024-10-22 04:49:06','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.35 pH\r\nmV                   25.0 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        97.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             22-10-24,00:18\r\n\r\n\r\n'),
(793,'2024-10-22','07:02:13','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005522','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','1.68 pH',1,'90778','B. Bhaskar','2024-10-22 07:24:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.64 pH\r\nmV                   308.5 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(794,'2024-10-22','07:04:49','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005522','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','4.01pH',1,'90778','B. Bhaskar','2024-10-22 07:24:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   169.9 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(795,'2024-10-22','07:07:08','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005522','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','7.00pH',1,'90778','B. Bhaskar','2024-10-22 07:25:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -4.9 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(796,'2024-10-22','07:09:44','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005522','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','10.01 pH',1,'90778','B. Bhaskar','2024-10-22 07:26:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.98 pH\r\nmV                   -177.4 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(797,'2024-10-22','07:12:20','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005522','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','12.46 pH',1,'90778','B. Bhaskar','2024-10-22 07:26:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -323.3 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(798,'2024-10-22','07:53:38','90.00000','105.00000','Aseptic Media Fill Validation','pH','DP24009','E4AMV24000162','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','pH  Test',1,'90778','B. Bhaskar','2024-10-22 08:55:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.32 pH\r\nmV                   -22.4 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(799,'2024-10-22','08:14:35','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0024101','E4IP24001275','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','Pipertazoinj',1,'90778','B. Bhaskar','2024-10-22 08:56:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.45 pH\r\nmV                   28.4 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(800,'2024-10-22','09:10:16','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005527','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.65 pH\r\nmV                   306.9 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             22-10-24,09:06\r\n\r\n\r\n'),
(801,'2024-10-22','09:12:54','90.00000','105.00000','pH Buffer','NA','pHverification','E4INS24005527','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.04 pH\r\nmV                   166.9 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             22-10-24,09:06\r\n\r\n\r\n'),
(802,'2024-10-22','09:14:52','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005527','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -7.8 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             22-10-24,09:06\r\nLog #                225\r\n\r\n\r\n'),
(803,'2024-10-22','09:19:59','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005527','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   9.97 pH\r\nmV                   -178.5 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             22-10-24,09:06\r\nLog #                226\r\n\r\n\r\n'),
(804,'2024-10-22','10:22:54','90.00000','105.00000','Penicillin G Benzathine','disso','PG0124002A','E4STFP24000563','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Penicillin G',1,'86253','D. Jayapal Reddy','2024-10-22 11:52:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.25 pH\r\nmV                   219.0 mV\r\nTemperature          29.5 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\nLog #                413\r\n\r\n\r\n'),
(805,'2024-10-22','10:28:49','90.00000','105.00000','Penicillin G Benzathine','disso','PG0124002A','E4STFP24000563','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Penicillin G',1,'86253','D. Jayapal Reddy','2024-10-22 11:52:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.48 pH\r\nmV                   -32.4 mV\r\nTemperature          31.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(806,'2024-10-22','11:33:00','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TA0323001B','E4STFP24000652','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-10-22 11:51:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.82 pH\r\nmV                   65.4 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(807,'2024-10-22','11:36:56','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TA0323001B','E4STFP24000653','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-10-22 11:52:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.66 pH\r\nmV                   74.6 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(808,'2024-10-22','11:40:41','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TA0323003A','E4STFP24000656','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-10-22 11:52:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   72.3 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(809,'2024-10-22','11:59:10','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005527','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.46 pH\r\nmV                   -306.0 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        96.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             22-10-24,11:58\r\nLog #                227\r\n\r\n\r\n'),
(810,'2024-10-22','12:10:11','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005696','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-10-22 13:13:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   71.3 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(811,'2024-10-22','12:13:20','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005697','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-10-22 13:14:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.59 pH\r\nmV                   78.4 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(812,'2024-10-22','12:15:37','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005698','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90778','B. Bhaskar','2024-10-22 13:14:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.65 pH\r\nmV                   75.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(813,'2024-10-22','12:19:29','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005699','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90778','B. Bhaskar','2024-10-22 13:14:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.67 pH\r\nmV                   74.1 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(814,'2024-10-22','12:21:54','90.00000','105.00000','Water For Injection','pH','WFI/SP13','E4RM24005700','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90778','B. Bhaskar','2024-10-22 13:15:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.59 pH\r\nmV                   78.6 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(815,'2024-10-22','12:24:22','90.00000','105.00000','Purified water','pH','PW/SP15','E4RM24005701','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-10-22 13:15:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   69.6 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(816,'2024-10-22','12:27:44','90.00000','105.00000','Water For Injection','pH','WFI/SP17','E4RM24005702','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90778','B. Bhaskar','2024-10-22 13:15:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.57 pH\r\nmV                   79.6 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(817,'2024-10-22','12:58:02','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS','PT0424111','E4FP24000696','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','Pipertazoinj',1,'90778','B. Bhaskar','2024-10-22 13:16:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.81 pH\r\nmV                   184.5 mV\r\nTemperature          27.8 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(818,'2024-10-22','15:01:32','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0322001A','E4STFP24000648','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-22 16:48:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.49 pH\r\nmV                   85.1 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(819,'2024-10-22','15:06:36','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0322002A','E4STFP24000649','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-22 16:48:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   72.1 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(820,'2024-10-22','15:14:18','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0222001A','E4STFP24000651','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-22 16:50:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.33 pH\r\nmV                   94.0 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\n\r\n\r\n'),
(821,'2024-10-22','15:19:26','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TA0323002B','E4STFP24000654','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-22 16:51:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   69.4 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_123\r\nLast Cal             22-10-24,06:58\r\nLog #                429\r\n\r\n\r\n'),
(822,'2024-10-22','15:33:19','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005527','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-22 15:56:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.64 pH\r\nmV                   311.4 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             22-10-24,15:27\r\n\r\n\r\n'),
(823,'2024-10-22','15:34:37','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005527','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-22 15:56:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   171.9 mV\r\nTemperature          26.7 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             22-10-24,15:27\r\nLog #                229\r\n\r\n\r\n'),
(824,'2024-10-22','15:41:12','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005527','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-22 15:56:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -4.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             22-10-24,15:27\r\n\r\n\r\n'),
(825,'2024-10-22','15:43:29','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005527','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-22 15:56:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -177.6 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             22-10-24,15:27\r\n\r\n\r\n'),
(826,'2024-10-22','15:46:04','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005527','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-22 15:56:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.43 pH\r\nmV                   -319.2 mV\r\nTemperature          26.5 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             22-10-24,15:27\r\n\r\n\r\n'),
(827,'2024-10-23','04:02:30','90.00000','105.00000','TAZOBACTAM','NA','TZ0024102','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','1',1,'84472','G. Ashok','2024-10-23 05:43:17','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.57 pH\r\nmV                   127.5 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        96.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             23-10-24,00:51\r\n\r\n\r\n'),
(828,'2024-10-23','04:52:25','90.00000','105.00000','TAZOBACTAM','NA','TZ0024102','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','1',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.56 pH\r\nmV                   127.3 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        96.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             23-10-24,00:51\r\n\r\n\r\n'),
(829,'2024-10-23','07:27:42','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005537','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-10-23 08:18:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.65 pH\r\nmV                   308.9 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(830,'2024-10-23','07:43:56','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005537','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-10-23 08:18:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.05 pH\r\nmV                   170.3 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(831,'2024-10-23','07:49:37','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005537','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-10-23 08:18:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -1.2 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(832,'2024-10-23','08:00:29','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005537','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-10-23 08:18:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.99 pH\r\nmV                   -174.0 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(833,'2024-10-23','08:07:22','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005537','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-10-23 08:18:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.48 pH\r\nmV                   -318.2 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(834,'2024-10-23','08:21:02','90.00000','105.00000','PIPERACILLIN','NA','TZ0024102','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-23 08:31:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.94 pH\r\nmV                   48.7 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        96.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             23-10-24,00:51\r\n\r\n\r\n'),
(835,'2024-10-23','09:09:28','90.00000','105.00000','PIPERACILLIN','NA','TZ0024102','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-23 10:02:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.97 pH\r\nmV                   47.1 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        96.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             23-10-24,00:51\r\n\r\n\r\n'),
(836,'2024-10-23','09:29:27','90.00000','105.00000','Sensor Calibration','NA','NA','E4INS24005545','Temperature Sensor','QC-PHMTZ-007','CX1-15733','1','NULL','87624','N. Nani babu',2,1,101,'pH','Purified Water',1,'82735','M.D. Babar Qhadri','2024-10-23 09:47:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.60 pH\r\nmV                   80.0 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(837,'2024-10-23','09:44:26','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005560','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-23 10:09:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.65 pH\r\nmV                   312.0 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             23-10-24,09:41\r\n\r\n\r\n'),
(838,'2024-10-23','09:45:36','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005560','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-23 10:10:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   173.7 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             23-10-24,09:41\r\nLog #                235\r\n\r\n\r\n'),
(839,'2024-10-23','09:48:40','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005560','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-23 10:10:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -2.0 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             23-10-24,09:41\r\nLog #                236\r\n\r\n\r\n'),
(840,'2024-10-23','09:51:44','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005560','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-23 10:10:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   9.97 pH\r\nmV                   -174.6 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             23-10-24,09:41\r\n\r\n\r\n'),
(841,'2024-10-23','09:52:41','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005560','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-23 10:11:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.45 pH\r\nmV                   -319.9 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             23-10-24,09:41\r\n\r\n\r\n'),
(842,'2024-10-23','09:54:25','90.00000','105.00000','PIPERACILLIN','NA','TZ0024102','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-23 10:02:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.25 pH\r\nmV                   31.5 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        96.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             23-10-24,00:51\r\n\r\n\r\n'),
(843,'2024-10-23','10:10:33','90.00000','105.00000','Ampicillin and Sulbactam','UOD','AS0324079','E4FP24000709','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','90017','G. Srishailam',2,1,101,'pH','AS',1,'82735','M.D. Babar Qhadri','2024-10-23 11:40:19','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.01 pH\r\nmV                   116.6 mV\r\nTemperature          31.2 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(844,'2024-10-23','10:27:50','90.00000','105.00000','Media Fill','pH','DP24009','E4AMV24000167','ASEPTIC MEDIA FILL','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Media fill pH',1,'82735','M.D. Babar Qhadri','2024-10-23 11:42:02','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.32 pH\r\nmV                   -19.7 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(845,'2024-10-23','10:36:12','90.00000','105.00000','Ampicillin and Sulbactam','RS BY HPLC MP A','AS0324077','E4FP24000706','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,1,101,'pH','AS',1,'205208','S. Siddhartha Kumar Reddy','2024-10-23 11:09:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.03 pH\r\nmV                   114.8 mV\r\nTemperature          29.5 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\nLog #                438\r\n\r\n\r\n'),
(846,'2024-10-23','10:43:44','90.00000','105.00000','PIPERACILLIN','NA','TZ0024102','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'88515','Prashanth B','2024-10-23 13:19:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.51 pH\r\nmV                   16.3 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        96.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             23-10-24,00:51\r\n\r\n\r\n'),
(847,'2024-10-23','11:01:25','90.00000','105.00000','Glass Vials for pH','pH(vial-1)','NA',' E4EQ24000379','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-10-23 11:43:13','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.90 pH\r\nmV                   235.5 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(848,'2024-10-23','11:03:34','90.00000','105.00000','Glass Vials for pH','pH(vial-2)','NA','E4EQ24000379','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'82735','M.D. Babar Qhadri','2024-10-23 11:43:44','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.72 pH\r\nmV                   245.9 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(849,'2024-10-23','11:39:28','90.00000','105.00000','Piperacillin And Tazobactam','IDENTIFICATION','2449200877','E4RM24005630','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205232','K. Narender Reddy',2,1,101,'pH','PT',1,'82735','M.D. Babar Qhadri','2024-10-23 11:45:37','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.47 pH\r\nmV                   90.3 mV\r\nTemperature          33.7 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(850,'2024-10-23','11:54:26','90.00000','105.00000','Ampicillin and Sulbactam for Injection','Buffer pH','AS0324080','E4CV24000296','Cleaning Validation','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,0,101,'pH','pH Buffer',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.01 pH\r\nmV                   116.8 mV\r\nTemperature          32.1 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(851,'2024-10-23','12:07:30','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324077','E4FP24000706','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-10-23 16:45:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.14 pH\r\nmV                   -124.4 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(852,'2024-10-23','12:11:25','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324078','E4FP24000707','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-10-23 16:46:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.11 pH\r\nmV                   -122.7 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(853,'2024-10-23','12:25:44','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024102','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','3',1,'88515','Prashanth B','2024-10-23 13:19:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.29 pH\r\nmV                   29.1 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        96.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             23-10-24,00:51\r\n\r\n\r\n'),
(854,'2024-10-23','12:25:52','90.00000','105.00000','Tryptone soya agar TSA with 0.05%soya lecithin&0.5% tween 80 and beta lactamase 55mm plates(RODAC)','NA','SPU408','E4SO24000717','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','RUTSA',1,'205143','Ch. Prasad','2024-10-25 09:59:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -2.3 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             23-10-24,09:41\r\nLog #                239\r\n\r\n\r\n'),
(855,'2024-10-23','12:39:45','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024102','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','3',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.29 pH\r\nmV                   29.6 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        96.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             23-10-24,00:51\r\n\r\n\r\n'),
(856,'2024-10-23','12:42:02','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024102','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','3',1,'88515','Prashanth B','2024-10-23 13:21:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.27 pH\r\nmV                   30.4 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        96.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             23-10-24,00:51\r\n\r\n\r\n'),
(857,'2024-10-23','12:48:51','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324079','E4FP24000709','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-10-23 16:46:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.18 pH\r\nmV                   -126.6 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\nLog #                445\r\n\r\n\r\n'),
(858,'2024-10-23','12:56:25','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324080','E4FP24000712','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-10-23 16:44:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.15 pH\r\nmV                   -125.2 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(859,'2024-10-23','13:48:13','90.00000','105.00000','FLUID-A','NA','RS012/24008','E4SO24000714','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','Fluid-A',1,'205143','Ch. Prasad','2024-10-25 09:59:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -2.4 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             23-10-24,09:41\r\n\r\n\r\n'),
(860,'2024-10-23','13:50:28','90.00000','105.00000','FLUID-A','NA','RS010/24009','E4SO24000722','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','Fluid-A',1,'205143','Ch. Prasad','2024-10-25 10:00:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -2.5 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             23-10-24,09:41\r\n\r\n\r\n'),
(861,'2024-10-23','14:29:30','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005708','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-23 15:02:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.80 pH\r\nmV                   68.4 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(862,'2024-10-23','14:31:40','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005709','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-23 15:03:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   71.3 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(863,'2024-10-23','14:34:11','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005710','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-10-23 15:03:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.76 pH\r\nmV                   70.4 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(864,'2024-10-23','14:36:18','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005711','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-10-23 15:04:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   73.4 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(865,'2024-10-23','14:38:35','90.00000','105.00000','Purified water','pH','PW/SP05','E4RM24005712','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-23 15:05:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   70.9 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(866,'2024-10-23','14:40:23','90.00000','105.00000','Purified water','pH','PW/SP18','E4RM24005713','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-23 15:05:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.76 pH\r\nmV                   70.5 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(867,'2024-10-23','14:42:02','90.00000','105.00000','Water For Injection','pH','WFI/SP19','E4RM24005714','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-10-23 15:05:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.78 pH\r\nmV                   69.4 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(868,'2024-10-23','14:55:59','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TA0323002B','E4STFP24000655','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-10-23 15:07:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   73.9 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(869,'2024-10-23','15:02:37','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TA0223001A','E4STFP24000657','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-10-23 15:18:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.68 pH\r\nmV                   75.2 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(870,'2024-10-23','15:06:35','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TA0223001A','E4STFP24000658','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-10-23 15:18:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.63 pH\r\nmV                   77.9 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(871,'2024-10-23','15:21:29','90.00000','105.00000','Ampicillin and Sulbactam','UOD','AS0324080','E4FP24000712','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-10-23 16:19:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.03 pH\r\nmV                   115.9 mV\r\nTemperature          32.7 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(872,'2024-10-23','15:35:04','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0424098','E4FP24000705','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-10-23 16:11:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.09 pH\r\nmV                   52.0 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(873,'2024-10-23','15:40:15','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0424099','E4FP24000711','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-10-23 16:11:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.06 pH\r\nmV                   53.4 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(874,'2024-10-23','15:45:05','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424113','E4FP24000708','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-10-23 16:12:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.89 pH\r\nmV                   63.6 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(875,'2024-10-23','15:48:47','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424114','E4FP24000710','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-10-23 16:12:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.91 pH\r\nmV                   61.9 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(876,'2024-10-23','16:25:36','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','pH','TZ0024102','E4IP24001278','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','3',1,'86253','D. Jayapal Reddy','2024-10-23 16:43:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.12 pH\r\nmV                   49.6 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        97.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_124\r\nLast Cal             23-10-24,07:16\r\n\r\n\r\n'),
(877,'2024-10-23','17:06:02','90.00000','105.00000','FLUID-A','NA','RS012/24008','E4SO24000714','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','Fluid-A',1,'205143','Ch. Prasad','2024-10-25 10:00:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.99 pH\r\nmV                   -0.1 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             23-10-24,09:41\r\nLog #                242\r\n\r\n\r\n'),
(878,'2024-10-23','17:10:17','90.00000','105.00000','Fluid-D','NA','RS010/24009','E4SO24000722','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','Fluid-D',1,'205143','Ch. Prasad','2024-10-25 10:00:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.98 pH\r\nmV                   0.5 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             23-10-24,09:41\r\nLog #                243\r\n\r\n\r\n'),
(879,'2024-10-23','22:39:39','90.00000','105.00000','Penicillin G','NA','PG0124001A','E4STFP24000551','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','92574','Abhinav.G',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-25 10:05:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.52 pH\r\nmV                   26.9 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             23-10-24,09:41\r\n\r\n\r\n'),
(880,'2024-10-23','22:41:48','90.00000','105.00000','Penicillin G','NA','PG0124002A','E4STFP24000558','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','92574','Abhinav.G',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-25 10:05:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.55 pH\r\nmV                   25.2 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             23-10-24,09:41\r\n\r\n\r\n'),
(881,'2024-10-23','22:46:08','90.00000','105.00000','Penicillin G','NA','PG0124003A','E4STFP24000566','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','92574','Abhinav.G',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-25 10:05:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.56 pH\r\nmV                   24.7 mV\r\nTemperature          22.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             23-10-24,09:41\r\n\r\n\r\n'),
(882,'2024-10-24','07:53:35','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005566','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-10-24 08:45:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.70 pH\r\nmV                   310.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(883,'2024-10-24','08:01:01','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005566','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-10-24 08:45:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.97 pH\r\nmV                   177.7 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(884,'2024-10-24','08:06:58','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005566','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-10-24 08:45:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -0.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(885,'2024-10-24','08:22:51','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005566','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-10-24 08:45:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.96 pH\r\nmV                   -173.2 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\nLog #                468\r\n\r\n\r\n'),
(886,'2024-10-24','08:35:03','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005566','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-10-24 08:46:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.41 pH\r\nmV                   -317.2 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\nLog #                469\r\n\r\n\r\n'),
(887,'2024-10-24','08:54:42','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005575','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-24 10:27:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.67 pH\r\nmV                   312.1 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             24-10-24,08:52\r\n\r\n\r\n'),
(888,'2024-10-24','08:56:37','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005575','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-24 10:28:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   173.1 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             24-10-24,08:52\r\n\r\n\r\n'),
(889,'2024-10-24','08:57:55','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005575','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-24 10:28:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -2.4 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             24-10-24,08:52\r\n\r\n\r\n'),
(890,'2024-10-24','08:59:20','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005575','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-24 10:28:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -179.3 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             24-10-24,08:52\r\n\r\n\r\n'),
(891,'2024-10-24','09:00:42','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005575','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-24 10:28:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.46 pH\r\nmV                   -322.8 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             24-10-24,08:52\r\n\r\n\r\n'),
(892,'2024-10-24','09:10:29','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC','TZ0424099','E4FP24000711','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-10-24 10:09:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   85.3 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(893,'2024-10-24','09:37:36','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC','TZ0424099','E4FP24000711','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-10-24 10:09:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   85.6 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(894,'2024-10-24','10:11:04','90.00000','105.00000','Ampcillin for injection','pH','AP0124001A','E4STFP24000664','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Ampi',1,'82735','M.D. Babar Qhadri','2024-10-24 11:35:51','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.19 pH\r\nmV                   -127.7 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(895,'2024-10-24','10:18:44','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0424001A','E4STFP24000665','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-24 11:36:30','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.29 pH\r\nmV                   41.6 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(896,'2024-10-24','10:32:32','90.00000','105.00000','Piperacillin and Tazobactam For injection','UOD','PT0424113','E4FP24000708','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','90017','G. Srishailam',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-10-24 11:40:24','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   89.9 mV\r\nTemperature          36.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(897,'2024-10-24','10:41:44','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0424002A','E4STFP24000666','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-24 11:37:04','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.87 pH\r\nmV                   66.7 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(898,'2024-10-24','10:47:54','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0424002A','E4STFP24000667','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-24 11:37:24','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.87 pH\r\nmV                   65.9 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(899,'2024-10-24','10:53:45','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0424004A','E4STFP24000668','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-24 11:37:55','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   73.7 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\nLog #                477\r\n\r\n\r\n'),
(900,'2024-10-24','11:05:55','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0424004A','E4STFP24000669','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-24 11:39:22','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.65 pH\r\nmV                   79.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(901,'2024-10-24','11:43:22','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS-II MP_A','TZ0424097','E4FP24000701','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,1,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 09:14:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   85.8 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(902,'2024-10-24','11:49:32','90.00000','105.00000','TAZOBACTAM','NA','TZ0024103','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','1',1,'88515','Prashanth B','2024-10-24 12:00:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.53 pH\r\nmV                   128.4 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        97.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             24-10-24,10:53\r\n\r\n\r\n'),
(903,'2024-10-24','12:08:58','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS_II MP_B','TZ0424097','E4FP24000701','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,1,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 09:31:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   85.6 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\nLog #                480\r\n\r\n\r\n'),
(904,'2024-10-24','12:18:35','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0322003A','E4STFP24000659','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-24 14:03:34','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.46 pH\r\nmV                   90.9 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(905,'2024-10-24','12:23:11','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0424001A','E4STFP24000661','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-24 14:03:57','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.94 pH\r\nmV                   62.5 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(906,'2024-10-24','12:26:42','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0424001A','E4STFP24000662','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-24 14:04:19','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.92 pH\r\nmV                   63.6 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(907,'2024-10-24','12:33:14','90.00000','105.00000','Penicillin G Benzathine (LECITHIN COATED)-Sterile','pH','2449104473','E4RM24005658','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','Peni G Benz',1,'82735','M.D. Babar Qhadri','2024-10-24 14:07:45','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.86 pH\r\nmV                   124.6 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\nLog #                484\r\n\r\n\r\n'),
(908,'2024-10-24','12:42:19','90.00000','105.00000','Soyabean casein digest broth for sterilty','NA','RS033/24009','E4SO24000728','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SCDBSIH',1,'205143','Ch. Prasad','2024-10-25 10:01:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.32 pH\r\nmV                   -20.9 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             24-10-24,08:52\r\n\r\n\r\n'),
(909,'2024-10-24','12:49:00','90.00000','105.00000','Peptone','NA','RS007/24040','E4SO24000730','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-25 10:02:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -2.7 mV\r\nTemperature          26.5 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             24-10-24,08:52\r\nLog #                253\r\n\r\n\r\n'),
(910,'2024-10-24','12:51:29','90.00000','105.00000','Penicillin G Benzathine (LECITHIN COATED)-Sterile','pH','2449105080','E4RM24005659','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','Peni G Benz',1,'82735','M.D. Babar Qhadri','2024-10-24 14:08:29','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.45 pH\r\nmV                   149.5 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\nLog #                485\r\n\r\n\r\n'),
(911,'2024-10-24','12:57:52','90.00000','105.00000','Penicillin G Benzathine (LECITHIN COATED)-Sterile','pH','2449105082','E4RM24005660','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','Peni G Benz',1,'82735','M.D. Babar Qhadri','2024-10-24 14:09:03','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.04 pH\r\nmV                   114.0 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(912,'2024-10-24','13:06:53','90.00000','105.00000','Penicillin G Benzathine (LECITHIN COATED)-Sterile','pH','2449105083','E4RM24005670','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','Peni G Benz',1,'90018','Ch. Satyam','2024-10-24 14:48:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.39 pH\r\nmV                   91.0 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(913,'2024-10-24','13:48:42','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005735','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-24 14:10:55','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   76.9 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(914,'2024-10-24','13:51:05','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005736','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-24 14:11:16','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.82 pH\r\nmV                   69.4 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(915,'2024-10-24','13:53:55','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005737','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-10-24 14:12:44','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.84 pH\r\nmV                   67.9 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(916,'2024-10-24','13:56:25','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005738','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-10-24 14:13:01','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.82 pH\r\nmV                   69.0 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(917,'2024-10-24','13:58:23','90.00000','105.00000','Water For Injection','pH','WFI/SP33','E4RM24005739','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-10-24 14:13:19','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.84 pH\r\nmV                   67.9 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(918,'2024-10-24','14:00:31','90.00000','105.00000','Purified water','pH','PW/SP41','E4RM24005740','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-24 14:14:11','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.85 pH\r\nmV                   67.3 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(919,'2024-10-24','14:02:23','90.00000','105.00000','Purified water','pH','PW/SP07','E4RM24005741','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-24 14:14:26','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.91 pH\r\nmV                   63.7 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(920,'2024-10-24','14:07:37','90.00000','105.00000','Penicillin G Benzathine (LECITHIN COATED)-Sterile','pH','2449105084','E4RM24005671','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Peni G Benz',1,'82735','M.D. Babar Qhadri','2024-10-24 14:15:11','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.00 pH\r\nmV                   117.3 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(921,'2024-10-24','14:13:04','90.00000','105.00000','Penicillin G Benzathine (LECITHIN COATED)-Sterile','pH','2449105085','E4RM24005672','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Peni G Benz',1,'90018','Ch. Satyam','2024-10-24 14:51:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.97 pH\r\nmV                   118.9 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(922,'2024-10-24','14:21:40','90.00000','105.00000','Fluid thioglycollate meddiuam','NA','RS026/24010','E4SO24000729','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','FTGMIH',1,'205143','Ch. Prasad','2024-10-25 10:02:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.20 pH\r\nmV                   -14.4 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             24-10-24,08:52\r\n\r\n\r\n'),
(923,'2024-10-24','14:25:06','90.00000','105.00000','PIPERACILLIN','NA','TZ0024103','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-24 22:10:07','Reveiwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.88 pH\r\nmV                   49.6 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        97.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             24-10-24,10:53\r\n\r\n\r\n'),
(924,'2024-10-24','14:19:21','90.00000','105.00000','Penicillin G Benzathine (LECITHIN COATED)-Sterile','pH','2449105086','E4RM24005673','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','Peni G Benz',1,'90018','Ch. Satyam','2024-10-24 14:48:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.83 pH\r\nmV                   126.8 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\nLog #                497\r\n\r\n\r\n'),
(925,'2024-10-24','14:32:56','90.00000','105.00000','Penicillin G Benzathine (LECITHIN COATED)-Sterile','pH','2449105087','E4RM24005674','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','Peni G Benz',1,'90018','Ch. Satyam','2024-10-24 14:48:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.93 pH\r\nmV                   121.2 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(926,'2024-10-24','15:04:36','90.00000','105.00000','PIPERACILLIN','NA','TZ0024103','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-24 22:10:35','Reveiwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.92 pH\r\nmV                   47.3 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        97.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             24-10-24,10:53\r\n\r\n\r\n'),
(927,'2024-10-24','15:02:25','90.00000','105.00000','Rubber Stopper','pH 3.5 acetate buffer','0041850250','E4PM24000806','Packing Material','QC-PHMTZ-007','CX1-15733','1','NULL','205196','P. Mounika',2,1,101,'pH','3.5 Acetate Buffer',1,'90018','Ch. Satyam','2024-10-24 16:30:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.50 pH\r\nmV                   206.3 mV\r\nTemperature          26.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\nLog #                499\r\n\r\n\r\n'),
(928,'2024-10-24','15:25:54','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PC0424001A','E4STFP24000660','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'90018','Ch. Satyam','2024-10-24 16:45:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.25 pH\r\nmV                   44.2 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(929,'2024-10-24','15:36:19','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PT0424074A','E4STFP24000670','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'90018','Ch. Satyam','2024-10-24 16:45:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.39 pH\r\nmV                   94.7 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(930,'2024-10-24','15:45:23','90.00000','105.00000','Tryptone soya agar TSA with 0.05%soya lecithin&0.5% tween 80 and beta lactamase 55mm plates(RODAC)','NA','SPU408','E4SO24000733','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','205207','P. Hanmanthu',2,0,103,'pH','RUTSA',1,'205143','Ch. Prasad','2024-10-25 10:02:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.22 pH\r\nmV                   -14.8 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             24-10-24,08:52\r\nLog #                255\r\n\r\n\r\n'),
(931,'2024-10-24','15:48:03','90.00000','105.00000','PIPERACILLIN','NA','TZ0024103','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-24 22:10:54','Reveiwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.27 pH\r\nmV                   26.8 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        97.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             24-10-24,10:53\r\nLog #                200\r\n\r\n\r\n'),
(932,'2024-10-24','15:51:17','90.00000','105.00000','Soyabean casein digest agar (SCDA)with beta-lactamase','NA','MPT0569','E4SO24000734','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','205207','P. Hanmanthu',2,0,103,'pH','RUSCDA',1,'205143','Ch. Prasad','2024-10-25 10:06:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.17 pH\r\nmV                   -12.1 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             24-10-24,08:52\r\nLog #                256\r\n\r\n\r\n'),
(933,'2024-10-24','15:46:01','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0424100','E4FP24000713','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Pipertazoinj',1,'90018','Ch. Satyam','2024-10-24 16:45:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.73 pH\r\nmV                   74.8 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\nLog #                502\r\n\r\n\r\n'),
(934,'2024-10-24','16:42:30','90.00000','105.00000','PIPERACILLIN','NA','TZ0024103','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-24 22:11:10','Reveiwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.48 pH\r\nmV                   14.8 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        97.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             24-10-24,10:53\r\n\r\n\r\n'),
(935,'2024-10-24','17:15:11','90.00000','105.00000','AMPISULBACTAM','Assay pH','2349100832','E4RM24005746','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','Assay buffer pH',1,'90018','Ch. Satyam','2024-10-25 09:50:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.02 pH\r\nmV                   119.5 mV\r\nTemperature          33.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_125\r\nLast Cal             24-10-24,07:45\r\n\r\n\r\n'),
(936,'2024-10-24','18:23:43','90.00000','105.00000','Soyabean casein digest broth for sterilty','NA','RS033/24009','E4SO24000728','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','SCDBSIH',1,'205143','Ch. Prasad','2024-10-25 10:07:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.38 pH\r\nmV                   -24.5 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             24-10-24,08:52\r\n\r\n\r\n'),
(937,'2024-10-24','18:27:51','90.00000','105.00000','Fluid thioglycollate meddiuam','NA','RS026/24010','E4SO24000729','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','FTGMIH',1,'205143','Ch. Prasad','2024-10-25 10:07:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.17 pH\r\nmV                   -12.1 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             24-10-24,08:52\r\n\r\n\r\n'),
(938,'2024-10-24','18:30:59','90.00000','105.00000','Peptone','NA','RS007/24040','E4SO24000730','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-25 10:08:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.60 pH\r\nmV                   21.5 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             24-10-24,08:52\r\n\r\n\r\n'),
(939,'2024-10-24','18:53:30','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024103','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','3',1,'88629','Raghuram Reddy','2024-10-24 22:18:54','Reveiwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.30 pH\r\nmV                   25.6 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        97.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             24-10-24,10:53\r\n\r\n\r\n'),
(940,'2024-10-24','19:08:06','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024103','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','3',1,'88629','Raghuram Reddy','2024-10-24 22:11:32','Reveiwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.32 pH\r\nmV                   24.2 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        97.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             24-10-24,10:53\r\n\r\n\r\n'),
(941,'2024-10-25','07:00:44','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005586','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','1.68 pH',1,'90018','Ch. Satyam','2024-10-25 07:56:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   308.7 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(942,'2024-10-25','07:04:34','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005586','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','4.01pH',1,'90018','Ch. Satyam','2024-10-25 07:56:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.05 pH\r\nmV                   170.8 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\nLog #                506\r\n\r\n\r\n'),
(943,'2024-10-25','07:11:04','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005586','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','7.00pH',1,'90018','Ch. Satyam','2024-10-25 07:56:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -2.5 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(944,'2024-10-25','07:23:46','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005586','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','10.01 pH',1,'90018','Ch. Satyam','2024-10-25 07:56:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.03 pH\r\nmV                   -178.7 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\nLog #                508\r\n\r\n\r\n'),
(945,'2024-10-25','07:33:55','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005586','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','12.46 pH',1,'90018','Ch. Satyam','2024-10-25 07:56:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -320.5 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(946,'2024-10-25','08:15:32','90.00000','105.00000','Piperacillin and Tazobactam for Injection','Assay','TA0323001B','E4STFP24000652','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,1,101,'pH','PIPERTAZO',1,'90018','Ch. Satyam','2024-10-25 11:42:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   87.2 mV\r\nTemperature          31.7 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(947,'2024-10-25','08:28:01','90.00000','105.00000','Ampicillin and Sulbactam for injection','IDENTIFICATION BUFFER pH','2449101496','E4RM24005669','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','Ampi&Sulbactam',1,'90018','Ch. Satyam','2024-10-25 11:41:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.00 pH\r\nmV                   118.6 mV\r\nTemperature          32.0 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(948,'2024-10-25','08:50:11','90.00000','105.00000','Piperacillin and Tazobactam For injection','Mobile phase A','TZ0424099','E4FP24000711','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,1,101,'pH','Pipertazoinj',1,'90018','Ch. Satyam','2024-10-25 09:06:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.57 pH\r\nmV                   82.4 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(949,'2024-10-25','09:51:05','90.00000','105.00000','pH Buffer',' NA','NA','E4INS24005602','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-25 10:09:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   312.2 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             25-10-24,09:46\r\n\r\n\r\n'),
(950,'2024-10-25','09:52:57','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005602','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-25 10:10:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   173.1 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             25-10-24,09:46\r\n\r\n\r\n'),
(951,'2024-10-25','09:55:03','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005602','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-25 10:10:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -3.2 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             25-10-24,09:46\r\n\r\n\r\n'),
(952,'2024-10-25','09:56:35','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005602','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-25 10:10:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.02 pH\r\nmV                   -179.2 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             25-10-24,09:46\r\n\r\n\r\n'),
(953,'2024-10-25','09:59:25','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005602','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-25 10:10:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.44 pH\r\nmV                   -321.4 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             25-10-24,09:46\r\n\r\n\r\n'),
(954,'2024-10-25','09:58:09','90.00000','105.00000','Piperacillin and Tazobactam For injection','ASSAY','TZ0424098','E4FP24000705','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','90017','G. Srishailam',2,1,101,'pH','Pipertazoinj',1,'90018','Ch. Satyam','2024-10-25 11:42:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   88.5 mV\r\nTemperature          32.3 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\nLog #                513\r\n\r\n\r\n'),
(955,'2024-10-25','11:02:56','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0024103','E4IP24001284','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','pH (Bulk)',1,'90018','Ch. Satyam','2024-10-25 11:42:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.13 pH\r\nmV                   49.5 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(956,'2024-10-25','12:10:01','90.00000','105.00000','Peptone','NA','RS007/24041','E4SO24000735','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-26 14:48:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.64 pH\r\nmV                   19.2 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             25-10-24,09:46\r\nLog #                265\r\n\r\n\r\n'),
(957,'2024-10-25','12:06:53','90.00000','105.00000','Ampicillin and Sulbactam for injection','pH','2349100832','E4RM24005746','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','Ampi&Sulbactam',1,'205208','S. Siddhartha Kumar Reddy','2024-10-25 13:33:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.96 pH\r\nmV                   -116.2 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\nLog #                515\r\n\r\n\r\n'),
(958,'2024-10-25','13:49:11','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005759','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-10-25 15:37:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   72.4 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(959,'2024-10-25','13:52:10','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005760','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-10-25 15:38:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.73 pH\r\nmV                   73.1 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(960,'2024-10-25','13:54:28','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005761','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90778','B. Bhaskar','2024-10-25 15:38:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.80 pH\r\nmV                   69.0 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(961,'2024-10-25','13:59:20','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005762','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90778','B. Bhaskar','2024-10-25 15:41:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.81 pH\r\nmV                   68.1 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(962,'2024-10-25','14:01:52','90.00000','105.00000','Purified water','pH','PW/SP34','E4RM24005763','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-10-25 15:42:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.72 pH\r\nmV                   73.6 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(963,'2024-10-25','14:04:04','90.00000','105.00000','Purified water','pH','PW/SP36','E4RM24005764','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-10-25 15:45:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   73.7 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(964,'2024-10-25','14:06:55','90.00000','105.00000','Purified water','pH','PW/SP43','E4RM24005765','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-10-25 15:45:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   72.0 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(965,'2024-10-25','14:09:38','90.00000','105.00000','Purified water','pH','PW/SP08','E4RM24005766','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-10-25 15:45:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.82 pH\r\nmV                   67.7 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(966,'2024-10-25','14:13:04','90.00000','105.00000','Pure steam','pH','PSC/SP10','E4RM24005767','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'90778','B. Bhaskar','2024-10-25 15:45:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   74.4 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(967,'2024-10-25','14:15:34','90.00000','105.00000','Pure steam','pH','PSC/SP20','E4RM24005768','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'90778','B. Bhaskar','2024-10-25 15:46:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.76 pH\r\nmV                   71.0 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(968,'2024-10-25','14:18:08','90.00000','105.00000','Pure steam','pH','PSC/SP21','E4RM24005769','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'90778','B. Bhaskar','2024-10-25 15:46:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.79 pH\r\nmV                   69.2 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(969,'2024-10-25','14:24:44','90.00000','105.00000','Glass Vials for pH','pH(Vial-1)','NA','E4EQ24000385','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'90778','B. Bhaskar','2024-10-25 15:46:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.51 pH\r\nmV                   261.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(970,'2024-10-25','14:27:16','90.00000','105.00000','Glass Vials for pH','pH(Vial-2)','NA','E4EQ24000385','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'90778','B. Bhaskar','2024-10-25 15:47:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.43 pH\r\nmV                   265.8 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(971,'2024-10-25','15:25:28','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0322007A','E4STFP24000671','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-10-25 16:28:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.82 pH\r\nmV                   67.7 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(972,'2024-10-25','15:29:44','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0424005A','E4STFP24000672','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-10-25 16:28:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.89 pH\r\nmV                   63.6 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(973,'2024-10-25','15:35:19','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0424005A','E4STFP24000673','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-10-25 16:28:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.67 pH\r\nmV                   76.1 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(974,'2024-10-25','15:41:44','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0424006A','E4STFP24000674','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-10-25 16:29:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   67.0 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(975,'2024-10-25','15:45:57','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0424006A','E4STFP24000675','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-10-25 16:30:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.87 pH\r\nmV                   64.8 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(976,'2024-10-25','16:11:21','90.00000','105.00000','Peptone','NA','RS007/24041','E4SO24000735','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-26 14:49:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.63 pH\r\nmV                   19.9 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             25-10-24,09:46\r\n\r\n\r\n'),
(977,'2024-10-25','16:17:14','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0424101','E4FP24000716','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','Pipertazoinj',1,'90778','B. Bhaskar','2024-10-25 16:30:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.65 pH\r\nmV                   19.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(978,'2024-10-25','17:06:23','90.00000','105.00000','NAFCILLIN FOR INJECTION','pH','NF0224014','E4FP24000714','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','NAFCILLIN',1,'90778','B. Bhaskar','2024-10-25 17:20:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.71 pH\r\nmV                   -43.0 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(979,'2024-10-25','17:11:11','90.00000','105.00000','NAFCILLIN FOR INJECTION','pH','NF0224015','E4FP24000717','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','NAFCILLIN',1,'90778','B. Bhaskar','2024-10-25 17:21:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.70 pH\r\nmV                   -42.2 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(980,'2024-10-25','17:59:50','90.00000','105.00000','Aseptic Media Fill Validation','PH','DP24010','E4AMV24000175','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','89853','J.Ravi',2,1,101,'pH','pH  Test',1,'90778','B. Bhaskar','2024-10-25 18:08:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.43 pH\r\nmV                   -26.4 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_126\r\nLast Cal             25-10-24,06:55\r\n\r\n\r\n'),
(981,'2024-10-26','07:18:05','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005615','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','1.68 pH',1,'86253','D. Jayapal Reddy','2024-10-26 08:24:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.64 pH\r\nmV                   303.3 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\nLog #                538\r\n\r\n\r\n'),
(982,'2024-10-26','07:24:45','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005615','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','4.01pH',1,'86253','D. Jayapal Reddy','2024-10-26 08:24:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.05 pH\r\nmV                   164.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\nLog #                539\r\n\r\n\r\n'),
