(983,'2024-10-26','07:33:08','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005615','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','7.00pH',1,'86253','D. Jayapal Reddy','2024-10-26 08:24:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.05 pH\r\nmV                   -8.3 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\nLog #                540\r\n\r\n\r\n'),
(984,'2024-10-26','07:55:41','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005615','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','10.01 pH',1,'86253','D. Jayapal Reddy','2024-10-26 08:24:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.02 pH\r\nmV                   -179.7 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\nLog #                541\r\n\r\n\r\n'),
(985,'2024-10-26','08:07:27','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005615','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','12.46 pH',1,'86253','D. Jayapal Reddy','2024-10-26 08:24:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.45 pH\r\nmV                   -319.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\nLog #                542\r\n\r\n\r\n'),
(986,'2024-10-26','08:47:41','90.00000','105.00000','Ampicillin and Sulbactam','RS','2349100832','E4RM24005746','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205189','P. Ravi Kumar',2,1,101,'pH','RS Mobile phase A',1,'86253','D. Jayapal Reddy','2024-10-26 09:46:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.00 pH\r\nmV                   110.5 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\n\r\n\r\n'),
(987,'2024-10-26','10:19:23','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005627','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','1.68 pH',1,'89384','M.V.V. Satish','2024-10-26 15:16:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   310.1 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             26-10-24,10:15\r\n\r\n\r\n'),
(988,'2024-10-26','10:21:52','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005627','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','4.01pH',1,'89384','M.V.V. Satish','2024-10-26 15:16:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.04 pH\r\nmV                   170.8 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             26-10-24,10:15\r\n\r\n\r\n'),
(989,'2024-10-26','10:36:41','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005627','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','7.00pH',1,'89384','M.V.V. Satish','2024-10-26 15:16:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -4.4 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             26-10-24,10:15\r\nLog #                269\r\n\r\n\r\n'),
(990,'2024-10-26','10:45:23','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005627','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'89384','M.V.V. Satish','2024-10-26 15:17:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.04 pH\r\nmV                   -180.1 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             26-10-24,10:15\r\n\r\n\r\n'),
(991,'2024-10-26','10:47:47','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005627','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'89384','M.V.V. Satish','2024-10-26 15:19:21','verified 12.45 pH buffer','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.45 pH\r\nmV                   -322.0 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             26-10-24,10:15\r\nLog #                271\r\n\r\n\r\n'),
(992,'2024-10-26','10:42:21','90.00000','105.00000','NAFCILLIN FOR INJECTION','RS BY HPLC MP A','NF0224016','E4FP24000718','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,1,101,'pH','NAFCILLIN',1,'86253','D. Jayapal Reddy','2024-10-26 14:12:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.51 pH\r\nmV                   201.1 mV\r\nTemperature          32.2 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\nLog #                544\r\n\r\n\r\n'),
(993,'2024-10-26','11:30:04','90.00000','105.00000','TAZOBACTAM','NA','PN0024099','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','1',1,'88515','Prashanth B','2024-10-26 12:05:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   3.92 pH\r\nmV                   163.6 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        96.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             26-10-24,07:06\r\n\r\n\r\n'),
(994,'2024-10-26','11:32:40','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC','PT0424115','E4FP24000715','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205058','C. Srinivasulu',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-10-26 14:12:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.03 pH\r\nmV                   51.4 mV\r\nTemperature          28.5 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\n\r\n\r\n'),
(995,'2024-10-26','12:23:13','90.00000','105.00000','TAZOBACTAM','NA','PN0024099','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','1',1,'88515','Prashanth B','2024-10-26 13:25:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.04 pH\r\nmV                   155.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        96.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             26-10-24,07:06\r\n\r\n\r\n'),
(996,'2024-10-26','13:13:49','90.00000','105.00000','TAZOBACTAM','NA','PN0024099','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','1',1,'88515','Prashanth B','2024-10-26 13:26:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.51 pH\r\nmV                   129.6 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        96.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             26-10-24,07:06\r\n\r\n\r\n'),
(997,'2024-10-26','14:55:33','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0024099','E4PP24001766','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','pH (Bulk)',1,'90778','B. Bhaskar','2024-10-26 16:45:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.76 pH\r\nmV                   123.4 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\nLog #                546\r\n\r\n\r\n'),
(998,'2024-10-26','15:20:09','90.00000','105.00000','PIPERACILLIN','NA','PN0024099','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','2',1,'84472','G. Ashok','2024-10-26 18:03:08','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.30 pH\r\nmV                   25.6 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        96.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             26-10-24,07:06\r\n\r\n\r\n'),
(999,'2024-10-26','15:24:02','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005791','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-26 17:59:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.60 pH\r\nmV                   75.4 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\n\r\n\r\n'),
(1000,'2024-10-26','15:25:49','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005792','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-26 18:00:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.59 pH\r\nmV                   75.7 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\n\r\n\r\n'),
(1001,'2024-10-26','15:28:35','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005793','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-26 18:01:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   62.0 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\n\r\n\r\n'),
(1002,'2024-10-26','15:30:48','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005794','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-26 18:15:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.64 pH\r\nmV                   72.7 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\n\r\n\r\n'),
(1003,'2024-10-26','15:33:27','90.00000','105.00000','Water For Injection','pH','WFI/SP34','E4RM24005795','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-26 18:15:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.31 pH\r\nmV                   92.0 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\n\r\n\r\n'),
(1004,'2024-10-26','15:35:48','90.00000','105.00000','Purified water','pH','PW/SP35','E4RM24005796','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-26 18:16:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.82 pH\r\nmV                   62.6 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\nLog #                552\r\n\r\n\r\n'),
(1005,'2024-10-26','15:41:05','90.00000','105.00000','Pure steam','pH','PSC/SP22','E4RM24005797','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'205208','S. Siddhartha Kumar Reddy','2024-10-26 18:17:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.65 pH\r\nmV                   72.0 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\n\r\n\r\n'),
(1006,'2024-10-26','15:42:13','90.00000','105.00000','Sensor Calibration','NA','NA','NA','Temperature Sensor','PR-PHMTZ-003','CW1-12031','1','NULL','85705','P. Prakash Rao',2,0,105,'pH','Purified Water',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   5.63 pH\r\nmV                   82.6 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             26-10-24,12:19\r\nLog #                18\r\n\r\n\r\n'),
(1007,'2024-10-26','16:08:56','90.00000','105.00000','Nafcillin for injection','Cleaning Verification Diluent pH','NF0124004','E4CV24000300','Cleaning Validation','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,0,101,'pH','CV',1,'205208','S. Siddhartha Kumar Reddy','2024-10-26 18:21:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -5.7 mV\r\nTemperature          31.1 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\n\r\n\r\n'),
(1008,'2024-10-26','16:34:10','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0024099','E4PP24001768','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','pH (Bulk)',1,'90778','B. Bhaskar','2024-10-26 16:44:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.42 pH\r\nmV                   27.8 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\n\r\n\r\n'),
(1009,'2024-10-26','16:53:07','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','PN0024099','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','3',1,'84472','G. Ashok','2024-10-26 18:03:36','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.16 pH\r\nmV                   34.6 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        96.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             26-10-24,07:06\r\n\r\n\r\n'),
(1010,'2024-10-26','17:15:30','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','PN0024099','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','3',1,'84472','G. Ashok','2024-10-26 18:03:56','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.20 pH\r\nmV                   32.2 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        96.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             26-10-24,07:06\r\nLog #                209\r\n\r\n\r\n'),
(1011,'2024-10-26','17:17:04','90.00000','105.00000','Nafcillin for injection','Cleaning Verification mobile phase pH','NF0124004','E4CV24000300','Cleaning Validation','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','CV',1,'205208','S. Siddhartha Kumar Reddy','2024-10-26 18:23:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.47 pH\r\nmV                   -33.0 mV\r\nTemperature          32.0 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\nLog #                556\r\n\r\n\r\n'),
(1012,'2024-10-26','17:23:45','90.00000','105.00000','Tryptone soya agar TSA with 0.05%soya lecithin&0.5% tween 80 and beta lactamase 55mm plates(RODAC)','NA','SPU408','E4SO24000717','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','RUTSA',1,'205143','Ch. Prasad','2024-11-05 17:13:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.35 pH\r\nmV                   -22.6 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             26-10-24,10:15\r\nLog #                273\r\n\r\n\r\n'),
(1013,'2024-10-26','17:35:59','90.00000','105.00000','Glass Vials for pH','pH(Vial-1)','NA','E4EQ24000391','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-26 17:51:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.50 pH\r\nmV                   253.6 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\n\r\n\r\n'),
(1014,'2024-10-26','17:37:34','90.00000','105.00000','Tryptone soya agar TSA with 0.05%soya lecithin&0.5% tween 80 and beta lactamase 55mm plates(RODAC)',' NA','SPU408','	E4SO24000746','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','205207','P. Hanmanthu',2,0,103,'pH','RUTSA',1,'205143','Ch. Prasad','2024-10-29 16:51:33','Consider as E4SO24000745','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.37 pH\r\nmV                   -24.5 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             26-10-24,10:15\r\nLog #                274\r\n\r\n\r\n'),
(1015,'2024-10-26','17:38:38','90.00000','105.00000','Glass Vials for pH','pH(Vial-2)','NA','E4EQ24000391','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-26 17:52:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.32 pH\r\nmV                   264.1 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\nLog #                558\r\n\r\n\r\n'),
(1016,'2024-10-26','17:45:33','90.00000','105.00000','Soyabean casein digest agar (SCDA)with beta-lactamase','NA','MPT0569','E4SO24000746','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','205207','P. Hanmanthu',2,0,103,'pH','RUSCDA',1,'89384','M.V.V. Satish','2024-10-26 18:24:35','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.24 pH\r\nmV                   -17.1 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             26-10-24,10:15\r\nLog #                275\r\n\r\n\r\n'),
(1017,'2024-10-26','17:58:27','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','ph','PN0024099','E4PP24001769','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-10-26 18:13:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.01 pH\r\nmV                   51.4 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        97.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_127\r\nLast Cal             26-10-24,07:11\r\n\r\n\r\n'),
(1018,'2024-10-27','07:32:31','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005633','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-10-27 08:33:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   306.2 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1019,'2024-10-27','07:41:43','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005633','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-10-27 08:33:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.05 pH\r\nmV                   169.6 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\nLog #                561\r\n\r\n\r\n'),
(1020,'2024-10-27','07:49:15','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005633','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-10-27 08:33:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.99 pH\r\nmV                   -1.2 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\nLog #                562\r\n\r\n\r\n'),
(1021,'2024-10-27','08:09:27','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005633','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-10-27 08:33:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.00 pH\r\nmV                   -176.5 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\nLog #                563\r\n\r\n\r\n'),
(1022,'2024-10-27','08:15:19','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005633','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-10-27 08:33:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.45 pH\r\nmV                   -318.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\nLog #                564\r\n\r\n\r\n'),
(1023,'2024-10-27','08:49:32','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','Assay','PN0024099','E4PP24001770','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-10-27 10:00:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   86.7 mV\r\nTemperature          29.5 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1024,'2024-10-27','09:03:35','90.00000','105.00000','NAFCILLIN FOR INJECTION','ASSAY','NF0224014','E4FP24000714','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','89853','J.Ravi',2,1,101,'pH','NAFCILLIN',1,'82735','M.D. Babar Qhadri','2024-10-27 10:03:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.49 pH\r\nmV                   -31.1 mV\r\nTemperature          30.8 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\nLog #                566\r\n\r\n\r\n'),
(1025,'2024-10-27','10:20:16','90.00000','105.00000','NAFCILLIN FOR INJECTION','Assay','NF0224014','E4FP24000714','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','89853','J.Ravi',2,1,101,'pH','NAFCILLIN',1,'82735','M.D. Babar Qhadri','2024-10-27 13:14:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -1.8 mV\r\nTemperature          29.0 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1026,'2024-10-27','11:24:56','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0324079','E4PP24001773','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-10-27 13:15:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.20 pH\r\nmV                   44.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\nLog #                569\r\n\r\n\r\n'),
(1027,'2024-10-27','11:31:01','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','pH','PN0024099','E4IP24001290','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','3',1,'82735','M.D. Babar Qhadri','2024-10-27 13:15:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.20 pH\r\nmV                   44.2 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1028,'2024-10-27','11:35:40','90.00000','105.00000','Media Fill','pH','DP24010','E4AMV24000179','ASEPTIC MEDIA FILL','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Media fill pH',1,'82735','M.D. Babar Qhadri','2024-10-27 13:15:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.40 pH\r\nmV                   -24.9 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1029,'2024-10-27','12:03:49','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424115','E4FP24000715','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-10-27 13:16:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.84 pH\r\nmV                   65.8 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1030,'2024-10-27','12:09:13','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424116','E4FP24000720','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-10-27 13:16:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   66.2 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1031,'2024-10-27','14:05:06','90.00000','105.00000','Glass Vials for pH','pH(Vial-1)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:21:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.64 pH\r\nmV                   251.9 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1032,'2024-10-27','14:07:58','90.00000','105.00000','Glass Vials for pH','pH(Vial-2)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:22:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.45 pH\r\nmV                   262.2 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1033,'2024-10-27','14:10:38','90.00000','105.00000','Glass Vials for pH','pH(Vial-3)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:22:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.40 pH\r\nmV                   265.3 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1034,'2024-10-27','14:12:53','90.00000','105.00000','Glass Vials for pH','pH(Vial-4)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:23:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.45 pH\r\nmV                   262.4 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1035,'2024-10-27','14:15:03','90.00000','105.00000','Glass Vials for pH','pH(Vial-5)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:23:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.52 pH\r\nmV                   258.3 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1036,'2024-10-27','14:17:06','90.00000','105.00000','Glass Vials for pH','pH(Vial-6)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:24:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.56 pH\r\nmV                   255.6 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1037,'2024-10-27','14:19:53','90.00000','105.00000','Glass Vials for pH','pH(Vial-7)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:24:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.62 pH\r\nmV                   252.3 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1038,'2024-10-27','14:22:01','90.00000','105.00000','Glass Vials for pH','pH(Vial-8)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:25:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.69 pH\r\nmV                   248.5 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1039,'2024-10-27','14:24:55','90.00000','105.00000','Glass Vials for pH','pH(Vial-9)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:26:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.71 pH\r\nmV                   246.8 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1040,'2024-10-27','14:27:33','90.00000','105.00000','Glass Vials for pH','pH(Vial-10)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:31:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.54 pH\r\nmV                   256.9 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1041,'2024-10-27','14:30:42','90.00000','105.00000','Glass Vials for pH','pH(Vial-11)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:31:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.40 pH\r\nmV                   264.8 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1042,'2024-10-27','14:32:51','90.00000','105.00000','Glass Vials for pH','pH(Vial-12)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:32:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.45 pH\r\nmV                   262.0 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1043,'2024-10-27','14:35:09','90.00000','105.00000','Glass Vials for pH','pH(Vial-13)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:32:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.52 pH\r\nmV                   258.4 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1044,'2024-10-27','14:37:33','90.00000','105.00000','Glass Vials for pH','pH(Vial-14)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:32:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.59 pH\r\nmV                   254.4 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1045,'2024-10-27','14:39:43','90.00000','105.00000','Glass Vials for pH','pH(Vial-15)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:33:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.40 pH\r\nmV                   264.7 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1046,'2024-10-27','14:42:36','90.00000','105.00000','Glass Vials for pH','pH(Vial-16)','NA','E4EQ24000381','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:33:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.70 pH\r\nmV                   247.4 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1047,'2024-10-27','14:57:35','90.00000','105.00000','Glass Vials for pH','pH(Vial-1)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:34:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.73 pH\r\nmV                   246.3 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1048,'2024-10-27','14:59:47','90.00000','105.00000','Glass Vials for pH','pH(Vial-2)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:35:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.35 pH\r\nmV                   268.2 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1049,'2024-10-27','15:02:06','90.00000','105.00000','Glass Vials for pH','pH(Vial-3)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:35:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.43 pH\r\nmV                   263.5 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1050,'2024-10-27','15:04:46','90.00000','105.00000','Glass Vials for pH','pH(Vial-4)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:36:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.87 pH\r\nmV                   238.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1051,'2024-10-27','15:06:52','90.00000','105.00000','Glass Vials for pH','pH(Vial-5)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:36:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.88 pH\r\nmV                   236.9 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1052,'2024-10-27','15:09:20','90.00000','105.00000','Glass Vials for pH','pH(Vial-6)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:36:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.69 pH\r\nmV                   248.0 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1053,'2024-10-27','15:11:50','90.00000','105.00000','Glass Vials for pH','pH(Vial-7)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:37:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.61 pH\r\nmV                   252.8 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1054,'2024-10-27','15:15:11','90.00000','105.00000','Glass Vials for pH','pH(Vial-8)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:37:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.32 pH\r\nmV                   269.4 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1055,'2024-10-27','15:18:07','90.00000','105.00000','Glass Vials for pH','pH(Vial-9)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:37:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.54 pH\r\nmV                   257.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1056,'2024-10-27','15:21:21','90.00000','105.00000','Glass Vials for pH','pH(Vial-10)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:38:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.31 pH\r\nmV                   270.3 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1057,'2024-10-27','15:24:33','90.00000','105.00000','Glass Vials for pH','pH(Vial-11)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:38:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.36 pH\r\nmV                   267.9 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1058,'2024-10-27','15:27:37','90.00000','105.00000','Glass Vials for pH','pH(Vial-12)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:39:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.60 pH\r\nmV                   253.9 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1059,'2024-10-27','15:30:20','90.00000','105.00000','Glass Vials for pH','pH(Vial-13)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:39:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.87 pH\r\nmV                   238.4 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1060,'2024-10-27','15:33:29','90.00000','105.00000','Glass Vials for pH','pH(Vial-14)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:40:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.52 pH\r\nmV                   258.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1061,'2024-10-27','15:35:40','90.00000','105.00000','Glass Vials for pH','pH(Vial-15)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:40:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.22 pH\r\nmV                   275.3 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1062,'2024-10-27','15:38:13','90.00000','105.00000','Glass Vials for pH','pH(Vial-16)','NA','E4EQ24000382','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:41:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.57 pH\r\nmV                   255.3 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1063,'2024-10-27','15:58:05','90.00000','105.00000','Glass Vials for pH','pH(Vial-1)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:42:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.86 pH\r\nmV                   239.1 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1064,'2024-10-27','15:59:54','90.00000','105.00000','Glass Vials for pH','pH(Vial-2)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:43:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.56 pH\r\nmV                   256.4 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1065,'2024-10-27','16:02:01','90.00000','105.00000','Glass Vials for pH','pH(Vial-3)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:43:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.38 pH\r\nmV                   266.5 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1066,'2024-10-27','16:05:08','90.00000','105.00000','Glass Vials for pH','pH(Vial-4)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:44:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.82 pH\r\nmV                   241.1 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1067,'2024-10-27','16:07:56','90.00000','105.00000','Glass Vials for pH','pH(Vial-5)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:44:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.56 pH\r\nmV                   255.8 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1068,'2024-10-27','16:10:04','90.00000','105.00000','Glass Vials for pH','pH(Vial-6)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:45:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.44 pH\r\nmV                   263.0 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1069,'2024-10-27','16:12:00','90.00000','105.00000','Glass Vials for pH','pH(Vial-7)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:45:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.52 pH\r\nmV                   258.3 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1070,'2024-10-27','16:13:35','90.00000','105.00000','Glass Vials for pH','pH(Vial-8)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:45:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.84 pH\r\nmV                   239.8 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1071,'2024-10-27','16:15:33','90.00000','105.00000','Glass Vials for pH','pH(Vial-9)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:45:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.38 pH\r\nmV                   266.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1072,'2024-10-27','16:17:30','90.00000','105.00000','Glass Vials for pH','pH(Vial-10)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:46:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.56 pH\r\nmV                   255.9 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1073,'2024-10-27','16:19:24','90.00000','105.00000','Glass Vials for pH','pH(Vial-11)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:46:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.34 pH\r\nmV                   268.8 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1074,'2024-10-27','16:22:48','90.00000','105.00000','Glass Vials for pH','pH(Vial-12)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:46:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.90 pH\r\nmV                   236.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1075,'2024-10-27','16:25:07','90.00000','105.00000','Glass Vials for pH','pH(Vial-13)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:47:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.63 pH\r\nmV                   251.8 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1076,'2024-10-27','16:27:21','90.00000','105.00000','Glass Vials for pH','pH(Vial-14)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:47:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.96 pH\r\nmV                   232.4 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1077,'2024-10-27','16:30:33','90.00000','105.00000','Glass Vials for pH','pH(Vial-15)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:47:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.49 pH\r\nmV                   259.9 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1078,'2024-10-27','16:33:07','90.00000','105.00000','Glass Vials for pH','pH(Vial-16)','NA','E4EQ24000387','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-10-28 07:48:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.32 pH\r\nmV                   269.5 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_128\r\nLast Cal             27-10-24,07:22\r\n\r\n\r\n'),
(1079,'2024-10-28','06:55:30','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005638','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-10-28 07:15:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.64 pH\r\nmV                   305.8 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1080,'2024-10-28','06:57:40','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005638','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-10-28 07:15:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   167.7 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1081,'2024-10-28','07:00:31','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005638','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-10-28 07:15:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -7.1 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1082,'2024-10-28','07:03:29','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005638','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-10-28 07:15:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.03 pH\r\nmV                   -183.8 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1083,'2024-10-28','07:06:09','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005638','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-10-28 07:16:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.44 pH\r\nmV                   -324.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1084,'2024-10-28','09:19:05','90.00000','105.00000','pH Buffer','NA','NA','E4INS240005645','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-28 10:38:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   311.4 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             28-10-24,09:16\r\n\r\n\r\n'),
(1085,'2024-10-28','09:28:46','90.00000','105.00000','pH Buffer','NA','NA','E4INS240005645','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-28 10:38:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.05 pH\r\nmV                   174.1 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             28-10-24,09:16\r\n\r\n\r\n'),
(1086,'2024-10-28','09:34:14','90.00000','105.00000','pH Buffer','NA','NA','E4INS240005645','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-28 10:38:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -0.6 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             28-10-24,09:16\r\n\r\n\r\n'),
(1087,'2024-10-28','09:37:21','90.00000','105.00000','pH Buffer','NA','NA','E4INS240005645','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-28 10:39:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.04 pH\r\nmV                   -176.6 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             28-10-24,09:16\r\n\r\n\r\n'),
(1088,'2024-10-28','09:39:54','90.00000','105.00000','pH Buffer','NA','NA','E4INS240005645','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-28 10:39:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -321.0 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             28-10-24,09:16\r\nLog #                280\r\n\r\n\r\n'),
(1089,'2024-10-28','10:00:17','90.00000','105.00000','Piperacillin and Tazobactam For injection','ASSAY','PT0424115','E4FP24000715','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','90017','G. Srishailam',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-10-28 10:58:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   82.6 mV\r\nTemperature          31.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1090,'2024-10-28','10:15:40','90.00000','105.00000','Piperacillin and Tazobactam For injection','UOD','PT0424115','E4FP24000715','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-10-28 10:59:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   82.3 mV\r\nTemperature          31.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1091,'2024-10-28','10:23:31','90.00000','105.00000','Piperacillin and Tazobactam For injection','ASSAY','TZ0424102','E4FP24000721','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205232','K. Narender Reddy',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-10-28 10:59:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   81.5 mV\r\nTemperature          32.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1092,'2024-10-28','10:37:18','90.00000','105.00000','Ampicillin Sodium USP','RS','2449102472','E4RM24005715','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205189','P. Ravi Kumar',2,1,101,'pH','pH 5.0 Phosphate Buffer',1,'82735','M.D. Babar Qhadri','2024-10-28 11:00:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.01 pH\r\nmV                   111.1 mV\r\nTemperature          28.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1093,'2024-10-28','11:14:00','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0324079','E4PP24001774','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-10-28 11:33:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.02 pH\r\nmV                   50.7 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1094,'2024-10-28','11:16:19','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0324079','E4PP24001775','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-10-28 11:35:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.03 pH\r\nmV                   49.8 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\nLog #                632\r\n\r\n\r\n'),
(1095,'2024-10-28','11:24:45','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0322008A','E4STFP24000676','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-28 11:35:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.59 pH\r\nmV                   75.5 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1096,'2024-10-28','12:31:21','90.00000','105.00000','Macckankey broth','NA','RS013/24008','E4SO24000748','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','NA',1,'205143','Ch. Prasad','2024-10-28 17:08:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.17 pH\r\nmV                   -9.0 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             28-10-24,09:16\r\n\r\n\r\n'),
(1097,'2024-10-28','12:35:15','90.00000','105.00000','Phosphate buffered saline','NA','RS014/24004','E4SO24000749','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','PBSIH',1,'205143','Ch. Prasad','2024-10-28 17:09:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.40 pH\r\nmV                   -22.4 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             28-10-24,09:16\r\n\r\n\r\n'),
(1098,'2024-10-28','13:43:21','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0424102','E4FP24000721','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-10-28 13:57:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.08 pH\r\nmV                   47.0 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1099,'2024-10-28','14:36:04','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005818','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-28 15:45:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.60 pH\r\nmV                   74.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1100,'2024-10-28','14:40:11','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005819','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-28 15:46:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   69.7 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1101,'2024-10-28','14:45:00','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005820','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','pH',1,'90018','Ch. Satyam','2024-10-28 15:46:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   69.8 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1102,'2024-10-28','14:49:55','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005821','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','pH',1,'90018','Ch. Satyam','2024-10-28 15:46:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   69.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1103,'2024-10-28','14:56:16','90.00000','105.00000','Macckankey broth','NA','RS013/24008','E4SO24000748','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','NA',1,'205143','Ch. Prasad','2024-10-28 17:09:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.29 pH\r\nmV                   -16.1 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             28-10-24,09:16\r\n\r\n\r\n'),
(1104,'2024-10-28','14:59:25','90.00000','105.00000','Phosphate buffered saline','NA','RS014/24004','E4SO24000749','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','PBSIH',1,'205143','Ch. Prasad','2024-10-28 17:09:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.42 pH\r\nmV                   -23.8 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             28-10-24,09:16\r\n\r\n\r\n'),
(1105,'2024-10-28','15:01:06','90.00000','105.00000','Purified water','pH','PW/SP03','E4RM24005822','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-28 15:46:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.73 pH\r\nmV                   67.1 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1106,'2024-10-28','15:06:12','90.00000','105.00000','Pure steam','pH','PSC/SP09','E4RM24005823','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','pH_Pure Steam',1,'90018','Ch. Satyam','2024-10-28 15:49:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.84 pH\r\nmV                   61.0 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1107,'2024-10-28','15:10:21','90.00000','105.00000','Purified water','pH','PW/SP11','E4RM24005824','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-28 15:49:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   68.6 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1108,'2024-10-28','15:15:14','90.00000','105.00000','Water For Injection','pH','WFI/SP15','E4RM24005825','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','pH',1,'90018','Ch. Satyam','2024-10-28 15:46:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   69.6 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1109,'2024-10-28','16:01:33','90.00000','105.00000','NAFCILLIN FOR INJECTION','pH','NF0224016','E4FP24000718','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','NAFCILLIN',1,'90778','B. Bhaskar','2024-10-29 08:59:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.71 pH\r\nmV                   -48.3 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1110,'2024-10-28','16:04:12','90.00000','105.00000','NAFCILLIN FOR INJECTION','pH','NF0124004','E4FP24000719','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','NAFCILLIN',1,'90778','B. Bhaskar','2024-10-29 09:00:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.63 pH\r\nmV                   -43.7 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1111,'2024-10-28','16:28:05','90.00000','105.00000','Soyabean casein digest agar (SCDA)with beta-lactamase','NA','MPT0677','E4SO24000753','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','RUSCDA',1,'205143','Ch. Prasad','2024-10-28 17:10:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.37 pH\r\nmV                   -20.3 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             28-10-24,09:16\r\n\r\n\r\n'),
(1112,'2024-10-28','16:31:17','90.00000','105.00000','Soyabean casean digest agar SCDA With 0.05%soya lecithin 05% tween 80 and beta lactamase (Finger dab)','NA','MPT0622','E4SO24000754','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','RUSCDAF',1,'205143','Ch. Prasad','2024-10-28 17:10:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.27 pH\r\nmV                   -15.0 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             28-10-24,09:16\r\n\r\n\r\n'),
(1113,'2024-10-28','16:33:44','90.00000','105.00000','Tryptone soya agar TSA with 0.05%soya lecithin&0.5% tween 80 and beta lactamase 55mm plates(RODAC)','NA','SPU408','E4SO24000755','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','RUTSA',1,'205143','Ch. Prasad','2024-10-28 17:11:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.32 pH\r\nmV                   -17.6 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             28-10-24,09:16\r\n\r\n\r\n'),
(1114,'2024-10-28','20:54:24','90.00000','105.00000','Media Fill','pH','LY24004','E4AMV24000186','ASEPTIC MEDIA FILL','QC-PHMTZ-007','CX1-15733','1','NULL','205058','C. Srinivasulu',2,1,101,'pH','Media fill pH',1,'90018','Ch. Satyam','2024-10-28 21:03:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.34 pH\r\nmV                   -26.5 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_129\r\nLast Cal             28-10-24,06:50\r\n\r\n\r\n'),
(1115,'2024-10-29','07:45:23','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005676','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205196','P. Mounika',2,1,101,'pH','1.68 pH',1,'90778','B. Bhaskar','2024-10-29 08:21:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   303.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1116,'2024-10-29','07:49:05','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005676','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205196','P. Mounika',2,1,101,'pH','4.01pH',1,'90778','B. Bhaskar','2024-10-29 08:21:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   167.1 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\nLog #                647\r\n\r\n\r\n'),
(1117,'2024-10-29','07:56:07','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005676','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205196','P. Mounika',2,1,101,'pH','7.00pH',1,'90778','B. Bhaskar','2024-10-29 08:21:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -8.5 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1118,'2024-10-29','08:04:47','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005676','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205196','P. Mounika',2,1,101,'pH','10.01 pH',1,'90778','B. Bhaskar','2024-10-29 08:22:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.06 pH\r\nmV                   -184.8 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1119,'2024-10-29','08:08:07','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005676','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205196','P. Mounika',2,1,101,'pH','12.46 pH',1,'90778','B. Bhaskar','2024-10-29 08:22:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -326.8 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1120,'2024-10-29','08:18:17','90.00000','105.00000','TAZOBACTAM','NA','PN0024100','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','1',1,'88629','Raghuram Reddy','2024-10-29 09:21:32','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   3.89 pH\r\nmV                   165.9 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        97.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             29-10-24,05:12\r\n\r\n\r\n'),
(1121,'2024-10-29','09:02:02','90.00000','105.00000','PIPERACILLIN','NA','PN0024100','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-29 09:22:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.53 pH\r\nmV                   14.6 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        97.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             29-10-24,05:12\r\n\r\n\r\n'),
(1122,'2024-10-29','09:10:58','90.00000','105.00000','TAZOBACTAM','NA','PN0024100','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','1',1,'88629','Raghuram Reddy','2024-10-29 09:22:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.16 pH\r\nmV                   151.0 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        97.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             29-10-24,05:12\r\n\r\n\r\n'),
(1123,'2024-10-29','09:22:34','90.00000','105.00000','Penicillin G Benzathine','disso','PG0124003A','E4STFP24000570','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Penicillin G',1,'90778','B. Bhaskar','2024-10-29 09:28:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.32 pH\r\nmV                   209.8 mV\r\nTemperature          26.8 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1124,'2024-10-29','09:38:46','90.00000','105.00000','Penicillin G Benzathine','disso','PG0124003A','E4STFP24000570','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Penicillin G',1,'90778','B. Bhaskar','2024-10-29 11:54:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.49 pH\r\nmV                   -35.3 mV\r\nTemperature          28.7 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1125,'2024-10-29','09:57:43','90.00000','105.00000','TAZOBACTAM','NA','PN0024100','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','1',1,'88515','Prashanth B','2024-10-29 12:38:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.43 pH\r\nmV                   134.8 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        97.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             29-10-24,05:12\r\n\r\n\r\n'),
(1126,'2024-10-29','10:29:57','90.00000','105.00000','Piperacillin and Tazobactam for Injection','IDENTIFICATION','2449200734','E4RM24005721','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','Mobile phase pH',1,'84378','Shivanand. W','2024-10-30 08:09:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   82.8 mV\r\nTemperature          31.4 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1127,'2024-10-29','10:35:51','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS BY HPLC','PN0322001A','E4STFP24000648','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,0,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-11-02 12:37:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.02 pH\r\nmV                   51.4 mV\r\nTemperature          28.2 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1128,'2024-10-29','10:47:24','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005688','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-29 11:11:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   310.7 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             29-10-24,10:44\r\n\r\n\r\n'),
(1129,'2024-10-29','10:49:29','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005688','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-29 11:12:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   172.9 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             29-10-24,10:44\r\n\r\n\r\n'),
(1130,'2024-10-29','10:50:54','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005688','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-29 11:13:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -1.0 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             29-10-24,10:44\r\n\r\n\r\n'),
(1131,'2024-10-29','10:52:21','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005688','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-29 11:13:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.03 pH\r\nmV                   -176.9 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             29-10-24,10:44\r\n\r\n\r\n'),
(1132,'2024-10-29','10:54:06','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005688','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-29 11:13:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.49 pH\r\nmV                   -320.7 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             29-10-24,10:44\r\n\r\n\r\n'),
(1133,'2024-10-29','10:55:17','90.00000','105.00000','Ampcillin for injection','RS BY HPLC MP A','AP0224001A','E4STFP24000647','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,1,101,'pH','Ampi',1,'90778','B. Bhaskar','2024-10-29 11:12:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.98 pH\r\nmV                   113.7 mV\r\nTemperature          29.9 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\nLog #                655\r\n\r\n\r\n'),
(1134,'2024-10-29','11:37:03','90.00000','105.00000','Ampicillin sodium USP','pH','2449102472','E4RM24005715','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','test pH',1,'90778','B. Bhaskar','2024-10-29 11:54:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.16 pH\r\nmV                   -132.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1135,'2024-10-29','11:41:03','90.00000','105.00000','Ampicillin sodium USP','pH','2449102473','E4RM24005716','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','test pH',1,'90778','B. Bhaskar','2024-10-29 11:54:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.14 pH\r\nmV                   -130.8 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\nLog #                657\r\n\r\n\r\n'),
(1136,'2024-10-29','11:44:52','90.00000','105.00000','Peptone','NA','RS007/24042','E4SO24000759','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-30 16:29:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.56 pH\r\nmV                   25.4 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             29-10-24,10:44\r\nLog #                293\r\n\r\n\r\n'),
(1137,'2024-10-29','11:46:52','90.00000','105.00000','Ampicillin sodium USP','pH','2449102474','E4RM24005717','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','test pH',1,'90778','B. Bhaskar','2024-10-29 11:54:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.21 pH\r\nmV                   -134.7 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1138,'2024-10-29','11:57:39','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','PN0024100','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','3',1,'88515','Prashanth B','2024-10-29 12:39:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.32 pH\r\nmV                   26.5 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        97.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             29-10-24,05:12\r\n\r\n\r\n'),
(1139,'2024-10-29','12:16:06','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','PN0024100','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','3',1,'88515','Prashanth B','2024-10-29 12:39:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.34 pH\r\nmV                   25.7 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        97.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             29-10-24,05:12\r\n\r\n\r\n'),
(1140,'2024-10-29','12:58:39','90.00000','105.00000','XLDA','NA','RS029/24007','E4SO24000761','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','XLDA',1,'205143','Ch. Prasad','2024-10-30 16:29:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.35 pH\r\nmV                   -20.9 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             29-10-24,10:44\r\n\r\n\r\n'),
(1141,'2024-10-29','13:01:05','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24018','E4SO24000762','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','SCDBIH',1,'205143','Ch. Prasad','2024-10-30 16:29:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.39 pH\r\nmV                   -22.9 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             29-10-24,10:44\r\n\r\n\r\n'),
(1142,'2024-10-29','13:36:02','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005842','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-29 16:02:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   80.8 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1143,'2024-10-29','13:37:34','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005843','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-29 16:03:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.59 pH\r\nmV                   75.5 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1144,'2024-10-29','13:39:23','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005844','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-29 16:04:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.78 pH\r\nmV                   64.9 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1145,'2024-10-29','13:40:55','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005845','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-29 16:05:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.68 pH\r\nmV                   70.4 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1146,'2024-10-29','13:42:41','90.00000','105.00000','Water For Injection','pH','WFI/SP13','E4RM24005846','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-29 16:05:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   69.7 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1147,'2024-10-29','13:44:29','90.00000','105.00000','Pure steam','pH','PSC/SP10','E4RM24005847','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'205208','S. Siddhartha Kumar Reddy','2024-10-29 16:07:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.73 pH\r\nmV                   67.3 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1148,'2024-10-29','14:48:08','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0024100','E4PP24001780','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-10-29 16:44:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.77 pH\r\nmV                   65.3 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1149,'2024-10-29','14:55:53','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0024100','E4PP24001777','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-10-29 16:09:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.34 pH\r\nmV                   148.9 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1150,'2024-10-29','15:02:36','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0024100','E4PP24001779','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-10-29 16:09:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.21 pH\r\nmV                   39.7 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1151,'2024-10-29','15:33:50','90.00000','105.00000','FLUID-A','NA','RS012/24010','E4SO24000743','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Fluid-A',1,'205143','Ch. Prasad','2024-10-30 16:29:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.80 pH\r\nmV                   11.1 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             29-10-24,10:44\r\nLog #                296\r\n\r\n\r\n'),
(1152,'2024-10-29','15:38:18','90.00000','105.00000','Peptone','NA','RS007/24042','E4SO24000759','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-30 16:29:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.71 pH\r\nmV                   16.5 mV\r\nTemperature          26.8 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             29-10-24,10:44\r\nLog #                297\r\n\r\n\r\n'),
(1153,'2024-10-29','16:01:32','90.00000','105.00000','Piperacillin and Tazobactam for Injection','ASSAY','TR0424001A','E4STFP24000640','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,1,101,'pH','PIPERTAZO',1,'205208','S. Siddhartha Kumar Reddy','2024-10-29 16:11:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   82.4 mV\r\nTemperature          30.8 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1154,'2024-10-29','16:54:50','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0024100','E4IP24001295','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'86253','D. Jayapal Reddy','2024-10-29 17:21:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.18 pH\r\nmV                   41.7 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1155,'2024-10-29','17:09:39','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0324080','E4PP24001784','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'86253','D. Jayapal Reddy','2024-10-29 17:22:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.16 pH\r\nmV                   43.1 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_130\r\nLast Cal             29-10-24,07:40\r\n\r\n\r\n'),
(1156,'2024-10-29','19:17:46','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24018','E4SO24000762','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDBIH',1,'205143','Ch. Prasad','2024-11-02 09:38:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.47 pH\r\nmV                   -27.0 mV\r\nTemperature          18.9 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             29-10-24,10:44\r\n\r\n\r\n'),
(1157,'2024-10-29','19:23:41','90.00000','105.00000','XLDA','NA','RS029/24007','E4SO24000761','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','XLDA',1,'205143','Ch. Prasad','2024-10-30 16:31:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.40 pH\r\nmV                   -23.4 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             29-10-24,10:44\r\n\r\n\r\n'),
(1158,'2024-10-29','21:41:57','90.00000','105.00000','FLUID-A','NA','RS012/24010','E4SO24000743','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Fluid-A',1,'205143','Ch. Prasad','2024-10-30 16:32:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.62 pH\r\nmV                   21.7 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             29-10-24,10:44\r\n\r\n\r\n'),
(1159,'2024-10-30','07:47:45','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005694','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','1.68 pH',1,'84378','Shivanand. W','2024-10-30 08:25:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.71 pH\r\nmV                   306.5 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1160,'2024-10-30','08:04:24','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005694','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','4.01pH',1,'84378','Shivanand. W','2024-10-30 08:25:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   170.9 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1161,'2024-10-30','08:10:01','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005694','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','7.00pH',1,'84378','Shivanand. W','2024-10-30 08:26:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -4.9 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1162,'2024-10-30','08:12:46','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005694','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','10.01 pH',1,'84378','Shivanand. W','2024-10-30 08:26:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.04 pH\r\nmV                   -181.6 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1163,'2024-10-30','08:15:18','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005694','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','12.46 pH',1,'84378','Shivanand. W','2024-10-30 08:27:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -325.8 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\nLog #                675\r\n\r\n\r\n'),
(1164,'2024-10-30','08:53:25','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005705','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','1.68 pH',1,'89384','M.V.V. Satish','2024-10-30 09:17:39','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   311.3 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-10-24,08:51\r\n\r\n\r\n'),
(1165,'2024-10-30','08:54:59','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005705','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','4.01pH',1,'89384','M.V.V. Satish','2024-10-30 09:17:58','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   174.0 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-10-24,08:51\r\n\r\n\r\n'),
(1166,'2024-10-30','08:56:29','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005705','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','7.00pH',1,'89384','M.V.V. Satish','2024-10-30 09:18:13','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -0.0 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-10-24,08:51\r\n\r\n\r\n'),
(1167,'2024-10-30','08:58:36','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005705','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','10.01 pH',1,'89384','M.V.V. Satish','2024-10-30 09:18:33','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   9.99 pH\r\nmV                   -174.5 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-10-24,08:51\r\nLog #                304\r\n\r\n\r\n'),
(1168,'2024-10-30','09:00:42','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005705','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','12.46 pH',1,'89384','M.V.V. Satish','2024-10-30 09:18:46','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.48 pH\r\nmV                   -320.3 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-10-24,08:51\r\n\r\n\r\n'),
(1169,'2024-10-30','08:59:29','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','ASSAY','PN0024100','E4PP24001781','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,0,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:33:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.49 pH\r\nmV                   86.3 mV\r\nTemperature          30.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\nLog #                676\r\n\r\n\r\n'),
(1170,'2024-10-30','11:03:18','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0424103','E4FP24000723','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-10-30 12:03:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.18 pH\r\nmV                   44.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1171,'2024-10-30','11:07:42','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424117','E4FP24000725','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-10-30 12:03:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.72 pH\r\nmV                   71.0 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1172,'2024-10-30','11:14:41','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0324080','E4PP24001785','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','pH (Bulk)',1,'86253','D. Jayapal Reddy','2024-10-30 12:04:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.01 pH\r\nmV                   54.3 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1173,'2024-10-30','11:19:35','90.00000','105.00000','AMPICILLIN FOR INJECTION','pH','AI0124009','E4FP24000724','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Ampicillin pH',1,'86253','D. Jayapal Reddy','2024-10-30 12:05:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.16 pH\r\nmV                   -129.3 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1174,'2024-10-30','11:37:42','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0324080','E4PP24001786','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','pH (Bulk)',1,'86253','D. Jayapal Reddy','2024-10-30 12:05:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.98 pH\r\nmV                   55.8 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1175,'2024-10-30','12:14:21','90.00000','105.00000','0.9% SALINE','NA','RS008/24013','E4SO24000766','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','SALIH',1,'89384','M.V.V. Satish','2024-10-30 14:27:27','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.90 pH\r\nmV                   5.5 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-10-24,08:51\r\n\r\n\r\n'),
(1176,'2024-10-30','12:20:55','90.00000','105.00000','Soyabean casein digest agar','NA','RS022/24008','E4SO24000767','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','SCDAIH',1,'89384','M.V.V. Satish','2024-10-30 14:26:33','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.41 pH\r\nmV                   -24.2 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-10-24,08:51\r\n\r\n\r\n'),
(1177,'2024-10-30','12:27:13','90.00000','105.00000','Sabourarud dextrose agar','NA','RS021/24008','E4SO24000768','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','SDAIH',1,'89384','M.V.V. Satish','2024-10-30 14:24:42','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.61 pH\r\nmV                   81.2 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-10-24,08:51\r\n\r\n\r\n'),
(1178,'2024-10-30','12:33:48','90.00000','105.00000','R2AIH AGAR','NA','RS020/24006','E4SO24000769','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','R2AIH',1,'89384','M.V.V. Satish','2024-10-30 14:25:52','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.13 pH\r\nmV                   -7.6 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-10-24,08:51\r\n\r\n\r\n'),
(1179,'2024-10-30','13:31:00','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005849','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-30 17:13:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.72 pH\r\nmV                   71.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1180,'2024-10-30','13:35:12','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005850','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-30 17:13:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.73 pH\r\nmV                   70.6 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1181,'2024-10-30','13:38:36','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005851','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-10-30 17:14:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.72 pH\r\nmV                   71.3 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1182,'2024-10-30','13:42:05','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005852','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-10-30 17:14:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.76 pH\r\nmV                   68.9 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1183,'2024-10-30','13:46:51','90.00000','105.00000','Purified water','pH','PW/SP05','E4RM24005853','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-30 17:14:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   71.9 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1184,'2024-10-30','13:51:00','90.00000','105.00000','Pure steam','pH','PSC/SP20','E4RM24005854','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'82735','M.D. Babar Qhadri','2024-10-30 17:15:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.78 pH\r\nmV                   67.7 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1185,'2024-10-30','15:22:19','90.00000','105.00000','Glass Vials for pH','pH(Vial-1)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:39:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.74 pH\r\nmV                   246.2 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1186,'2024-10-30','15:25:03','90.00000','105.00000','Glass Vials for pH','pH(Vial-2)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:40:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.42 pH\r\nmV                   264.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1187,'2024-10-30','15:27:15','90.00000','105.00000','Glass Vials for pH','pH(Vial-3)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:40:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.47 pH\r\nmV                   261.4 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1188,'2024-10-30','15:29:42','90.00000','105.00000','Glass Vials for pH','pH(Vial-4)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:40:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.27 pH\r\nmV                   273.4 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1189,'2024-10-30','15:32:37','90.00000','105.00000','Glass Vials for pH','pH(Vial-5)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:41:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.47 pH\r\nmV                   261.8 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1190,'2024-10-30','15:35:22','90.00000','105.00000','Glass Vials for pH','pH(Vial-6)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:41:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.62 pH\r\nmV                   252.9 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1191,'2024-10-30','15:38:17','90.00000','105.00000','Glass Vials for pH','pH(Vial-7)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:41:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.71 pH\r\nmV                   247.1 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1192,'2024-10-30','15:40:36','90.00000','105.00000','Glass Vials for pH','pH(Vial-8)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:42:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.51 pH\r\nmV                   259.0 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1193,'2024-10-30','15:43:17','90.00000','105.00000','Glass Vials for pH','pH(Vial-9)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:42:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.60 pH\r\nmV                   254.5 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1194,'2024-10-30','15:45:44','90.00000','105.00000','Glass Vials for pH','pH(Vial-10)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:42:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.44 pH\r\nmV                   263.2 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1195,'2024-10-30','15:49:21','90.00000','105.00000','Glass Vials for pH','pH(Vial-11)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:43:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.56 pH\r\nmV                   256.2 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1196,'2024-10-30','15:51:30','90.00000','105.00000','Glass Vials for pH','pH(Vial-12)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:43:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.63 pH\r\nmV                   252.1 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1197,'2024-10-30','15:55:14','90.00000','105.00000','Glass Vials for pH','pH(Vial-13)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:43:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.56 pH\r\nmV                   256.1 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1198,'2024-10-30','15:58:21','90.00000','105.00000','Glass Vials for pH','pH(Vial-14)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:44:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.63 pH\r\nmV                   252.5 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1199,'2024-10-30','16:01:21','90.00000','105.00000','Glass Vials for pH','pH(Vial-15)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:44:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.48 pH\r\nmV                   260.8 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1200,'2024-10-30','16:04:15','90.00000','105.00000','Glass Vials for pH','pH(Vial-16)','NA','E4EQ24000388','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:44:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.62 pH\r\nmV                   252.8 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1201,'2024-10-30','16:21:52','90.00000','105.00000','Peptone','NA','RS007/24043','E4SO24000771','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-30 16:35:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.65 pH\r\nmV                   20.6 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-10-24,08:51\r\nLog #                310\r\n\r\n\r\n'),
(1202,'2024-10-30','16:21:08','90.00000','105.00000','Glass Vials for pH','pH(Vial-1)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:57:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.46 pH\r\nmV                   262.6 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\nLog #                704\r\n\r\n\r\n'),
(1203,'2024-10-30','16:30:46','90.00000','105.00000','Glass Vials for pH','pH(Vial-2)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:58:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.61 pH\r\nmV                   253.8 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1204,'2024-10-30','16:33:39','90.00000','105.00000','Glass Vials for pH','pH(Vial-3)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:58:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.10 pH\r\nmV                   283.0 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\nLog #                706\r\n\r\n\r\n'),
(1205,'2024-10-30','16:35:41','90.00000','105.00000','Glass Vials for pH','pH(Vial-4)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:58:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.19 pH\r\nmV                   277.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1206,'2024-10-30','16:38:44','90.00000','105.00000','Glass Vials for pH','pH(Vial-5)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:59:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.31 pH\r\nmV                   271.1 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1207,'2024-10-30','16:43:29','90.00000','105.00000','TAZOBACTAM','NA','PN0024101','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','1',1,'84472','G. Ashok','2024-10-30 18:45:37','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   3.96 pH\r\nmV                   161.9 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        97.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             30-10-24,07:49\r\n\r\n\r\n'),
(1208,'2024-10-30','16:41:04','90.00000','105.00000','Glass Vials for pH','pH(Vial-6)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:59:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.38 pH\r\nmV                   266.9 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\nLog #                709\r\n\r\n\r\n'),
(1209,'2024-10-30','16:48:01','90.00000','105.00000','Glass Vials for pH','pH(Vial-7)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 08:59:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.48 pH\r\nmV                   261.1 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1210,'2024-10-30','16:49:44','90.00000','105.00000','Glass Vials for pH','pH(Vial-8)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:01:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.58 pH\r\nmV                   254.9 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1211,'2024-10-30','16:52:15','90.00000','105.00000','Glass Vials for pH','pH(Vial-9)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:01:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.63 pH\r\nmV                   252.0 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1212,'2024-10-30','16:54:56','90.00000','105.00000','Glass Vials for pH','pH(Vial-10)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:02:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.72 pH\r\nmV                   246.9 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1213,'2024-10-30','16:57:13','90.00000','105.00000','Glass Vials for pH','pH(Vial-11)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:02:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.74 pH\r\nmV                   245.4 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1214,'2024-10-30','16:59:42','90.00000','105.00000','Glass Vials for pH','pH(Vial-12)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:02:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.40 pH\r\nmV                   265.2 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1215,'2024-10-30','17:02:59','90.00000','105.00000','Glass Vials for pH','pH(Vial-13)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:02:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.51 pH\r\nmV                   258.5 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1216,'2024-10-30','17:05:04','90.00000','105.00000','Glass Vials for pH','pH(Vial-14)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:06:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.61 pH\r\nmV                   252.9 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1217,'2024-10-30','17:07:48','90.00000','105.00000','Glass Vials for pH','pH(Vial-15)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:07:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.65 pH\r\nmV                   250.7 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1218,'2024-10-30','17:10:18','90.00000','105.00000','Glass Vials for pH','pH(Vial-16)','NA','E4EQ24000393','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:08:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.86 pH\r\nmV                   238.6 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\nLog #                719\r\n\r\n\r\n'),
(1219,'2024-10-30','17:28:09','90.00000','105.00000','TAZOBACTAM','NA','PN0024101','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','1',1,'84472','G. Ashok','2024-10-30 18:45:58','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.14 pH\r\nmV                   152.1 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        97.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             30-10-24,07:49\r\n\r\n\r\n'),
(1220,'2024-10-30','17:42:22','90.00000','105.00000','Sabourarud dextrose agar','NA','RS021/24008','E4SO24000768','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SDAIH',1,'85152','V. Chandra Shekara Achari','2024-11-01 08:04:41','FOR REVIEW','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.61 pH\r\nmV                   80.6 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-10-24,08:51\r\nLog #                311\r\n\r\n\r\n'),
(1221,'2024-10-30','17:46:28','90.00000','105.00000','Glass Vials for pH','pH(Vial-1)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:14:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.19 pH\r\nmV                   277.2 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1222,'2024-10-30','17:49:19','90.00000','105.00000','Glass Vials for pH','pH(Vial-2)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:15:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.31 pH\r\nmV                   270.5 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1223,'2024-10-30','17:53:36','90.00000','105.00000','Glass Vials for pH','pH(Vial-3)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:16:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.46 pH\r\nmV                   261.9 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1224,'2024-10-30','17:57:06','90.00000','105.00000','Glass Vials for pH','pH(Vial-4)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:16:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.89 pH\r\nmV                   237.0 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1225,'2024-10-30','18:00:07','90.00000','105.00000','Glass Vials for pH','pH(Vial-5)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:17:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.52 pH\r\nmV                   259.0 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1226,'2024-10-30','17:49:03','90.00000','105.00000','R2AIH AGAR','NA','RS020/24006','E4SO24000769','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','R2AIH',1,'85152','V. Chandra Shekara Achari','2024-11-01 08:07:02','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -1.8 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-10-24,08:51\r\nLog #                312\r\n\r\n\r\n'),
(1227,'2024-10-30','18:02:36','90.00000','105.00000','Glass Vials for pH','pH(Vial-6)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:17:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.61 pH\r\nmV                   253.2 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\nLog #                725\r\n\r\n\r\n'),
(1228,'2024-10-30','18:06:28','90.00000','105.00000','Soyabean casein digest agar','NA','RS022/24008','E4SO24000767','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDAIH',1,'85152','V. Chandra Shekara Achari','2024-11-01 08:08:08','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.46 pH\r\nmV                   -26.9 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-10-24,08:51\r\nLog #                313\r\n\r\n\r\n'),
(1229,'2024-10-30','18:07:16','90.00000','105.00000','Glass Vials for pH','pH(Vial-7)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:20:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.79 pH\r\nmV                   242.4 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\nLog #                726\r\n\r\n\r\n'),
(1230,'2024-10-30','18:14:40','90.00000','105.00000','0.9% SALINE','NA','RS008/24013','E4SO24000766','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SALIH',1,'85152','V. Chandra Shekara Achari','2024-11-01 08:09:00','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.37 pH\r\nmV                   36.8 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-10-24,08:51\r\n\r\n\r\n'),
(1231,'2024-10-30','18:16:48','90.00000','105.00000','TAZOBACTAM','NA','PN0024101','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','1',1,'84472','G. Ashok','2024-10-30 18:46:11','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.61 pH\r\nmV                   124.4 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        97.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             30-10-24,07:49\r\n\r\n\r\n'),
(1232,'2024-10-30','18:12:50','90.00000','105.00000','Glass Vials for pH','pH(Vial-8)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:22:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.92 pH\r\nmV                   234.9 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\nLog #                727\r\n\r\n\r\n'),
(1233,'2024-10-30','18:19:26','90.00000','105.00000','Glass Vials for pH','pH(Vial-9)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:22:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.55 pH\r\nmV                   256.9 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1234,'2024-10-30','18:23:15','90.00000','105.00000','Glass Vials for pH','pH(Vial-10)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:23:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.37 pH\r\nmV                   267.0 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1235,'2024-10-30','18:25:40','90.00000','105.00000','Glass Vials for pH','pH(Vial-11)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:24:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.69 pH\r\nmV                   247.9 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1236,'2024-10-30','18:28:35','90.00000','105.00000','Glass Vials for pH','pH(Vial-12)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:24:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.46 pH\r\nmV                   261.6 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1237,'2024-10-30','18:31:46','90.00000','105.00000','Glass Vials for pH','pH(Vial-13)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:25:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.92 pH\r\nmV                   234.8 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1238,'2024-10-30','18:34:27','90.00000','105.00000','Glass Vials for pH','pH(Vial-14)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:26:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.44 pH\r\nmV                   263.2 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1239,'2024-10-30','18:37:34','90.00000','105.00000','Glass Vials for pH','pH(Vial-15)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:26:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.45 pH\r\nmV                   262.9 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1240,'2024-10-30','18:39:35','90.00000','105.00000','Glass Vials for pH','pH(Vial-16)','NA','E4EQ24000394','Miscellaneous pH Test','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','Miscellaneous pH Tests',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:27:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   2.77 pH\r\nmV                   243.9 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1241,'2024-10-30','21:09:10','90.00000','105.00000','PIPERACILLIN','NA','PN0024101','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'84472','G. Ashok','2024-10-30 21:19:19','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.41 pH\r\nmV                   20.6 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        97.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             30-10-24,07:49\r\n\r\n\r\n'),
(1242,'2024-10-30','21:32:23','90.00000','105.00000','Peptone','NA','RS007/24043','E4SO24000771','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'85152','V. Chandra Shekara Achari','2024-11-01 08:10:29','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.81 pH\r\nmV                   10.8 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-10-24,08:51\r\n\r\n\r\n'),
(1243,'2024-10-30','22:43:05','90.00000','105.00000','Media Fill','pH','LY24005','E4AMV24000191','ASEPTIC MEDIA FILL','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','Media fill pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 09:28:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.41 pH\r\nmV                   -27.6 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_101\r\nLast Cal             30-10-24,07:44\r\n\r\n\r\n'),
(1244,'2024-10-31','01:30:33','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','PN0024101','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','3',1,'88515','Prashanth B','2024-10-31 03:08:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.27 pH\r\nmV                   28.8 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        97.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             31-10-24,00:50\r\nLog #                225\r\n\r\n\r\n'),
(1245,'2024-11-01','03:27:10','90.00000','105.00000','TAZOBACTAM','NA','PN0024102','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','1',1,'95681','T. Sivanarayana','2024-11-01 05:50:44','Revirwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   3.87 pH\r\nmV                   167.7 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        97.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             01-11-24,03:05\r\n\r\n\r\n'),
(1246,'2024-11-01','04:10:49','90.00000','105.00000','TAZOBACTAM','NA','PN0024102','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','1',1,'95681','T. Sivanarayana','2024-11-01 05:51:08','Revirwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.13 pH\r\nmV                   151.3 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        97.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             01-11-24,03:05\r\n\r\n\r\n'),
(1247,'2024-11-01','05:05:26','90.00000','105.00000','TAZOBACTAM','NA','PN0024102','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','1',1,'95681','T. Sivanarayana','2024-11-01 05:51:59','Revirwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.30 pH\r\nmV                   141.6 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        97.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             01-11-24,03:05\r\n\r\n\r\n'),
(1248,'2024-11-01','05:51:38','90.00000','105.00000','PIPERACILLIN','NA','PN0024102','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-11-01 05:59:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.10 pH\r\nmV                   38.7 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        97.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             01-11-24,03:05\r\n\r\n\r\n'),
(1249,'2024-11-01','06:36:40','90.00000','105.00000','PIPERACILLIN','NA','PN0024102','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-11-01 14:18:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.05 pH\r\nmV                   38.6 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        97.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             01-11-24,03:05\r\n\r\n\r\n'),
(1250,'2024-11-01','07:10:45','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005721','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','1.68 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 07:45:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   304.7 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1251,'2024-11-01','07:15:59','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005721','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','4.01pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 07:45:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.05 pH\r\nmV                   164.5 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\nLog #                738\r\n\r\n\r\n'),
(1252,'2024-11-01','07:19:30','90.00000','105.00000','pH Buffer','verification','Nil','E4INS24005721','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','7.00pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 07:46:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -8.0 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1253,'2024-11-01','07:28:02','90.00000','105.00000','PIPERACILLIN','NA','PN0024102','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-11-01 14:17:26','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.47 pH\r\nmV                   17.4 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        97.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             01-11-24,03:05\r\n\r\n\r\n'),
(1254,'2024-11-01','07:22:51','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005721','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','10.01 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 07:46:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.98 pH\r\nmV                   -182.2 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\nLog #                740\r\n\r\n\r\n'),
(1255,'2024-11-01','07:34:07','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005721','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','12.46 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 07:47:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.46 pH\r\nmV                   -327.5 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1256,'2024-11-01','08:22:25','90.00000','105.00000','Piperacillin and Tazobactam for Injection','IDENTIFICATION','2449201773','E4RM24005816','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,0,101,'pH','Mobile phase pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 14:04:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.49 pH\r\nmV                   82.4 mV\r\nTemperature          31.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1257,'2024-11-01','09:18:15','90.00000','105.00000','Piperacillin and Tazobactam for injection','MOBILE PHASE_A','TZ0424025A','E4CS24000502','Diluent compatibility','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,1,101,'pH','Piper Tazo',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 14:06:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.55 pH\r\nmV                   77.5 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1258,'2024-11-01','09:36:26','90.00000','105.00000','Piperacillin and Tazobactam for injection','Mobile phase B','TZ0424025A','E4CS24000502','Diluent compatibility','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,1,101,'pH','Piper Tazo',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 14:06:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   77.8 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1259,'2024-11-01','12:18:07','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','PN0024102','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205224','K. Uma Mahesh',2,0,104,'pH','3',1,'88629','Raghuram Reddy','2024-11-01 14:17:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.31 pH\r\nmV                   26.3 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        97.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             01-11-24,03:05\r\n\r\n\r\n'),
(1260,'2024-11-01','14:11:18','90.00000','105.00000','Ampcillin for injection','RS BY HPLC MP A','AP0224001A','E4STFP24000647','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,1,101,'pH','Ampi',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 14:35:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.97 pH\r\nmV                   110.8 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\nLog #                745\r\n\r\n\r\n'),
(1261,'2024-11-01','14:21:04','90.00000','105.00000','Piperacillin and Tazobactam For injection','UOD','PT0424117','E4FP24000725','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205232','K. Narender Reddy',2,1,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 14:38:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.49 pH\r\nmV                   81.1 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1262,'2024-11-01','14:40:24','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS','TA0223001A','E4STFP24000657','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205043','T.V. Subbaiah',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-11-01 15:30:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.04 pH\r\nmV                   48.2 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1263,'2024-11-01','15:01:54','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0024102','E4PP24001844','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','pH (Bulk)',1,'90778','B. Bhaskar','2024-11-01 15:28:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.63 pH\r\nmV                   72.2 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1264,'2024-11-01','15:13:10','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0024102','E4PP24001841','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','pH (Bulk)',1,'90778','B. Bhaskar','2024-11-01 15:29:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.29 pH\r\nmV                   150.6 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1265,'2024-11-01','15:19:30','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0024102','E4PP24001843','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','pH (Bulk)',1,'90778','B. Bhaskar','2024-11-01 15:29:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.13 pH\r\nmV                   43.0 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1266,'2024-11-01','15:41:26','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0024101','E4PP24001788','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-11-01 17:02:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.62 pH\r\nmV                   131.6 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1267,'2024-11-01','16:04:31','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0024101','E4PP24001790','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-11-01 17:03:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.06 pH\r\nmV                   47.4 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1268,'2024-11-01','16:11:31','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0024101','E4PP24001791','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-11-01 17:03:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.05 pH\r\nmV                   47.8 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1269,'2024-11-01','16:17:33','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0224001','E4PP24001795','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-11-01 17:03:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.56 pH\r\nmV                   76.9 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1270,'2024-11-01','16:23:16','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0224001','E4PP24001821','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-11-01 17:03:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.72 pH\r\nmV                   67.1 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1271,'2024-11-01','16:30:17','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0224001','E4PP24001839','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-11-01 17:04:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   69.2 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1272,'2024-11-01','16:42:13','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0024101','E4IP24001298','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-11-01 17:01:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.97 pH\r\nmV                   52.4 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1273,'2024-11-01','17:13:23','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0024102','E4IP24001305','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','pH (Bulk)',1,'205208','S. Siddhartha Kumar Reddy','2024-11-01 17:22:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.10 pH\r\nmV                   45.1 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1274,'2024-11-01','17:41:12','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005876','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-11-01 18:03:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   68.5 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1275,'2024-11-01','17:44:19','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005877','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-11-01 18:03:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   65.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1276,'2024-11-01','17:47:20','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005878','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90778','B. Bhaskar','2024-11-01 18:03:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.72 pH\r\nmV                   67.2 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1277,'2024-11-01','17:49:12','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005879','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90778','B. Bhaskar','2024-11-01 18:04:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   65.3 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1278,'2024-11-01','17:52:21','90.00000','105.00000','Purified water','pH','PW/SP34','E4RM24005880','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-11-01 18:04:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   65.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1279,'2024-11-01','17:54:08','90.00000','105.00000','Purified water','pH','PW/SP36','E4RM24005881','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-11-01 18:04:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.78 pH\r\nmV                   63.3 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1280,'2024-11-01','17:57:14','90.00000','105.00000','Purified water','pH','PW/SP33','E4RM24005882','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90778','B. Bhaskar','2024-11-01 18:05:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.68 pH\r\nmV                   69.0 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1281,'2024-11-01','18:00:17','90.00000','105.00000','Water For Injection','pH','WFI/SP40','E4RM24005883','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90778','B. Bhaskar','2024-11-01 18:05:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   65.5 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1282,'2024-11-01','18:02:50','90.00000','105.00000','Water For Injection','pH','WFI/SP33','E4RM24005884','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'90778','B. Bhaskar','2024-11-01 18:20:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   65.2 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1283,'2024-11-01','18:05:29','90.00000','105.00000','Pure steam','pH','PSC/SP21','E4RM24005885','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'90778','B. Bhaskar','2024-11-01 18:20:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.77 pH\r\nmV                   63.8 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_102\r\nLast Cal             01-11-24,07:06\r\n\r\n\r\n'),
(1284,'2024-11-02','07:02:11','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005736','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','1.68 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 07:22:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.67 pH\r\nmV                   303.7 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1285,'2024-11-02','07:04:41','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005736','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','4.01pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 07:22:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   165.9 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1286,'2024-11-02','07:07:01','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005736','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','7.00pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 07:22:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -8.9 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1287,'2024-11-02','07:10:09','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005736','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','10.01 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 07:23:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.96 pH\r\nmV                   -181.7 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\nLog #                772\r\n\r\n\r\n'),
(1288,'2024-11-02','07:13:13','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005736','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','12.46 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 07:23:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -328.8 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1289,'2024-11-02','07:50:32','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005746','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-11-02 09:42:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.65 pH\r\nmV                   309.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             02-11-24,07:47\r\n\r\n\r\n'),
(1290,'2024-11-02','07:53:42','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005746','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-11-02 09:42:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   172.1 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             02-11-24,07:47\r\n\r\n\r\n'),
(1291,'2024-11-02','07:56:32','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005746','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-11-02 09:43:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.98 pH\r\nmV                   -1.1 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             02-11-24,07:47\r\n\r\n\r\n'),
(1292,'2024-11-02','08:00:29','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005746','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-11-02 09:43:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -176.2 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             02-11-24,07:47\r\n\r\n\r\n'),
(1293,'2024-11-02','08:03:42','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005746','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-11-02 09:44:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.46 pH\r\nmV                   -319.4 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             02-11-24,07:47\r\n\r\n\r\n'),
(1294,'2024-11-02','10:36:26','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','ASSAY BY HPLC','PN0024102','E4PP24001845','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,1,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 12:21:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   81.5 mV\r\nTemperature          31.9 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1295,'2024-11-02','12:00:46','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0224002','E4PP24001848','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 12:22:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.98 pH\r\nmV                   51.3 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1296,'2024-11-02','12:07:56','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0224002','E4PP24001882','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 12:23:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.94 pH\r\nmV                   54.0 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1297,'2024-11-02','12:17:46','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24019','E4SO24000778','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SCDBIH',1,'205143','Ch. Prasad','2024-11-02 12:38:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.32 pH\r\nmV                   -20.6 mV\r\nTemperature          26.7 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             02-11-24,07:47\r\n\r\n\r\n'),
(1298,'2024-11-02','12:21:54','90.00000','105.00000','SCDM WITH 4%Tween80 &0.5 Soya lecthin','NA','RS011/24009','E4SO24000779','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','NSCDBIH',1,'205143','Ch. Prasad','2024-11-02 12:39:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.30 pH\r\nmV                   -19.7 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             02-11-24,07:47\r\n\r\n\r\n'),
(1299,'2024-11-02','12:24:54','90.00000','105.00000','SCDM with 1% Tween 80','NA','RS034/24008','E4SO24000780','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','NSSCDBSIH',1,'205143','Ch. Prasad','2024-11-02 12:40:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.29 pH\r\nmV                   -18.8 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             02-11-24,07:47\r\n\r\n\r\n'),
(1300,'2024-11-02','13:17:55','90.00000','105.00000','potable water pH','pH','PR/SP11','E4RM24005897','potable water ','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','potable water',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 14:20:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.66 pH\r\nmV                   -47.0 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1301,'2024-11-02','13:22:09','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005890','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 14:20:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.63 pH\r\nmV                   71.9 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1302,'2024-11-02','13:25:47','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005891','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 14:21:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.60 pH\r\nmV                   73.5 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1303,'2024-11-02','13:28:34','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005892','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 14:22:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   76.8 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1304,'2024-11-02','13:31:45','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005893','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 14:23:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   79.2 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1305,'2024-11-02','13:37:34','90.00000','105.00000','Water For Injection','pH','WFI/SP34','E4RM24005894','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 14:24:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.43 pH\r\nmV                   83.9 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1306,'2024-11-02','13:40:47','90.00000','105.00000','Purified water','pH','PW/SP20','E4RM24005895','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 14:25:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.32 pH\r\nmV                   90.1 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1307,'2024-11-02','13:43:39','90.00000','105.00000','Water For Injection','pH','WFI/SP21','E4RM24005896','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 14:26:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.36 pH\r\nmV                   87.7 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1308,'2024-11-02','13:56:43','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','PN0224002','E4PP24001909','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-11-02 14:19:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   60.4 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1309,'2024-11-02','14:35:46','90.00000','105.00000','Ampicillin for injection','pH','AP0224021','E4AQ24000057','Analyst Qualification','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','AQ',1,'86253','D. Jayapal Reddy','2024-11-02 14:59:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.20 pH\r\nmV                   -137.1 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1310,'2024-11-02','15:21:46','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24019','E4SO24000778','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDBIH',1,'205143','Ch. Prasad','2024-11-02 17:19:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.38 pH\r\nmV                   -24.3 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             02-11-24,07:47\r\n\r\n\r\n'),
(1311,'2024-11-02','15:24:07','90.00000','105.00000','SCDM WITH 4%Tween80 &0.5 Soya lecthin','NA','RS011/24009','E4SO24000779','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','NSCDBIH',1,'205143','Ch. Prasad','2024-11-02 17:20:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.35 pH\r\nmV                   -22.8 mV\r\nTemperature          26.5 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             02-11-24,07:47\r\n\r\n\r\n'),
(1312,'2024-11-02','15:26:39','90.00000','105.00000','SCDM with 1% Tween 80','NA','RS034/24008','E4SO24000780','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','NSSCDBSIH',1,'205143','Ch. Prasad','2024-11-02 17:20:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.39 pH\r\nmV                   -24.7 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             02-11-24,07:47\r\n\r\n\r\n'),
(1313,'2024-11-02','16:29:41','90.00000','105.00000','Tryptone soya agar TSA with 0.05%soya lecithin&0.5% tween 80 and beta lactamase 55mm plates(RODAC)','NA','RS039/24014','E4SO24000782','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','RUTSA',1,'205143','Ch. Prasad','2024-11-02 17:21:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.22 pH\r\nmV                   -14.9 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             02-11-24,07:47\r\n\r\n\r\n'),
(1314,'2024-11-02','16:33:43','90.00000','105.00000','Soyabean casean digest agar SCDA With 0.05%soya lecithin 05% tween 80 and beta lactamase (Finger dab)','NA','RS041/24007','E4SO24000783','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','RUSCDAF',1,'205143','Ch. Prasad','2024-11-02 17:21:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.20 pH\r\nmV                   -13.9 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             02-11-24,07:47\r\n\r\n\r\n'),
(1315,'2024-11-02','16:39:52','90.00000','105.00000','Soyabean casein digest agar (SCDA)with beta-lactamase','NA','RS040/24008','E4SO24000784','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','RUSCDA',1,'205143','Ch. Prasad','2024-11-02 17:22:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.21 pH\r\nmV                   -14.5 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             02-11-24,07:47\r\n\r\n\r\n'),
(1316,'2024-11-02','16:46:12','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS BY HPLC','PN0322007A','E4STFP24000671','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,1,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-11-02 16:53:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.03 pH\r\nmV                   49.0 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             02-11-24,06:57\r\n\r\n\r\n'),
(1317,'2024-11-03','02:34:04','90.00000','105.00000','TAZOBACTAM','NA','TT0024013','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','1',1,'95681','T. Sivanarayana','2024-11-03 06:11:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.41 pH\r\nmV                   134.9 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        96.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             03-11-24,00:32\r\n\r\n\r\n'),
(1318,'2024-11-03','08:08:47','90.00000','105.00000','PIPERACILLIN','NA','TT0024013','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-03 14:18:39','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.96 pH\r\nmV                   45.9 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        96.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             03-11-24,00:32\r\n\r\n\r\n'),
(1319,'2024-11-03','08:48:37','90.00000','105.00000','PIPERACILLIN','NA','TT0024013','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-03 14:18:58','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.99 pH\r\nmV                   44.0 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        96.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             03-11-24,00:32\r\n\r\n\r\n'),
(1320,'2024-11-03','09:29:14','90.00000','105.00000','PIPERACILLIN','NA','TT0024013','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-03 14:19:15','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.34 pH\r\nmV                   24.8 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        96.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             03-11-24,00:32\r\n\r\n\r\n'),
(1321,'2024-11-03','10:09:14','90.00000','105.00000','PIPERACILLIN','NA','TT0024013','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-03 14:19:34','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.44 pH\r\nmV                   18.6 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        96.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             03-11-24,00:32\r\n\r\n\r\n'),
(1322,'2024-11-03','13:16:11','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TT0024013','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','3',1,'84472','G. Ashok','2024-11-03 14:19:58','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.25 pH\r\nmV                   29.2 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        96.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             03-11-24,00:32\r\n\r\n\r\n'),
(1323,'2024-11-03','13:31:13','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TT0024013','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','3',1,'84472','G. Ashok','2024-11-03 14:20:37','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.27 pH\r\nmV                   27.8 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        96.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             03-11-24,00:32\r\n\r\n\r\n'),
(1324,'2024-11-04','09:14:56','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005772','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-11-04 11:15:07','Approved','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.70 pH\r\nmV                   308.3 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1325,'2024-11-04','09:20:16','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005772','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-11-04 11:15:23','Approved','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.05 pH\r\nmV                   170.6 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1326,'2024-11-04','09:24:21','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005772','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-11-04 11:15:31','Approved','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -3.2 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1327,'2024-11-04','09:26:31','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005772','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-11-04 11:15:38','Approved','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -177.7 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1328,'2024-11-04','09:28:45','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005772','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-11-04 11:15:45','Approved','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -321.3 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\nLog #                334\r\n\r\n\r\n'),
(1329,'2024-11-04','11:46:43','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC MP A','TZ0424103','E4FP24000723','Finished Product','QC-PHMTZ-008','CW1-12012','1','NULL','87697','M. Raghavendr Reddy',2,0,103,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 14:30:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   83.7 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\nLog #                335\r\n\r\n\r\n'),
(1330,'2024-11-04','11:56:50','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC MP B','TZ0424103','E4FP24000723','Finished Product','QC-PHMTZ-008','CW1-12012','1','NULL','87697','M. Raghavendr Reddy',2,0,103,'pH','Pipertazoinj',1,'90778','B. Bhaskar','2024-11-04 14:12:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   84.4 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\nLog #                336\r\n\r\n\r\n'),
(1331,'2024-11-04','11:56:50','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC MP B','TZ0424103','E4FP24000723','Finished Product','QC-PHMTZ-008','CW1-12012','1','NULL','87697','M. Raghavendr Reddy',2,0,103,'pH','Pipertazoinj',1,'90778','B. Bhaskar','2024-11-04 14:13:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   84.4 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\nLog #                336\r\n\r\n\r\n'),
(1332,'2024-11-04','12:13:00','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS','PT0424117','E4FP24000725','Finished Product','QC-PHMTZ-008','CW1-12012','1','NULL','205232','K. Narender Reddy',2,0,103,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 14:31:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.02 pH\r\nmV                   55.6 mV\r\nTemperature          28.1 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1333,'2024-11-04','12:30:31','90.00000','105.00000','PIPERACILLIN MONOHYDRATE','RS Mobile phase A','2449103414','E4AQR24000014','Raw Material','QC-PHMTZ-008','CW1-12012','1','NULL','205189','P. Ravi Kumar',2,1,103,'pH','RS Mobile phase',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 14:32:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   85.7 mV\r\nTemperature          33.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1334,'2024-11-04','12:35:07','90.00000','105.00000','PIPERACILLIN MONOHYDRATE','RS MOBILE PHASE B','2449103414','E4AQR24000014','Raw Material','QC-PHMTZ-008','CW1-12012','1','NULL','205189','P. Ravi Kumar',2,1,103,'pH','RS Mobile phase',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 14:33:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   86.0 mV\r\nTemperature          33.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\nLog #                339\r\n\r\n\r\n'),
(1335,'2024-11-04','12:47:13','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TT0024013','E4IP24001310','BULK','QC-PHMTZ-008','CW1-12012','1','NULL','205210','B. Anitha',2,1,103,'pH','pH (Bulk)',1,'90778','B. Bhaskar','2024-11-04 12:57:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.19 pH\r\nmV                   45.2 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\nLog #                340\r\n\r\n\r\n'),
(1336,'2024-11-04','13:50:16','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS_II MP_A','TZ0424102','E4FP24000721','Finished Product','QC-PHMTZ-008','CW1-12012','1','NULL','89180','R. Harish Reddy',2,0,103,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 14:35:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   83.7 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1337,'2024-11-04','14:02:01','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS-II MP_B','TZ0424102','E4FP24000721','Finished Product','QC-PHMTZ-008','CW1-12012','1','NULL','89180','R. Harish Reddy',2,0,103,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 14:35:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   83.7 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1338,'2024-11-04','14:09:39','90.00000','105.00000','Peptone','RS007/24044','NA','E4SO24000788','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-11-04 14:26:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.75 pH\r\nmV                   12.6 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1339,'2024-11-04','14:19:31','90.00000','105.00000','0.9% SALINE','RS008/24014','NA','E4SO24000790','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SALIH',1,'205143','Ch. Prasad','2024-11-04 14:27:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.96 pH\r\nmV                   0.0 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1340,'2024-11-04','14:57:12','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005926','Purified water','QC-PHMTZ-008','CW1-12012','1','NULL','205066','Sreeram Jakkaraiah',2,1,103,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 16:10:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.91 pH\r\nmV                   61.8 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1341,'2024-11-04','15:01:05','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005927','Purified water','QC-PHMTZ-008','CW1-12012','1','NULL','205066','Sreeram Jakkaraiah',2,1,103,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 16:11:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   66.3 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1342,'2024-11-04','15:04:25','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005928','Water For Injection','QC-PHMTZ-008','CW1-12012','1','NULL','205066','Sreeram Jakkaraiah',2,1,103,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 16:11:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.77 pH\r\nmV                   69.5 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1343,'2024-11-04','15:07:30','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005929','Water For Injection','QC-PHMTZ-008','CW1-12012','1','NULL','205066','Sreeram Jakkaraiah',2,1,103,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 16:12:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   71.5 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1344,'2024-11-04','15:10:52','90.00000','105.00000','Purified water','pH','PW/SP03','E4RM24005930','Purified water','QC-PHMTZ-008','CW1-12012','1','NULL','205066','Sreeram Jakkaraiah',2,1,103,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 16:12:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   73.4 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\nLog #                349\r\n\r\n\r\n'),
(1345,'2024-11-04','15:15:21','90.00000','105.00000','Pure steam','pH','PSC/SP09','E4RM24005931','Pure steam Condensate','QC-PHMTZ-008','CW1-12012','1','NULL','205066','Sreeram Jakkaraiah',2,1,103,'pH','pH_Pure Steam',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 16:13:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   73.9 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1346,'2024-11-04','15:18:54','90.00000','105.00000','Purified water','pH','PW/SP38','E4RM24005932','Purified water','QC-PHMTZ-008','CW1-12012','1','NULL','205066','Sreeram Jakkaraiah',2,1,103,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 16:13:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   74.4 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1347,'2024-11-04','15:21:49','90.00000','105.00000','Pure steam','pH','PSC/SP07','E4RM24005933','Pure steam Condensate','QC-PHMTZ-008','CW1-12012','1','NULL','205066','Sreeram Jakkaraiah',2,1,103,'pH','pH_Pure Steam',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 16:14:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.66 pH\r\nmV                   75.8 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1348,'2024-11-04','15:33:50','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005762','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','1.68 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 17:23:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   303.4 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             04-11-24,15:22\r\n\r\n\r\n'),
(1349,'2024-11-04','15:36:46','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005762','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','4.01pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 17:23:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   165.3 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             04-11-24,15:22\r\n\r\n\r\n'),
(1350,'2024-11-04','15:39:18','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005762','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','7.00pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 17:23:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -9.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             04-11-24,15:22\r\n\r\n\r\n'),
(1351,'2024-11-04','15:44:33','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005762','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','10.01 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 17:24:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.05 pH\r\nmV                   -185.2 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             04-11-24,15:22\r\n\r\n\r\n'),
(1352,'2024-11-04','15:46:52','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005762','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','12.46 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 17:24:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.49 pH\r\nmV                   -328.0 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             04-11-24,15:22\r\n\r\n\r\n'),
(1353,'2024-11-04','16:03:56','90.00000','105.00000','TAZOBACTAM','NA','TT0024014','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','1',1,'88515','Prashanth B','2024-11-04 17:42:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.31 pH\r\nmV                   140.9 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        96.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,12:33\r\n\r\n\r\n'),
(1354,'2024-11-04','16:17:49','90.00000','105.00000','Piperacillin and Tazobactam For injection','Assay','PT0424117','E4FP24000725','Finished Product','QC-PHMTZ-008','CW1-12012','1','NULL','205119','S.M. Subhani',2,1,103,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 17:26:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   87.9 mV\r\nTemperature          32.1 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1355,'2024-11-04','18:06:10','90.00000','105.00000','Flucloxacillin Sodium','NA','2305103726','E4RM24005898','Raw Material','QC-PHMTZ-008','CW1-12012','1','NULL','205048','P. Nagesh',2,0,103,'pH','5.0 Phosphate Buffer',1,'205208','S. Siddhartha Kumar Reddy','2024-11-04 18:19:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.02 pH\r\nmV                   115.2 mV\r\nTemperature          28.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\nLog #                354\r\n\r\n\r\n'),
(1356,'2024-11-04','18:55:12','90.00000','105.00000','Peptone','NA','RS007/24044','E4SO24000788','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-11-05 09:46:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.92 pH\r\nmV                   2.5 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1357,'2024-11-04','19:01:27','90.00000','105.00000','0.9% SALINE','NA','RS008/24014','E4SO24000790','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SALIH',1,'205143','Ch. Prasad','2024-11-05 09:48:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.83 pH\r\nmV                   8.0 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,09:11\r\n\r\n\r\n'),
(1358,'2024-11-04','19:20:17','90.00000','105.00000','PIPERACILLIN','NA','TT0024014','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-11-04 22:20:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.87 pH\r\nmV                   51.2 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        96.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,12:33\r\n\r\n\r\n'),
(1359,'2024-11-04','20:05:30','90.00000','105.00000','PIPERACILLIN','NA','TT0024014','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-11-04 22:20:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.91 pH\r\nmV                   48.7 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        96.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,12:33\r\n\r\n\r\n'),
(1360,'2024-11-04','20:50:16','90.00000','105.00000','PIPERACILLIN','NA','TT0024014','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-11-04 22:20:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   27.2 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        96.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,12:33\r\n\r\n\r\n'),
(1361,'2024-11-04','21:35:18','90.00000','105.00000','PIPERACILLIN','NA','TT0024014','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-11-04 22:21:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.37 pH\r\nmV                   22.5 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        96.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             04-11-24,12:33\r\n\r\n\r\n'),
(1362,'2024-11-05','07:15:16','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005792','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','1.68 pH',1,'90778','B. Bhaskar','2024-11-05 07:48:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.67 pH\r\nmV                   303.0 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1363,'2024-11-05','07:19:02','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005792','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','4.01pH',1,'90778','B. Bhaskar','2024-11-05 07:48:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   165.2 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1364,'2024-11-05','07:23:07','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005792','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','7.00pH',1,'90778','B. Bhaskar','2024-11-05 07:49:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -10.3 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1365,'2024-11-05','07:26:38','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005792','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','10.01 pH',1,'90778','B. Bhaskar','2024-11-05 07:49:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.03 pH\r\nmV                   -186.5 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1366,'2024-11-05','07:30:12','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005792','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','12.46 pH',1,'90778','B. Bhaskar','2024-11-05 07:49:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.42 pH\r\nmV                   -327.2 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1367,'2024-11-05','07:55:21','90.00000','105.00000','Sensor Calibration','NA','NA','E4INS24005797','Temperature Sensor','QC-PHMTZ-007','CX1-15733','1','NULL','87624','N. Nani babu',2,1,101,'pH','Purified Water',1,'90778','B. Bhaskar','2024-11-05 08:09:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.49 pH\r\nmV                   79.3 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1368,'2024-11-05','09:01:53','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005799','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-11-05 09:52:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.64 pH\r\nmV                   311.5 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             05-11-24,08:53\r\nLog #                357\r\n\r\n\r\n'),
(1369,'2024-11-05','09:05:18','90.00000','105.00000','pH Buffer','NA','pH verification','E4SO24005799','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-11-05 09:52:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   172.8 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             05-11-24,08:53\r\nLog #                358\r\n\r\n\r\n'),
(1370,'2024-11-05','09:09:26','90.00000','105.00000','pH Buffer','NA','pH verification','E4SO24005799','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-11-05 09:53:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -1.4 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             05-11-24,08:53\r\nLog #                359\r\n\r\n\r\n'),
(1371,'2024-11-05','08:53:24','90.00000','105.00000','Flucloxacillin Sodium','Assay pH','2305103726','E4RM24005898','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','5.0 Phosphate Buffer',1,'90778','B. Bhaskar','2024-11-05 09:29:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.00 pH\r\nmV                   110.1 mV\r\nTemperature          29.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\nLog #                799\r\n\r\n\r\n'),
(1372,'2024-11-05','09:14:28','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005799','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-11-05 09:53:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.03 pH\r\nmV                   -177.2 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             05-11-24,08:53\r\nLog #                360\r\n\r\n\r\n'),
(1373,'2024-11-05','09:21:06','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005799','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-11-05 09:53:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.48 pH\r\nmV                   -319.4 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             05-11-24,08:53\r\n\r\n\r\n'),
(1374,'2024-11-05','09:58:48','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TT0024014','E4IP24001316','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','pH (Bulk)',1,'90778','B. Bhaskar','2024-11-05 10:11:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.02 pH\r\nmV                   48.6 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1375,'2024-11-05','11:14:02','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS','PT0424117','E4FP24000725','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205232','K. Narender Reddy',2,0,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-11-05 14:20:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.00 pH\r\nmV                   50.6 mV\r\nTemperature          29.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1376,'2024-11-05','11:36:59','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0322009A','E4STFP24000677','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-11-05 12:37:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.61 pH\r\nmV                   72.5 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1377,'2024-11-05','11:40:17','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0322010A','E4STFP24000678','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-11-05 12:39:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.79 pH\r\nmV                   62.1 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1378,'2024-11-05','11:48:36','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TA0223002A','E4STFP24000679','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-11-05 12:40:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.55 pH\r\nmV                   75.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1379,'2024-11-05','11:58:44','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TA0223002A','E4STFP24000680','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-11-05 12:40:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.48 pH\r\nmV                   80.2 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1380,'2024-11-05','12:02:34','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TA0223004A','E4STFP24000681','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-11-05 12:40:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.64 pH\r\nmV                   70.0 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1381,'2024-11-05','12:06:35','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TA0223004A','E4STFP24000682','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-11-05 12:41:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   77.5 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1382,'2024-11-05','12:14:04','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PC0422001A','E4STFP24000685','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-11-05 12:41:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.33 pH\r\nmV                   88.9 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1383,'2024-11-05','12:17:51','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TE0424001A','E4STFP24000689','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'90778','B. Bhaskar','2024-11-05 12:42:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   67.3 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1384,'2024-11-05','12:22:29','90.00000','105.00000','Ampicillin and sulbactam for Injection','pH','AS0123009A','E4STFP24000683','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','Ampisulbactam',1,'90778','B. Bhaskar','2024-11-05 12:42:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.90 pH\r\nmV                   -120.5 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1385,'2024-11-05','12:26:00','90.00000','105.00000','Ampcillin for injection','pH','AP1024001A','E4STFP24000690','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','Ampi',1,'90778','B. Bhaskar','2024-11-05 12:42:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.94 pH\r\nmV                   -123.0 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1386,'2024-11-05','13:33:06','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005942','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-05 15:04:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.56 pH\r\nmV                   75.1 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1387,'2024-11-05','13:35:34','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005943','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-05 15:05:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.65 pH\r\nmV                   69.9 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1388,'2024-11-05','13:38:38','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005944','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-05 15:06:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.57 pH\r\nmV                   74.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1389,'2024-11-05','13:42:08','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005945','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-05 15:06:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   77.2 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\nLog #                815\r\n\r\n\r\n'),
(1390,'2024-11-05','13:45:34','90.00000','105.00000','Water For Injection','pH','WFI/SP13','E4RM24005946','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-05 15:06:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.59 pH\r\nmV                   73.6 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1391,'2024-11-05','13:49:25','90.00000','105.00000','Pure steam','pH','PSC/SP10','E4RM24005947','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'86253','D. Jayapal Reddy','2024-11-05 15:07:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   78.2 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1392,'2024-11-05','13:53:02','90.00000','105.00000','Purified water','pH','PW/SP13','E4RM24005948','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-05 15:07:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.56 pH\r\nmV                   75.2 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1393,'2024-11-05','14:06:58','90.00000','105.00000','Purified water','pH','PW/SP48','E4RM24005949','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-05 15:08:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   67.0 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1394,'2024-11-05','14:10:14','90.00000','105.00000','Water For Injection','pH','WFI/SP16','E4RM24005950','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-05 15:08:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.60 pH\r\nmV                   72.8 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1395,'2024-11-05','14:33:36','90.00000','105.00000','FLUID-A','NA','RS012/24012','E4SO24000795','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Fluid-A',1,'205143','Ch. Prasad','2024-11-05 15:59:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.05 pH\r\nmV                   -3.8 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             05-11-24,08:53\r\n\r\n\r\n'),
(1396,'2024-11-05','14:36:40','90.00000','105.00000','Fluid-D','NA','RS010/24010','E4SO24000796','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Fluid-D',1,'205143','Ch. Prasad','2024-11-05 16:25:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.07 pH\r\nmV                   -5.1 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             05-11-24,08:53\r\n\r\n\r\n'),
(1397,'2024-11-05','16:13:22','90.00000','105.00000','AMPICILLIN FOR INJECTION','pH','AI0124010','E4FP24000726','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Ampicillin pH',1,'86253','D. Jayapal Reddy','2024-11-05 16:53:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.14 pH\r\nmV                   -134.7 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1398,'2024-11-05','16:19:52','90.00000','105.00000','AMPICILLIN FOR INJECTION','pH','AI0124011 (LOT-A)','E4FP24000732','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Ampicillin pH',1,'86253','D. Jayapal Reddy','2024-11-05 16:53:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.10 pH\r\nmV                   -132.0 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1399,'2024-11-05','16:24:38','90.00000','105.00000','AMPICILLIN FOR INJECTION','pH','AI0124011 (LOT-B)','E4FP24000733','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Ampicillin pH',1,'86253','D. Jayapal Reddy','2024-11-05 16:54:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.15 pH\r\nmV                   -134.8 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1400,'2024-11-05','16:28:51','90.00000','105.00000','Nafcillin For Injection','pH','NF0224001A','E4STFP24000684','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','pH_Test',1,'86253','D. Jayapal Reddy','2024-11-05 17:06:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.07 pH\r\nmV                   -13.3 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1401,'2024-11-05','16:33:22','90.00000','105.00000','Nafcillin For Injection','pH','NF0222013A','E4STFP24000687','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','pH_Test',1,'86253','D. Jayapal Reddy','2024-11-05 17:06:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.75 pH\r\nmV                   5.6 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1402,'2024-11-05','16:39:01','90.00000','105.00000','Nafcillin For Injection','pH','NF1024001A','E4STFP24000688','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','pH_Test',1,'86253','D. Jayapal Reddy','2024-11-05 17:07:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.44 pH\r\nmV                   -34.5 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\nLog #                826\r\n\r\n\r\n'),
(1403,'2024-11-05','16:46:56','90.00000','105.00000','Nafcillin For Injection','pH','NF1024007A','E4STFP24000691','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','pH_Test',1,'86253','D. Jayapal Reddy','2024-11-05 17:07:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.34 pH\r\nmV                   -28.7 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1404,'2024-11-05','16:50:25','90.00000','105.00000','Ampicillin and sulbactam for Injection','pH','AS1523001A','E4STFP24000693','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','Ampisulbactam',1,'86253','D. Jayapal Reddy','2024-11-05 17:08:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.96 pH\r\nmV                   -124.4 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1405,'2024-11-05','17:16:47','90.00000','105.00000','Media Fill','pH','LY24006','E4AMV24000200','ASEPTIC MEDIA FILL','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Media fill pH',1,'86253','D. Jayapal Reddy','2024-11-05 17:27:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.34 pH\r\nmV                   -29.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             05-11-24,07:10\r\n\r\n\r\n'),
(1406,'2024-11-05','18:35:55','90.00000','105.00000','FLUID-A','NA','RS012/24012','E4SO24000795','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Fluid-A',1,'85152','V. Chandra Shekara Achari','2024-11-05 19:28:10','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.98 pH\r\nmV                   -0.1 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             05-11-24,08:53\r\n\r\n\r\n'),
(1407,'2024-11-05','18:38:30','90.00000','105.00000','Fluid-D','NA','RS010/24010','E4SO24000796','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','Fluid-D',1,'85152','V. Chandra Shekara Achari','2024-11-05 19:28:24','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.08 pH\r\nmV                   -5.6 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             05-11-24,08:53\r\n\r\n\r\n'),
(1408,'2024-11-06','01:51:37','90.00000','105.00000','TAZOBACTAM','NA','TT0024015','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','1',1,'88629','Raghuram Reddy','2024-11-06 06:12:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.63 pH\r\nmV                   123.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        96.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             06-11-24,00:31\r\n\r\n\r\n'),
(1409,'2024-11-06','05:42:28','90.00000','105.00000','PIPERACILLIN','NA','TT0024015','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-11-06 06:13:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.95 pH\r\nmV                   47.6 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        96.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             06-11-24,00:31\r\n\r\n\r\n'),
(1410,'2024-11-06','06:30:17','90.00000','105.00000','PIPERACILLIN','NA','TT0024015','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-06 13:05:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.97 pH\r\nmV                   46.3 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        96.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             06-11-24,00:31\r\n\r\n\r\n'),
(1411,'2024-11-06','07:12:03','90.00000','105.00000','PIPERACILLIN','NA','TT0024015','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-06 13:06:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   28.6 mV\r\nTemperature          26.7 C (ATC)\r\nAverage Slope        96.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             06-11-24,00:31\r\n\r\n\r\n'),
(1412,'2024-11-06','07:19:24','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005810','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-11-06 07:47:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   302.8 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1413,'2024-11-06','07:22:26','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005810','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-11-06 07:48:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   164.9 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1414,'2024-11-06','07:25:45','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005810','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-11-06 07:48:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -10.9 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1415,'2024-11-06','07:29:22','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005810','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-11-06 07:48:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.04 pH\r\nmV                   -186.4 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1416,'2024-11-06','07:34:44','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005810','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-11-06 07:49:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.45 pH\r\nmV                   -327.6 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1417,'2024-11-06','07:51:27','90.00000','105.00000','PIPERACILLIN','NA','TT0024015','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-06 13:06:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.47 pH\r\nmV                   17.6 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        96.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             06-11-24,00:31\r\n\r\n\r\n'),
(1418,'2024-11-06','08:23:47','90.00000','105.00000','Piperacillin And Tazobactam','Assay pH','2449203210','E4RM24005934','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','PT',1,'82735','M.D. Babar Qhadri','2024-11-06 09:08:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   79.4 mV\r\nTemperature          31.0 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1419,'2024-11-06','08:41:05','90.00000','105.00000','Piperacillin and Tazobactam For injection','ASSAY','PN0324079','E4FP24000728','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','89853','J.Ravi',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-06 09:08:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   78.9 mV\r\nTemperature          30.2 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1420,'2024-11-06','09:44:43','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TT0024015','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','3',1,'84472','G. Ashok','2024-11-06 13:06:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.18 pH\r\nmV                   34.3 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        96.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             06-11-24,00:31\r\n\r\n\r\n'),
(1421,'2024-11-06','09:37:02','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS','2449203210','E4RM24005934','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205189','P. Ravi Kumar',2,1,101,'pH','Mobile phase pH',1,'82735','M.D. Babar Qhadri','2024-11-06 10:10:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.04 pH\r\nmV                   47.4 mV\r\nTemperature          28.1 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\nLog #                837\r\n\r\n\r\n'),
(1422,'2024-11-06','09:58:19','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TT0024015','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','3',1,'84472','G. Ashok','2024-11-06 13:07:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   28.3 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        96.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             06-11-24,00:31\r\n\r\n\r\n'),
(1423,'2024-11-06','10:11:07','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005820','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-11-06 11:24:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   309.5 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-11-24,10:08\r\n\r\n\r\n'),
(1424,'2024-11-06','10:13:08','90.00000','105.00000','pH Buffer','NA','pHverification','E4INS24005820','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-11-06 11:25:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   172.4 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-11-24,10:08\r\n\r\n\r\n'),
(1425,'2024-11-06','10:14:48','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005820','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-11-06 11:25:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.98 pH\r\nmV                   -2.3 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-11-24,10:08\r\n\r\n\r\n'),
(1426,'2024-11-06','10:17:02','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005820','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-11-06 11:26:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   9.98 pH\r\nmV                   -179.8 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-11-24,10:08\r\n\r\n\r\n'),
(1427,'2024-11-06','10:19:10','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005820','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-11-06 11:26:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -326.7 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-11-24,10:08\r\n\r\n\r\n'),
(1428,'2024-11-06','13:32:58','90.00000','105.00000','Soyabean casein digest broth for sterilty','NA','RS033/24010','E4SO24000802','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDBSIH',1,'205143','Ch. Prasad','2024-11-06 16:04:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.32 pH\r\nmV                   -22.4 mV\r\nTemperature          26.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-11-24,10:08\r\n\r\n\r\n'),
(1429,'2024-11-06','13:38:39','90.00000','105.00000','Peptone',' NA','RS007/24045','E4SO24000804','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-11-06 16:04:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.99 pH\r\nmV                   -2.9 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-11-24,10:08\r\n\r\n\r\n'),
(1430,'2024-11-06','13:49:05','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC','PN0324079','E4FP24000728','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205058','C. Srinivasulu',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-06 13:58:36','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.04 pH\r\nmV                   47.6 mV\r\nTemperature          27.6 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1431,'2024-11-06','14:02:04','90.00000','105.00000','Fluid thioglycollate meddiuam','NA','RS026/24011','E4SO24000803','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','FTGMIH',1,'205143','Ch. Prasad','2024-11-06 16:05:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.28 pH\r\nmV                   -20.1 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-11-24,10:08\r\n\r\n\r\n'),
(1432,'2024-11-06','14:54:59','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TT0024015','E4IP24001322','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','pH (Bulk)',1,'86253','D. Jayapal Reddy','2024-11-06 15:25:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.11 pH\r\nmV                   43.1 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1433,'2024-11-06','15:08:24','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324081','E4FP24000736','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-11-06 15:26:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.09 pH\r\nmV                   -131.1 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1434,'2024-11-06','15:16:52','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PN0224002','E4FP24000735','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-11-06 15:26:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.78 pH\r\nmV                   62.4 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1435,'2024-11-06','17:01:12','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005967','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-06 17:34:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.01 pH\r\nmV                   48.8 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1436,'2024-11-06','17:04:18','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005968','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-06 17:34:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.67 pH\r\nmV                   68.6 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1437,'2024-11-06','17:07:03','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005969','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-06 17:34:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.62 pH\r\nmV                   71.2 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1438,'2024-11-06','17:09:47','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005970','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-06 17:35:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.61 pH\r\nmV                   71.9 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1439,'2024-11-06','17:11:58','90.00000','105.00000','Purified water','pH','PW/SP05','E4RM24005971','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-06 17:36:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.62 pH\r\nmV                   71.4 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1440,'2024-11-06','17:14:23','90.00000','105.00000','Purified water','pH','PW/SP23','E4RM24005972','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-06 17:35:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   67.4 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1441,'2024-11-06','17:17:27','90.00000','105.00000','Water For Injection','pH','WFI/SP24','E4RM24005973','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-06 17:35:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.58 pH\r\nmV                   73.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1442,'2024-11-06','17:19:44','90.00000','105.00000','Water For Injection','pH','WFI/SP41','E4RM24005974','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-06 17:35:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.64 pH\r\nmV                   70.1 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1443,'2024-11-06','17:22:01','90.00000','105.00000','Pure steam','pH','PSC/SP20','E4RM24005975','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'86253','D. Jayapal Reddy','2024-11-06 17:35:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.56 pH\r\nmV                   75.0 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             06-11-24,07:15\r\n\r\n\r\n'),
(1444,'2024-11-06','18:54:41','90.00000','105.00000','Soyabean casein digest broth for sterilty','NA','RS033/24010','E4SO24000802','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDBSIH',1,'205143','Ch. Prasad','2024-11-07 11:26:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.31 pH\r\nmV                   -21.8 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-11-24,10:08\r\n\r\n\r\n'),
(1445,'2024-11-06','18:57:18','90.00000','105.00000','Fluid thioglycollate meddiuam','NA','RS026/24011','E4SO24000803','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','FTGMIH',1,'205143','Ch. Prasad','2024-11-07 11:27:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.26 pH\r\nmV                   -19.0 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-11-24,10:08\r\n\r\n\r\n'),
(1446,'2024-11-06','18:59:48','90.00000','105.00000','Peptone','NA','RS007/24045','E4SO24000804','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-11-07 11:28:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -5.0 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-11-24,10:08\r\n\r\n\r\n'),
(1447,'2024-11-06','22:39:15','90.00000','105.00000','Penicillin G','NA','PG0124001A','E4STFP24000551','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','92574','Abhinav.G',2,0,103,'pH','Penicillin G Micro',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.52 pH\r\nmV                   24.8 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-11-24,10:08\r\n\r\n\r\n'),
(1448,'2024-11-06','22:42:35','90.00000','105.00000','Penicillin G','NA','PG0124002A','E4STFP24000558','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','92574','Abhinav.G',2,0,103,'pH','Penicillin G Micro',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.57 pH\r\nmV                   24.5 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-11-24,10:08\r\n\r\n\r\n'),
(1449,'2024-11-06','22:46:04','90.00000','105.00000','Penicillin G','NA','PG0124003A','E4STFP24000566','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','92574','Abhinav.G',2,0,103,'pH','Penicillin G Micro',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.50 pH\r\nmV                   25.9 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-11-24,10:08\r\n\r\n\r\n'),
(1450,'2024-11-07','07:08:35','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005829','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,1,101,'pH','1.68 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 07:44:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   303.0 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\nLog #                851\r\n\r\n\r\n'),
(1451,'2024-11-07','07:11:01','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005829','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,1,101,'pH','4.01pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 07:44:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   164.8 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\nLog #                852\r\n\r\n\r\n'),
(1452,'2024-11-07','07:12:38','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005829','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,1,101,'pH','7.00pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 07:45:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -10.5 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\nLog #                853\r\n\r\n\r\n'),
(1453,'2024-11-07','07:15:11','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005829','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,1,101,'pH','10.01 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 07:45:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.98 pH\r\nmV                   -186.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\nLog #                854\r\n\r\n\r\n'),
(1454,'2024-11-07','07:17:07','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005829','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,1,101,'pH','12.46 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 07:45:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.48 pH\r\nmV                   -333.6 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\nLog #                855\r\n\r\n\r\n'),
(1455,'2024-11-07','08:36:22','90.00000','105.00000','Ampicillin and Sulbactam','Buffer pH','AS0324081','E4FP24000736','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','AS',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 08:56:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.00 pH\r\nmV                   110.4 mV\r\nTemperature          31.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1456,'2024-11-07','09:18:46','90.00000','105.00000','Piperacillin and Tazobactam for injection','Mobile phase A RS BY HPLC','TZ0424025A','E4CS24000488','Diluent compatibility','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,0,101,'pH','Piper Tazo',1,'90778','B. Bhaskar','2024-11-07 10:28:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   76.3 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1457,'2024-11-07','09:32:19','90.00000','105.00000','Piperacillin and Tazobactam for injection','RS BY HPLC Mobile phase B','TZ0424025A','E4CS24000488','Diluent compatibility','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,0,101,'pH','Piper Tazo',1,'90778','B. Bhaskar','2024-11-07 10:28:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   76.6 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1458,'2024-11-07','09:44:15','90.00000','105.00000','Piperacillin and Tazobactam for injection','RS BY HPLC MP_ A','TZ0424025A','E4CS24000508','Diluent compatibility','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,1,101,'pH','Piper Tazo',1,'90778','B. Bhaskar','2024-11-07 10:28:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   76.3 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\nLog #                860\r\n\r\n\r\n'),
(1459,'2024-11-07','10:11:14','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005837','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-11-07 11:22:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   308.2 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-11-24,10:03\r\n\r\n\r\n'),
(1460,'2024-11-07','10:13:22','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005837','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-11-07 11:22:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.04 pH\r\nmV                   169.4 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-11-24,10:03\r\n\r\n\r\n'),
(1461,'2024-11-07','10:15:40','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005837','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-11-07 11:22:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -3.2 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-11-24,10:03\r\n\r\n\r\n'),
(1462,'2024-11-07','10:19:12','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005837','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-11-07 11:24:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.04 pH\r\nmV                   -179.1 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-11-24,10:03\r\n\r\n\r\n'),
(1463,'2024-11-07','10:25:33','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005837','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-11-07 11:24:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.49 pH\r\nmV                   -322.9 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-11-24,10:03\r\n\r\n\r\n'),
(1464,'2024-11-07','11:35:36','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS','TA0323001B','E4STFP24000653','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205043','T.V. Subbaiah',2,1,101,'pH','PIPERTAZO',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 12:34:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.03 pH\r\nmV                   46.8 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1465,'2024-11-07','11:51:44','90.00000','105.00000','Ampicillin and Sulbactam','UOD','AS0324082','E4FP24000737','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205232','K. Narender Reddy',2,0,101,'pH','AS',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 12:35:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.02 pH\r\nmV                   108.4 mV\r\nTemperature          29.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1466,'2024-11-07','12:03:18','90.00000','105.00000','Flucloxacillin Sodium','Assay pH','NA','MCL24000110','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','5.0 Phosphate Buffer',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 12:39:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.00 pH\r\nmV                   110.4 mV\r\nTemperature          32.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1467,'2024-11-07','12:19:15','90.00000','105.00000','Ampicillin and Sulbactam','SST pH UOD TEST','AS0324081','E4FP24000736','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,0,101,'pH','AS',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 12:38:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.02 pH\r\nmV                   106.5 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1468,'2024-11-07','13:16:37','90.00000','105.00000','Peptone','NA','RS007/24046','E4SO24000808','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','PEPIH',1,'85152','V. Chandra Shekara Achari','2024-11-07 13:45:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.90 pH\r\nmV                   3.7 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-11-24,10:03\r\n\r\n\r\n'),
(1469,'2024-11-07','13:27:28','90.00000','105.00000','Piperacillin and Tazobactam for Injection','Assay','TT0024015','E4IP24001323','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,1,101,'pH','pH (Bulk)',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 14:04:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   80.0 mV\r\nTemperature          31.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1470,'2024-11-07','13:47:31','90.00000','105.00000','PIPERACILLIN MONOHYDRATE','RS MOBILE PHASE_A','2449103414','E4AQ24000062','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205266','K. Vara prasad',2,1,101,'pH','RS Mobile phase',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 14:05:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   77.9 mV\r\nTemperature          33.1 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1471,'2024-11-07','13:53:44','90.00000','105.00000','PIPERACILLIN MONOHYDRATE','RS MOBILE PHASE _B','2449103414','E4AQ24000062','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205266','K. Vara prasad',2,1,101,'pH','RS Mobile phase',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 14:05:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   78.4 mV\r\nTemperature          32.4 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1472,'2024-11-07','14:04:03','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424118','E4FP24000727','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 14:26:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.86 pH\r\nmV                   56.8 mV\r\nTemperature          26.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1473,'2024-11-07','14:12:51','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424119','E4FP24000730','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205211','S. Sailatha',2,1,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-11-07 14:26:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.10 pH\r\nmV                   42.8 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1474,'2024-11-07','14:55:52','90.00000','105.00000','Sensor Calibration','NA','NA','NA','Temperature Sensor','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','Purified Water',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   54.4 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             07-11-24,11:00\r\n\r\n\r\n'),
(1475,'2024-11-07','15:10:00','90.00000','105.00000','Piperacillin And Tazobactam','pH','2449203210','E4RM24005934','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','PT',1,'90018','Ch. Satyam','2024-11-07 15:30:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.31 pH\r\nmV                   30.1 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1476,'2024-11-07','15:15:41','90.00000','105.00000','Piperacillin And Tazobactam','pH','2449203228','E4RM24005935','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','PT',1,'90018','Ch. Satyam','2024-11-07 15:30:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.50 pH\r\nmV                   19.0 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1477,'2024-11-07','15:20:06','90.00000','105.00000','Piperacillin And Tazobactam','pH','2449203240','E4RM24005936','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','PT',1,'90018','Ch. Satyam','2024-11-07 15:30:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.10 pH\r\nmV                   42.7 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1478,'2024-11-07','15:33:14','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PT0424034A','E4STFP24000696','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-11-07 16:50:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.31 pH\r\nmV                   89.3 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1479,'2024-11-07','15:39:24','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PT0424034A','E4STFP24000697','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-11-07 16:52:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.57 pH\r\nmV                   73.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1480,'2024-11-07','15:45:40','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PN0324080','E4FP24000729','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-11-07 16:57:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   58.7 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1481,'2024-11-07','15:53:32','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PN0324079','E4FP24000728','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-11-07 16:58:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.27 pH\r\nmV                   32.9 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1482,'2024-11-07','15:58:54','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PN0224001','E4FP24000731','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-11-07 17:01:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.58 pH\r\nmV                   73.7 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1483,'2024-11-07','16:07:42','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005988','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-07 17:01:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.58 pH\r\nmV                   73.0 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1484,'2024-11-07','16:10:34','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005989','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-07 17:02:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.59 pH\r\nmV                   72.6 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1485,'2024-11-07','16:13:17','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005990','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-07 17:02:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.55 pH\r\nmV                   75.4 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1486,'2024-11-07','16:16:26','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005991','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-07 17:03:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   76.7 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1487,'2024-11-07','16:19:39','90.00000','105.00000','Water For Injection','pH','WFI/SP33','E4RM24005992','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-07 17:06:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   77.4 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1488,'2024-11-07','16:23:12','90.00000','105.00000','Pure steam','pH','PSC/SP21','E4RM24005993','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'86253','D. Jayapal Reddy','2024-11-07 17:06:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   77.9 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1489,'2024-11-07','16:26:17','90.00000','105.00000','Purified water','pH','PW/SP17','E4RM24005994','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-07 17:06:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   75.9 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1490,'2024-11-07','16:29:28','90.00000','105.00000','Water For Injection','pH','WFI/SP18','E4RM24005995','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-07 17:06:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   76.1 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             07-11-24,07:04\r\n\r\n\r\n'),
(1491,'2024-11-07','17:12:36','90.00000','105.00000','Peptone','NA','RS007/24046','E4SO24000808','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-11-08 10:49:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.99 pH\r\nmV                   -1.4 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-11-24,10:03\r\n\r\n\r\n'),
(1492,'2024-11-08','01:32:12','90.00000','105.00000','TAZOBACTAM','NA','TT0024016','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','1',1,'88629','Raghuram Reddy','2024-11-08 05:14:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.52 pH\r\nmV                   125.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             08-11-24,00:29\r\n\r\n\r\n'),
(1493,'2024-11-08','04:49:23','90.00000','105.00000','PIPERACILLIN','NA','TT0024016','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-11-08 05:15:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.86 pH\r\nmV                   47.6 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             08-11-24,00:29\r\n\r\n\r\n'),
(1494,'2024-11-08','05:32:03','90.00000','105.00000','PIPERACILLIN','NA','TT0024016','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-11-08 06:32:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.89 pH\r\nmV                   45.7 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             08-11-24,00:29\r\n\r\n\r\n'),
(1495,'2024-11-08','06:18:13','90.00000','105.00000','PIPERACILLIN','NA','TT0024016','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-11-08 06:32:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.24 pH\r\nmV                   25.2 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             08-11-24,00:29\r\n\r\n\r\n'),
(1496,'2024-11-08','07:18:16','90.00000','105.00000','PIPERACILLIN','NA','TT0024016','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-11-08 08:31:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.43 pH\r\nmV                   13.8 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             08-11-24,00:29\r\n\r\n\r\n'),
(1497,'2024-11-08','07:15:32','90.00000','105.00000','pH Buffer','  NA','NA','E4INS24005847','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','1.68 pH',1,'84378','Shivanand. W','2024-11-08 07:44:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   301.3 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\nLog #                886\r\n\r\n\r\n'),
(1498,'2024-11-08','07:22:17','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005847','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','4.01pH',1,'84378','Shivanand. W','2024-11-08 07:45:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   162.5 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1499,'2024-11-08','07:25:47','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005847','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','7.00pH',1,'84378','Shivanand. W','2024-11-08 07:45:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -12.5 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1500,'2024-11-08','07:29:17','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005847','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','10.01 pH',1,'84378','Shivanand. W','2024-11-08 07:46:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.96 pH\r\nmV                   -186.8 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1501,'2024-11-08','07:32:29','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005847','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','12.46 pH',1,'84378','Shivanand. W','2024-11-08 07:46:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.45 pH\r\nmV                   -333.4 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1502,'2024-11-08','08:46:58','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424120','E4FP24000734','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Pipertazoinj',1,'84378','Shivanand. W','2024-11-08 09:33:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   63.2 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1503,'2024-11-08','08:55:01','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424121','E4FP24000739','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Pipertazoinj',1,'84378','Shivanand. W','2024-11-08 09:34:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.61 pH\r\nmV                   69.1 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1504,'2024-11-08','08:59:11','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TT0024016','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','3',1,'95681','T. Sivanarayana','2024-11-08 09:20:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.29 pH\r\nmV                   22.2 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             08-11-24,00:29\r\n\r\n\r\n'),
(1505,'2024-11-08','09:07:07','90.00000','105.00000','Media Fill','pH','LY24007','E4AMV24000209','ASEPTIC MEDIA FILL','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Media fill pH',1,'84378','Shivanand. W','2024-11-08 09:35:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.33 pH\r\nmV                   -32.0 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1506,'2024-11-08','09:15:15','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TT0024016','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','3',1,'95681','T. Sivanarayana','2024-11-08 09:21:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.26 pH\r\nmV                   23.1 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             08-11-24,00:29\r\n\r\n\r\n'),
(1507,'2024-11-08','09:39:39','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005856','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-11-08 10:44:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.70 pH\r\nmV                   308.0 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1508,'2024-11-08','09:43:29','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005856','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-11-08 10:45:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.04 pH\r\nmV                   171.5 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1509,'2024-11-08','09:45:29','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005856','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-11-08 10:46:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -2.9 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1510,'2024-11-08','09:53:39','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005856','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-11-08 10:47:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.00 pH\r\nmV                   -175.2 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1511,'2024-11-08','09:57:46','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005856','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-11-08 10:47:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -319.0 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1512,'2024-11-08','10:24:26','90.00000','105.00000','Piperacillin and Tazobactam For injection','UOD','PT0424120','E4FP24000734','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,0,101,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 14:35:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   77.4 mV\r\nTemperature          31.6 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1513,'2024-11-08','10:34:21','90.00000','105.00000','Piperacillin and Tazobactam for Injection','IDENTIFICATION','2449202490','E4RM24005889','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','Mobile phase pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 14:26:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   77.4 mV\r\nTemperature          31.8 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1514,'2024-11-08','14:13:34','90.00000','105.00000','Nafcillin for Injection','pH','NF0124001A','E4STFP24000698','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Nafcillin pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 14:25:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.21 pH\r\nmV                   -25.0 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1515,'2024-11-08','14:16:16','90.00000','105.00000','Nafcillin for Injection','pH','NF0124001A','E4STFP24000699','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Nafcillin pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 14:25:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.13 pH\r\nmV                   -19.9 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1516,'2024-11-08','14:25:55','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TT0024016','E4IP24001328','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','pH (Bulk)',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 14:36:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.90 pH\r\nmV                   52.1 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1517,'2024-11-08','14:41:10','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24020','E4SO24000813','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SCDBIH',1,'85152','V. Chandra Shekara Achari','2024-11-08 21:31:23','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.44 pH\r\nmV                   -26.4 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1518,'2024-11-08','14:44:42','90.00000','105.00000','Macckankey broth','NA','RS009/24020','E4SO24000814','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','NA',1,'85152','V. Chandra Shekara Achari','2024-11-08 21:32:05','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.38 pH\r\nmV                   -23.0 mV\r\nTemperature          22.8 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1519,'2024-11-08','15:33:27','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24006019','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:23:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   64.5 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1520,'2024-11-08','15:36:25','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24006020','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:25:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.61 pH\r\nmV                   68.8 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\nLog #                900\r\n\r\n\r\n'),
(1521,'2024-11-08','15:41:53','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24006021','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:26:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   73.4 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1522,'2024-11-08','15:44:41','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24006022','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:26:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   73.1 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1523,'2024-11-08','15:47:16','90.00000','105.00000','Purified water','pH','PW/SP34','E4RM24006023','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:27:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.57 pH\r\nmV                   71.1 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1524,'2024-11-08','15:51:01','90.00000','105.00000','Purified water','pH','PW/SP36','E4RM24006024','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:28:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.20 pH\r\nmV                   93.1 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1525,'2024-11-08','15:52:55','90.00000','105.00000','Purified water','pH','PW/SP25','E4RM24006025','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:28:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.29 pH\r\nmV                   87.5 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1526,'2024-11-08','15:55:12','90.00000','105.00000','Water For Injection','pH','WFI/SP26','E4RM24006026','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-11-08 16:29:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.34 pH\r\nmV                   84.8 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1527,'2024-11-08','17:14:29','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS','PT0424074A','E4STFP24000670','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-11-09 13:34:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   3.82 pH\r\nmV                   177.7 mV\r\nTemperature          29.1 C (ATC)\r\nAverage Slope        99.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             08-11-24,07:08\r\n\r\n\r\n'),
(1528,'2024-11-08','18:25:38','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24020','E4SO24000813','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','SCDBIH',1,'85152','V. Chandra Shekara Achari','2024-11-08 21:32:19','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.48 pH\r\nmV                   -28.9 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1529,'2024-11-08','18:29:37','90.00000','105.00000','Macckankey broth','NA','RS013/24009','E4SO24000814','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','NA',1,'85152','V. Chandra Shekara Achari','2024-11-08 21:32:33','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.34 pH\r\nmV                   -20.6 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             08-11-24,09:19\r\n\r\n\r\n'),
(1530,'2024-11-09','07:02:43','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005869','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','1.68 pH',1,'86253','D. Jayapal Reddy','2024-11-09 07:28:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   297.6 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1531,'2024-11-09','07:04:54','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005869','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','4.01pH',1,'86253','D. Jayapal Reddy','2024-11-09 07:29:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   159.2 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1532,'2024-11-09','07:07:36','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005869','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','7.00pH',1,'86253','D. Jayapal Reddy','2024-11-09 07:29:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -16.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1533,'2024-11-09','07:10:21','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005869','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','10.01 pH',1,'86253','D. Jayapal Reddy','2024-11-09 07:29:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.98 pH\r\nmV                   -189.5 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1534,'2024-11-09','07:12:52','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005869','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','12.46 pH',1,'86253','D. Jayapal Reddy','2024-11-09 07:29:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -335.9 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1535,'2024-11-09','08:13:55','90.00000','105.00000','Piperacillin and Tazobactam for Injection','Assay pH','2449203350','E4RM24006009','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','Mobile phase pH',1,'86253','D. Jayapal Reddy','2024-11-09 12:27:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   74.0 mV\r\nTemperature          31.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1536,'2024-11-09','11:16:46','90.00000','105.00000','Ampicillin and Sulbactam','RS BY HPLC MP A','AS0324081','E4FP24000736','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,0,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-11-09 12:27:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.03 pH\r\nmV                   101.7 mV\r\nTemperature          28.9 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\nLog #                914\r\n\r\n\r\n'),
(1537,'2024-11-09','11:31:06','90.00000','105.00000','Ampicillin and Sulbactam','UOD','AS0324083','E4FP24000740','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205232','K. Narender Reddy',2,0,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-11-09 12:27:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.99 pH\r\nmV                   105.5 mV\r\nTemperature          32.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1538,'2024-11-09','11:40:25','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS','2449203350','E4RM24006009','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205189','P. Ravi Kumar',2,1,101,'pH','Mobile phase pH',1,'86253','D. Jayapal Reddy','2024-11-09 12:28:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.02 pH\r\nmV                   43.3 mV\r\nTemperature          28.0 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\nLog #                916\r\n\r\n\r\n'),
(1539,'2024-11-09','11:53:23','90.00000','105.00000','Ampicillin and Sulbactam for injection','Identification','2449101498','E4RM24005952','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,0,101,'pH','Ampi&Sulbactam',1,'86253','D. Jayapal Reddy','2024-11-09 12:28:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.02 pH\r\nmV                   103.5 mV\r\nTemperature          31.4 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1540,'2024-11-09','11:57:25','90.00000','105.00000','TAZOBACTAM','NA','TZ0024105','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','92936','Sameer',2,0,104,'pH','1',1,'95681','T. Sivanarayana','2024-11-09 14:00:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.69 pH\r\nmV                   117.6 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             09-11-24,10:25\r\nLog #                277\r\n\r\n\r\n'),
(1541,'2024-11-09','12:28:12','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324082','E4FP24000737','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-11-09 13:42:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.11 pH\r\nmV                   -139.3 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1542,'2024-11-09','12:32:40','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324083','E4FP24000740','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-11-09 13:43:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.07 pH\r\nmV                   -136.6 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1543,'2024-11-09','12:37:18','90.00000','105.00000','Ampicillin and Sulbactam','pH','AS0324084','E4FP24000742','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','AS',1,'86253','D. Jayapal Reddy','2024-11-09 13:43:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.04 pH\r\nmV                   -134.7 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1544,'2024-11-09','12:45:01','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TT0424013','E4FP24000738','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-11-09 13:43:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.98 pH\r\nmV                   44.9 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1545,'2024-11-09','12:48:46','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TT0424014','E4FP24000741','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205187','B. Ramvikas',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-11-09 13:44:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.97 pH\r\nmV                   45.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1546,'2024-11-09','13:23:03','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0222004B','E4STFP24000694','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-11-09 14:06:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.86 pH\r\nmV                   52.0 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1547,'2024-11-09','13:27:02','90.00000','105.00000','Nafcillin For Injection','pH','NF1024006A','E4STFP24000695','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','pH_Test',1,'86253','D. Jayapal Reddy','2024-11-09 14:06:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.37 pH\r\nmV                   -36.8 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1548,'2024-11-09','13:39:12','90.00000','105.00000','Flucloxacillin powder for solution for injection','pH','FL0122001A','E4STFP24000686','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','ph Flucloxacillin',1,'86253','D. Jayapal Reddy','2024-11-09 14:07:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.63 pH\r\nmV                   65.3 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1549,'2024-11-09','13:42:17','90.00000','105.00000','Flucloxacillin powder for solution for injection','pH','FL0124001B','E4STFP24000692','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','ph Flucloxacillin',1,'86253','D. Jayapal Reddy','2024-11-09 14:07:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.66 pH\r\nmV                   63.3 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1550,'2024-11-09','13:46:30','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24006048','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-09 14:25:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.61 pH\r\nmV                   66.2 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1551,'2024-11-09','13:48:50','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24006049','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-09 14:25:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.55 pH\r\nmV                   69.7 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1552,'2024-11-09','13:51:30','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24006050','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-09 14:25:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   72.8 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1553,'2024-11-09','13:54:04','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24006051','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-09 14:26:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   72.3 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1554,'2024-11-09','13:57:04','90.00000','105.00000','Water For Injection','pH','WFI/SP34','E4RM24006052','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-09 14:27:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.44 pH\r\nmV                   76.2 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1555,'2024-11-09','13:59:58','90.00000','105.00000','Purified water','pH','PW/SP26','E4RM24006053','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-11-09 14:28:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.57 pH\r\nmV                   68.8 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1556,'2024-11-09','14:02:46','90.00000','105.00000','Water For Injection','pH','WFI/SP27','E4RM24006054','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-11-09 14:28:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.50 pH\r\nmV                   73.0 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1557,'2024-11-09','15:29:56','90.00000','105.00000','Ampicillin and Sulbactam','UOD TEST','AS0324084','E4FP24000742','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,0,101,'pH','AS',1,'82735','M.D. Babar Qhadri','2024-11-11 14:51:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.01 pH\r\nmV                   104.9 mV\r\nTemperature          32.4 C (ATC)\r\nAverage Slope        99.2 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             09-11-24,06:58\r\n\r\n\r\n'),
(1558,'2024-11-09','17:18:50','90.00000','105.00000','PIPERACILLIN','NA','TZ0024105','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'88515','Prashanth B','2024-11-09 21:14:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.91 pH\r\nmV                   46.1 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             09-11-24,10:25\r\n\r\n\r\n'),
(1559,'2024-11-09','18:00:18','90.00000','105.00000','PIPERACILLIN','NA','TZ0024105','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'88515','Prashanth B','2024-11-09 21:15:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.92 pH\r\nmV                   45.0 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             09-11-24,10:25\r\n\r\n\r\n'),
(1560,'2024-11-09','18:43:06','90.00000','105.00000','PIPERACILLIN','NA','TZ0024105','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'88515','Prashanth B','2024-11-09 21:15:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.20 pH\r\nmV                   29.0 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             09-11-24,10:25\r\n\r\n\r\n'),
(1561,'2024-11-09','19:25:01','90.00000','105.00000','PIPERACILLIN','NA','TZ0024105','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','2',1,'88515','Prashanth B','2024-11-09 21:16:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.37 pH\r\nmV                   19.2 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             09-11-24,10:25\r\n\r\n\r\n'),
(1562,'2024-11-09','21:22:16','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024105','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','3',1,'88515','Prashanth B','2024-11-09 21:39:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   24.0 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             09-11-24,10:25\r\n\r\n\r\n'),
(1563,'2024-11-09','21:35:02','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TT0024105','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','3',1,'88515','Prashanth B','2024-11-09 21:40:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.31 pH\r\nmV                   22.7 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             09-11-24,10:25\r\n\r\n\r\n'),
(1564,'2024-11-10','07:51:07','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005891','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-11-10 08:29:34','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   301.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             10-11-24,07:35\r\n\r\n\r\n'),
(1565,'2024-11-10','07:55:19','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005891','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-11-10 08:29:46','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   163.6 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             10-11-24,07:35\r\n\r\n\r\n'),
(1566,'2024-11-10','08:00:30','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005891','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-11-10 08:29:56','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -12.5 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             10-11-24,07:35\r\n\r\n\r\n'),
(1567,'2024-11-10','08:10:15','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005891','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-11-10 08:30:06','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.04 pH\r\nmV                   -188.4 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             10-11-24,07:35\r\n\r\n\r\n'),
(1568,'2024-11-10','08:15:35','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005891','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-11-10 08:30:17','REVIEWED','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.49 pH\r\nmV                   -332.4 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             10-11-24,07:35\r\n\r\n\r\n'),
(1569,'2024-11-10','12:12:55','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0024105','E4IP24001334','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205252','Y. Sivaiah',2,1,101,'pH','pH (Bulk)',1,'82735','M.D. Babar Qhadri','2024-11-10 12:30:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.05 pH\r\nmV                   44.9 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             10-11-24,07:35\r\n\r\n\r\n'),
(1570,'2024-11-10','21:12:54','90.00000','105.00000','TAZOBACTAM','NA','TZ0024106','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','1',1,'88629','Raghuram Reddy','2024-11-10 21:33:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.40 pH\r\nmV                   132.8 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             10-11-24,10:18\r\n\r\n\r\n'),
(1571,'2024-11-11','00:37:07','90.00000','105.00000','PIPERACILLIN','NA','TZ0024106','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-11 04:39:05','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.90 pH\r\nmV                   45.8 mV\r\nTemperature          26.9 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-11-24,00:18\r\n\r\n\r\n'),
(1572,'2024-11-11','01:19:07','90.00000','105.00000','PIPERACILLIN','NA','TZ0024106','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-11 04:47:12','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.89 pH\r\nmV                   44.0 mV\r\nTemperature          23.1 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-11-24,00:18\r\n\r\n\r\n'),
(1573,'2024-11-11','01:21:03','90.00000','105.00000','PIPERACILLIN','NA','TZ0024106','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-11 04:47:31','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.91 pH\r\nmV                   44.2 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-11-24,00:18\r\n\r\n\r\n'),
(1574,'2024-11-11','02:04:15','90.00000','105.00000','PIPERACILLIN','NA','TZ0024106','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-11 04:47:52','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.22 pH\r\nmV                   26.5 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-11-24,00:18\r\n\r\n\r\n'),
(1575,'2024-11-11','02:47:43','90.00000','105.00000','PIPERACILLIN','NA','TZ0024106','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'84472','G. Ashok','2024-11-11 04:48:15','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.42 pH\r\nmV                   14.8 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-11-24,00:18\r\n\r\n\r\n'),
(1576,'2024-11-11','05:07:52','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024106','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','3',1,'84472','G. Ashok','2024-11-11 06:12:11','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.22 pH\r\nmV                   26.2 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-11-24,00:18\r\n\r\n\r\n'),
(1577,'2024-11-11','05:27:31','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024106','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','3',1,'84472','G. Ashok','2024-11-11 06:12:28','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.23 pH\r\nmV                   25.8 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_119\r\nLast Cal             11-11-24,00:18\r\n\r\n\r\n'),
(1578,'2024-11-11','06:53:00','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005900','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','1.68 pH',1,'86253','D. Jayapal Reddy','2024-11-11 07:28:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   302.3 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1579,'2024-11-11','06:57:23','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005900','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','4.01pH',1,'86253','D. Jayapal Reddy','2024-11-11 07:29:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   164.1 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1580,'2024-11-11','07:01:18','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005900','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','7.00pH',1,'86253','D. Jayapal Reddy','2024-11-11 07:30:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -11.8 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1581,'2024-11-11','07:08:00','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005900','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','10.01 pH',1,'86253','D. Jayapal Reddy','2024-11-11 07:30:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.06 pH\r\nmV                   -188.1 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1582,'2024-11-11','07:11:55','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005900','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','12.46 pH',1,'86253','D. Jayapal Reddy','2024-11-11 07:31:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -331.4 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1583,'2024-11-11','10:18:10','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005905','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-11-11 12:43:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   309.1 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-11-24,10:14\r\n\r\n\r\n'),
(1584,'2024-11-11','10:20:14','90.00000','105.00000','Piperacillin and Tazobactam for Injection','Assay pH','2449203357','E4RM24006016','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','Mobile phase pH',1,'90018','Ch. Satyam','2024-11-11 12:21:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   79.6 mV\r\nTemperature          32.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1585,'2024-11-11','10:20:54','90.00000','105.00000','pH Buffer','NA','	NA','E4INS24005905','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-11-11 12:45:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   171.0 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-11-24,10:14\r\nLog #                398\r\n\r\n\r\n'),
(1586,'2024-11-11','10:45:41','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005905','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-11-11 12:47:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -6.3 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-11-24,10:14\r\n\r\n\r\n'),
(1587,'2024-11-11','10:48:09','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005905','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-11-11 12:49:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   9.99 pH\r\nmV                   -179.8 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-11-24,10:14\r\n\r\n\r\n'),
(1588,'2024-11-11','10:49:54','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005905','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-11-11 12:52:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.49 pH\r\nmV                   -327.0 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             11-11-24,10:14\r\n\r\n\r\n'),
(1589,'2024-11-11','11:53:41','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS','TT0424013','E4FP24000738','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205232','K. Narender Reddy',2,0,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-11-11 14:45:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.99 pH\r\nmV                   50.1 mV\r\nTemperature          27.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
(1590,'2024-11-11','12:25:17','90.00000','105.00000','Piperacillin and Tazobactam for Injection','TZ0024106','NA','E4IP24001339','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','pH (Bulk)',1,'90018','Ch. Satyam','2024-11-11 12:35:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.48 pH\r\nmV                   20.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             11-11-24,06:48\r\n\r\n\r\n'),
