Thermo Scientific (c) 2011

          Versa Star Pro    

Meter S/N               V122329
SW Rev                  11.50

Channel                 1
Module S/n              VA36273
Method                  100
Calibration Time        03-10-24,00:53

-- pH Calibration Report -- 

Point                   1
pH                      1.68
mV                      310.9
Temperature             23.5 C (ATC)
Calibration Type        Auto

Point                   2
pH                      4.01
mV                      172.7
Temperature             23.6 C (ATC)
Calibration Type        Auto

Point                   3
pH                      7.00
mV                      2.5
Temperature             23.9 C (ATC)
Calibration Type        Auto

Point                   4
pH                      10.01
mV                      -169.6
Temperature             23.8 C (ATC)
Calibration Type        Manual

Point                   5
pH                      12.46
mV                      -312.6
Temperature             23.7 C (ATC)
Calibration Type        Manual

Slope 1                100.8 %
Slope 2                96.6 %
Slope 3                97.1 %
Slope 4                99.1 %
Eo    1                -4.9 mV
Eo    2                2.5 mV
Eo    3                2.5 mV
Eo    4                6.1 mV
Average Slope           98.4 %
Cal #                   106

