
CREATE TABLE `tbl_calibration_master_ph_ise` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `PrDate` date DEFAULT NULL,
  `PrTime` time DEFAULT NULL,
  `MeterID` varchar(500) DEFAULT 'NULL',
  `ProbID` varchar(500) DEFAULT 'NULL',
  `NegTol` varchar(30) DEFAULT NULL,
  `PosTol` varchar(30) DEFAULT NULL,
  `DP` int(5) DEFAULT '0',
  `AvgSlope` varchar(30) DEFAULT '0',
  `Blank` varchar(50) DEFAULT 'NULL',
  `Remark` varchar(50) DEFAULT 'NULL',
  `UserID` varchar(500) DEFAULT 'NULL',
  `UserName` varchar(500) DEFAULT 'NULL',
  `ChannelNo` int(11) DEFAULT '0',
  `IDSNo` int(11) DEFAULT '0',
  `ProbType` varchar(50) DEFAULT 'pH/ISE',
  `ProdID` varchar(500) DEFAULT 'NULL',
  `PrinterName` varchar(500) DEFAULT 'NULL',
  `PrintNo` int(11) DEFAULT '0',
  `CheckedByID` varchar(25) DEFAULT 'NULL',
  `CheckedByName` varchar(200) DEFAULT 'NULL',
  `CheckedByDate` datetime DEFAULT '1992-12-08 00:00:00',
  `VerifyRemark` varchar(500) DEFAULT 'NULL',
  `AllData` text,
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_calibration_master_ph_ise` */

/*Table structure for table `tbl_calibration_master_rdo` */


CREATE TABLE `tbl_calibration_master_rdo` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `PrDate` date DEFAULT NULL,
  `PrTime` time DEFAULT NULL,
  `MeterID` varchar(500) DEFAULT 'NULL',
  `ProbID` varchar(500) DEFAULT 'NULL',
  `NegTol` varchar(30) DEFAULT NULL,
  `PosTol` varchar(30) DEFAULT NULL,
  `DP` int(5) DEFAULT '0',
  `Slope` varchar(30) DEFAULT '0',
  `Remark` varchar(50) DEFAULT 'NULL',
  `UserID` varchar(500) DEFAULT 'NULL',
  `UserName` varchar(500) DEFAULT 'NULL',
  `ChannelNo` int(11) DEFAULT '0',
  `IDSNo` int(11) DEFAULT '0',
  `ProbType` varchar(50) DEFAULT 'RDO',
  `ProdID` varchar(500) DEFAULT 'NULL',
  `PrinterName` varchar(500) DEFAULT 'NULL',
  `PrintNo` int(11) DEFAULT '0',
  `Saturation` varchar(50) DEFAULT '0',
  `Current` varchar(50) DEFAULT '0',
  `ZeroCurrent` varchar(50) DEFAULT '0',
  `TempSolution` varchar(50) DEFAULT '0',
  `TempMembrane` varchar(50) DEFAULT '0',
  `CalibType` varchar(100) DEFAULT 'NULL',
  `BioPressure` varchar(50) DEFAULT '0',
  `PartialPressure` varchar(50) DEFAULT '0',
  `CheckedByID` varchar(25) DEFAULT 'NULL',
  `CheckedByName` varchar(200) DEFAULT 'NULL',
  `CheckedByDate` datetime DEFAULT '1992-12-08 00:00:00',
  `VerifyRemark` varchar(500) DEFAULT 'NULL',
  `AllData` text,
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_calibration_master_rdo` */

/*Table structure for table `tbl_calibration_master_temperature` */


CREATE TABLE `tbl_calibration_master_temperature` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `PrDate` date DEFAULT NULL,
  `PrTime` time DEFAULT NULL,
  `MeterID` varchar(500) DEFAULT 'NULL',
  `ProbID` varchar(500) DEFAULT 'NULL',
  `NegTol` varchar(30) DEFAULT NULL,
  `PosTol` varchar(30) DEFAULT NULL,
  `DP` int(5) DEFAULT '0',
  `Remark` varchar(50) DEFAULT 'NULL',
  `UserID` varchar(500) DEFAULT 'NULL',
  `UserName` varchar(500) DEFAULT 'NULL',
  `ChannelNo` int(11) DEFAULT '0',
  `IDSNo` int(11) DEFAULT '0',
  `ProbType` varchar(50) NOT NULL DEFAULT 'Temperature',
  `ProdID` varchar(500) DEFAULT 'NULL',
  `PrinterName` varchar(500) DEFAULT 'NULL',
  `PrintNo` int(11) DEFAULT '0',
  `CheckedByID` varchar(25) DEFAULT 'NULL',
  `CheckedByName` varchar(200) DEFAULT 'NULL',
  `CheckedByDate` datetime DEFAULT '1992-12-08 00:00:00',
  `VerifyRemark` varchar(500) DEFAULT 'NULL',
  `AllData` text,
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_calibration_master_temperature` */

/*Table structure for table `tbl_calibration_print` */


CREATE TABLE `tbl_calibration_print` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `MeterID` varchar(500) NOT NULL DEFAULT 'NULL',
  `RepSerNo` int(11) NOT NULL,
  `ProbType` varchar(500) DEFAULT NULL,
  `PrintNo` int(11) NOT NULL DEFAULT '0',
  `Reason` varchar(300) NOT NULL DEFAULT 'NULL',
  `Title` varchar(20) NOT NULL DEFAULT 'NULL',
  `UserID` varchar(50) NOT NULL DEFAULT 'NULL',
  `UserName` varchar(100) NOT NULL DEFAULT 'NULL',
  `Print_Dt` date NOT NULL,
  `Print_Tm` time NOT NULL,
  `Incomplete` int(1) NOT NULL DEFAULT '0' COMMENT '0:C,1:IC',
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB AUTO_INCREMENT=63 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_calibration_print` */

insert  into `tbl_calibration_print`(`RecNo`,`MeterID`,`RepSerNo`,`ProbType`,`PrintNo`,`Reason`,`Title`,`UserID`,`UserName`,`Print_Dt`,`Print_Tm`,`Incomplete`) values 
(1,'QC-PHMTZ-007',1,'pH',1,'','Title','V-trident','Trident','2024-09-13','10:25:25',0),
(2,'QC-PHMTZ-007',1,'pH',2,'As part of OQ','Title','V-trident','Trident','2024-09-13','10:26:54',0),
(7,'QC-PHMTZ-007',3,'pH',1,'','Title','V-trident','Trident','2024-09-17','16:10:57',0),
(8,'QC-PHMTZ-009',4,'pH',1,'','Title','V-trident','Trident','2024-09-17','16:11:42',0),
(9,'QC-PHMTZ-008',5,'pH',1,'','Title','V-trident','Trident','2024-09-17','16:41:07',0),
(10,'PR-PHMTZ-003',6,'pH',1,'','Title','V-trident','Trident','2024-09-18','15:49:48',0),
(11,'PR-PHMTZ-004',7,'pH',1,'','Title','V-trident','Trident','2024-09-18','15:50:16',0),
(12,'QC-PHMTZ-009',13,'pH',1,'','Title','205066','Sreeram Jakkaraiah','2024-09-30','17:47:30',0),
(13,'QC-PHMTZ-007',12,'pH',1,'','Title','205089','M. Venkata Ramana','2024-09-30','18:01:06',0),
(14,'QC-PHMTZ-009',15,'pH',1,'','Title','205089','M. Venkata Ramana','2024-10-01','08:00:14',0),
(15,'QC-PHMTZ-007',14,'pH',1,'','Title','88571','V. Ramu','2024-10-01','08:13:39',0),
(16,'QC-PHMTZ-009',25,'pH',1,'','Title','V-trident','Trident','2024-10-04','12:42:02',0),
(17,'QC-PHMTZ-009',33,'pH',1,'','Title','205187','B. Ramvikas','2024-10-06','10:01:19',0),
(18,'QC-PHMTZ-007',36,'pH',1,'','Title','205089','M. Venkata Ramana','2024-10-07','09:30:33',0),
(19,'QC-PHMTZ-007',39,'pH',1,'','Title','90018','Ch. Satyam','2024-10-08','08:03:29',0),
(20,'QC-PHMTZ-007',43,'pH',1,'','Title','88599','MD Mahamad','2024-10-09','08:02:31',0),
(21,'QC-PHMTZ-007',46,'pH',1,'','Title','205075','G. Shyam Kumar','2024-10-10','08:06:36',0),
(22,'QC-PHMTZ-007',48,'pH',1,'','Title','88571','V. Ramu','2024-10-11','07:00:12',0),
(23,'QC-PHMTZ-009',51,'pH',1,'','Title','87624','N. Nani babu','2024-10-11','17:09:30',0),
(24,'QC-PHMTZ-007',54,'pH',1,'','Title','88655','V. Prashanth Kumar','2024-10-13','11:34:09',0),
(25,'QC-PHMTZ-007',56,'pH',1,'','Title','88571','V. Ramu','2024-10-14','08:06:17',0),
(26,'QC-PHMTZ-007',58,'pH',1,'','Title','88599','MD Mahamad','2024-10-15','07:53:22',0),
(27,'QC-PHMTZ-009',60,'pH',1,'','Title','88599','MD Mahamad','2024-10-15','12:33:47',0),
(28,'QC-PHMTZ-007',61,'pH',1,'','Title','205090','M. Srikrishnudu','2024-10-16','07:37:05',0),
(29,'QC-PHMTZ-007',64,'pH',1,'','Title','205211','S. Sailatha','2024-10-17','07:52:44',0),
(30,'QC-PHMTZ-007',66,'pH',1,'','Title','205247','R. Puneetha','2024-10-18','08:01:41',0),
(31,'QC-PHMTZ-007',69,'pH',1,'','Title','205210','B. Anitha','2024-10-19','08:47:59',0),
(32,'QC-PHMTZ-007',72,'pH',1,'','Title','205089','M. Venkata Ramana','2024-10-21','07:40:03',0),
(33,'QC-PHMTZ-008',73,'pH',1,'','Title','92513','Ch. Chinna Rao','2024-10-21','11:32:13',0),
(34,'QC-PHMTZ-007',76,'pH',1,'','Title','205089','M. Venkata Ramana','2024-10-22','07:28:12',0),
(35,'QC-PHMTZ-007',80,'pH',1,'','Title','88599','MD Mahamad','2024-10-23','08:19:25',0),
(36,'QC-PHMTZ-007',82,'pH',1,'','Title','205211','S. Sailatha','2024-10-24','08:47:40',0),
(37,'QC-PHMTZ-007',85,'pH',1,'','Title','205247','R. Puneetha','2024-10-25','07:44:31',0),
(38,'QC-PHMTZ-007',90,'pH',1,'','Title','205210','B. Anitha','2024-10-26','08:24:10',0),
(39,'QC-PHMTZ-007',93,'pH',1,'','Title','205187','B. Ramvikas','2024-10-27','08:35:03',0),
(40,'QC-PHMTZ-007',97,'pH',1,'','Title','205089','M. Venkata Ramana','2024-10-28','07:17:38',0),
(41,'QC-PHMTZ-007',101,'pH',1,'','Title','205196','P. Mounika','2024-10-29','08:31:29',0),
(42,'QC-PHMTZ-007',104,'pH',1,'','Title','88599','MD Mahamad','2024-10-30','08:28:16',0),
(43,'QC-PHMTZ-007',111,'pH',1,'','Title','205187','B. Ramvikas','2024-11-01','07:48:07',0),
(44,'QC-PHMTZ-007',113,'pH',1,'','Title','205210','B. Anitha','2024-11-02','07:22:35',0),
(45,'QC-PHMTZ-007',118,'pH',1,'','Title','205089','M. Venkata Ramana','2024-11-04','17:25:59',0),
(46,'QC-PHMTZ-007',122,'pH',1,'','Title','205089','M. Venkata Ramana','2024-11-04','17:26:14',0),
(47,'QC-PHMTZ-007',123,'pH',1,'','Title','88571','V. Ramu','2024-11-05','07:45:52',0),
(48,'QC-PHMTZ-007',128,'pH',1,'','Title','205089','M. Venkata Ramana','2024-11-06','07:50:31',0),
(49,'QC-PHMTZ-007',131,'pH',1,'','Title','205119','S.M. Subhani','2024-11-07','07:56:07',0),
(50,'QC-PHMTZ-007',136,'pH',1,'','Title','205247','R. Puneetha','2024-11-08','07:52:35',0),
(51,'QC-PHMTZ-007',139,'pH',1,'','Title','205210','B. Anitha','2024-11-09','07:31:02',0),
(52,'QC-PHMTZ-007',139,'pH',2,'For Documentation purpose','Title','205210','B. Anitha','2024-11-09','07:45:31',0),
(53,'QC-PHMTZ-007',142,'pH',1,'','Title','205252','Y. Sivaiah','2024-11-10','08:32:27',0),
(54,'QC-PHMTZ-007',146,'pH',1,'','Title','205089','M. Venkata Ramana','2024-11-11','07:33:26',0),
(55,'QC-PHMTZ-007',149,'pH',1,'','Title','88571','V. Ramu','2024-11-12','07:42:01',0),
(56,'QC-PHMTZ-007',154,'pH',1,'','Title','88599','MD Mahamad','2024-11-13','07:23:29',0),
(57,'QC-PHMTZ-007',159,'pH',1,'','Title','205187','B. Ramvikas','2024-11-14','07:42:32',0),
(58,'QC-PHMTZ-007',161,'pH',1,'','Title','205210','B. Anitha','2024-11-15','07:31:54',0),
(59,'QC-PHMTZ-007',166,'pH',1,'','Title','205247','R. Puneetha','2024-11-16','07:38:30',0),
(60,'QC-PHMTZ-007',169,'pH',1,'','Title','205252','Y. Sivaiah','2024-11-17','09:10:03',0),
(61,'QC-PHMTZ-007',172,'pH',1,'','Title','205089','M. Venkata Ramana','2024-11-18','08:02:26',0),
(62,'QC-PHMTZ-007',183,'pH',1,'','Title','205089','M. Venkata Ramana','2024-11-20','08:52:56',0);

/*Table structure for table `tbl_conductivity_buffer` */


CREATE TABLE `tbl_conductivity_buffer` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `CB_bufferName` varchar(300) DEFAULT 'NULL',
  `CB_standardValue` decimal(10,5) DEFAULT '0.00000',
  `CB_IsUsed` int(1) DEFAULT '0' COMMENT '0:not used for calibration, 1: used for calibration',
  `CB_DP` int(10) DEFAULT '0',
  `locked` bit(1) DEFAULT b'0',
  `editCounter` int(11) DEFAULT NULL,
  `CB_NegTol` decimal(10,5) DEFAULT '0.00000',
  `CB_PosTol` decimal(10,5) DEFAULT '0.00000',
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_conductivity_buffer` */

/*Table structure for table `tbl_daq_master` */


CREATE TABLE `tbl_daq_master` (
  `SrNo` int(11) NOT NULL AUTO_INCREMENT,
  `Daq_SrNo` varchar(50) NOT NULL,
  `Port_Count` int(11) DEFAULT '0',
  `Port1_Id` int(11) DEFAULT NULL,
  `Port1_Instrument_Type` varchar(50) DEFAULT NULL,
  `Port2_Id` int(11) DEFAULT NULL,
  `Port2_Instrument_Type` varchar(50) DEFAULT NULL,
  `Port3_Id` int(11) DEFAULT NULL,
  `Port3_Instrument_Type` varchar(50) DEFAULT NULL,
  `Port4_Id` int(11) DEFAULT NULL,
  `Port4_Instrument_Type` varchar(50) DEFAULT NULL,
  `Port5_Id` int(11) DEFAULT NULL,
  `Port5_Instrument_Type` varchar(50) DEFAULT NULL,
  `Port6_Id` int(11) DEFAULT NULL,
  `Port6_Instrument_Type` varchar(50) DEFAULT NULL,
  `Port7_Id` int(11) DEFAULT NULL,
  `Port7_Instrument_Type` varchar(50) DEFAULT NULL,
  `Port8_Id` int(11) DEFAULT NULL,
  `Port8_Instrument_Type` varchar(50) DEFAULT NULL,
  PRIMARY KEY (`SrNo`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_daq_master` */

/*Table structure for table `tbl_department` */


CREATE TABLE `tbl_department` (
  `department_id` int(11) NOT NULL AUTO_INCREMENT,
  `department_name` varchar(50) NOT NULL DEFAULT 'NULL',
  `locked` bit(1) NOT NULL DEFAULT b'0',
  PRIMARY KEY (`department_id`),
  UNIQUE KEY `department_name` (`department_name`),
  UNIQUE KEY `department_name_2` (`department_name`)
) ENGINE=InnoDB AUTO_INCREMENT=6 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_department` */

insert  into `tbl_department`(`department_id`,`department_name`,`locked`) values 
(1,'QC-Chemical','\0'),
(2,'Production','\0'),
(3,'Service','\0'),
(4,'Testing','\0'),
(5,'Softwawre','\0');

/*Table structure for table `tbl_edit_status` */


CREATE TABLE `tbl_edit_status` (
  `id` int(11) NOT NULL AUTO_INCREMENT,
  `userID` varchar(50) NOT NULL DEFAULT 'NULL',
  `Type` varchar(50) NOT NULL DEFAULT 'NULL',
  `Item` varchar(400) NOT NULL DEFAULT 'NULL',
  `fieldName` varchar(100) NOT NULL DEFAULT 'NULL',
  `Ip` varchar(50) NOT NULL,
  PRIMARY KEY (`id`)
) ENGINE=InnoDB AUTO_INCREMENT=12 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_edit_status` */

/*Table structure for table `tbl_fp` */


CREATE TABLE `tbl_fp` (
  `id` int(11) NOT NULL AUTO_INCREMENT,
  `HMI_ip` varchar(10) DEFAULT '0',
  `IDS_ip` varchar(10) DEFAULT '0',
  `isBiometric` int(10) DEFAULT '0',
  `biometricMode` varchar(50) DEFAULT 'HMI' COMMENT 'HMI OR IDS',
  `isRIMode` varchar(10) DEFAULT 'R' COMMENT 'Registration Or Iddentification',
  PRIMARY KEY (`id`)
) ENGINE=InnoDB AUTO_INCREMENT=2 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_fp` */

insert  into `tbl_fp`(`id`,`HMI_ip`,`IDS_ip`,`isBiometric`,`biometricMode`,`isRIMode`) values 
(1,'171','154',0,'HMI','R');

/*Table structure for table `tbl_ids_setting` */


CREATE TABLE `tbl_ids_setting` (
  `Sys_RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `Sys_IDSNo` int(11) DEFAULT NULL,
  `Sys_Meter` varchar(200) DEFAULT NULL,
  `Sys_Printer` varchar(500) DEFAULT 'NA',
  `Sys_Port1_Type` varchar(200) DEFAULT 'None',
  `Sys_Port2_Type` varchar(200) DEFAULT 'None',
  `Sys_Port3_Type` varchar(200) DEFAULT 'None',
  `Sys_Port4_Type` varchar(200) DEFAULT 'None',
  `Sys_Port1_ID` varchar(200) DEFAULT 'None',
  `Sys_Port2_ID` varchar(200) DEFAULT 'None',
  `Sys_Port3_ID` varchar(200) DEFAULT 'None',
  `Sys_Port4_ID` varchar(200) DEFAULT 'None',
  `Sys_ISRecalib1` int(1) DEFAULT '0',
  `Sys_ISRecalib2` int(1) DEFAULT '0',
  `Sys_ISRecalib3` int(1) DEFAULT '0',
  `Sys_ISRecalib4` int(1) DEFAULT '0',
  `Sys_RecalibDate1` date DEFAULT NULL,
  `Sys_RecalibDate2` date DEFAULT NULL,
  `Sys_RecalibDate3` date DEFAULT NULL,
  `Sys_RecalibDate4` date DEFAULT NULL,
  `Sys_CalibInProcess` int(1) DEFAULT '0',
  `Sys_WeighmentInProcess` int(1) DEFAULT '0',
  `locked` bit(1) DEFAULT b'0',
  `editCounter` int(11) DEFAULT NULL,
  `Sys_Port1_unit` varchar(20) DEFAULT 'NULL',
  `Sys_Port2_unit` varchar(20) DEFAULT 'NULL',
  `Sys_Port3_unit` varchar(20) DEFAULT 'NULL',
  `Sys_Port4_unit` varchar(20) DEFAULT 'NULL',
  `Sys_HMIID` varchar(50) DEFAULT 'NULL',
  PRIMARY KEY (`Sys_RecNo`)
) ENGINE=InnoDB AUTO_INCREMENT=3 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_ids_setting` */

insert  into `tbl_ids_setting`(`Sys_RecNo`,`Sys_IDSNo`,`Sys_Meter`,`Sys_Printer`,`Sys_Port1_Type`,`Sys_Port2_Type`,`Sys_Port3_Type`,`Sys_Port4_Type`,`Sys_Port1_ID`,`Sys_Port2_ID`,`Sys_Port3_ID`,`Sys_Port4_ID`,`Sys_ISRecalib1`,`Sys_ISRecalib2`,`Sys_ISRecalib3`,`Sys_ISRecalib4`,`Sys_RecalibDate1`,`Sys_RecalibDate2`,`Sys_RecalibDate3`,`Sys_RecalibDate4`,`Sys_CalibInProcess`,`Sys_WeighmentInProcess`,`locked`,`editCounter`,`Sys_Port1_unit`,`Sys_Port2_unit`,`Sys_Port3_unit`,`Sys_Port4_unit`,`Sys_HMIID`) values 
(1,122,'12345','NA','pH','Conductivity','None','None','123PH','123COND','None','None',0,0,0,0,NULL,NULL,NULL,NULL,0,0,'\0',NULL,'pH','uS/cm','None','None','10000000303bf2a4'),
(2,123,'TS1062','NA','pH','None','None','None','1062','None','None','None',0,0,0,0,NULL,NULL,NULL,NULL,0,0,'\0',NULL,'pH','None','None','None','10000000f7824c6d');

/*Table structure for table `tbl_instrumentlog` */


CREATE TABLE `tbl_instrumentlog` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `MeterID` varchar(50) DEFAULT 'NULL',
  `ProbType` varchar(100) DEFAULT 'NULL',
  `ProbID` varchar(100) DEFAULT 'NULL',
  `FromDT` date DEFAULT NULL,
  `ToDT` date DEFAULT NULL,
  `FromTM` time DEFAULT NULL,
  `ToTM` time DEFAULT NULL,
  `Activity` varchar(50) DEFAULT 'NULL',
  `UserId` varchar(50) DEFAULT 'NULL',
  `UserName` varchar(50) DEFAULT 'NULL',
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB AUTO_INCREMENT=298 DEFAULT CHARSET=utf8;

/*Data for the table `tbl_instrumentlog` */

insert  into `tbl_instrumentlog`(`RecNo`,`MeterID`,`ProbType`,`ProbID`,`FromDT`,`ToDT`,`FromTM`,`ToTM`,`Activity`,`UserId`,`UserName`) values 
(1,'CX1-15733','pH','NULL','2024-09-13',NULL,'09:35:49',NULL,'pH Calibration','V-trident','Trident'),
(2,'CX1-15733','pH','NULL','2024-09-13','2024-09-13','09:36:30','09:36:30','pH Calibration','V-trident','Trident'),
(3,'CX1-15733','pH','NULL','2024-09-13','2024-09-13','09:47:50','09:47:50','pH Calibration','V-trident','Trident'),
(4,'CX1-15733','pH','NULL','2024-09-17','2024-09-17','15:03:17','15:03:17','pH Calibration','V-trident','Trident'),
(5,'CX1-16015','pH','NULL','2024-09-17','2024-09-17','15:15:56','15:15:56','pH Calibration','V-trident','Trident'),
(6,'CW1-12012','pH','NULL','2024-09-17','2024-09-17','16:34:53','16:34:53','pH Calibration','V-trident','Trident'),
(7,'CW1-12031','pH','NULL','2024-09-18','2024-09-18','14:30:09','14:30:09','pH Calibration','V-trident','Trident'),
(8,'CX1-15689','pH','NULL','2024-09-18','2024-09-18','15:41:50','15:41:50','pH Calibration','V-trident','Trident'),
(9,'CX1-16015','pH','NULL','2024-09-19','2024-09-19','10:39:12','10:39:13','pH Calibration','V-trident','Trident'),
(10,'CX1-15733','pH','NULL','2024-09-19','2024-09-19','12:01:55','12:01:55','pH Calibration','V-trident','Trident'),
(11,'CX1-15733','pH','NULL','2024-09-19','2024-09-19','12:30:37','12:30:37','pH Calibration','V-trident','Trident'),
(12,'CW1-12012','pH','NULL','2024-09-30','2024-09-30','16:13:13','16:13:13','pH Calibration','V-trident','Trident'),
(13,'CX1-15733','pH','NULL','2024-09-30','2024-09-30','16:51:44','16:51:45','pH Calibration','205089','M. Venkata Ramana'),
(14,'CX1-16015','pH','NULL','2024-09-30','2024-09-30','16:56:45','16:56:45','pH Calibration','205066','Sreeram Jakkaraiah'),
(15,'CX1-15733','pH','NULL','2024-10-01','2024-10-01','07:14:25','07:14:25','pH Calibration','88571','V. Ramu'),
(16,'CX1-16015','pH','NULL','2024-10-01','2024-10-01','07:20:21','07:20:21','pH Calibration','205089','M. Venkata Ramana'),
(17,'CX1-15689','pH','NULL','2024-10-01','2024-10-01','09:13:46','09:13:46','pH Calibration','205044','Prem kumar'),
(18,'CW1-12012','pH','NULL','2024-10-01','2024-10-01','09:23:43','09:23:43','pH Calibration','95422','M. Shankar'),
(19,'CW1-12012','pH','NULL','2024-10-01','2024-10-01','09:49:18','09:49:18','pH Calibration','95422','M. Shankar'),
(20,'CX1-15689','pH','NULL','2024-10-02','2024-10-02','08:22:35','08:22:35','pH Calibration','205044','Prem kumar'),
(21,'CX1-15689','pH','NULL','2024-10-03','2024-10-03','00:54:27','00:54:27','pH Calibration','205104','Siva Shankar Reddy'),
(22,'CX1-16015','pH','NULL','2024-10-03','2024-10-03','07:46:53','07:46:53','pH Calibration','205211','S. Sailatha'),
(23,'CW1-12012','pH','NULL','2024-10-03','2024-10-03','10:34:18','10:34:18','pH Calibration','92513','Ch. Chinna Rao'),
(24,'CW1-12012','pH','NULL','2024-10-03','2024-10-03','11:48:47','11:48:48','pH Calibration','92513','Ch. Chinna Rao'),
(25,'CX1-15689','pH','NULL','2024-10-04','2024-10-04','03:36:08','03:36:08','pH Calibration','205223','Jithendra'),
(26,'CX1-16015','pH','NULL','2024-10-04','2024-10-04','07:48:29','07:48:29','pH Calibration','205187','B. Ramvikas'),
(27,'CW1-12012','pH','NULL','2024-10-04','2024-10-04','10:28:36','10:28:36','pH Calibration','92513','Ch. Chinna Rao'),
(28,'CX1-16015','pH','NULL','2024-10-05','2024-10-05','07:52:03','07:52:03','pH Calibration','205210','B. Anitha'),
(29,'CX1-15733','pH','NULL','2024-10-05','2024-10-05','08:51:16','08:51:16','pH Calibration','205210','B. Anitha'),
(30,'CW1-12012','pH','NULL','2024-10-05','2024-10-05','10:41:49','10:41:49','pH Calibration','88819','M. Ramya'),
(31,'CX1-15689','pH','NULL','2024-10-05','2024-10-05','12:26:01','12:26:01','pH Calibration','84342','G.S.A. Naidu'),
(32,'CX1-16015','pH','NULL','2024-10-05','2024-10-05','14:44:50','14:44:51','pH Calibration','205210','B. Anitha'),
(33,'CX1-15689','pH','NULL','2024-10-06','2024-10-06','00:40:01','00:40:01','pH Calibration','205190','B. Prabhudas'),
(34,'CX1-16015','pH','NULL','2024-10-06','2024-10-06','08:46:26','08:46:26','pH Calibration','205187','B. Ramvikas'),
(35,'CX1-16015','pH','NULL','2024-10-07','2024-10-07','07:10:43','07:10:43','pH Calibration','205089','M. Venkata Ramana'),
(36,'CX1-15689','pH','NULL','2024-10-07','2024-10-07','07:42:12','07:42:12','pH Calibration','205223','Jithendra'),
(37,'CX1-15733','pH','NULL','2024-10-07','2024-10-07','08:28:28','08:28:28','pH Calibration','205089','M. Venkata Ramana'),
(38,'CW1-12012','pH','NULL','2024-10-07','2024-10-07','11:49:25','11:49:25','pH Calibration','92513','Ch. Chinna Rao'),
(39,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:15:07',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(40,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:16:40',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(41,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:18:36',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(42,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:19:33',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(43,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:20:01',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(44,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:20:19',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(45,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:22:53',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(46,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:25:08',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(47,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:25:36',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(48,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:25:57',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(49,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:31:07',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(50,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:32:06',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(51,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:33:30',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(52,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:34:20',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(53,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:35:33',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(54,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:36:24',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(55,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:36:53',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(56,'CW1-12012','pH','NULL','2024-10-07',NULL,'16:40:13',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(57,'CW1-12012','pH','NULL','2024-10-07',NULL,'17:00:41',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(58,'CW1-12012','pH','NULL','2024-10-07',NULL,'17:07:27',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(59,'CW1-12012','pH','NULL','2024-10-07',NULL,'17:08:47',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(60,'CW1-12012','pH','NULL','2024-10-07',NULL,'17:16:16',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(61,'CW1-12012','pH','NULL','2024-10-07',NULL,'17:50:25',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(62,'CW1-12012','pH','NULL','2024-10-07','2024-10-07','18:01:12','18:01:12','pH Calibration','92513','Ch. Chinna Rao'),
(63,'CX1-15733','pH','NULL','2024-10-08','2024-10-08','07:18:36','07:18:36','pH Calibration','205089','M. Venkata Ramana'),
(64,'CW1-12012','pH','NULL','2024-10-08','2024-10-08','10:10:38','10:10:38','pH Calibration','88819','M. Ramya'),
(65,'CX1-15689','pH','NULL','2024-10-08',NULL,'15:36:15',NULL,'pH Calibration','205104','Siva Shankar Reddy'),
(66,'CX1-15689','pH','NULL','2024-10-08',NULL,'16:05:08',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(67,'CX1-15689','pH','NULL','2024-10-08',NULL,'16:05:23',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(68,'CX1-15689','pH','NULL','2024-10-08',NULL,'16:07:59',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(69,'CX1-15689','pH','NULL','2024-10-08',NULL,'16:09:01',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(70,'CX1-15689','pH','NULL','2024-10-08',NULL,'16:09:20',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(71,'CX1-15689','pH','NULL','2024-10-08','2024-10-08','16:56:45','16:56:45','pH Calibration','205104','N. Siva Sekhar Reddy'),
(72,'CX1-15689','pH','NULL','2024-10-09','2024-10-09','00:32:51','00:32:51','pH Calibration','205044','Prem kumar'),
(73,'CX1-15733','pH','NULL','2024-10-09','2024-10-09','07:15:50','07:15:50','pH Calibration','88599','MD Mahamad'),
(74,'CW1-12012','pH','NULL','2024-10-09','2024-10-09','08:47:12','08:47:12','pH Calibration','92513','Ch. Chinna Rao'),
(75,'CX1-15689','pH','NULL','2024-10-10','2024-10-10','05:18:39','05:18:39','pH Calibration','205223','Jithendra'),
(76,'CX1-15733','pH','NULL','2024-10-10','2024-10-10','07:33:03','07:33:04','pH Calibration','205075','G. Shyam Kumar'),
(77,'CW1-12012','pH','NULL','2024-10-10','2024-10-10','08:47:49','08:47:49','pH Calibration','92513','Ch. Chinna Rao'),
(78,'CX1-15733','pH','NULL','2024-10-11','2024-10-11','00:30:28','00:30:29','pH Calibration','88571','V. Ramu'),
(79,'CW1-12012','pH','NULL','2024-10-11','2024-10-11','07:14:59','07:14:59','pH Calibration','88819','M. Ramya'),
(80,'CX1-15689','pH','NULL','2024-10-11','2024-10-11','14:19:44','14:19:44','pH Calibration','205104','N. Siva Sekhar Reddy'),
(81,'CX1-16015','pH','NULL','2024-10-11','2024-10-11','15:25:35','15:25:35','pH Calibration','87624','N. Nani babu'),
(82,'CX1-15689','pH','NULL','2024-10-12','2024-10-12','15:32:10','15:32:10','pH Calibration','84342','G.S.A. Naidu'),
(83,'CX1-15689','pH','NULL','2024-10-13',NULL,'00:26:48',NULL,'pH Calibration','205190','B. Prabhudas'),
(84,'CX1-15689','pH','NULL','2024-10-13',NULL,'00:27:27',NULL,'pH Calibration','205190','B. Prabhudas'),
(85,'CX1-15689','pH','NULL','2024-10-13',NULL,'00:28:37',NULL,'pH Calibration','205190','B. Prabhudas'),
(86,'CX1-15689','pH','NULL','2024-10-13','2024-10-13','00:30:25','00:30:25','pH Calibration','205190','B. Prabhudas'),
(87,'CX1-15733','pH','NULL','2024-10-13','2024-10-13','10:46:34','10:46:34','pH Calibration','88655','V. Prashanth Kumar'),
(88,'CX1-15689','pH','NULL','2024-10-14','2024-10-14','06:34:46','06:34:46','pH Calibration','205223','Jithendra'),
(89,'CX1-15733','pH','NULL','2024-10-14','2024-10-14','07:30:17','07:30:17','pH Calibration','88571','V. Ramu'),
(90,'CW1-12012','pH','NULL','2024-10-14','2024-10-14','11:51:21','11:51:21','pH Calibration','92513','Ch. Chinna Rao'),
(91,'CX1-15733','pH','NULL','2024-10-15','2024-10-15','07:10:03','07:10:03','pH Calibration','88599','MD Mahamad'),
(92,'CX1-15689','pH','NULL','2024-10-15','2024-10-15','10:30:28','10:30:28','pH Calibration','205190','B. Prabhudas'),
(93,'CX1-16015','pH','NULL','2024-10-15','2024-10-15','12:00:49','12:00:49','pH Calibration','88599','MD Mahamad'),
(94,'CX1-15733','pH','NULL','2024-10-16','2024-10-16','07:06:13','07:06:13','pH Calibration','205090','M. Srikrishnudu'),
(95,'CW1-12012','pH','NULL','2024-10-16',NULL,'11:48:45',NULL,'pH Calibration','95422','M. Shankar'),
(96,'CW1-12012','pH','NULL','2024-10-16','2024-10-16','11:59:17','11:59:17','pH Calibration','95422','M. Shankar'),
(97,'CX1-15689','pH','NULL','2024-10-17','2024-10-17','03:03:44','03:03:44','pH Calibration','205223','Jithendra'),
(98,'CX1-15733','pH','NULL','2024-10-17','2024-10-17','07:15:07','07:15:07','pH Calibration','205211','S. Sailatha'),
(99,'CW1-12012','pH','NULL','2024-10-17','2024-10-17','08:46:59','08:46:59','pH Calibration','95422','M. Shankar'),
(100,'CX1-15733','pH','NULL','2024-10-18','2024-10-18','07:30:40','07:30:40','pH Calibration','205247','R. Puneetha'),
(101,'CW1-12012','pH','NULL','2024-10-18','2024-10-18','08:58:01','08:58:01','pH Calibration','95422','M. Shankar'),
(102,'CX1-15689','pH','NULL','2024-10-19','2024-10-19','00:35:57','00:35:57','pH Calibration','205190','B. Prabhudas'),
(103,'CX1-15733','pH','NULL','2024-10-19','2024-10-19','07:16:31','07:16:31','pH Calibration','205210','B. Anitha'),
(104,'CW1-12012','pH','NULL','2024-10-19','2024-10-19','09:51:31','09:51:31','pH Calibration','88819','M. Ramya'),
(105,'CX1-15689','pH','NULL','2024-10-20','2024-10-20','02:54:30','02:54:30','pH Calibration','205104','N. Siva Sekhar Reddy'),
(106,'CX1-15733','pH','NULL','2024-10-21','2024-10-21','07:02:34','07:02:34','pH Calibration','205089','M. Venkata Ramana'),
(107,'CW1-12012','pH','NULL','2024-10-21','2024-10-21','09:56:32','09:56:32','pH Calibration','92513','Ch. Chinna Rao'),
(108,'CX1-15689','pH','NULL','2024-10-21','2024-10-21','17:50:11','17:50:11','pH Calibration','205223','Jithendra'),
(109,'CX1-15689','pH','NULL','2024-10-22',NULL,'00:19:40',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(110,'CX1-15689','pH','NULL','2024-10-22',NULL,'00:20:38',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(111,'CX1-15689','pH','NULL','2024-10-22',NULL,'00:21:26',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(112,'CX1-15689','pH','NULL','2024-10-22',NULL,'00:22:45',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(113,'CX1-15689','pH','NULL','2024-10-22',NULL,'00:27:22',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(114,'CX1-15689','pH','NULL','2024-10-22',NULL,'00:29:31',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(115,'CX1-15689','pH','NULL','2024-10-22',NULL,'00:29:52',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(116,'CX1-15689','pH','NULL','2024-10-22','2024-10-22','00:47:01','00:47:01','pH Calibration','205104','N. Siva Sekhar Reddy'),
(117,'CX1-15733','pH','NULL','2024-10-22','2024-10-22','07:02:23','07:02:23','pH Calibration','205089','M. Venkata Ramana'),
(118,'CW1-12012','pH','NULL','2024-10-22','2024-10-22','09:08:47','09:08:47','pH Calibration','92513','Ch. Chinna Rao'),
(119,'CW1-12012','pH','NULL','2024-10-22',NULL,'12:46:02',NULL,'pH Calibration','92513','Ch. Chinna Rao'),
(120,'CW1-12012','pH','NULL','2024-10-22','2024-10-22','15:29:06','15:29:06','pH Calibration','95422','M. Shankar'),
(121,'CX1-15689','pH','NULL','2024-10-23','2024-10-23','00:52:38','00:52:38','pH Calibration','205104','N. Siva Sekhar Reddy'),
(122,'CX1-15733','pH','NULL','2024-10-23','2024-10-23','07:20:13','07:20:14','pH Calibration','88599','MD Mahamad'),
(123,'CW1-12012','pH','NULL','2024-10-23','2024-10-23','09:43:17','09:43:17','pH Calibration','95422','M. Shankar'),
(124,'CX1-15733','pH','NULL','2024-10-24','2024-10-24','07:49:07','07:49:07','pH Calibration','205211','S. Sailatha'),
(125,'CW1-12012','pH','NULL','2024-10-24','2024-10-24','08:54:22','08:54:22','pH Calibration','88819','M. Ramya'),
(126,'CX1-15689','pH','NULL','2024-10-24','2024-10-24','10:56:28','10:56:28','pH Calibration','205223','Jithendra'),
(127,'CX1-15733','pH','NULL','2024-10-25','2024-10-25','06:59:10','06:59:10','pH Calibration','205247','R. Puneetha'),
(128,'CW1-12012','pH','NULL','2024-10-25','2024-10-25','09:48:39','09:48:39','pH Calibration','88819','M. Ramya'),
(129,'CX1-15689','pH','NULL','2024-10-25','2024-10-25','19:29:42','19:29:42','pH Calibration','205224','K. Uma Mahesh'),
(130,'CW1-12031','pH','NULL','2024-10-25','2024-10-25','20:05:25','20:05:25','pH Calibration','205169','M. Venkatesham'),
(131,'CX1-15689','pH','NULL','2024-10-26','2024-10-26','07:08:10','07:08:10','pH Calibration','205223','Jithendra'),
(132,'CX1-15733','pH','NULL','2024-10-26','2024-10-26','07:16:06','07:16:06','pH Calibration','205210','B. Anitha'),
(133,'CW1-12012','pH','NULL','2024-10-26','2024-10-26','10:17:47','10:17:47','pH Calibration','88819','M. Ramya'),
(134,'CW1-12031','pH','NULL','2024-10-26','2024-10-26','12:46:38','12:46:38','pH Calibration','205117','V. Seenaiah'),
(135,'CX1-15733','pH','NULL','2024-10-27','2024-10-27','07:26:18','07:26:18','pH Calibration','205187','B. Ramvikas'),
(136,'CX1-15689','pH','NULL','2024-10-27','2024-10-27','08:59:52','08:59:52','pH Calibration','84342','G.S.A. Naidu'),
(137,'CW1-12031','pH','NULL','2024-10-27','2024-10-27','09:25:55','09:25:55','pH Calibration','205169','M. Venkatesham'),
(138,'CW1-12031','pH','NULL','2024-10-28',NULL,'00:23:59',NULL,'pH Calibration','205117','V. Seenaiah'),
(139,'CW1-12031','pH','NULL','2024-10-28',NULL,'00:26:21',NULL,'pH Calibration','205117','V. Seenaiah'),
(140,'CW1-12031','pH','NULL','2024-10-28','2024-10-28','00:31:21','00:31:21','pH Calibration','205117','V. Seenaiah'),
(141,'CX1-15733','pH','NULL','2024-10-28','2024-10-28','06:54:42','06:54:42','pH Calibration','205089','M. Venkata Ramana'),
(142,'CW1-12012','pH','NULL','2024-10-28','2024-10-28','09:18:48','09:18:48','pH Calibration','95422','M. Shankar'),
(143,'CX1-15689','pH','NULL','2024-10-28','2024-10-28','19:41:17','19:41:17','pH Calibration','205224','K. Uma Mahesh'),
(144,'CX1-15689','pH','NULL','2024-10-29','2024-10-29','05:14:00','05:14:00','pH Calibration','92936','Sameer'),
(145,'CX1-15733','pH','NULL','2024-10-29','2024-10-29','07:44:03','07:44:03','pH Calibration','205196','P. Mounika'),
(146,'CW1-12012','pH','NULL','2024-10-29','2024-10-29','10:47:36','10:47:36','pH Calibration','95422','M. Shankar'),
(147,'CW1-12031','pH','NULL','2024-10-29','2024-10-29','15:46:50','15:46:50','pH Calibration','92595','K. Murali Krishna'),
(148,'CX1-15733','pH','NULL','2024-10-30','2024-10-30','07:48:40','07:48:40','pH Calibration','88599','MD Mahamad'),
(149,'CX1-15689','pH','NULL','2024-10-30','2024-10-30','07:53:51','07:53:51','pH Calibration','205044','Prem kumar'),
(150,'CW1-12012','pH','NULL','2024-10-30','2024-10-30','08:53:51','08:53:51','pH Calibration','95422','M. Shankar'),
(151,'CW1-12031','pH','NULL','2024-10-30','2024-10-30','09:19:53','09:19:53','pH Calibration','91700','Ch. Ramarao'),
(152,'CX1-15689','pH','NULL','2024-10-31','2024-10-31','00:51:41','00:51:41','pH Calibration','92936','Sameer'),
(153,'CW1-12031','pH','NULL','2024-10-31','2024-10-31','15:14:14','15:14:14','pH Calibration','205169','M. Venkatesham'),
(154,'CX1-15689','pH','NULL','2024-11-01','2024-11-01','03:07:14','03:07:14','pH Calibration','205044','Prem kumar'),
(155,'CX1-15733','pH','NULL','2024-11-01','2024-11-01','07:10:53','07:10:53','pH Calibration','205187','B. Ramvikas'),
(156,'CW1-12031','pH','NULL','2024-11-01','2024-11-01','10:18:46','10:18:46','pH Calibration','92595','K. Murali Krishna'),
(157,'CX1-15733','pH','NULL','2024-11-02','2024-11-02','07:01:54','07:01:54','pH Calibration','205210','B. Anitha'),
(158,'CW1-12012','pH','NULL','2024-11-02','2024-11-02','07:50:14','07:50:14','pH Calibration','88819','M. Ramya'),
(159,'CW1-12031','pH','NULL','2024-11-02','2024-11-02','10:02:25','10:02:25','pH Calibration','92595','K. Murali Krishna'),
(160,'CX1-15689','pH','NULL','2024-11-02','2024-11-02','10:24:16','10:24:16','pH Calibration','205224','K. Uma Mahesh'),
(161,'CX1-15689','pH','NULL','2024-11-03',NULL,'00:19:56',NULL,'pH Calibration','205223','Jithendra'),
(162,'CX1-15689','pH','NULL','2024-11-03',NULL,'00:20:15',NULL,'pH Calibration','205223','Jithendra'),
(163,'CX1-15689','pH','NULL','2024-11-03',NULL,'00:20:56',NULL,'pH Calibration','205223','Jithendra'),
(164,'CX1-15689','pH','NULL','2024-11-03',NULL,'00:21:24',NULL,'pH Calibration','205223','Jithendra'),
(165,'CX1-15689','pH','NULL','2024-11-03',NULL,'00:22:04',NULL,'pH Calibration','205223','Jithendra'),
(166,'CX1-15689','pH','NULL','2024-11-03','2024-11-03','00:34:24','00:34:24','pH Calibration','205223','Jithendra'),
(167,'CX1-15733','pH','NULL','2024-11-04','2024-11-04','07:04:02','07:04:02','pH Calibration','205089','M. Venkata Ramana'),
(168,'CW1-12031','pH','NULL','2024-11-04','2024-11-04','07:18:01','07:18:01','pH Calibration','205169','M. Venkatesham'),
(169,'CW1-12012','pH','NULL','2024-11-04','2024-11-04','09:13:31','09:13:31','pH Calibration','88819','M. Ramya'),
(170,'CX1-15689','pH','NULL','2024-11-04','2024-11-04','12:35:33','12:35:33','pH Calibration','205104','N. Siva Sekhar Reddy'),
(171,'CX1-15733','pH','NULL','2024-11-04','2024-11-04','15:35:12','15:35:12','pH Calibration','205089','M. Venkata Ramana'),
(172,'CX1-15689','pH','NULL','2024-11-05',NULL,'00:19:06',NULL,'pH Calibration','205223','Jithendra'),
(173,'CX1-15689','pH','NULL','2024-11-05',NULL,'00:19:59',NULL,'pH Calibration','205223','Jithendra'),
(174,'CX1-15689','pH','NULL','2024-11-05',NULL,'00:20:36',NULL,'pH Calibration','205223','Jithendra'),
(175,'CX1-15689','pH','NULL','2024-11-05',NULL,'00:21:28',NULL,'pH Calibration','205223','Jithendra'),
(176,'CX1-15689','pH','NULL','2024-11-05',NULL,'00:22:04',NULL,'pH Calibration','205223','Jithendra'),
(177,'CX1-15689','pH','NULL','2024-11-05',NULL,'00:23:04',NULL,'pH Calibration','205223','Jithendra'),
(178,'CX1-15689','pH','NULL','2024-11-05',NULL,'00:23:35',NULL,'pH Calibration','205223','Jithendra'),
(179,'CX1-15689','pH','NULL','2024-11-05',NULL,'00:28:51',NULL,'pH Calibration','205223','Jithendra'),
(180,'CX1-15689','pH','NULL','2024-11-05',NULL,'01:27:56',NULL,'pH Calibration','205223','Jithendra'),
(181,'CX1-15689','pH','NULL','2024-11-05',NULL,'01:30:56',NULL,'pH Calibration','205223','Jithendra'),
(182,'CX1-15689','pH','NULL','2024-11-05',NULL,'01:33:43',NULL,'pH Calibration','205223','Jithendra'),
(183,'CX1-15689','pH','NULL','2024-11-05',NULL,'01:37:18',NULL,'pH Calibration','205223','Jithendra'),
(184,'CX1-15689','pH','NULL','2024-11-05',NULL,'01:38:18',NULL,'pH Calibration','205223','Jithendra'),
(185,'CX1-15733','pH','NULL','2024-11-05','2024-11-05','07:15:13','07:15:13','pH Calibration','88571','V. Ramu'),
(186,'CX1-15689','pH','NULL','2024-11-05',NULL,'08:51:20',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(187,'CW1-12012','pH','NULL','2024-11-05','2024-11-05','08:59:16','08:59:16','pH Calibration','92513','Ch. Chinna Rao'),
(188,'CX1-15689','pH','NULL','2024-11-05',NULL,'09:01:17',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(189,'CX1-15689','pH','NULL','2024-11-05',NULL,'09:39:05',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(190,'CX1-15689','pH','NULL','2024-11-05',NULL,'10:05:38',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(191,'CX1-15689','pH','NULL','2024-11-05',NULL,'10:05:53',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(192,'CX1-15689','pH','NULL','2024-11-05',NULL,'10:06:43',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(193,'CX1-15689','pH','NULL','2024-11-05',NULL,'10:09:59',NULL,'pH Calibration','205104','N. Siva Sekhar Reddy'),
(194,'CX1-15689','pH','NULL','2024-11-05','2024-11-05','11:12:41','11:12:41','pH Calibration','205104','N. Siva Sekhar Reddy'),
(195,'CW1-12031','pH','NULL','2024-11-05','2024-11-05','16:43:36','16:43:36','pH Calibration','85705','P. Prakash Rao'),
(196,'CX1-15689','pH','NULL','2024-11-06','2024-11-06','00:39:50','00:39:50','pH Calibration','84342','G.S.A. Naidu'),
(197,'CX1-15733','pH','NULL','2024-11-06','2024-11-06','07:20:19','07:20:19','pH Calibration','205089','M. Venkata Ramana'),
(198,'CW1-12031','pH','NULL','2024-11-06','2024-11-06','09:49:49','09:49:49','pH Calibration','85705','P. Prakash Rao'),
(199,'CW1-12012','pH','NULL','2024-11-06','2024-11-06','10:11:34','10:11:34','pH Calibration','92513','Ch. Chinna Rao'),
(200,'CX1-15733','pH','NULL','2024-11-07','2024-11-07','07:09:15','07:09:15','pH Calibration','205119','S.M. Subhani'),
(201,'CW1-12031','pH','NULL','2024-11-07','2024-11-07','10:05:11','10:05:11','pH Calibration','85705','P. Prakash Rao'),
(202,'CW1-12012','pH','NULL','2024-11-07','2024-11-07','10:06:14','10:06:15','pH Calibration','92513','Ch. Chinna Rao'),
(203,'CX1-15689','pH','NULL','2024-11-07',NULL,'10:53:42',NULL,'pH Calibration','205190','B. Prabhudas'),
(204,'CX1-15689','pH','NULL','2024-11-07','2024-11-07','11:05:37','11:05:37','pH Calibration','205190','B. Prabhudas'),
(205,'CX1-15689','pH','NULL','2024-11-08','2024-11-08','00:31:31','00:31:31','pH Calibration','205223','Jithendra'),
(206,'CX1-15733','pH','NULL','2024-11-08','2024-11-08','07:13:05','07:13:05','pH Calibration','205247','R. Puneetha'),
(207,'CW1-12012','pH','NULL','2024-11-08','2024-11-08','09:22:55','09:22:55','pH Calibration','92513','Ch. Chinna Rao'),
(208,'CW1-12031','pH','NULL','2024-11-08','2024-11-08','15:39:29','15:39:29','pH Calibration','88332','P.V.A.K. Raju'),
(209,'CX1-15733','pH','NULL','2024-11-09','2024-11-09','07:03:01','07:03:01','pH Calibration','205210','B. Anitha'),
(210,'CW1-12031','pH','NULL','2024-11-09','2024-11-09','09:52:09','09:52:09','pH Calibration','85705','P. Prakash Rao'),
(211,'CX1-15689','pH','NULL','2024-11-09','2024-11-09','10:27:15','10:27:15','pH Calibration','205190','B. Prabhudas'),
(212,'CX1-15733','pH','NULL','2024-11-10','2024-11-10','07:44:14','07:44:14','pH Calibration','205252','Y. Sivaiah'),
(213,'CX1-15689','pH','NULL','2024-11-10','2024-11-10','10:20:09','10:20:09','pH Calibration','92936','Sameer'),
(214,'CW1-12031','pH','NULL','2024-11-10','2024-11-10','11:15:50','11:15:50','pH Calibration','205169','M. Venkatesham'),
(215,'CX1-15689','pH','NULL','2024-11-11',NULL,'00:09:56',NULL,'pH Calibration','205190','B. Prabhudas'),
(216,'CX1-15689','pH','NULL','2024-11-11',NULL,'00:11:33',NULL,'pH Calibration','205190','B. Prabhudas'),
(217,'CX1-15689','pH','NULL','2024-11-11',NULL,'00:11:51',NULL,'pH Calibration','205190','B. Prabhudas'),
(218,'CX1-15689','pH','NULL','2024-11-11',NULL,'00:13:57',NULL,'pH Calibration','205190','B. Prabhudas'),
(219,'CX1-15689','pH','NULL','2024-11-11',NULL,'00:20:45',NULL,'pH Calibration','205190','B. Prabhudas'),
(220,'CX1-15689','pH','NULL','2024-11-11',NULL,'00:22:59',NULL,'pH Calibration','205190','B. Prabhudas'),
(221,'CX1-15689','pH','NULL','2024-11-11',NULL,'00:23:25',NULL,'pH Calibration','205190','B. Prabhudas'),
(222,'CX1-15689','pH','NULL','2024-11-11',NULL,'00:23:40',NULL,'pH Calibration','205190','B. Prabhudas'),
(223,'CX1-15689','pH','NULL','2024-11-11',NULL,'00:24:11',NULL,'pH Calibration','205190','B. Prabhudas'),
(224,'CX1-15689','pH','NULL','2024-11-11',NULL,'00:24:45',NULL,'pH Calibration','205190','B. Prabhudas'),
(225,'CX1-15689','pH','NULL','2024-11-11',NULL,'00:26:42',NULL,'pH Calibration','205190','B. Prabhudas'),
(226,'CX1-15689','pH','NULL','2024-11-11',NULL,'00:28:44',NULL,'pH Calibration','88332','P.V.A.K. Raju'),
(227,'CX1-15689','pH','NULL','2024-11-11',NULL,'00:29:31',NULL,'pH Calibration','88332','P.V.A.K. Raju'),
(228,'CX1-15689','pH','NULL','2024-11-11',NULL,'00:29:42',NULL,'pH Calibration','88332','P.V.A.K. Raju'),
(229,'CX1-15689','pH','NULL','2024-11-11','2024-11-11','00:32:44','00:32:44','pH Calibration','205190','B. Prabhudas'),
(230,'CX1-15733','pH','NULL','2024-11-11','2024-11-11','06:52:39','06:52:39','pH Calibration','205089','M. Venkata Ramana'),
(231,'CW1-12012','pH','NULL','2024-11-11','2024-11-11','10:15:03','10:15:03','pH Calibration','88819','M. Ramya'),
(232,'CW1-12031','pH','NULL','2024-11-11','2024-11-11','11:35:00','11:35:00','pH Calibration','205007','T. Siva Ranganath'),
(233,'CX1-15733','pH','NULL','2024-11-12','2024-11-12','07:09:10','07:09:10','pH Calibration','88571','V. Ramu'),
(234,'CX1-15689','pH','NULL','2024-11-12','2024-11-12','08:26:18','08:26:18','pH Calibration','205223','Jithendra'),
(235,'CW1-12031','pH','NULL','2024-11-12','2024-11-12','09:49:30','09:49:30','pH Calibration','91700','Ch. Ramarao'),
(236,'CW1-12012','pH','NULL','2024-11-12','2024-11-12','14:21:15','14:21:15','pH Calibration','88819','M. Ramya'),
(237,'CW1-12031','pH','NULL','2024-11-13','2024-11-13','01:01:30','01:01:30','pH Calibration','85705','P. Prakash Rao'),
(238,'CX1-15733','pH','NULL','2024-11-13','2024-11-13','07:02:33','07:02:33','pH Calibration','88599','MD Mahamad'),
(239,'CW1-12012','pH','NULL','2024-11-13','2024-11-13','08:31:41','08:31:41','pH Calibration','88819','M. Ramya'),
(240,'CX1-15689','pH','NULL','2024-11-13','2024-11-13','08:37:40','08:37:40','pH Calibration','205190','B. Prabhudas'),
(241,'CX1-15689','pH','NULL','2024-11-14',NULL,'00:10:49',NULL,'pH Calibration','205044','Prem kumar'),
(242,'CX1-15689','pH','NULL','2024-11-14',NULL,'00:11:36',NULL,'pH Calibration','205044','Prem kumar'),
(243,'CX1-15689','pH','NULL','2024-11-14',NULL,'00:12:48',NULL,'pH Calibration','205044','Prem kumar'),
(244,'CX1-15689','pH','NULL','2024-11-14',NULL,'00:13:08',NULL,'pH Calibration','205044','Prem kumar'),
(245,'CX1-15689','pH','NULL','2024-11-14',NULL,'00:15:19',NULL,'pH Calibration','205044','Prem kumar'),
(246,'CX1-15689','pH','NULL','2024-11-14',NULL,'00:17:40',NULL,'pH Calibration','205044','Prem kumar'),
(247,'CX1-15689','pH','NULL','2024-11-14',NULL,'00:18:22',NULL,'pH Calibration','205044','Prem kumar'),
(248,'CX1-15689','pH','NULL','2024-11-14','2024-11-14','00:38:40','00:38:40','pH Calibration','205044','Prem kumar'),
(249,'CW1-12031','pH','NULL','2024-11-14','2024-11-14','01:31:36','01:31:36','pH Calibration','205117','V. Seenaiah'),
(250,'CX1-15733','pH','NULL','2024-11-14','2024-11-14','07:08:35','07:08:35','pH Calibration','205187','B. Ramvikas'),
(251,'CW1-12012','pH','NULL','2024-11-14','2024-11-14','12:53:10','12:53:10','pH Calibration','88819','M. Ramya'),
(252,'CX1-15733','pH','NULL','2024-11-15','2024-11-15','07:04:52','07:04:52','pH Calibration','205210','B. Anitha'),
(253,'CX1-15689','pH','NULL','2024-11-15','2024-11-15','07:09:14','07:09:14','pH Calibration','92936','Sameer'),
(254,'CW1-12012','pH','NULL','2024-11-15','2024-11-15','10:20:36','10:20:36','pH Calibration','95422','M. Shankar'),
(255,'CW1-12031','pH','NULL','2024-11-15','2024-11-15','10:52:57','10:52:57','pH Calibration','88332','P.V.A.K. Raju'),
(256,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:10:55',NULL,'pH Calibration','85705','P. Prakash Rao'),
(257,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:11:14',NULL,'pH Calibration','85705','P. Prakash Rao'),
(258,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:11:39',NULL,'pH Calibration','85705','P. Prakash Rao'),
(259,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:11:54',NULL,'pH Calibration','85705','P. Prakash Rao'),
(260,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:12:38',NULL,'pH Calibration','85705','P. Prakash Rao'),
(261,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:14:25',NULL,'pH Calibration','85705','P. Prakash Rao'),
(262,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:14:48',NULL,'pH Calibration','85705','P. Prakash Rao'),
(263,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:15:08',NULL,'pH Calibration','85705','P. Prakash Rao'),
(264,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:15:17',NULL,'pH Calibration','85705','P. Prakash Rao'),
(265,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:15:29',NULL,'pH Calibration','85705','P. Prakash Rao'),
(266,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:17:10',NULL,'pH Calibration','85705','P. Prakash Rao'),
(267,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:17:57',NULL,'pH Calibration','85705','P. Prakash Rao'),
(268,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:18:25',NULL,'pH Calibration','85705','P. Prakash Rao'),
(269,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:20:40',NULL,'pH Calibration','85705','P. Prakash Rao'),
(270,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:21:41',NULL,'pH Calibration','85705','P. Prakash Rao'),
(271,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:25:19',NULL,'pH Calibration','85705','P. Prakash Rao'),
(272,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:25:58',NULL,'pH Calibration','85705','P. Prakash Rao'),
(273,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:26:25',NULL,'pH Calibration','85705','P. Prakash Rao'),
(274,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:26:35',NULL,'pH Calibration','85705','P. Prakash Rao'),
(275,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:26:41',NULL,'pH Calibration','85705','P. Prakash Rao'),
(276,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:26:47',NULL,'pH Calibration','85705','P. Prakash Rao'),
(277,'CW1-12031','pH','NULL','2024-11-16',NULL,'00:27:09',NULL,'pH Calibration','85705','P. Prakash Rao'),
(278,'CW1-12031','pH','NULL','2024-11-16','2024-11-16','00:31:38','00:31:38','pH Calibration','85705','P. Prakash Rao'),
(279,'CX1-15733','pH','NULL','2024-11-16','2024-11-16','07:12:25','07:12:25','pH Calibration','205247','R. Puneetha'),
(280,'CW1-12012','pH','NULL','2024-11-16','2024-11-16','10:42:03','10:42:03','pH Calibration','95422','M. Shankar'),
(281,'CX1-15689','pH','NULL','2024-11-16','2024-11-16','23:25:37','23:25:37','pH Calibration','205190','B. Prabhudas'),
(282,'CX1-15733','pH','NULL','2024-11-17','2024-11-17','08:04:10','08:04:10','pH Calibration','205252','Y. Sivaiah'),
(283,'CW1-12031','pH','NULL','2024-11-17','2024-11-17','09:58:13','09:58:13','pH Calibration','205223','Jithendra'),
(284,'CX1-15689','pH','NULL','2024-11-17','2024-11-17','23:16:03','23:16:04','pH Calibration','205190','B. Prabhudas'),
(285,'CX1-15733','pH','NULL','2024-11-18','2024-11-18','06:48:06','06:48:06','pH Calibration','205089','M. Venkata Ramana'),
(286,'CX1-15689','pH','NULL','2024-11-18','2024-11-18','07:18:28','07:18:28','pH Calibration','205223','Jithendra'),
(287,'CW1-12031','pH','NULL','2024-11-18','2024-11-18','09:27:45','09:27:45','pH Calibration','88332','P.V.A.K. Raju'),
(288,'CW1-12012','pH','NULL','2024-11-18','2024-11-18','11:17:40','11:17:40','pH Calibration','92513','Ch. Chinna Rao'),
(289,'CW1-12031','pH','NULL','2024-11-19','2024-11-19','00:49:12','00:49:12','pH Calibration','205007','T. Siva Ranganath'),
(290,'CX1-15733','pH','NULL','2024-11-19','2024-11-19','07:11:17','07:11:17','pH Calibration','205089','M. Venkata Ramana'),
(291,'CW1-12012','pH','NULL','2024-11-19','2024-11-19','10:04:16','10:04:16','pH Calibration','92513','Ch. Chinna Rao'),
(292,'CX1-15689','pH','NULL','2024-11-19','2024-11-19','11:51:40','11:51:40','pH Calibration','205104','N. Siva Sekhar Reddy'),
(293,'CW1-12012','pH','NULL','2024-11-19','2024-11-19','14:38:01','14:38:01','pH Calibration','92513','Ch. Chinna Rao'),
(294,'CX1-15689','pH','NULL','2024-11-20','2024-11-20','00:37:59','00:37:59','pH Calibration','92936','Sameer'),
(295,'CW1-12031','pH','NULL','2024-11-20','2024-11-20','05:51:10','05:51:10','pH Calibration','205117','V. Seenaiah'),
(296,'CX1-15733','pH','NULL','2024-11-20','2024-11-20','07:03:08','07:03:08','pH Calibration','205089','M. Venkata Ramana'),
(297,'CW1-12012','pH','NULL','2024-11-20','2024-11-20','12:23:17','12:23:17','pH Calibration','88819','M. Ramya');

/*Table structure for table `tbl_menulist` */


CREATE TABLE `tbl_menulist` (
  `id` int(11) NOT NULL AUTO_INCREMENT,
  `SeqNo` smallint(6) NOT NULL DEFAULT '0',
  `MenuName` varchar(50) NOT NULL DEFAULT 'NULL',
  `ProductType` int(11) NOT NULL DEFAULT '0',
  `InstruId` smallint(6) NOT NULL DEFAULT '0',
  PRIMARY KEY (`id`),
  KEY `SeqNo` (`SeqNo`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_menulist` */

/*Table structure for table `tbl_meter` */


CREATE TABLE `tbl_meter` (
  `Id` int(11) NOT NULL AUTO_INCREMENT,
  `meter_ID` varchar(50) NOT NULL DEFAULT 'NULL',
  `meter_Make` varchar(50) NOT NULL DEFAULT 'NULL',
  `meter_Model` varchar(50) NOT NULL DEFAULT 'NULL',
  `meter_SerialNo` varchar(50) NOT NULL DEFAULT 'NULL',
  `meter_locked` int(1) DEFAULT '0',
  `meter_EditCounter` int(11) DEFAULT '0',
  `meter_isActivate` int(1) DEFAULT '0',
  `meter_UserID` varchar(200) DEFAULT 'NULL',
  `meter_IsApproved` int(1) DEFAULT '0',
  `meter_ApprovedBy` varchar(200) DEFAULT 'NULL',
  PRIMARY KEY (`Id`)
) ENGINE=InnoDB AUTO_INCREMENT=4 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_meter` */

insert  into `tbl_meter`(`Id`,`meter_ID`,`meter_Make`,`meter_Model`,`meter_SerialNo`,`meter_locked`,`meter_EditCounter`,`meter_isActivate`,`meter_UserID`,`meter_IsApproved`,`meter_ApprovedBy`) values 
(1,'MeterID','TFS','Orion Versa Star','1234',0,0,0,'1001',0,'NULL'),
(2,'12345','TFS','Versa Star Pro','V14010',0,0,1,'1001',1,'1002'),
(3,'TS1062','1062','1062','1062',0,0,1,'1001',1,'1002');

/*Table structure for table `tbl_meter_weights` */


CREATE TABLE `tbl_meter_weights` (
  `Id` int(11) NOT NULL AUTO_INCREMENT,
  `meterwt_ID` varchar(50) NOT NULL DEFAULT 'NULL',
  `meterwt_ChannelNo` int(10) DEFAULT '0',
  `meterwt_ProbID` varchar(100) DEFAULT 'NULL',
  `meterwt_ProbType` varchar(100) DEFAULT 'NULL',
  `meterwt_NegTol` decimal(10,5) DEFAULT '0.00000',
  `meterwt_PosTol` decimal(10,5) DEFAULT '0.00000',
  `meterwt_Daily` int(1) DEFAULT '0',
  `meterwt_dp` int(5) DEFAULT NULL,
  `meterwt_unit` varchar(50) DEFAULT 'NULL',
  `meterwt_type` varchar(50) DEFAULT 'NA',
  PRIMARY KEY (`Id`)
) ENGINE=InnoDB AUTO_INCREMENT=5 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_meter_weights` */

insert  into `tbl_meter_weights`(`Id`,`meterwt_ID`,`meterwt_ChannelNo`,`meterwt_ProbID`,`meterwt_ProbType`,`meterwt_NegTol`,`meterwt_PosTol`,`meterwt_Daily`,`meterwt_dp`,`meterwt_unit`,`meterwt_type`) values 
(1,'MeterID',1,'1234','pH',90.00000,105.00000,1,0,'%','NA'),
(2,'12345',1,'123PH','pH',90.00000,105.00000,1,0,'%','NA'),
(3,'12345',2,'123COND','Conductivity',0.47500,0.50000,0,3,'K Avg','NA'),
(4,'TS1062',1,'1062','pH',10.00000,30.00000,1,0,'%','NA');

/*Table structure for table `tbl_nomenclature` */


CREATE TABLE `tbl_nomenclature` (
  `Id` int(11) NOT NULL AUTO_INCREMENT,
  `BFGCode` varchar(200) NOT NULL,
  `ProductName` varchar(200) NOT NULL,
  `HardnessID` varchar(200) NOT NULL,
  `DisintegrationTester` varchar(200) NOT NULL,
  `FriabilatorID` varchar(200) NOT NULL,
  `TappedDensityID` varchar(200) NOT NULL,
  `MoistureAnalyzerID` varchar(200) NOT NULL,
  `BalanceID` varchar(200) NOT NULL,
  `VernierID` varchar(200) NOT NULL,
  `BinBalanceID` varchar(200) NOT NULL,
  `SieveShakerID` varchar(200) NOT NULL,
  `CubicleNo` varchar(200) NOT NULL,
  `MachineCode` varchar(200) NOT NULL,
  `Tablet_Compression` varchar(200) DEFAULT NULL,
  `BinText` varchar(200) DEFAULT NULL,
  `BatchSummary` int(1) DEFAULT '0' COMMENT '0:Increncyv4,1:Allparameter',
  `Instrument` varchar(200) DEFAULT NULL,
  `dateFormat` varchar(50) DEFAULT NULL,
  `timeFormat` varchar(50) DEFAULT NULL,
  `Capsule_Capsulation` varchar(200) DEFAULT NULL,
  PRIMARY KEY (`Id`)
) ENGINE=InnoDB AUTO_INCREMENT=2 DEFAULT CHARSET=utf8;

/*Data for the table `tbl_nomenclature` */

insert  into `tbl_nomenclature`(`Id`,`BFGCode`,`ProductName`,`HardnessID`,`DisintegrationTester`,`FriabilatorID`,`TappedDensityID`,`MoistureAnalyzerID`,`BalanceID`,`VernierID`,`BinBalanceID`,`SieveShakerID`,`CubicleNo`,`MachineCode`,`Tablet_Compression`,`BinText`,`BatchSummary`,`Instrument`,`dateFormat`,`timeFormat`,`Capsule_Capsulation`) values 
(1,'BFG Code No.','Product Name','Hardness Tester Code No.','Disintegration Tester Code No.','Friabilator Code No.','Tap Density Tester Code No.','Moisture analyzer code No.','Meter ID','Vernier Code No.','BinBalanceID','SSID','CubicleNo','Equipment Code No.','Monolayer','IPC',0,'Instrument Code No.','dd-MM-yyyy','HH:mm:ss','Capsule Filling');

/*Table structure for table `tbl_product` */


CREATE TABLE `tbl_product` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `productID` varchar(300) DEFAULT 'NULL',
  `productName` varchar(300) DEFAULT 'NULL',
  `productType` varchar(300) DEFAULT 'NULL' COMMENT 'From Product Type Table',
  `locked` bit(1) DEFAULT b'0',
  `editCounter` int(11) DEFAULT '0',
  `InProcess` int(1) DEFAULT '0' COMMENT '0:No,1:yes',
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB AUTO_INCREMENT=2 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_product` */

insert  into `tbl_product`(`RecNo`,`productID`,`productName`,`productType`,`locked`,`editCounter`,`InProcess`) values 
(1,'123','Para Main','Test Para','\0',0,0);

/*Table structure for table `tbl_producttype` */


CREATE TABLE `tbl_producttype` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `PT_productType` varchar(300) DEFAULT NULL,
  `locked` bit(1) DEFAULT b'0',
  `editCounter` int(11) DEFAULT NULL,
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB AUTO_INCREMENT=2 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_producttype` */

insert  into `tbl_producttype`(`RecNo`,`PT_productType`,`locked`,`editCounter`) values 
(1,'Test Para','\0',NULL);

/*Table structure for table `tbl_pwd_complexity` */


CREATE TABLE `tbl_pwd_complexity` (
  `Pwd_id` int(1) NOT NULL AUTO_INCREMENT,
  `Pwd_Length` int(10) NOT NULL DEFAULT '0',
  `Pwd_SpecialChr` int(1) NOT NULL DEFAULT '0',
  `Pwd_Alphabate` int(1) NOT NULL DEFAULT '0',
  `Pwd_Digit` int(1) NOT NULL DEFAULT '0',
  PRIMARY KEY (`Pwd_id`)
) ENGINE=InnoDB AUTO_INCREMENT=2 DEFAULT CHARSET=utf8;

/*Data for the table `tbl_pwd_complexity` */

insert  into `tbl_pwd_complexity`(`Pwd_id`,`Pwd_Length`,`Pwd_SpecialChr`,`Pwd_Alphabate`,`Pwd_Digit`) values 
(1,4,1,2,1);

/*Table structure for table `tbl_pwd_history` */


CREATE TABLE `tbl_pwd_history` (
  `id` int(11) NOT NULL AUTO_INCREMENT,
  `userid` varchar(50) NOT NULL DEFAULT 'NULL',
  `password` varchar(255) NOT NULL DEFAULT 'NULL',
  `pwdChgDate` date NOT NULL,
  `addedTime` timestamp NOT NULL DEFAULT CURRENT_TIMESTAMP,
  `realPassword` varchar(200) NOT NULL DEFAULT 'NULL',
  PRIMARY KEY (`id`)
) ENGINE=InnoDB AUTO_INCREMENT=3 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_pwd_history` */

insert  into `tbl_pwd_history`(`id`,`userid`,`password`,`pwdChgDate`,`addedTime`,`realPassword`) values 
(1,'1002','0a0aebbca5a3b2109190bd70c126eaa3ae4327a3','2025-07-21','2025-07-21 00:00:00','TF@12345'),
(2,'1002','a2be6b593ae82bd1e8fc1953a65c227705ec8668','2025-07-28','2025-07-28 00:00:00','TFS@12345');

/*Table structure for table `tbl_rights` */


CREATE TABLE `tbl_rights` (
  `Id` int(5) NOT NULL AUTO_INCREMENT,
  `Right_Name` varchar(50) NOT NULL DEFAULT 'NULL',
  `Visibility` bit(1) NOT NULL DEFAULT b'0',
  PRIMARY KEY (`Id`)
) ENGINE=InnoDB AUTO_INCREMENT=32 DEFAULT CHARSET=latin1;



/*Table structure for table `tbl_rights_removed` */


CREATE TABLE `tbl_rights_removed` (
  `id` int(11) NOT NULL AUTO_INCREMENT,
  `userid` varchar(50) NOT NULL DEFAULT 'NULL',
  `removed_right` varchar(50) NOT NULL DEFAULT 'NULL',
  `role_name` varchar(200) DEFAULT 'NULL',
  PRIMARY KEY (`id`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_rights_removed` */

/*Table structure for table `tbl_rights_special` */


CREATE TABLE `tbl_rights_special` (
  `id` int(11) NOT NULL AUTO_INCREMENT,
  `userid` varchar(50) NOT NULL DEFAULT 'NULL',
  `spl_right` varchar(50) NOT NULL DEFAULT 'NULL',
  PRIMARY KEY (`id`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_rights_special` */

/*Table structure for table `tbl_role` */


CREATE TABLE `tbl_role` (
  `id` int(11) NOT NULL AUTO_INCREMENT,
  `role_name` varchar(50) NOT NULL DEFAULT 'NULL',
  `role_rights` varchar(50) NOT NULL DEFAULT 'NULL',
  `locked` bit(1) NOT NULL DEFAULT b'0',
  `editCounter` int(11) NOT NULL DEFAULT '0',
  PRIMARY KEY (`id`)
) ENGINE=InnoDB AUTO_INCREMENT=3277 DEFAULT CHARSET=latin1;



/*Table structure for table `tbl_rpi` */


CREATE TABLE `tbl_rpi` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `RPIID` varchar(100) DEFAULT 'NULL',
  `IDSNo` varchar(100) DEFAULT 'NULL',
  `TSH_IP` varchar(50) DEFAULT NULL,
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB AUTO_INCREMENT=12 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_rpi` */

insert  into `tbl_rpi`(`RecNo`,`RPIID`,`IDSNo`,`TSH_IP`) values 
(1,'1000000072008a1f','101','198.168.1.102'),
(2,'100000009e7c8be2','102','10.16.60.52'),
(3,'100000006dad3d4c','103','10.16.60.53'),
(4,'10000000ed326860','104','10.16.60.54'),
(5,'10000000df0de673','105','10.16.60.55'),
(6,'10000000f225514c','106','10.16.60.56'),
(7,'100000008c93a5a9','107','192.168.1.91'),
(8,'00000000a92f3339','108','147.93.130.206'),
(9,'10000000303bf2a4','122','192.168.1.4'),
(10,'10000000062fc7c2','121','192.168.1.5'),
(11,'10000000f7824c6d','123','192.168.1.153');

/*Table structure for table `tbl_rpt_path` */


CREATE TABLE `tbl_rpt_path` (
  `id` int(11) NOT NULL,
  `path` varchar(250) DEFAULT 'NULL',
  `Comp_name` varchar(100) DEFAULT 'NULL',
  `proj_name` varchar(100) DEFAULT 'NULL',
  PRIMARY KEY (`id`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_rpt_path` */

insert  into `tbl_rpt_path`(`id`,`path`,`Comp_name`,`proj_name`) values 
(1,'C:\\\\Apache24\\\\htdocs\\\\Incrency_PHMeter_API\\\\API_V1\\\\jasperFiles\\\\ILP\\\\','C:\\\\Apache24\\\\htdocs\\\\Incrency_PHMeter_API\\\\DeveloperPanel_IonLinkPro.json','ILP');

/*Table structure for table `tbl_serverconfig` */


CREATE TABLE `tbl_serverconfig` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `Path` varchar(500) DEFAULT 'NULL',
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_serverconfig` */

/*Table structure for table `tbl_setallparameter` */


CREATE TABLE `tbl_setallparameter` (
  `config_id` int(11) NOT NULL AUTO_INCREMENT,
  `tbl_config_TimeoutPeriod` varchar(10) NOT NULL DEFAULT 'NULL',
  `tbl_config_PasswordExpPeriod` varchar(10) NOT NULL DEFAULT 'NULL',
  `tbl_config_ReminderPassword` varchar(10) NOT NULL DEFAULT 'NULL',
  `tbl_config_ArchivePeriod` varchar(10) NOT NULL DEFAULT 'NULL',
  `tbl_config_AutoDisablePeriod` varchar(10) NOT NULL DEFAULT 'NULL',
  `tbl_config_LoginAttempts` varchar(10) NOT NULL DEFAULT 'NULL',
  `tbl_config_AutoEnableChances` int(11) NOT NULL DEFAULT '0' COMMENT 'No.of times that a system can automatically enable a user after unsuccessful login attempts.',
  `tbl_config_DisabledTime` int(11) NOT NULL DEFAULT '0' COMMENT 'Time in seconds that a user will get temporary disabled after unsuccessful login attempts.',
  `tbl_config_PwdHistoryCount` int(11) NOT NULL DEFAULT '0',
  `tbl_config_PeriodicCalibDays` int(10) NOT NULL DEFAULT '0',
  `tbl_PeriodicCalbVer` int(10) NOT NULL DEFAULT '0',
  `tbl_calibration_Reminder` int(3) NOT NULL DEFAULT '0',
  `tbl_PrintingMode` varchar(10) NOT NULL DEFAULT 'NULL',
  `tbl_biometricMode` varchar(20) NOT NULL DEFAULT 'NULL',
  `tbl_config_logoUploaded` int(10) DEFAULT NULL,
  PRIMARY KEY (`config_id`)
) ENGINE=InnoDB AUTO_INCREMENT=2 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_setallparameter` */

insert  into `tbl_setallparameter`(`config_id`,`tbl_config_TimeoutPeriod`,`tbl_config_PasswordExpPeriod`,`tbl_config_ReminderPassword`,`tbl_config_ArchivePeriod`,`tbl_config_AutoDisablePeriod`,`tbl_config_LoginAttempts`,`tbl_config_AutoEnableChances`,`tbl_config_DisabledTime`,`tbl_config_PwdHistoryCount`,`tbl_config_PeriodicCalibDays`,`tbl_PeriodicCalbVer`,`tbl_calibration_Reminder`,`tbl_PrintingMode`,`tbl_biometricMode`,`tbl_config_logoUploaded`) values 
(1,'5','2','1','1','2','3',2,4,1,4,3,999,'Auto','Registration',0);

/*Table structure for table `tbl_user_statusreport` */


CREATE TABLE `tbl_user_statusreport` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `UserID` varchar(100) DEFAULT 'NULL',
  `UserName` varchar(300) DEFAULT 'NULL',
  `Role` varchar(100) DEFAULT 'NULL',
  `Dept` varchar(100) DEFAULT 'NULL',
  `Status` varchar(200) DEFAULT 'NULL',
  `HMIID` int(3) DEFAULT '0',
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB AUTO_INCREMENT=5 DEFAULT CHARSET=utf8;

/*Data for the table `tbl_user_statusreport` */

insert  into `tbl_user_statusreport`(`RecNo`,`UserID`,`UserName`,`Role`,`Dept`,`Status`,`HMIID`) values 
(1,'1001','Thermofisher','SuperAdmin','NA','Active',66),
(2,'1002','User1002','Service engineer','Service','Active',66),
(3,'1003','User1003','Service engineer','QC-Chemical','Active',66),
(4,'1004','User1004','Reviewer-QA','QC-Chemical','Active',66);

/*Table structure for table `tbl_users` */


CREATE TABLE `tbl_users` (
  `Id` int(11) NOT NULL AUTO_INCREMENT,
  `UserID` varchar(50) NOT NULL DEFAULT 'NULL',
  `UserName` varchar(100) NOT NULL DEFAULT 'NULL',
  `Pwd` varchar(50) DEFAULT 'NULL',
  `userType` int(11) NOT NULL DEFAULT '0',
  `PwdChgDate` date NOT NULL,
  `PwdExpStauts` bit(1) NOT NULL DEFAULT b'0' COMMENT '0=Active',
  `Status` int(11) NOT NULL DEFAULT '0' COMMENT '0 : Enabled || 1: Temp Disabled || 2:Permanent Disable || 4: Auto Disable || 6: User Locked',
  `Reason` varchar(50) DEFAULT 'NULL',
  `PwdChg` bit(1) NOT NULL DEFAULT b'0' COMMENT 'By defaullt value is 1,1 means this password is set by admin.0 means set by user',
  `isadmin` tinyint(1) NOT NULL DEFAULT '0',
  `LastLoginDt` date NOT NULL,
  `UserInitials` varchar(25) NOT NULL DEFAULT 'NULL',
  `loginatmpt` int(11) NOT NULL DEFAULT '0',
  `suspensionPeriod` timestamp NULL DEFAULT NULL,
  `autoEnbl` int(11) NOT NULL DEFAULT '0',
  `active` tinyint(1) NOT NULL DEFAULT '0' COMMENT 'To check logged in already or not',
  `realPassword` varchar(100) DEFAULT 'NULL',
  `lstActvtyTime` datetime DEFAULT NULL,
  `Department` varchar(250) NOT NULL DEFAULT 'NULL',
  `Role` varchar(250) NOT NULL DEFAULT 'NULL',
  `HostName` varchar(100) DEFAULT 'NULL',
  `locked` bit(1) NOT NULL DEFAULT b'0',
  `editCounter` int(11) NOT NULL DEFAULT '0',
  `loginCounter` int(11) DEFAULT '0',
  `source` varchar(15) NOT NULL DEFAULT 'NULL',
  `PREV_STATUS` int(10) DEFAULT '0' COMMENT 'This will store the previous status when user is autodisable and he made unsuccessfull attempts',
  `Domain` varchar(30) DEFAULT 'NULL',
  `failedAtmpTime` datetime DEFAULT NULL,
  PRIMARY KEY (`Id`),
  UNIQUE KEY `userid` (`UserID`)
) ENGINE=InnoDB AUTO_INCREMENT=114 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_users` */

insert  into `tbl_users`(`Id`,`UserID`,`UserName`,`Pwd`,`userType`,`PwdChgDate`,`PwdExpStauts`,`Status`,`Reason`,`PwdChg`,`isadmin`,`LastLoginDt`,`UserInitials`,`loginatmpt`,`suspensionPeriod`,`autoEnbl`,`active`,`realPassword`,`lstActvtyTime`,`Department`,`Role`,`HostName`,`locked`,`editCounter`,`loginCounter`,`source`,`PREV_STATUS`,`Domain`,`failedAtmpTime`) values 
(30,'1001','Thermofisher','a388edd1b11e6880a3a41f90bb835cb7b9823809',1,'2024-08-27','\0',0,'0','\0',0,'2025-07-28','Thermofisher',0,NULL,0,0,'TS@12345','2025-07-28 14:23:41','NA','SuperAdmin','::ffff:147.93.130.206','\0',0,1,'0',0,'0','2025-07-22 10:00:31'),
(31,'1002','User1002','a2be6b593ae82bd1e8fc1953a65c227705ec8668',2,'2025-07-28','\0',0,'0','\0',0,'2025-07-28','Trident',0,'0000-00-00 00:00:00',0,0,'TFS@12345','2025-07-28 13:52:48','Service','Service engineer','::ffff:103.60.214.170','\0',0,0,'SOFTWARE',0,'0','2025-07-21 17:38:44'),
(32,'1003','User1003','83ed9c23386c0e57f07876c5e716525367d0bbd0',2,'2025-06-26','',0,'0','\0',0,'2025-07-21','Sreeram Jakkaraiah',0,'0000-00-00 00:00:00',0,0,'TS@1003','2025-07-21 16:41:00','QC-Chemical','Service engineer','','\0',2,2,'',0,'0','2025-01-18 17:22:01'),
(33,'1004','User1004','b8968ba47dfa01df711aa18d975378d28133ccb3',2,'2024-12-28','',0,'0','\0',0,'2025-01-29','S. Siddhartha Kumar Reddy',1,'0000-00-00 00:00:00',0,0,'TS@1004','2025-07-28 13:59:34','QC-Chemical','Reviewer-QA','','\0',0,5,'0',0,'0','2025-07-28 13:59:10');

/*Table structure for table `tbl_versionhistory` */


CREATE TABLE `tbl_versionhistory` (
  `id` int(11) NOT NULL AUTO_INCREMENT,
  `A` tinyint(4) DEFAULT NULL COMMENT 'Major',
  `B` tinyint(4) DEFAULT NULL COMMENT 'Minor',
  `C` tinyint(4) DEFAULT NULL COMMENT 'Build',
  `D` smallint(6) DEFAULT NULL COMMENT 'All Bugs from client',
  `E` smallint(6) DEFAULT NULL COMMENT 'All Bugs from tester',
  `F` varchar(2000) DEFAULT 'NULL' COMMENT 'Description',
  `DeveloperName` varchar(200) DEFAULT 'NULL',
  `DateTime` timestamp NOT NULL DEFAULT CURRENT_TIMESTAMP ON UPDATE CURRENT_TIMESTAMP,
  PRIMARY KEY (`id`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_versionhistory` */

/*Table structure for table `tbl_weighment_detail_conductivity` */


CREATE TABLE `tbl_weighment_detail_conductivity` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `RepSerNo` int(11) DEFAULT NULL,
  `RecSeqNo` int(11) DEFAULT NULL,
  `Conductivity` varchar(100) DEFAULT '0',
  `Conductance` varchar(100) DEFAULT '0',
  `Temp` varchar(100) DEFAULT '0',
  `TempCoeff` varchar(100) DEFAULT '0',
  `TempRef` varchar(100) DEFAULT '0',
  `K` varchar(100) DEFAULT '0',
  `K_Flag` bit(1) DEFAULT b'1',
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB AUTO_INCREMENT=3 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_weighment_detail_conductivity` */

/*Table structure for table `tbl_weighment_detail_conductivity_archived` */


CREATE TABLE `tbl_weighment_detail_conductivity_archived` (
  `RecNo` int(11) DEFAULT NULL,
  `RepSerNo` int(11) DEFAULT NULL,
  `RecSeqNo` int(11) DEFAULT NULL,
  `Conductivity` varchar(100) DEFAULT '0',
  `Conductance` varchar(100) DEFAULT '0',
  `Temp` varchar(100) DEFAULT '0',
  `TempCoeff` varchar(100) DEFAULT '0',
  `TempRef` varchar(100) DEFAULT '0',
  `K` varchar(100) DEFAULT '0',
  `K_Flag` bit(1) DEFAULT b'1'
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_weighment_detail_conductivity_archived` */

insert  into `tbl_weighment_detail_conductivity_archived`(`RecNo`,`RepSerNo`,`RecSeqNo`,`Conductivity`,`Conductance`,`Temp`,`TempCoeff`,`TempRef`,`K`,`K_Flag`) values 
(1,1,1,'98.25 uS/cm','1056.489 uS','24.7 C (ATC)','Off','0','0.0930 /cm','\0'),
(2,2,1,'100.5 uS/cm','1080.608 uS','26.1 C (ATC)','Off','0','0.0930 /cm','\0');

/*Table structure for table `tbl_weighment_detail_ph` */


CREATE TABLE `tbl_weighment_detail_ph` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `RepSerNo` int(11) DEFAULT NULL,
  `RecSeqNo` int(11) DEFAULT NULL,
  `PhValue` varchar(100) DEFAULT '0',
  `MvValue` varchar(100) DEFAULT '0',
  `Temp` varchar(100) DEFAULT '0',
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB AUTO_INCREMENT=2029 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_weighment_detail_ph` */

/*Table structure for table `tbl_weighment_detail_ph_archived` */


CREATE TABLE `tbl_weighment_detail_ph_archived` (
  `RecNo` int(11) DEFAULT NULL,
  `RepSerNo` int(11) DEFAULT NULL,
  `RecSeqNo` int(11) DEFAULT NULL,
  `PhValue` varchar(100) DEFAULT '0',
  `MvValue` varchar(100) DEFAULT '0',
  `Temp` varchar(100) DEFAULT '0'
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_weighment_detail_ph_archived` */

insert  into `tbl_weighment_detail_ph_archived`(`RecNo`,`RepSerNo`,`RecSeqNo`,`PhValue`,`MvValue`,`Temp`) values 
(1,1,1,'7.00 pH','0.0 mV','25.0 C (MAN)'),
(2,2,1,'7.00 pH','0.0 mV','25.0 C (MAN)'),
(3,3,1,'7.00 pH','0.0 mV','25.0 C (MAN)'),
(4,4,1,'7.00 pH','0.0 mV','25.0 C (MAN)'),
(5,5,1,'7.00 pH','0.0 mV','25.0 C (MAN)'),
(6,6,1,'7.00 pH','0.0 mV','25.0 C (MAN)'),
(7,7,1,'6.96 pH','13.9 mV','22.9 C (ATC)'),
(8,8,1,'6.97 pH','-1.5 mV','24.0 C (ATC)'),
(9,9,1,'7.00 pH','-0.3 mV','24.0 C (ATC)'),
(10,10,1,'7.00 pH','-0.3 mV','24.0 C (ATC)'),
(11,11,1,'7.00 pH','-0.2 mV','23.4 C (ATC)'),
(12,12,1,'1.69 pH','309.7 mV','23.5 C (ATC)'),
(13,13,1,'1.69 pH','311.5 mV','23.4 C (ATC)'),
(14,14,1,'4.03 pH','173.6 mV','24.0 C (ATC)'),
(15,15,1,'4.00 pH','176.4 mV','23.6 C (ATC)'),
(16,16,1,'6.98 pH','-2.6 mV','23.6 C (ATC)'),
(17,17,1,'7.00 pH','0.4 mV','23.4 C (ATC)'),
(18,18,1,'9.99 pH','-170.9 mV','23.7 C (ATC)'),
(19,19,1,'10.03 pH','-174.5 mV','24.0 C (ATC)'),
(20,20,1,'12.50 pH','-319.6 mV','23.8 C (ATC)'),
(21,21,1,'12.49 pH','-318.6 mV','24.7 C (ATC)'),
(22,22,1,'6.37 pH','37.3 mV','24.9 C (ATC)'),
(23,23,1,'1.71 pH','312.3 mV','24.3 C (ATC)'),
(24,24,1,'4.02 pH','173.7 mV','24.1 C (ATC)'),
(25,25,1,'7.02 pH','-0.6 mV','24.4 C (ATC)'),
(26,26,1,'10.03 pH','-173.1 mV','25.4 C (ATC)'),
(27,27,1,'12.44 pH','-317.5 mV','24.8 C (ATC)'),
(28,28,1,'1.67 pH','310.1 mV','25.6 C (ATC)'),
(29,29,1,'4.01 pH','170.7 mV','24.3 C (ATC)'),
(30,30,1,'6.98 pH','-2.9 mV','24.7 C (ATC)'),
(31,31,1,'10.02 pH','-173.2 mV','25.4 C (ATC)'),
(32,32,1,'12.50 pH','-315.8 mV','24.8 C (ATC)'),
(33,33,1,'1.68 pH','319.7 mV','26.5 C (ATC)'),
(34,34,1,'4.01 pH','180.3 mV','26.5 C (ATC)'),
(35,35,1,'6.98 pH','4.9 mV','26.3 C (ATC)'),
(36,36,1,'9.98 pH','-168.7 mV','26.1 C (ATC)'),
(37,37,1,'12.43 pH','-307.7 mV','26.7 C (ATC)'),
(38,38,1,'6.64 pH','21.4 mV','24.8 C (ATC)'),
(39,39,1,'5.85 pH','68.9 mV','24.5 C (ATC)'),
(40,40,1,'5.90 pH','65.8 mV','24.5 C (ATC)'),
(41,41,1,'5.95 pH','62.6 mV','24.6 C (ATC)'),
(42,42,1,'4.48 pH','149.4 mV','23.5 C (ATC)'),
(43,43,1,'3.52 pH','205.3 mV','26.3 C (ATC)'),
(44,44,1,'6.44 pH','33.7 mV','24.8 C (ATC)'),
(45,45,1,'6.43 pH','34.2 mV','24.8 C (ATC)'),
(46,46,1,'5.00 pH','118.0 mV','27.5 C (ATC)'),
(47,47,1,'4.99 pH','118.1 mV','27.3 C (ATC)'),
(48,48,1,'5.66 pH','82.5 mV','25.4 C (ATC)'),
(49,49,1,'5.77 pH','72.1 mV','24.6 C (ATC)'),
(50,50,1,'5.73 pH','73.8 mV','23.3 C (ATC)'),
(51,51,1,'5.76 pH','72.4 mV','24.1 C (ATC)'),
(52,52,1,'5.74 pH','73.4 mV','24.3 C (ATC)'),
(53,53,1,'5.67 pH','77.7 mV','24.2 C (ATC)'),
(54,54,1,'5.70 pH','75.9 mV','24.4 C (ATC)'),
(55,55,1,'5.75 pH','72.7 mV','24.0 C (ATC)'),
(56,56,1,'6.25 pH','45.3 mV','24.7 C (ATC)'),
(57,57,1,'5.77 pH','72.0 mV','24.5 C (ATC)'),
(58,58,1,'5.70 pH','75.9 mV','24.4 C (ATC)'),
(59,59,1,'6.27 pH','43.5 mV','24.1 C (ATC)'),
(60,60,1,'5.51 pH','87.3 mV','25.4 C (ATC)'),
(61,61,1,'7.31 pH','-14.1 mV','24.9 C (ATC)'),
(62,62,1,'7.17 pH','-6.1 mV','25.0 C (ATC)'),
(63,63,1,'4.31 pH','156.5 mV','24.1 C (ATC)'),
(64,64,1,'6.32 pH','42.9 mV','26.0 C (ATC)'),
(65,65,1,'6.16 pH','50.6 mV','23.8 C (ATC)'),
(66,66,1,'6.18 pH','49.2 mV','24.3 C (ATC)'),
(67,67,1,'1.68 pH','311.9 mV','25.8 C (ATC)'),
(68,68,1,'4.00 pH','173.2 mV','25.8 C (ATC)'),
(69,69,1,'7.01 pH','-3.1 mV','25.4 C (ATC)'),
(70,70,1,'10.03 pH','-174.9 mV','25.1 C (ATC)'),
(71,71,1,'12.45 pH','-321.2 mV','25.3 C (ATC)'),
(72,72,1,'6.04 pH','53.4 mV','25.5 C (ATC)'),
(73,73,1,'5.51 pH','87.6 mV','35.1 C (ATC)'),
(74,74,1,'3.31 pH','217.5 mV','30.1 C (ATC)'),
(75,75,1,'7.54 pH','-33.7 mV','31.2 C (ATC)'),
(76,76,1,'1.68 pH','318.5 mV','25.4 C (ATC)'),
(77,77,1,'5.52 pH','85.8 mV','31.4 C (ATC)'),
(78,78,1,'4.02 pH','180.1 mV','25.5 C (ATC)'),
(79,79,1,'7.00 pH','4.8 mV','25.8 C (ATC)'),
(80,80,1,'10.01 pH','-168.6 mV','25.8 C (ATC)'),
(81,81,1,'12.44 pH','-310.7 mV','25.8 C (ATC)'),
(82,82,1,'3.31 pH','216.4 mV','28.1 C (ATC)'),
(83,83,1,'5.53 pH','82.9 mV','24.3 C (ATC)'),
(84,84,1,'5.53 pH','82.9 mV','24.1 C (ATC)'),
(85,85,1,'7.07 pH','0.7 mV','28.8 C (ATC)'),
(86,86,1,'5.71 pH','72.7 mV','24.0 C (ATC)'),
(87,87,1,'6.22 pH','42.7 mV','24.8 C (ATC)'),
(88,88,1,'5.78 pH','68.8 mV','24.6 C (ATC)'),
(89,89,1,'5.75 pH','70.7 mV','24.7 C (ATC)'),
(90,90,1,'5.81 pH','67.0 mV','24.8 C (ATC)'),
(91,91,1,'5.75 pH','70.2 mV','23.5 C (ATC)'),
(92,92,1,'4.23 pH','158.8 mV','23.7 C (ATC)'),
(93,93,1,'6.25 pH','41.4 mV','24.8 C (ATC)'),
(94,94,1,'6.13 pH','48.5 mV','24.6 C (ATC)'),
(95,95,1,'6.33 pH','36.2 mV','24.4 C (ATC)'),
(96,96,1,'6.33 pH','36.5 mV','24.5 C (ATC)'),
(97,97,1,'6.33 pH','36.3 mV','24.6 C (ATC)'),
(98,98,1,'6.46 pH','28.8 mV','24.5 C (ATC)'),
(99,99,1,'5.74 pH','70.8 mV','23.5 C (ATC)'),
(100,100,1,'5.79 pH','67.9 mV','24.0 C (ATC)'),
(101,101,1,'5.80 pH','67.1 mV','23.6 C (ATC)'),
(102,102,1,'5.79 pH','67.6 mV','23.6 C (ATC)'),
(103,103,1,'5.76 pH','69.7 mV','23.6 C (ATC)'),
(104,104,1,'5.81 pH','66.8 mV','23.6 C (ATC)'),
(105,105,1,'5.79 pH','67.8 mV','23.8 C (ATC)'),
(106,106,1,'5.72 pH','71.9 mV','23.6 C (ATC)'),
(107,107,1,'5.79 pH','67.7 mV','23.6 C (ATC)'),
(108,108,1,'5.79 pH','68.1 mV','23.6 C (ATC)'),
(109,109,1,'7.23 pH','-13.6 mV','25.9 C (ATC)'),
(110,110,1,'7.23 pH','-13.4 mV','23.2 C (ATC)'),
(111,111,1,'7.03 pH','-1.6 mV','25.9 C (ATC)'),
(112,112,1,'7.33 pH','-21.5 mV','24.1 C (ATC)'),
(113,113,1,'7.04 pH','-2.2 mV','24.7 C (ATC)'),
(114,114,1,'7.01 pH','-1.2 mV','25.5 C (ATC)'),
(115,115,1,'7.12 pH','-7.1 mV','25.1 C (ATC)'),
(116,116,1,'7.12 pH','-7.5 mV','24.6 C (ATC)'),
(117,117,1,'7.18 pH','-10.9 mV','24.4 C (ATC)'),
(118,118,1,'7.21 pH','-12.6 mV','24.3 C (ATC)'),
(119,119,1,'6.80 pH','9.2 mV','24.0 C (ATC)'),
(120,120,1,'7.26 pH','-15.4 mV','25.5 C (ATC)'),
(121,121,1,'7.01 pH','-0.8 mV','26.0 C (ATC)'),
(122,122,1,'7.10 pH','-5.7 mV','25.4 C (ATC)'),
(123,123,1,'4.63 pH','137.3 mV','23.8 C (ATC)'),
(124,124,1,'1.71 pH','311.5 mV','24.0 C (ATC)'),
(125,125,1,'4.02 pH','173.3 mV','23.2 C (ATC)'),
(126,126,1,'7.03 pH','-1.8 mV','23.5 C (ATC)'),
(127,127,1,'10.02 pH','-168.9 mV','24.1 C (ATC)'),
(128,128,1,'12.48 pH','-321.7 mV','24.8 C (ATC)'),
(129,129,1,'5.51 pH','87.9 mV','27.2 C (ATC)'),
(130,130,1,'1.69 pH','316.1 mV','24.7 C (ATC)'),
(131,131,1,'4.02 pH','179.5 mV','24.9 C (ATC)'),
(132,132,1,'7.01 pH','4.4 mV','24.8 C (ATC)'),
(133,133,1,'10.01 pH','-167.9 mV','25.0 C (ATC)'),
(134,134,1,'12.46 pH','-313.0 mV','24.9 C (ATC)'),
(135,135,1,'5.96 pH','60.4 mV','23.2 C (ATC)'),
(136,136,1,'6.72 pH','21.1 mV','25.2 C (ATC)'),
(137,137,1,'4.70 pH','133.8 mV','23.3 C (ATC)'),
(138,138,1,'6.38 pH','38.0 mV','25.6 C (ATC)'),
(139,139,1,'7.48 pH','-22.5 mV','25.3 C (ATC)'),
(140,140,1,'6.36 pH','42.2 mV','24.3 C (ATC)'),
(141,141,1,'5.75 pH','78.0 mV','25.5 C (ATC)'),
(142,142,1,'7.40 pH','-18.2 mV','25.6 C (ATC)'),
(143,143,1,'7.24 pH','-8.8 mV','25.3 C (ATC)'),
(144,144,1,'6.43 pH','33.3 mV','23.8 C (ATC)'),
(145,145,1,'6.49 pH','30.0 mV','23.6 C (ATC)'),
(146,146,1,'5.84 pH','67.6 mV','24.3 C (ATC)'),
(147,147,1,'5.83 pH','68.4 mV','24.0 C (ATC)'),
(148,148,1,'5.62 pH','80.6 mV','24.7 C (ATC)'),
(149,149,1,'5.61 pH','81.3 mV','24.3 C (ATC)'),
(150,150,1,'6.42 pH','35.8 mV','23.4 C (ATC)'),
(151,151,1,'5.78 pH','71.6 mV','24.7 C (ATC)'),
(152,152,1,'5.76 pH','72.4 mV','24.6 C (ATC)'),
(153,153,1,'5.81 pH','69.1 mV','23.7 C (ATC)'),
(154,154,1,'5.52 pH','86.3 mV','24.1 C (ATC)'),
(155,155,1,'6.49 pH','33.5 mV','23.1 C (ATC)'),
(156,156,1,'7.34 pH','-14.8 mV','26.5 C (ATC)'),
(157,157,1,'6.77 pH','13.3 mV','25.0 C (ATC)'),
(158,158,1,'6.91 pH','9.9 mV','25.1 C (ATC)'),
(159,159,1,'7.13 pH','-4.5 mV','25.4 C (ATC)'),
(160,160,1,'5.63 pH','85.0 mV','23.5 C (ATC)'),
(161,161,1,'7.21 pH','-7.2 mV','23.6 C (ATC)'),
(162,162,1,'7.40 pH','-22.7 mV','25.2 C (ATC)'),
(163,163,1,'1.67 pH','309.9 mV','24.4 C (ATC)'),
(164,164,1,'4.01 pH','171.6 mV','24.4 C (ATC)'),
(165,165,1,'7.00 pH','-2.8 mV','24.6 C (ATC)'),
(166,166,1,'10.02 pH','-174.6 mV','24.7 C (ATC)'),
(167,167,1,'12.45 pH','-320.1 mV','25.0 C (ATC)'),
(168,168,1,'7.53 pH','-32.8 mV','25.7 C (ATC)'),
(169,169,1,'5.54 pH','84.3 mV','29.7 C (ATC)'),
(170,170,1,'5.52 pH','84.1 mV','28.0 C (ATC)'),
(171,171,1,'5.53 pH','82.8 mV','25.7 C (ATC)'),
(172,172,1,'5.52 pH','83.6 mV','25.4 C (ATC)'),
(173,173,1,'1.70 pH','318.5 mV','25.2 C (ATC)'),
(174,174,1,'4.02 pH','179.8 mV','25.7 C (ATC)'),
(175,175,1,'7.00 pH','4.6 mV','25.7 C (ATC)'),
(176,176,1,'10.01 pH','-168.1 mV','25.5 C (ATC)'),
(177,177,1,'12.46 pH','-313.7 mV','25.4 C (ATC)'),
(178,178,1,'6.42 pH','31.3 mV','25.0 C (ATC)'),
(179,179,1,'6.45 pH','29.0 mV','25.0 C (ATC)'),
(180,180,1,'6.39 pH','31.8 mV','25.0 C (ATC)'),
(181,181,1,'6.34 pH','35.8 mV','24.8 C (ATC)'),
(182,182,1,'6.50 pH','26.3 mV','24.7 C (ATC)'),
(183,183,1,'5.70 pH','72.9 mV','23.4 C (ATC)'),
(184,184,1,'5.64 pH','75.9 mV','24.0 C (ATC)'),
(185,185,1,'7.31 pH','-13.4 mV','26.5 C (ATC)'),
(186,186,1,'5.73 pH','71.3 mV','24.3 C (ATC)'),
(187,187,1,'5.74 pH','70.2 mV','24.4 C (ATC)'),
(188,188,1,'7.59 pH','-29.7 mV','25.1 C (ATC)'),
(189,189,1,'5.72 pH','71.6 mV','24.8 C (ATC)'),
(190,190,1,'5.65 pH','76.4 mV','24.6 C (ATC)'),
(191,191,1,'5.63 pH','72.0 mV','24.3 C (ATC)'),
(192,192,1,'1.69 pH','311.6 mV','25.0 C (ATC)'),
(193,193,1,'4.02 pH','172.6 mV','25.1 C (ATC)'),
(194,194,1,'6.99 pH','-1.6 mV','25.6 C (ATC)'),
(195,195,1,'10.00 pH','-172.3 mV','25.1 C (ATC)'),
(196,196,1,'12.43 pH','-320.7 mV','25.5 C (ATC)'),
(197,197,1,'4.99 pH','115.0 mV','27.1 C (ATC)'),
(198,198,1,'6.41 pH','31.5 mV','23.7 C (ATC)'),
(199,199,1,'7.01 pH','-3.6 mV','23.1 C (ATC)'),
(200,200,1,'7.40 pH','-18.4 mV','24.8 C (ATC)'),
(201,201,1,'7.18 pH','-5.6 mV','24.7 C (ATC)'),
(202,202,1,'7.29 pH','-12.0 mV','25.6 C (ATC)'),
(203,203,1,'7.56 pH','-27.6 mV','24.6 C (ATC)'),
(204,204,1,'4.70 pH','111.4 mV','23.4 C (ATC)'),
(205,205,1,'5.95 pH','43.9 mV','26.6 C (ATC)'),
(206,206,1,'6.09 pH','35.4 mV','25.2 C (ATC)'),
(207,207,1,'6.18 pH','30.3 mV','23.7 C (ATC)'),
(208,208,1,'6.47 pH','13.4 mV','24.0 C (ATC)'),
(209,209,1,'1.69 pH','311.0 mV','24.1 C (ATC)'),
(210,210,1,'3.99 pH','174.6 mV','24.5 C (ATC)'),
(211,211,1,'6.98 pH','-1.3 mV','24.2 C (ATC)'),
(212,212,1,'10.01 pH','-173.5 mV','24.0 C (ATC)'),
(213,213,1,'12.47 pH','-322.8 mV','24.1 C (ATC)'),
(214,214,1,'6.25 pH','25.9 mV','24.3 C (ATC)'),
(215,215,1,'6.28 pH','24.4 mV','24.7 C (ATC)'),
(216,216,1,'5.92 pH','61.0 mV','25.0 C (ATC)'),
(217,217,1,'5.87 pH','63.7 mV','24.9 C (ATC)'),
(218,218,1,'6.11 pH','49.5 mV','24.2 C (ATC)'),
(219,219,1,'1.66 pH','310.4 mV','23.4 C (ATC)'),
(220,220,1,'4.02 pH','174.3 mV','23.5 C (ATC)'),
(221,221,1,'7.03 pH','-1.5 mV','25.0 C (ATC)'),
(222,222,1,'10.01 pH','-173.2 mV','24.8 C (ATC)'),
(223,223,1,'12.49 pH','-321.5 mV','24.5 C (ATC)'),
(224,224,1,'1.70 pH','308.6 mV','23.2 C (ATC)'),
(225,225,1,'4.01 pH','174.2 mV','24.4 C (ATC)'),
(226,226,1,'7.01 pH','-2.0 mV','25.1 C (ATC)'),
(227,227,1,'10.03 pH','-172.0 mV','24.8 C (ATC)'),
(228,228,1,'12.46 pH','-319.8 mV','24.3 C (ATC)'),
(229,229,1,'3.26 pH','217.8 mV','25.3 C (ATC)'),
(230,230,1,'7.48 pH','-27.7 mV','26.8 C (ATC)'),
(231,231,1,'7.51 pH','-29.4 mV','28.5 C (ATC)'),
(232,232,1,'7.01 pH','-0.4 mV','28.6 C (ATC)'),
(233,233,1,'1.68 pH','316.9 mV','25.1 C (ATC)'),
(234,234,1,'4.61 pH','143.2 mV','25.5 C (ATC)'),
(235,235,1,'5.81 pH','68.8 mV','25.0 C (ATC)'),
(236,236,1,'5.86 pH','66.4 mV','24.7 C (ATC)'),
(237,237,1,'4.01 pH','178.5 mV','25.8 C (ATC)'),
(238,238,1,'5.89 pH','63.7 mV','24.5 C (ATC)'),
(239,239,1,'6.98 pH','3.3 mV','25.9 C (ATC)'),
(240,240,1,'5.85 pH','64.9 mV','24.8 C (ATC)'),
(241,241,1,'5.82 pH','59.4 mV','24.4 C (ATC)'),
(242,242,1,'5.91 pH','62.3 mV','24.5 C (ATC)'),
(243,243,1,'4.51 pH','124.1 mV','23.0 C (ATC)'),
(244,244,1,'5.86 pH','65.4 mV','24.1 C (ATC)'),
(245,245,1,'5.90 pH','63.7 mV','23.9 C (ATC)'),
(246,246,1,'10.04 pH','-173.7 mV','26.1 C (ATC)'),
(247,247,1,'12.48 pH','-317.1 mV','26.3 C (ATC)'),
(248,248,1,'5.79 pH','51.9 mV','26.0 C (ATC)'),
(249,249,1,'6.43 pH','31.7 mV','25.0 C (ATC)'),
(250,250,1,'5.83 pH','49.7 mV','26.6 C (ATC)'),
(251,251,1,'6.14 pH','31.7 mV','25.8 C (ATC)'),
(252,252,1,'1.64 pH','314.4 mV','23.5 C (ATC)'),
(253,253,1,'4.01 pH','177.2 mV','24.6 C (ATC)'),
(254,254,1,'7.00 pH','2.0 mV','24.5 C (ATC)'),
(255,255,1,'9.99 pH','-173.7 mV','25.7 C (ATC)'),
(256,256,1,'12.43 pH','-316.5 mV','25.5 C (ATC)'),
(257,257,1,'6.42 pH','16.0 mV','24.4 C (ATC)'),
(258,258,1,'7.26 pH','-13.0 mV','25.6 C (ATC)'),
(259,259,1,'7.01 pH','1.5 mV','26.9 C (ATC)'),
(260,260,1,'7.25 pH','-12.5 mV','26.4 C (ATC)'),
(261,261,1,'7.30 pH','-15.7 mV','26.2 C (ATC)'),
(262,262,1,'6.32 pH','21.7 mV','26.1 C (ATC)'),
(263,263,1,'6.34 pH','20.8 mV','24.5 C (ATC)'),
(264,264,1,'7.26 pH','-13.1 mV','25.6 C (ATC)'),
(265,265,1,'7.02 pH','0.6 mV','26.7 C (ATC)'),
(266,266,1,'7.29 pH','-14.7 mV','24.8 C (ATC)'),
(267,267,1,'7.30 pH','-15.6 mV','26.3 C (ATC)'),
(268,268,1,'1.68 pH','310.3 mV','24.8 C (ATC)'),
(269,269,1,'4.01 pH','173.9 mV','24.7 C (ATC)'),
(270,270,1,'7.03 pH','-2.0 mV','24.6 C (ATC)'),
(271,271,1,'10.00 pH','-175.6 mV','24.5 C (ATC)'),
(272,272,1,'12.50 pH','-321.8 mV','25.0 C (ATC)'),
(273,273,1,'6.03 pH','56.6 mV','24.9 C (ATC)'),
(274,274,1,'1.69 pH','312.5 mV','25.4 C (ATC)'),
(275,275,1,'4.03 pH','175.5 mV','26.0 C (ATC)'),
(276,276,1,'7.02 pH','-0.9 mV','26.3 C (ATC)'),
(277,277,1,'10.00 pH','-176.1 mV','25.1 C (ATC)'),
(278,278,1,'12.43 pH','-319.7 mV','25.9 C (ATC)'),
(279,279,1,'5.53 pH','85.9 mV','25.3 C (ATC)'),
(280,280,1,'5.53 pH','84.9 mV','23.4 C (ATC)'),
(281,281,1,'4.99 pH','119.0 mV','29.8 C (ATC)'),
(282,282,1,'6.05 pH','55.0 mV','24.2 C (ATC)'),
(283,283,1,'5.53 pH','88.2 mV','35.2 C (ATC)'),
(284,284,1,'7.01 pH','-0.3 mV','23.5 C (ATC)'),
(285,285,1,'5.79 pH','70.4 mV','26.0 C (ATC)'),
(286,286,1,'5.54 pH','85.1 mV','25.9 C (ATC)'),
(287,287,1,'5.68 pH','76.8 mV','25.8 C (ATC)'),
(288,288,1,'5.86 pH','66.6 mV','25.7 C (ATC)'),
(289,289,1,'5.82 pH','68.0 mV','24.9 C (ATC)'),
(290,290,1,'5.79 pH','69.5 mV','24.8 C (ATC)'),
(291,291,1,'5.69 pH','76.4 mV','24.8 C (ATC)'),
(292,292,1,'5.69 pH','75.8 mV','24.5 C (ATC)'),
(293,293,1,'5.70 pH','75.4 mV','24.5 C (ATC)'),
(294,294,1,'5.69 pH','74.9 mV','24.6 C (ATC)'),
(295,295,1,'5.70 pH','75.3 mV','24.5 C (ATC)'),
(296,296,1,'5.77 pH','71.3 mV','24.7 C (ATC)'),
(297,297,1,'5.74 pH','72.8 mV','24.5 C (ATC)'),
(298,298,1,'9.20 pH','-129.0 mV','24.9 C (ATC)'),
(299,299,1,'5.88 pH','64.9 mV','24.4 C (ATC)'),
(300,300,1,'5.82 pH','68.5 mV','24.4 C (ATC)'),
(301,301,1,'6.61 pH','23.2 mV','25.6 C (ATC)'),
(302,302,1,'5.94 pH','61.4 mV','25.5 C (ATC)'),
(303,303,1,'5.00 pH','119.8 mV','33.3 C (ATC)'),
(304,304,1,'5.92 pH','62.6 mV','25.7 C (ATC)'),
(305,305,1,'7.03 pH','-1.6 mV','26.4 C (ATC)'),
(306,306,1,'6.25 pH','43.1 mV','25.7 C (ATC)'),
(307,307,1,'6.12 pH','51.8 mV','25.9 C (ATC)'),
(308,308,1,'1.68 pH','293.7 mV','23.7 C (ATC)'),
(309,309,1,'4.01 pH','159.0 mV','25.4 C (ATC)'),
(310,310,1,'7.02 pH','-16.8 mV','23.7 C (ATC)'),
(311,311,1,'9.98 pH','-190.4 mV','25.7 C (ATC)'),
(312,312,1,'12.42 pH','-334.1 mV','26.9 C (ATC)'),
(313,313,1,'6.88 pH','7.0 mV','24.6 C (ATC)'),
(314,314,1,'4.54 pH','127.2 mV','23.8 C (ATC)'),
(315,315,1,'1.68 pH','294.7 mV','23.6 C (ATC)'),
(316,316,1,'4.02 pH','159.2 mV','24.3 C (ATC)'),
(317,317,1,'7.03 pH','-16.8 mV','25.7 C (ATC)'),
(318,318,1,'9.97 pH','-189.0 mV','26.0 C (ATC)'),
(319,319,1,'12.47 pH','-335.0 mV','25.4 C (ATC)'),
(320,320,1,'5.85 pH','52.0 mV','23.7 C (ATC)'),
(321,321,1,'5.87 pH','51.3 mV','24.3 C (ATC)'),
(322,322,1,'6.47 pH','16.0 mV','24.2 C (ATC)'),
(323,323,1,'6.28 pH','27.1 mV','24.0 C (ATC)'),
(324,324,1,'6.28 pH','27.3 mV','24.3 C (ATC)'),
(325,325,1,'6.27 pH','28.0 mV','24.4 C (ATC)'),
(326,326,1,'1.69 pH','309.3 mV','24.9 C (ATC)'),
(327,327,1,'4.02 pH','172.9 mV','24.9 C (ATC)'),
(328,328,1,'7.02 pH','-1.8 mV','24.9 C (ATC)'),
(329,329,1,'9.97 pH','-174.4 mV','25.7 C (ATC)'),
(330,330,1,'12.50 pH','-321.8 mV','24.9 C (ATC)'),
(331,331,1,'5.52 pH','87.7 mV','31.7 C (ATC)'),
(332,332,1,'1.65 pH','310.8 mV','24.0 C (ATC)'),
(333,333,1,'4.02 pH','173.5 mV','26.2 C (ATC)'),
(334,334,1,'7.01 pH','-2.2 mV','25.8 C (ATC)'),
(335,335,1,'10.01 pH','-177.6 mV','25.4 C (ATC)'),
(336,336,1,'12.43 pH','-321.0 mV','26.7 C (ATC)'),
(337,337,1,'5.87 pH','64.7 mV','24.3 C (ATC)'),
(338,338,1,'5.51 pH','88.3 mV','32.5 C (ATC)'),
(339,339,1,'6.97 pH','1.7 mV','25.0 C (ATC)'),
(340,340,1,'8.16 pH','-68.2 mV','24.3 C (ATC)'),
(341,341,1,'6.61 pH','21.0 mV','24.1 C (ATC)'),
(342,342,1,'5.78 pH','70.4 mV','23.9 C (ATC)'),
(343,343,1,'5.74 pH','72.3 mV','24.0 C (ATC)'),
(344,344,1,'5.81 pH','68.4 mV','23.9 C (ATC)'),
(345,345,1,'5.83 pH','67.3 mV','23.8 C (ATC)'),
(346,346,1,'5.80 pH','69.3 mV','23.9 C (ATC)'),
(347,347,1,'5.84 pH','66.9 mV','24.1 C (ATC)'),
(348,348,1,'5.75 pH','71.9 mV','24.1 C (ATC)'),
(349,349,1,'5.78 pH','70.0 mV','23.9 C (ATC)'),
(350,350,1,'5.94 pH','60.4 mV','22.0 C (ATC)'),
(351,351,1,'6.19 pH','46.7 mV','24.4 C (ATC)'),
(352,352,1,'6.10 pH','51.9 mV','24.6 C (ATC)'),
(353,353,1,'6.66 pH','18.7 mV','24.9 C (ATC)'),
(354,354,1,'6.69 pH','16.2 mV','23.3 C (ATC)'),
(355,355,1,'9.23 pH','-131.2 mV','25.1 C (ATC)'),
(356,356,1,'9.15 pH','-126.1 mV','24.9 C (ATC)'),
(357,357,1,'9.22 pH','-130.3 mV','24.5 C (ATC)'),
(358,358,1,'7.53 pH','-32.6 mV','25.4 C (ATC)'),
(359,359,1,'7.14 pH','-9.3 mV','24.9 C (ATC)'),
(360,360,1,'7.01 pH','-1.8 mV','26.6 C (ATC)'),
(361,361,1,'6.79 pH','11.0 mV','26.3 C (ATC)'),
(362,362,1,'6.44 pH','32.7 mV','26.5 C (ATC)'),
(363,363,1,'6.38 pH','35.0 mV','26.6 C (ATC)'),
(364,364,1,'4.39 pH','136.0 mV','23.3 C (ATC)'),
(365,365,1,'1.69 pH','308.4 mV','24.2 C (ATC)'),
(366,366,1,'4.03 pH','172.2 mV','24.2 C (ATC)'),
(367,367,1,'7.03 pH','-2.8 mV','24.1 C (ATC)'),
(368,368,1,'9.99 pH','-175.6 mV','24.6 C (ATC)'),
(369,369,1,'12.49 pH','-321.5 mV','24.7 C (ATC)'),
(370,370,1,'1.66 pH','310.9 mV','24.3 C (ATC)'),
(371,371,1,'4.02 pH','172.4 mV','23.6 C (ATC)'),
(372,372,1,'7.02 pH','-2.9 mV','23.8 C (ATC)'),
(373,373,1,'10.00 pH','-177.5 mV','25.2 C (ATC)'),
(374,374,1,'12.45 pH','-321.5 mV','25.5 C (ATC)'),
(375,375,1,'7.50 pH','-31.0 mV','32.9 C (ATC)'),
(376,376,1,'6.47 pH','29.0 mV','25.7 C (ATC)'),
(377,377,1,'6.50 pH','27.9 mV','24.5 C (ATC)'),
(378,378,1,'6.50 pH','27.5 mV','25.1 C (ATC)'),
(379,379,1,'5.51 pH','85.9 mV','25.4 C (ATC)'),
(380,380,1,'5.04 pH','115.1 mV','29.4 C (ATC)'),
(381,381,1,'5.86 pH','50.2 mV','24.0 C (ATC)'),
(382,382,1,'6.03 pH','56.2 mV','29.9 C (ATC)'),
(383,383,1,'5.88 pH','49.8 mV','28.5 C (ATC)'),
(384,384,1,'5.97 pH','43.5 mV','23.4 C (ATC)'),
(385,385,1,'7.43 pH','-27.2 mV','26.8 C (ATC)'),
(386,386,1,'5.70 pH','74.6 mV','25.9 C (ATC)'),
(387,387,1,'6.26 pH','27.0 mV','24.3 C (ATC)'),
(388,388,1,'6.42 pH','18.0 mV','25.3 C (ATC)'),
(389,389,1,'5.79 pH','69.3 mV','24.5 C (ATC)'),
(390,390,1,'5.65 pH','77.5 mV','24.6 C (ATC)'),
(391,391,1,'5.87 pH','64.6 mV','24.3 C (ATC)'),
(392,392,1,'5.71 pH','74.0 mV','24.4 C (ATC)'),
(393,393,1,'5.73 pH','72.5 mV','24.3 C (ATC)'),
(394,394,1,'5.69 pH','75.2 mV','24.5 C (ATC)'),
(395,395,1,'5.72 pH','73.3 mV','24.2 C (ATC)'),
(396,396,1,'5.75 pH','71.5 mV','24.1 C (ATC)'),
(397,397,1,'6.28 pH','25.9 mV','25.3 C (ATC)'),
(398,398,1,'6.30 pH','25.1 mV','23.5 C (ATC)'),
(399,399,1,'6.25 pH','42.6 mV','26.4 C (ATC)'),
(400,400,1,'6.43 pH','31.9 mV','26.7 C (ATC)'),
(401,401,1,'7.42 pH','-27.1 mV','25.2 C (ATC)'),
(402,402,1,'5.58 pH','81.8 mV','26.3 C (ATC)'),
(403,403,1,'2.73 pH','248.3 mV','24.9 C (ATC)'),
(404,404,1,'2.26 pH','275.4 mV','24.7 C (ATC)'),
(405,405,1,'6.23 pH','43.2 mV','22.2 C (ATC)'),
(406,406,1,'6.67 pH','18.0 mV','26.1 C (ATC)'),
(407,407,1,'6.80 pH','9.9 mV','25.1 C (ATC)'),
(408,408,1,'1.64 pH','313.0 mV','25.0 C (ATC)'),
(409,409,1,'4.04 pH','173.0 mV','25.2 C (ATC)'),
(410,410,1,'7.03 pH','-1.9 mV','25.4 C (ATC)'),
(411,411,1,'10.02 pH','-176.6 mV','24.9 C (ATC)'),
(412,412,1,'12.46 pH','-318.4 mV','24.2 C (ATC)'),
(413,413,1,'1.66 pH','312.4 mV','26.2 C (ATC)'),
(414,414,1,'4.01 pH','173.1 mV','26.0 C (ATC)'),
(415,415,1,'7.01 pH','-2.9 mV','26.3 C (ATC)'),
(416,416,1,'10.00 pH','-178.4 mV','26.1 C (ATC)'),
(417,417,1,'12.43 pH','-318.6 mV','23.5 C (ATC)'),
(418,418,1,'3.31 pH','219.4 mV','30.4 C (ATC)'),
(419,419,1,'7.50 pH','-29.6 mV','31.4 C (ATC)'),
(420,420,1,'6.81 pH','11.3 mV','25.4 C (ATC)'),
(421,421,1,'6.50 pH','29.4 mV','25.6 C (ATC)'),
(422,422,1,'6.08 pH','54.0 mV','25.4 C (ATC)'),
(423,423,1,'6.06 pH','55.1 mV','24.9 C (ATC)'),
(424,424,1,'5.90 pH','64.4 mV','25.8 C (ATC)'),
(425,425,1,'6.42 pH','32.2 mV','26.0 C (ATC)'),
(426,426,1,'7.33 pH','-21.6 mV','25.3 C (ATC)'),
(427,427,1,'7.23 pH','-15.4 mV','26.7 C (ATC)'),
(428,428,1,'9.24 pH','-131.1 mV','25.4 C (ATC)'),
(429,429,1,'9.14 pH','-125.8 mV','26.1 C (ATC)'),
(430,430,1,'6.66 pH','17.7 mV','26.7 C (ATC)'),
(431,431,1,'7.03 pH','-4.0 mV','26.9 C (ATC)'),
(432,432,1,'6.87 pH','5.2 mV','26.6 C (ATC)'),
(433,433,1,'6.81 pH','9.1 mV','25.8 C (ATC)'),
(434,434,1,'5.89 pH','64.7 mV','24.9 C (ATC)'),
(435,435,1,'5.85 pH','67.0 mV','24.8 C (ATC)'),
(436,436,1,'5.83 pH','68.3 mV','25.0 C (ATC)'),
(437,437,1,'5.83 pH','68.2 mV','25.0 C (ATC)'),
(438,438,1,'5.73 pH','74.2 mV','25.0 C (ATC)'),
(439,439,1,'5.85 pH','67.4 mV','24.9 C (ATC)'),
(440,440,1,'5.86 pH','66.6 mV','24.6 C (ATC)'),
(441,441,1,'5.80 pH','69.9 mV','24.7 C (ATC)'),
(442,442,1,'6.42 pH','33.8 mV','24.8 C (ATC)'),
(443,443,1,'5.75 pH','73.0 mV','24.7 C (ATC)'),
(444,444,1,'5.81 pH','69.3 mV','24.6 C (ATC)'),
(445,445,1,'6.94 pH','1.2 mV','25.4 C (ATC)'),
(446,446,1,'7.32 pH','-21.0 mV','26.5 C (ATC)'),
(447,447,1,'6.98 pH','-0.9 mV','25.2 C (ATC)'),
(448,448,1,'1.71 pH','312.1 mV','25.1 C (ATC)'),
(449,449,1,'4.00 pH','174.4 mV','24.5 C (ATC)'),
(450,450,1,'6.99 pH','-1.1 mV','24.5 C (ATC)'),
(451,451,1,'10.01 pH','-171.1 mV','25.0 C (ATC)'),
(452,452,1,'12.47 pH','-318.4 mV','24.7 C (ATC)'),
(453,453,1,'4.54 pH','127.0 mV','24.2 C (ATC)'),
(454,454,1,'5.83 pH','51.9 mV','25.4 C (ATC)'),
(455,455,1,'5.85 pH','51.0 mV','25.4 C (ATC)'),
(456,456,1,'6.20 pH','31.0 mV','26.0 C (ATC)'),
(457,457,1,'6.46 pH','15.4 mV','24.0 C (ATC)'),
(458,458,1,'6.25 pH','27.4 mV','23.5 C (ATC)'),
(459,459,1,'6.21 pH','29.6 mV','23.6 C (ATC)'),
(460,460,1,'4.53 pH','128.1 mV','23.2 C (ATC)'),
(461,461,1,'5.91 pH','48.7 mV','23.7 C (ATC)'),
(462,462,1,'5.93 pH','47.4 mV','24.3 C (ATC)'),
(463,463,1,'6.20 pH','32.0 mV','23.8 C (ATC)'),
(464,464,1,'6.45 pH','17.3 mV','24.5 C (ATC)'),
(465,465,1,'6.31 pH','25.3 mV','23.2 C (ATC)'),
(466,466,1,'6.29 pH','26.5 mV','24.7 C (ATC)'),
(467,467,1,'1.67 pH','310.4 mV','24.5 C (ATC)'),
(468,468,1,'4.03 pH','173.1 mV','24.7 C (ATC)'),
(469,469,1,'7.02 pH','-1.8 mV','24.9 C (ATC)'),
(470,470,1,'10.01 pH','-176.9 mV','24.8 C (ATC)'),
(471,471,1,'12.49 pH','-322.1 mV','24.7 C (ATC)'),
(472,472,1,'5.85 pH','66.5 mV','24.4 C (ATC)'),
(473,473,1,'6.05 pH','54.7 mV','24.7 C (ATC)'),
(474,474,1,'4.62 pH','123.1 mV','23.1 C (ATC)'),
(475,475,1,'1.65 pH','310.0 mV','24.7 C (ATC)'),
(476,476,1,'4.01 pH','171.7 mV','24.4 C (ATC)'),
(477,477,1,'7.00 pH','-2.3 mV','24.6 C (ATC)'),
(478,478,1,'10.02 pH','-178.1 mV','24.5 C (ATC)'),
(479,479,1,'12.50 pH','-322.9 mV','24.6 C (ATC)'),
(480,480,1,'5.51 pH','84.3 mV','23.9 C (ATC)'),
(481,481,1,'5.93 pH','46.7 mV','24.8 C (ATC)'),
(482,482,1,'5.51 pH','84.7 mV','25.6 C (ATC)'),
(483,483,1,'3.32 pH','215.0 mV','28.7 C (ATC)'),
(484,484,1,'7.50 pH','-32.0 mV','30.8 C (ATC)'),
(485,485,1,'5.95 pH','45.7 mV','23.6 C (ATC)'),
(486,486,1,'3.80 pH','185.5 mV','26.9 C (ATC)'),
(487,487,1,'6.24 pH','29.3 mV','26.0 C (ATC)'),
(488,488,1,'1.68 pH','310.8 mV','25.0 C (ATC)'),
(489,489,1,'6.40 pH','19.5 mV','24.5 C (ATC)'),
(490,490,1,'4.05 pH','170.8 mV','23.6 C (ATC)'),
(491,491,1,'7.04 pH','-3.9 mV','25.0 C (ATC)'),
(492,492,1,'10.01 pH','-177.9 mV','25.6 C (ATC)'),
(493,493,1,'12.45 pH','-321.5 mV','25.5 C (ATC)'),
(494,494,1,'5.70 pH','73.3 mV','24.8 C (ATC)'),
(495,495,1,'5.73 pH','71.8 mV','24.7 C (ATC)'),
(496,496,1,'5.74 pH','71.2 mV','24.7 C (ATC)'),
(497,497,1,'5.69 pH','73.9 mV','24.6 C (ATC)'),
(498,498,1,'6.28 pH','26.6 mV','24.6 C (ATC)'),
(499,499,1,'5.71 pH','72.7 mV','24.5 C (ATC)'),
(500,500,1,'5.79 pH','68.4 mV','24.4 C (ATC)'),
(501,501,1,'5.74 pH','71.0 mV','24.2 C (ATC)'),
(502,502,1,'5.77 pH','69.4 mV','24.3 C (ATC)'),
(503,503,1,'6.06 pH','52.3 mV','24.2 C (ATC)'),
(504,504,1,'6.25 pH','30.0 mV','24.6 C (ATC)'),
(505,505,1,'7.43 pH','-26.6 mV','25.2 C (ATC)'),
(506,506,1,'5.58 pH','80.5 mV','24.4 C (ATC)'),
(507,507,1,'6.61 pH','21.8 mV','26.2 C (ATC)'),
(508,508,1,'6.29 pH','38.9 mV','24.9 C (ATC)'),
(509,509,1,'7.45 pH','-27.8 mV','25.8 C (ATC)'),
(510,510,1,'5.51 pH','84.6 mV','24.6 C (ATC)'),
(511,511,1,'5.52 pH','84.0 mV','24.8 C (ATC)'),
(512,512,1,'6.58 pH','23.4 mV','25.6 C (ATC)'),
(513,513,1,'1.69 pH','311.8 mV','24.9 C (ATC)'),
(514,514,1,'4.04 pH','173.6 mV','24.8 C (ATC)'),
(515,515,1,'7.04 pH','-1.6 mV','24.7 C (ATC)'),
(516,516,1,'10.05 pH','-178.4 mV','24.7 C (ATC)'),
(517,517,1,'12.50 pH','-321.7 mV','24.5 C (ATC)'),
(518,518,1,'5.50 pH','91.1 mV','35.2 C (ATC)'),
(519,519,1,'5.51 pH','90.5 mV','34.6 C (ATC)'),
(520,520,1,'5.52 pH','88.8 mV','31.5 C (ATC)'),
(521,521,1,'1.67 pH','311.5 mV','24.8 C (ATC)'),
(522,522,1,'4.01 pH','175.4 mV','24.9 C (ATC)'),
(523,523,1,'7.01 pH','0.5 mV','24.8 C (ATC)'),
(524,524,1,'10.05 pH','-177.4 mV','25.5 C (ATC)'),
(525,525,1,'12.49 pH','-318.8 mV','24.7 C (ATC)'),
(526,526,1,'5.87 pH','66.7 mV','25.2 C (ATC)'),
(527,527,1,'5.40 pH','94.3 mV','25.2 C (ATC)'),
(528,528,1,'5.20 pH','105.7 mV','24.9 C (ATC)'),
(529,529,1,'5.52 pH','87.4 mV','24.7 C (ATC)'),
(530,530,1,'5.51 pH','90.2 mV','33.1 C (ATC)'),
(531,531,1,'4.57 pH','125.4 mV','23.3 C (ATC)'),
(532,532,1,'5.72 pH','74.7 mV','24.4 C (ATC)'),
(533,533,1,'5.69 pH','77.3 mV','24.4 C (ATC)'),
(534,534,1,'5.73 pH','74.6 mV','24.3 C (ATC)'),
(535,535,1,'5.76 pH','73.1 mV','24.3 C (ATC)'),
(536,536,1,'5.67 pH','78.3 mV','24.4 C (ATC)'),
(537,537,1,'5.67 pH','78.1 mV','24.5 C (ATC)'),
(538,538,1,'5.66 pH','78.7 mV','24.3 C (ATC)'),
(539,539,1,'5.75 pH','73.3 mV','24.2 C (ATC)'),
(540,540,1,'5.70 pH','76.6 mV','24.2 C (ATC)'),
(541,541,1,'5.72 pH','75.3 mV','24.1 C (ATC)'),
(542,542,1,'6.04 pH','56.9 mV','25.1 C (ATC)'),
(543,543,1,'5.90 pH','64.6 mV','24.7 C (ATC)'),
(544,544,1,'5.89 pH','48.9 mV','25.2 C (ATC)'),
(545,545,1,'5.91 pH','47.6 mV','24.7 C (ATC)'),
(546,546,1,'2.28 pH','277.3 mV','24.7 C (ATC)'),
(547,547,1,'2.27 pH','277.1 mV','24.6 C (ATC)'),
(548,548,1,'2.32 pH','274.2 mV','24.5 C (ATC)'),
(549,549,1,'2.39 pH','270.4 mV','24.5 C (ATC)'),
(550,550,1,'6.29 pH','25.9 mV','24.2 C (ATC)'),
(551,551,1,'2.31 pH','275.1 mV','24.3 C (ATC)'),
(552,552,1,'2.61 pH','257.6 mV','24.4 C (ATC)'),
(553,553,1,'2.60 pH','257.5 mV','24.4 C (ATC)'),
(554,554,1,'2.79 pH','246.7 mV','24.4 C (ATC)'),
(555,555,1,'2.54 pH','261.8 mV','24.4 C (ATC)'),
(556,556,1,'2.27 pH','277.3 mV','24.3 C (ATC)'),
(557,557,1,'2.42 pH','268.3 mV','24.4 C (ATC)'),
(558,558,1,'2.56 pH','260.1 mV','24.5 C (ATC)'),
(559,559,1,'2.71 pH','252.1 mV','25.0 C (ATC)'),
(560,560,1,'2.46 pH','266.6 mV','24.8 C (ATC)'),
(561,561,1,'2.55 pH','261.0 mV','24.7 C (ATC)'),
(562,562,1,'2.76 pH','248.4 mV','24.6 C (ATC)'),
(563,563,1,'6.48 pH','14.3 mV','23.4 C (ATC)'),
(564,564,1,'2.28 pH','277.2 mV','24.7 C (ATC)'),
(565,565,1,'2.38 pH','271.1 mV','24.6 C (ATC)'),
(566,566,1,'2.13 pH','285.7 mV','24.8 C (ATC)'),
(567,567,1,'2.34 pH','273.7 mV','24.8 C (ATC)'),
(568,568,1,'2.36 pH','271.9 mV','24.7 C (ATC)'),
(569,569,1,'2.48 pH','265.1 mV','24.6 C (ATC)'),
(570,570,1,'2.63 pH','256.5 mV','24.5 C (ATC)'),
(571,571,1,'2.78 pH','247.0 mV','24.4 C (ATC)'),
(572,572,1,'2.52 pH','262.4 mV','24.4 C (ATC)'),
(573,573,1,'2.58 pH','259.6 mV','24.4 C (ATC)'),
(574,574,1,'2.65 pH','255.0 mV','24.4 C (ATC)'),
(575,575,1,'2.40 pH','269.2 mV','24.3 C (ATC)'),
(576,576,1,'2.56 pH','260.0 mV','24.3 C (ATC)'),
(577,577,1,'2.71 pH','251.5 mV','24.3 C (ATC)'),
(578,578,1,'2.51 pH','263.0 mV','24.3 C (ATC)'),
(579,579,1,'2.76 pH','248.5 mV','24.3 C (ATC)'),
(580,580,1,'6.39 pH','20.2 mV','23.5 C (ATC)'),
(581,581,1,'6.36 pH','21.3 mV','23.5 C (ATC)'),
(582,582,1,'1.67 pH','312.5 mV','25.8 C (ATC)'),
(583,583,1,'4.03 pH','173.0 mV','25.6 C (ATC)'),
(584,584,1,'7.01 pH','-1.9 mV','25.5 C (ATC)'),
(585,585,1,'10.03 pH','-179.1 mV','25.0 C (ATC)'),
(586,586,1,'12.46 pH','-321.2 mV','24.4 C (ATC)'),
(587,587,1,'3.32 pH','218.9 mV','30.8 C (ATC)'),
(588,588,1,'7.49 pH','-31.1 mV','32.2 C (ATC)'),
(589,589,1,'4.98 pH','120.3 mV','32.3 C (ATC)'),
(590,590,1,'3.51 pH','208.7 mV','32.2 C (ATC)'),
(591,591,1,'5.52 pH','87.2 mV','31.9 C (ATC)'),
(592,592,1,'6.16 pH','47.5 mV','24.2 C (ATC)'),
(593,593,1,'8.09 pH','-65.0 mV','24.8 C (ATC)'),
(594,594,1,'8.11 pH','-66.6 mV','25.0 C (ATC)'),
(595,595,1,'8.12 pH','-67.1 mV','25.3 C (ATC)'),
(596,596,1,'1.66 pH','311.4 mV','26.6 C (ATC)'),
(597,597,1,'9.18 pH','-129.8 mV','25.7 C (ATC)'),
(598,598,1,'9.10 pH','-125.1 mV','25.6 C (ATC)'),
(599,599,1,'4.06 pH','170.4 mV','26.3 C (ATC)'),
(600,600,1,'7.05 pH','-4.1 mV','26.6 C (ATC)'),
(601,601,1,'9.99 pH','-176.5 mV','26.4 C (ATC)'),
(602,602,1,'12.45 pH','-320.9 mV','26.1 C (ATC)'),
(603,603,1,'5.03 pH','114.0 mV','24.4 C (ATC)'),
(604,604,1,'9.26 pH','-134.0 mV','24.3 C (ATC)'),
(605,605,1,'6.31 pH','39.0 mV','25.2 C (ATC)'),
(606,606,1,'5.74 pH','72.1 mV','23.8 C (ATC)'),
(607,607,1,'6.05 pH','53.9 mV','23.6 C (ATC)'),
(608,608,1,'5.77 pH','70.2 mV','23.7 C (ATC)'),
(609,609,1,'5.71 pH','74.1 mV','23.7 C (ATC)'),
(610,610,1,'5.71 pH','73.9 mV','23.8 C (ATC)'),
(611,611,1,'6.40 pH','33.3 mV','25.6 C (ATC)'),
(612,612,1,'5.81 pH','68.2 mV','23.8 C (ATC)'),
(613,613,1,'5.75 pH','71.9 mV','23.9 C (ATC)'),
(614,614,1,'5.71 pH','74.0 mV','23.7 C (ATC)'),
(615,615,1,'5.73 pH','72.8 mV','23.7 C (ATC)'),
(616,616,1,'5.74 pH','72.1 mV','23.7 C (ATC)'),
(617,617,1,'6.13 pH','49.9 mV','26.0 C (ATC)'),
(618,618,1,'6.02 pH','55.7 mV','24.0 C (ATC)'),
(619,619,1,'6.07 pH','53.0 mV','24.2 C (ATC)'),
(620,620,1,'6.09 pH','51.6 mV','24.2 C (ATC)'),
(621,621,1,'9.23 pH','-132.3 mV','24.3 C (ATC)'),
(622,622,1,'9.24 pH','-132.4 mV','23.2 C (ATC)'),
(623,623,1,'5.52 pH','88.7 mV','36.3 C (ATC)'),
(624,624,1,'6.59 pH','22.2 mV','24.3 C (ATC)'),
(625,625,1,'6.53 pH','25.7 mV','25.6 C (ATC)'),
(626,626,1,'6.43 pH','31.5 mV','25.5 C (ATC)'),
(627,627,1,'6.54 pH','25.2 mV','25.6 C (ATC)'),
(628,628,1,'1.69 pH','311.0 mV','25.1 C (ATC)'),
(629,629,1,'4.04 pH','172.2 mV','24.9 C (ATC)'),
(630,630,1,'7.02 pH','-2.6 mV','24.8 C (ATC)'),
(631,631,1,'10.03 pH','-179.5 mV','24.6 C (ATC)'),
(632,632,1,'12.43 pH','-320.2 mV','24.7 C (ATC)'),
(633,633,1,'5.03 pH','117.0 mV','31.1 C (ATC)'),
(634,634,1,'1.65 pH','307.9 mV','25.4 C (ATC)'),
(635,635,1,'4.04 pH','168.4 mV','25.5 C (ATC)'),
(636,636,1,'7.04 pH','-6.6 mV','25.6 C (ATC)'),
(637,637,1,'10.00 pH','-179.0 mV','25.5 C (ATC)'),
(638,638,1,'12.42 pH','-319.6 mV','25.1 C (ATC)'),
(639,639,1,'5.01 pH','118.4 mV','32.4 C (ATC)'),
(640,640,1,'7.34 pH','-21.5 mV','24.5 C (ATC)'),
(641,641,1,'4.97 pH','119.4 mV','28.3 C (ATC)'),
(642,642,1,'7.04 pH','-6.4 mV','26.3 C (ATC)'),
(643,643,1,'7.25 pH','-18.3 mV','24.3 C (ATC)'),
(644,644,1,'7.36 pH','-25.4 mV','25.4 C (ATC)'),
(645,645,1,'5.85 pH','66.2 mV','25.0 C (ATC)'),
(646,646,1,'5.82 pH','67.7 mV','24.7 C (ATC)'),
(647,647,1,'5.97 pH','59.0 mV','24.5 C (ATC)'),
(648,648,1,'5.82 pH','67.3 mV','24.3 C (ATC)'),
(649,649,1,'5.74 pH','71.1 mV','24.6 C (ATC)'),
(650,650,1,'5.81 pH','68.6 mV','24.5 C (ATC)'),
(651,651,1,'5.78 pH','70.4 mV','24.5 C (ATC)'),
(652,652,1,'5.84 pH','66.6 mV','24.2 C (ATC)'),
(653,653,1,'7.41 pH','-27.8 mV','23.8 C (ATC)'),
(654,654,1,'6.13 pH','46.9 mV','26.3 C (ATC)'),
(655,655,1,'5.95 pH','60.1 mV','24.0 C (ATC)'),
(656,656,1,'5.72 pH','73.8 mV','24.3 C (ATC)'),
(657,657,1,'5.88 pH','64.4 mV','24.7 C (ATC)'),
(658,658,1,'7.39 pH','-26.4 mV','23.3 C (ATC)'),
(659,659,1,'1.64 pH','305.8 mV','24.6 C (ATC)'),
(660,660,1,'4.01 pH','167.0 mV','24.6 C (ATC)'),
(661,661,1,'7.03 pH','-9.2 mV','24.5 C (ATC)'),
(662,662,1,'10.01 pH','-183.8 mV','25.1 C (ATC)'),
(663,663,1,'12.43 pH','-324.9 mV','24.8 C (ATC)'),
(664,664,1,'4.98 pH','112.9 mV','31.2 C (ATC)'),
(665,665,1,'5.04 pH','107.4 mV','25.3 C (ATC)'),
(666,666,1,'1.68 pH','307.4 mV','25.0 C (ATC)'),
(667,667,1,'5.03 pH','107.5 mV','25.0 C (ATC)'),
(668,668,1,'4.04 pH','169.1 mV','25.2 C (ATC)'),
(669,669,1,'7.04 pH','-6.2 mV','25.2 C (ATC)'),
(670,670,1,'10.00 pH','-179.4 mV','25.2 C (ATC)'),
(671,671,1,'12.44 pH','-322.5 mV','25.7 C (ATC)'),
(672,672,1,'5.52 pH','82.0 mV','35.5 C (ATC)'),
(673,673,1,'5.98 pH','52.8 mV','29.4 C (ATC)'),
(674,674,1,'6.26 pH','35.9 mV','24.0 C (ATC)'),
(675,675,1,'6.24 pH','36.5 mV','24.1 C (ATC)'),
(676,676,1,'9.19 pH','-135.4 mV','24.5 C (ATC)'),
(677,677,1,'9.06 pH','-128.1 mV','25.3 C (ATC)'),
(678,678,1,'7.30 pH','-24.9 mV','24.5 C (ATC)'),
(679,679,1,'5.95 pH','54.0 mV','25.0 C (ATC)'),
(680,680,1,'5.81 pH','62.1 mV','24.8 C (ATC)'),
(681,681,1,'5.74 pH','65.7 mV','24.8 C (ATC)'),
(682,682,1,'5.83 pH','61.3 mV','24.9 C (ATC)'),
(683,683,1,'5.81 pH','61.8 mV','24.6 C (ATC)'),
(684,684,1,'5.78 pH','64.1 mV','24.7 C (ATC)'),
(685,685,1,'5.75 pH','64.3 mV','24.6 C (ATC)'),
(686,686,1,'5.83 pH','60.8 mV','24.6 C (ATC)'),
(687,687,1,'5.78 pH','63.9 mV','24.6 C (ATC)'),
(688,688,1,'5.76 pH','65.1 mV','24.4 C (ATC)'),
(689,689,1,'5.01 pH','112.0 mV','32.5 C (ATC)'),
(690,690,1,'4.62 pH','126.7 mV','26.9 C (ATC)'),
(691,691,1,'5.94 pH','49.3 mV','26.2 C (ATC)'),
(692,692,1,'6.02 pH','45.2 mV','24.3 C (ATC)'),
(693,693,1,'6.23 pH','32.8 mV','23.9 C (ATC)'),
(694,694,1,'6.38 pH','23.8 mV','26.2 C (ATC)'),
(695,695,1,'6.31 pH','27.9 mV','23.7 C (ATC)'),
(696,696,1,'6.32 pH','27.2 mV','24.5 C (ATC)'),
(697,697,1,'1.63 pH','304.9 mV','25.6 C (ATC)'),
(698,698,1,'4.02 pH','165.0 mV','24.5 C (ATC)'),
(699,699,1,'7.01 pH','-8.7 mV','24.6 C (ATC)'),
(700,700,1,'9.99 pH','-182.4 mV','25.9 C (ATC)'),
(701,701,1,'12.48 pH','-328.0 mV','25.7 C (ATC)'),
(702,702,1,'5.46 pH','84.3 mV','34.6 C (ATC)'),
(703,703,1,'5.02 pH','111.4 mV','35.7 C (ATC)'),
(704,704,1,'5.00 pH','112.0 mV','35.0 C (ATC)'),
(705,705,1,'5.00 pH','111.3 mV','32.6 C (ATC)'),
(706,706,1,'1.68 pH','308.5 mV','25.6 C (ATC)'),
(707,707,1,'4.05 pH','169.6 mV','26.0 C (ATC)'),
(708,708,1,'7.05 pH','-5.8 mV','25.7 C (ATC)'),
(709,709,1,'9.99 pH','-178.2 mV','25.9 C (ATC)'),
(710,710,1,'12.44 pH','-321.7 mV','26.0 C (ATC)'),
(711,711,1,'9.00 pH','-123.9 mV','23.6 C (ATC)'),
(712,712,1,'8.91 pH','-119.1 mV','23.8 C (ATC)'),
(713,713,1,'6.10 pH','44.2 mV','25.5 C (ATC)'),
(714,714,1,'5.93 pH','54.1 mV','25.1 C (ATC)'),
(715,715,1,'5.52 pH','80.3 mV','32.7 C (ATC)'),
(716,716,1,'9.08 pH','-129.1 mV','24.6 C (ATC)'),
(717,717,1,'9.08 pH','-128.6 mV','23.7 C (ATC)'),
(718,718,1,'9.05 pH','-127.5 mV','24.5 C (ATC)'),
(719,719,1,'5.48 pH','79.9 mV','24.4 C (ATC)'),
(720,720,1,'5.47 pH','80.9 mV','24.7 C (ATC)'),
(721,721,1,'5.44 pH','82.5 mV','24.8 C (ATC)'),
(722,722,1,'5.40 pH','84.6 mV','24.6 C (ATC)'),
(723,723,1,'5.51 pH','78.1 mV','24.2 C (ATC)'),
(724,724,1,'5.41 pH','83.6 mV','24.4 C (ATC)'),
(725,725,1,'5.46 pH','81.2 mV','25.0 C (ATC)'),
(726,726,1,'5.41 pH','84.3 mV','24.5 C (ATC)'),
(727,727,1,'9.03 pH','-126.6 mV','25.0 C (ATC)'),
(728,728,1,'8.99 pH','-123.7 mV','24.9 C (ATC)'),
(729,729,1,'7.31 pH','-21.1 mV','24.0 C (ATC)'),
(730,730,1,'7.35 pH','-23.4 mV','25.4 C (ATC)'),
(731,731,1,'7.29 pH','-20.1 mV','25.6 C (ATC)'),
(732,732,1,'7.48 pH','-31.2 mV','25.4 C (ATC)'),
(733,733,1,'5.76 pH','69.2 mV','25.8 C (ATC)'),
(734,734,1,'7.25 pH','-17.3 mV','24.9 C (ATC)'),
(735,735,1,'7.32 pH','-21.6 mV','26.2 C (ATC)'),
(736,736,1,'7.37 pH','-24.5 mV','24.8 C (ATC)'),
(737,737,1,'7.42 pH','-27.3 mV','23.9 C (ATC)'),
(738,738,1,'7.17 pH','-13.0 mV','25.9 C (ATC)'),
(739,739,1,'5.72 pH','71.8 mV','26.4 C (ATC)'),
(740,740,1,'7.41 pH','-26.9 mV','25.8 C (ATC)'),
(741,741,1,'4.63 pH','126.5 mV','26.3 C (ATC)'),
(742,742,1,'5.95 pH','49.9 mV','25.0 C (ATC)'),
(743,743,1,'6.02 pH','45.2 mV','24.4 C (ATC)'),
(744,744,1,'6.28 pH','29.4 mV','25.4 C (ATC)'),
(745,745,1,'6.45 pH','20.4 mV','23.6 C (ATC)'),
(746,746,1,'6.28 pH','30.6 mV','24.2 C (ATC)'),
(747,747,1,'6.26 pH','31.5 mV','23.4 C (ATC)'),
(748,748,1,'1.68 pH','307.3 mV','24.2 C (ATC)'),
(749,749,1,'4.05 pH','168.5 mV','24.2 C (ATC)'),
(750,750,1,'7.04 pH','-6.3 mV','24.4 C (ATC)'),
(751,751,1,'10.03 pH','-181.3 mV','24.7 C (ATC)'),
(752,752,1,'12.50 pH','-325.5 mV','24.5 C (ATC)'),
(753,753,1,'4.99 pH','116.4 mV','31.7 C (ATC)'),
(754,754,1,'5.53 pH','82.3 mV','26.1 C (ATC)'),
(755,755,1,'1.65 pH','306.8 mV','24.2 C (ATC)'),
(756,756,1,'5.53 pH','82.1 mV','24.5 C (ATC)'),
(757,757,1,'4.06 pH','167.0 mV','24.5 C (ATC)'),
(758,758,1,'5.52 pH','82.4 mV','24.5 C (ATC)'),
(759,759,1,'7.01 pH','-5.3 mV','24.4 C (ATC)'),
(760,760,1,'9.97 pH','-177.5 mV','23.9 C (ATC)'),
(761,761,1,'5.74 pH','69.7 mV','25.3 C (ATC)'),
(762,762,1,'3.31 pH','215.3 mV','29.4 C (ATC)'),
(763,763,1,'12.44 pH','-322.4 mV','24.9 C (ATC)'),
(764,764,1,'5.52 pH','84.4 mV','30.8 C (ATC)'),
(765,765,1,'5.01 pH','112.7 mV','25.1 C (ATC)'),
(766,766,1,'5.87 pH','62.4 mV','24.9 C (ATC)'),
(767,767,1,'6.74 pH','10.4 mV','25.3 C (ATC)'),
(768,768,1,'5.84 pH','63.8 mV','25.0 C (ATC)'),
(769,769,1,'5.81 pH','66.3 mV','24.5 C (ATC)'),
(770,770,1,'5.78 pH','67.2 mV','24.2 C (ATC)'),
(771,771,1,'5.66 pH','73.7 mV','24.4 C (ATC)'),
(772,772,1,'5.75 pH','68.8 mV','24.2 C (ATC)'),
(773,773,1,'5.65 pH','74.5 mV','24.3 C (ATC)'),
(774,774,1,'5.73 pH','68.4 mV','24.3 C (ATC)'),
(775,775,1,'5.04 pH','111.2 mV','26.3 C (ATC)'),
(776,776,1,'6.11 pH','48.0 mV','24.8 C (ATC)'),
(777,777,1,'5.94 pH','58.4 mV','25.3 C (ATC)'),
(778,778,1,'6.02 pH','53.6 mV','25.2 C (ATC)'),
(779,779,1,'5.03 pH','114.1 mV','32.6 C (ATC)'),
(780,780,1,'5.80 pH','66.4 mV','24.8 C (ATC)'),
(781,781,1,'9.09 pH','-126.5 mV','25.3 C (ATC)'),
(782,782,1,'5.01 pH','115.7 mV','32.2 C (ATC)'),
(783,783,1,'6.44 pH','27.7 mV','24.1 C (ATC)'),
(784,784,1,'4.55 pH','126.6 mV','25.2 C (ATC)'),
(785,785,1,'6.85 pH','3.9 mV','23.5 C (ATC)'),
(786,786,1,'6.67 pH','14.5 mV','25.7 C (ATC)'),
(787,787,1,'5.99 pH','43.7 mV','25.7 C (ATC)'),
(788,788,1,'6.01 pH','44.7 mV','24.8 C (ATC)'),
(789,789,1,'6.32 pH','26.8 mV','25.0 C (ATC)'),
(790,790,1,'6.41 pH','21.6 mV','26.4 C (ATC)'),
(791,791,1,'6.36 pH','24.8 mV','26.4 C (ATC)'),
(792,792,1,'6.35 pH','25.0 mV','23.5 C (ATC)'),
(793,793,1,'1.64 pH','308.5 mV','24.1 C (ATC)'),
(794,794,1,'4.02 pH','169.9 mV','24.1 C (ATC)'),
(795,795,1,'7.02 pH','-4.9 mV','24.4 C (ATC)'),
(796,796,1,'9.98 pH','-177.4 mV','25.1 C (ATC)'),
(797,797,1,'12.47 pH','-323.3 mV','25.2 C (ATC)'),
(798,798,1,'7.32 pH','-22.4 mV','24.9 C (ATC)'),
(799,799,1,'6.45 pH','28.4 mV','23.6 C (ATC)'),
(800,800,1,'1.65 pH','306.9 mV','25.4 C (ATC)'),
(801,801,1,'4.04 pH','166.9 mV','25.2 C (ATC)'),
(802,802,1,'7.04 pH','-7.8 mV','25.1 C (ATC)'),
(803,803,1,'9.97 pH','-178.5 mV','24.9 C (ATC)'),
(804,804,1,'3.25 pH','219.0 mV','29.5 C (ATC)'),
(805,805,1,'7.48 pH','-32.4 mV','31.6 C (ATC)'),
(806,806,1,'5.82 pH','65.4 mV','25.7 C (ATC)'),
(807,807,1,'5.66 pH','74.6 mV','25.6 C (ATC)'),
(808,808,1,'5.70 pH','72.3 mV','25.3 C (ATC)'),
(809,809,1,'12.46 pH','-306.0 mV','25.0 C (ATC)'),
(810,810,1,'5.71 pH','71.3 mV','24.0 C (ATC)'),
(811,811,1,'5.59 pH','78.4 mV','23.9 C (ATC)'),
(812,812,1,'5.65 pH','75.1 mV','24.6 C (ATC)'),
(813,813,1,'5.67 pH','74.1 mV','23.6 C (ATC)'),
(814,814,1,'5.59 pH','78.6 mV','24.0 C (ATC)'),
(815,815,1,'5.74 pH','69.6 mV','24.4 C (ATC)'),
(816,816,1,'5.57 pH','79.6 mV','24.3 C (ATC)'),
(817,817,1,'3.81 pH','184.5 mV','27.8 C (ATC)'),
(818,818,1,'5.49 pH','85.1 mV','25.9 C (ATC)'),
(819,819,1,'5.71 pH','72.1 mV','25.7 C (ATC)'),
(820,820,1,'5.33 pH','94.0 mV','25.7 C (ATC)'),
(821,821,1,'5.75 pH','69.4 mV','25.6 C (ATC)'),
(822,822,1,'1.64 pH','311.4 mV','26.2 C (ATC)'),
(823,823,1,'4.03 pH','171.9 mV','26.7 C (ATC)'),
(824,824,1,'7.04 pH','-4.1 mV','24.6 C (ATC)'),
(825,825,1,'10.01 pH','-177.6 mV','26.4 C (ATC)'),
(826,826,1,'12.43 pH','-319.2 mV','26.5 C (ATC)'),
(827,827,1,'4.57 pH','127.5 mV','25.4 C (ATC)'),
(828,828,1,'4.56 pH','127.3 mV','23.5 C (ATC)'),
(829,829,1,'1.65 pH','308.9 mV','25.0 C (ATC)'),
(830,830,1,'4.05 pH','170.3 mV','25.4 C (ATC)'),
(831,831,1,'7.01 pH','-1.2 mV','24.8 C (ATC)'),
(832,832,1,'9.99 pH','-174.0 mV','25.2 C (ATC)'),
(833,833,1,'12.48 pH','-318.2 mV','25.4 C (ATC)'),
(834,834,1,'5.94 pH','48.7 mV','23.5 C (ATC)'),
(835,835,1,'5.97 pH','47.1 mV','23.8 C (ATC)'),
(836,836,1,'5.60 pH','80.0 mV','24.5 C (ATC)'),
(837,837,1,'1.65 pH','312.0 mV','25.0 C (ATC)'),
(838,838,1,'4.02 pH','173.7 mV','25.1 C (ATC)'),
(839,839,1,'7.02 pH','-2.0 mV','25.1 C (ATC)'),
(840,840,1,'9.97 pH','-174.6 mV','25.1 C (ATC)'),
(841,841,1,'12.45 pH','-319.9 mV','25.4 C (ATC)'),
(842,842,1,'6.25 pH','31.5 mV','24.6 C (ATC)'),
(843,843,1,'5.01 pH','116.6 mV','31.2 C (ATC)'),
(844,844,1,'7.32 pH','-19.7 mV','24.9 C (ATC)'),
(845,845,1,'5.03 pH','114.8 mV','29.5 C (ATC)'),
(846,846,1,'6.51 pH','16.3 mV','23.7 C (ATC)'),
(847,847,1,'2.90 pH','235.5 mV','24.4 C (ATC)'),
(848,848,1,'2.72 pH','245.9 mV','24.4 C (ATC)'),
(849,849,1,'5.47 pH','90.3 mV','33.7 C (ATC)'),
(850,850,1,'5.01 pH','116.8 mV','32.1 C (ATC)'),
(851,851,1,'9.14 pH','-124.4 mV','24.0 C (ATC)'),
(852,852,1,'9.11 pH','-122.7 mV','24.2 C (ATC)'),
(853,853,1,'6.29 pH','29.1 mV','24.8 C (ATC)'),
(854,854,1,'7.03 pH','-2.3 mV','26.2 C (ATC)'),
(855,855,1,'6.29 pH','29.6 mV','24.8 C (ATC)'),
(856,856,1,'6.27 pH','30.4 mV','24.5 C (ATC)'),
(857,857,1,'9.18 pH','-126.6 mV','24.4 C (ATC)'),
(858,858,1,'9.15 pH','-125.2 mV','24.3 C (ATC)'),
(859,859,1,'7.03 pH','-2.4 mV','26.2 C (ATC)'),
(860,860,1,'7.03 pH','-2.5 mV','26.3 C (ATC)'),
(861,861,1,'5.80 pH','68.4 mV','24.3 C (ATC)'),
(862,862,1,'5.75 pH','71.3 mV','24.0 C (ATC)'),
(863,863,1,'5.76 pH','70.4 mV','23.9 C (ATC)'),
(864,864,1,'5.71 pH','73.4 mV','24.0 C (ATC)'),
(865,865,1,'5.75 pH','70.9 mV','23.9 C (ATC)'),
(866,866,1,'5.76 pH','70.5 mV','24.1 C (ATC)'),
(867,867,1,'5.78 pH','69.4 mV','24.0 C (ATC)'),
(868,868,1,'5.70 pH','73.9 mV','24.2 C (ATC)'),
(869,869,1,'5.68 pH','75.2 mV','24.5 C (ATC)'),
(870,870,1,'5.63 pH','77.9 mV','24.2 C (ATC)'),
(871,871,1,'5.03 pH','115.9 mV','32.7 C (ATC)'),
(872,872,1,'6.09 pH','52.0 mV','25.7 C (ATC)'),
(873,873,1,'6.06 pH','53.4 mV','25.4 C (ATC)'),
(874,874,1,'5.89 pH','63.6 mV','25.6 C (ATC)'),
(875,875,1,'5.91 pH','61.9 mV','25.4 C (ATC)'),
(876,876,1,'6.12 pH','49.6 mV','23.4 C (ATC)'),
(877,877,1,'6.99 pH','-0.1 mV','24.9 C (ATC)'),
(878,878,1,'6.98 pH','0.5 mV','23.8 C (ATC)'),
(879,879,1,'6.52 pH','26.9 mV','24.0 C (ATC)'),
(880,880,1,'6.55 pH','25.2 mV','23.3 C (ATC)'),
(881,881,1,'6.56 pH','24.7 mV','22.9 C (ATC)'),
(882,882,1,'1.70 pH','310.3 mV','24.3 C (ATC)'),
(883,883,1,'3.97 pH','177.7 mV','24.4 C (ATC)'),
(884,884,1,'7.01 pH','-0.3 mV','24.3 C (ATC)'),
(885,885,1,'9.96 pH','-173.2 mV','25.1 C (ATC)'),
(886,886,1,'12.41 pH','-317.2 mV','25.4 C (ATC)'),
(887,887,1,'1.67 pH','312.1 mV','26.2 C (ATC)'),
(888,888,1,'4.02 pH','173.1 mV','26.0 C (ATC)'),
(889,889,1,'7.00 pH','-2.4 mV','25.5 C (ATC)'),
(890,890,1,'10.01 pH','-179.3 mV','25.7 C (ATC)'),
(891,891,1,'12.46 pH','-322.8 mV','25.3 C (ATC)'),
(892,892,1,'5.54 pH','85.3 mV','25.0 C (ATC)'),
(893,893,1,'5.54 pH','85.6 mV','23.5 C (ATC)'),
(894,894,1,'9.19 pH','-127.7 mV','23.7 C (ATC)'),
(895,895,1,'6.29 pH','41.6 mV','23.6 C (ATC)'),
(896,896,1,'5.52 pH','89.9 mV','36.4 C (ATC)'),
(897,897,1,'5.87 pH','66.7 mV','25.6 C (ATC)'),
(898,898,1,'5.87 pH','65.9 mV','23.9 C (ATC)'),
(899,899,1,'5.74 pH','73.7 mV','24.0 C (ATC)'),
(900,900,1,'5.65 pH','79.3 mV','24.3 C (ATC)'),
(901,901,1,'5.53 pH','85.8 mV','23.7 C (ATC)'),
(902,902,1,'4.53 pH','128.4 mV','26.3 C (ATC)'),
(903,903,1,'5.54 pH','85.6 mV','24.7 C (ATC)'),
(904,904,1,'5.46 pH','90.9 mV','25.9 C (ATC)'),
(905,905,1,'5.94 pH','62.5 mV','26.2 C (ATC)'),
(906,906,1,'5.92 pH','63.6 mV','25.9 C (ATC)'),
(907,907,1,'4.86 pH','124.6 mV','24.0 C (ATC)'),
(908,908,1,'7.32 pH','-20.9 mV','26.0 C (ATC)'),
(909,909,1,'7.01 pH','-2.7 mV','26.5 C (ATC)'),
(910,910,1,'4.45 pH','149.5 mV','24.5 C (ATC)'),
(911,911,1,'5.04 pH','114.0 mV','24.8 C (ATC)'),
(912,912,1,'5.39 pH','91.0 mV','24.5 C (ATC)'),
(913,913,1,'5.69 pH','76.9 mV','24.7 C (ATC)'),
(914,914,1,'5.82 pH','69.4 mV','25.0 C (ATC)'),
(915,915,1,'5.84 pH','67.9 mV','24.6 C (ATC)'),
(916,916,1,'5.82 pH','69.0 mV','25.0 C (ATC)'),
(917,917,1,'5.84 pH','67.9 mV','24.5 C (ATC)'),
(918,918,1,'5.85 pH','67.3 mV','24.9 C (ATC)'),
(919,919,1,'5.91 pH','63.7 mV','25.0 C (ATC)'),
(920,920,1,'5.00 pH','117.3 mV','24.6 C (ATC)'),
(921,921,1,'4.97 pH','118.9 mV','24.4 C (ATC)'),
(922,922,1,'7.20 pH','-14.4 mV','26.1 C (ATC)'),
(923,923,1,'5.88 pH','49.6 mV','25.1 C (ATC)'),
(924,924,1,'4.83 pH','126.8 mV','24.5 C (ATC)'),
(925,925,1,'4.93 pH','121.2 mV','24.5 C (ATC)'),
(926,926,1,'5.92 pH','47.3 mV','25.0 C (ATC)'),
(927,927,1,'3.50 pH','206.3 mV','26.8 C (ATC)'),
(928,928,1,'6.25 pH','44.2 mV','25.2 C (ATC)'),
(929,929,1,'5.39 pH','94.7 mV','25.6 C (ATC)'),
(930,930,1,'7.22 pH','-14.8 mV','25.3 C (ATC)'),
(931,931,1,'6.27 pH','26.8 mV','23.7 C (ATC)'),
(932,932,1,'7.17 pH','-12.1 mV','25.7 C (ATC)'),
(933,933,1,'5.73 pH','74.8 mV','25.4 C (ATC)'),
(934,934,1,'6.48 pH','14.8 mV','23.7 C (ATC)'),
(935,935,1,'5.02 pH','119.5 mV','33.2 C (ATC)'),
(936,936,1,'7.38 pH','-24.5 mV','25.7 C (ATC)'),
(937,937,1,'7.17 pH','-12.1 mV','26.0 C (ATC)'),
(938,938,1,'6.60 pH','21.5 mV','25.7 C (ATC)'),
(939,939,1,'6.30 pH','25.6 mV','25.3 C (ATC)'),
(940,940,1,'6.32 pH','24.2 mV','23.7 C (ATC)'),
(941,941,1,'1.69 pH','308.7 mV','24.6 C (ATC)'),
(942,942,1,'4.05 pH','170.8 mV','24.5 C (ATC)'),
(943,943,1,'7.02 pH','-2.5 mV','24.4 C (ATC)'),
(944,944,1,'10.03 pH','-178.7 mV','24.9 C (ATC)'),
(945,945,1,'12.47 pH','-320.5 mV','24.3 C (ATC)'),
(946,946,1,'5.52 pH','87.2 mV','31.7 C (ATC)'),
(947,947,1,'5.00 pH','118.6 mV','32.0 C (ATC)'),
(948,948,1,'5.57 pH','82.4 mV','25.1 C (ATC)'),
(949,949,1,'1.66 pH','312.2 mV','25.9 C (ATC)'),
(950,950,1,'4.03 pH','173.1 mV','26.3 C (ATC)'),
(951,951,1,'7.02 pH','-3.2 mV','26.4 C (ATC)'),
(952,952,1,'10.02 pH','-179.2 mV','25.2 C (ATC)'),
(953,953,1,'12.44 pH','-321.4 mV','25.3 C (ATC)'),
(954,954,1,'5.50 pH','88.5 mV','32.3 C (ATC)'),
(955,955,1,'6.13 pH','49.5 mV','24.7 C (ATC)'),
(956,956,1,'6.64 pH','19.2 mV','26.1 C (ATC)'),
(957,957,1,'8.96 pH','-116.2 mV','24.5 C (ATC)'),
(958,958,1,'5.74 pH','72.4 mV','25.0 C (ATC)'),
(959,959,1,'5.73 pH','73.1 mV','24.9 C (ATC)'),
(960,960,1,'5.80 pH','69.0 mV','25.0 C (ATC)'),
(961,961,1,'5.81 pH','68.1 mV','24.9 C (ATC)'),
(962,962,1,'5.72 pH','73.6 mV','24.7 C (ATC)'),
(963,963,1,'5.71 pH','73.7 mV','24.7 C (ATC)'),
(964,964,1,'5.74 pH','72.0 mV','24.6 C (ATC)'),
(965,965,1,'5.82 pH','67.7 mV','25.0 C (ATC)'),
(966,966,1,'5.70 pH','74.4 mV','24.5 C (ATC)'),
(967,967,1,'5.76 pH','71.0 mV','24.7 C (ATC)'),
(968,968,1,'5.79 pH','69.2 mV','24.6 C (ATC)'),
(969,969,1,'2.51 pH','261.1 mV','24.5 C (ATC)'),
(970,970,1,'2.43 pH','265.8 mV','24.4 C (ATC)'),
(971,971,1,'5.82 pH','67.7 mV','24.2 C (ATC)'),
(972,972,1,'5.89 pH','63.6 mV','25.0 C (ATC)'),
(973,973,1,'5.67 pH','76.1 mV','24.7 C (ATC)'),
(974,974,1,'5.83 pH','67.0 mV','24.8 C (ATC)'),
(975,975,1,'5.87 pH','64.8 mV','25.0 C (ATC)'),
(976,976,1,'6.63 pH','19.9 mV','26.4 C (ATC)'),
(977,977,1,'6.65 pH','19.1 mV','24.5 C (ATC)'),
(978,978,1,'7.71 pH','-43.0 mV','24.5 C (ATC)'),
(979,979,1,'7.70 pH','-42.2 mV','24.5 C (ATC)'),
(980,980,1,'7.43 pH','-26.4 mV','25.5 C (ATC)'),
(981,981,1,'1.64 pH','303.3 mV','24.4 C (ATC)'),
(982,982,1,'4.05 pH','164.3 mV','24.3 C (ATC)'),
(983,983,1,'7.05 pH','-8.3 mV','24.5 C (ATC)'),
(984,984,1,'10.02 pH','-179.7 mV','24.9 C (ATC)'),
(985,985,1,'12.45 pH','-319.7 mV','24.8 C (ATC)'),
(986,986,1,'5.00 pH','110.5 mV','26.0 C (ATC)'),
(987,987,1,'1.66 pH','310.1 mV','25.4 C (ATC)'),
(988,988,1,'4.04 pH','170.8 mV','24.8 C (ATC)'),
(989,989,1,'7.03 pH','-4.4 mV','25.4 C (ATC)'),
(990,990,1,'10.04 pH','-180.1 mV','24.4 C (ATC)'),
(991,991,1,'12.45 pH','-322.0 mV','25.6 C (ATC)'),
(992,992,1,'3.51 pH','201.1 mV','32.2 C (ATC)'),
(993,993,1,'3.92 pH','163.6 mV','26.3 C (ATC)'),
(994,994,1,'6.03 pH','51.4 mV','28.5 C (ATC)'),
(995,995,1,'4.04 pH','155.8 mV','24.6 C (ATC)'),
(996,996,1,'4.51 pH','129.6 mV','26.6 C (ATC)'),
(997,997,1,'4.76 pH','123.4 mV','24.0 C (ATC)'),
(998,998,1,'6.30 pH','25.6 mV','24.3 C (ATC)'),
(999,999,1,'5.60 pH','75.4 mV','24.6 C (ATC)'),
(1000,1000,1,'5.59 pH','75.7 mV','24.5 C (ATC)'),
(1001,1001,1,'5.83 pH','62.0 mV','24.7 C (ATC)'),
(1002,1002,1,'5.64 pH','72.7 mV','24.4 C (ATC)'),
(1003,1003,1,'5.31 pH','92.0 mV','24.6 C (ATC)'),
(1004,1004,1,'5.82 pH','62.6 mV','24.5 C (ATC)'),
(1005,1005,1,'5.65 pH','72.0 mV','24.1 C (ATC)'),
(1006,1006,1,'5.63 pH','82.6 mV','25.0 C (ATC)'),
(1007,1007,1,'7.01 pH','-5.7 mV','31.1 C (ATC)'),
(1008,1008,1,'6.42 pH','27.8 mV','23.5 C (ATC)'),
(1009,1009,1,'6.16 pH','34.6 mV','26.1 C (ATC)'),
(1010,1010,1,'6.20 pH','32.2 mV','25.9 C (ATC)'),
(1011,1011,1,'7.47 pH','-33.0 mV','32.0 C (ATC)'),
(1012,1012,1,'7.35 pH','-22.6 mV','24.8 C (ATC)'),
(1013,1013,1,'2.50 pH','253.6 mV','24.1 C (ATC)'),
(1014,1014,1,'7.37 pH','-24.5 mV','24.9 C (ATC)'),
(1015,1015,1,'2.32 pH','264.1 mV','24.1 C (ATC)'),
(1016,1016,1,'7.24 pH','-17.1 mV','23.8 C (ATC)'),
(1017,1017,1,'6.01 pH','51.4 mV','24.0 C (ATC)'),
(1018,1018,1,'1.69 pH','306.2 mV','24.3 C (ATC)'),
(1019,1019,1,'4.05 pH','169.6 mV','24.9 C (ATC)'),
(1020,1020,1,'6.99 pH','-1.2 mV','24.4 C (ATC)'),
(1021,1021,1,'10.00 pH','-176.5 mV','25.6 C (ATC)'),
(1022,1022,1,'12.45 pH','-318.3 mV','24.3 C (ATC)'),
(1023,1023,1,'5.50 pH','86.7 mV','29.5 C (ATC)'),
(1024,1024,1,'7.49 pH','-31.1 mV','30.8 C (ATC)'),
(1025,1025,1,'7.00 pH','-1.8 mV','29.0 C (ATC)'),
(1026,1026,1,'6.20 pH','44.3 mV','24.3 C (ATC)'),
(1027,1027,1,'6.20 pH','44.2 mV','23.3 C (ATC)'),
(1028,1028,1,'7.40 pH','-24.9 mV','24.6 C (ATC)'),
(1029,1029,1,'5.84 pH','65.8 mV','25.7 C (ATC)'),
(1030,1030,1,'5.83 pH','66.2 mV','25.5 C (ATC)'),
(1031,1031,1,'2.64 pH','251.9 mV','25.0 C (ATC)'),
(1032,1032,1,'2.45 pH','262.2 mV','24.1 C (ATC)'),
(1033,1033,1,'2.40 pH','265.3 mV','24.2 C (ATC)'),
(1034,1034,1,'2.45 pH','262.4 mV','24.8 C (ATC)'),
(1035,1035,1,'2.52 pH','258.3 mV','24.7 C (ATC)'),
(1036,1036,1,'2.56 pH','255.6 mV','24.0 C (ATC)'),
(1037,1037,1,'2.62 pH','252.3 mV','24.5 C (ATC)'),
(1038,1038,1,'2.69 pH','248.5 mV','24.6 C (ATC)'),
(1039,1039,1,'2.71 pH','246.8 mV','24.2 C (ATC)'),
(1040,1040,1,'2.54 pH','256.9 mV','24.1 C (ATC)'),
(1041,1041,1,'2.40 pH','264.8 mV','24.0 C (ATC)'),
(1042,1042,1,'2.45 pH','262.0 mV','24.3 C (ATC)'),
(1043,1043,1,'2.52 pH','258.4 mV','24.6 C (ATC)'),
(1044,1044,1,'2.59 pH','254.4 mV','24.7 C (ATC)'),
(1045,1045,1,'2.40 pH','264.7 mV','23.7 C (ATC)'),
(1046,1046,1,'2.70 pH','247.4 mV','24.3 C (ATC)'),
(1047,1047,1,'2.73 pH','246.3 mV','24.7 C (ATC)'),
(1048,1048,1,'2.35 pH','268.2 mV','24.6 C (ATC)'),
(1049,1049,1,'2.43 pH','263.5 mV','24.6 C (ATC)'),
(1050,1050,1,'2.87 pH','238.1 mV','24.5 C (ATC)'),
(1051,1051,1,'2.88 pH','236.9 mV','24.0 C (ATC)'),
(1052,1052,1,'2.69 pH','248.0 mV','24.0 C (ATC)'),
(1053,1053,1,'2.61 pH','252.8 mV','24.0 C (ATC)'),
(1054,1054,1,'2.32 pH','269.4 mV','23.9 C (ATC)'),
(1055,1055,1,'2.54 pH','257.1 mV','24.6 C (ATC)'),
(1056,1056,1,'2.31 pH','270.3 mV','24.5 C (ATC)'),
(1057,1057,1,'2.36 pH','267.9 mV','24.6 C (ATC)'),
(1058,1058,1,'2.60 pH','253.9 mV','24.6 C (ATC)'),
(1059,1059,1,'2.87 pH','238.4 mV','24.8 C (ATC)'),
(1060,1060,1,'2.52 pH','258.1 mV','24.5 C (ATC)'),
(1061,1061,1,'2.22 pH','275.3 mV','24.4 C (ATC)'),
(1062,1062,1,'2.57 pH','255.3 mV','24.8 C (ATC)'),
(1063,1063,1,'2.86 pH','239.1 mV','25.0 C (ATC)'),
(1064,1064,1,'2.56 pH','256.4 mV','24.8 C (ATC)'),
(1065,1065,1,'2.38 pH','266.5 mV','24.6 C (ATC)'),
(1066,1066,1,'2.82 pH','241.1 mV','24.7 C (ATC)'),
(1067,1067,1,'2.56 pH','255.8 mV','24.5 C (ATC)'),
(1068,1068,1,'2.44 pH','263.0 mV','24.4 C (ATC)'),
(1069,1069,1,'2.52 pH','258.3 mV','24.5 C (ATC)'),
(1070,1070,1,'2.84 pH','239.8 mV','24.8 C (ATC)'),
(1071,1071,1,'2.38 pH','266.8 mV','24.6 C (ATC)'),
(1072,1072,1,'2.56 pH','255.9 mV','24.6 C (ATC)'),
(1073,1073,1,'2.34 pH','268.8 mV','24.5 C (ATC)'),
(1074,1074,1,'2.90 pH','236.1 mV','24.5 C (ATC)'),
(1075,1075,1,'2.63 pH','251.8 mV','24.4 C (ATC)'),
(1076,1076,1,'2.96 pH','232.4 mV','24.2 C (ATC)'),
(1077,1077,1,'2.49 pH','259.9 mV','24.2 C (ATC)'),
(1078,1078,1,'2.32 pH','269.5 mV','24.1 C (ATC)'),
(1079,1079,1,'1.64 pH','305.8 mV','24.4 C (ATC)'),
(1080,1080,1,'4.01 pH','167.7 mV','24.4 C (ATC)'),
(1081,1081,1,'7.01 pH','-7.1 mV','24.4 C (ATC)'),
(1082,1082,1,'10.03 pH','-183.8 mV','24.7 C (ATC)'),
(1083,1083,1,'12.44 pH','-324.7 mV','24.8 C (ATC)'),
(1084,1084,1,'1.69 pH','311.4 mV','25.3 C (ATC)'),
(1085,1085,1,'4.05 pH','174.1 mV','26.0 C (ATC)'),
(1086,1086,1,'7.03 pH','-0.6 mV','25.1 C (ATC)'),
(1087,1087,1,'10.04 pH','-176.6 mV','25.1 C (ATC)'),
(1088,1088,1,'12.50 pH','-321.0 mV','26.0 C (ATC)'),
(1089,1089,1,'5.50 pH','82.6 mV','31.9 C (ATC)'),
(1090,1090,1,'5.51 pH','82.3 mV','31.5 C (ATC)'),
(1091,1091,1,'5.52 pH','81.5 mV','32.0 C (ATC)'),
(1092,1092,1,'5.01 pH','111.1 mV','28.9 C (ATC)'),
(1093,1093,1,'6.02 pH','50.7 mV','25.2 C (ATC)'),
(1094,1094,1,'6.03 pH','49.8 mV','24.9 C (ATC)'),
(1095,1095,1,'5.59 pH','75.5 mV','25.7 C (ATC)'),
(1096,1096,1,'7.17 pH','-9.0 mV','23.9 C (ATC)'),
(1097,1097,1,'7.40 pH','-22.4 mV','23.9 C (ATC)'),
(1098,1098,1,'6.08 pH','47.0 mV','25.8 C (ATC)'),
(1099,1099,1,'5.60 pH','74.8 mV','24.6 C (ATC)'),
(1100,1100,1,'5.69 pH','69.7 mV','24.5 C (ATC)'),
(1101,1101,1,'5.69 pH','69.8 mV','24.5 C (ATC)'),
(1102,1102,1,'5.70 pH','69.1 mV','24.5 C (ATC)'),
(1103,1103,1,'7.29 pH','-16.1 mV','25.6 C (ATC)'),
(1104,1104,1,'7.42 pH','-23.8 mV','26.0 C (ATC)'),
(1105,1105,1,'5.73 pH','67.1 mV','24.3 C (ATC)'),
(1106,1106,1,'5.84 pH','61.0 mV','24.3 C (ATC)'),
(1107,1107,1,'5.71 pH','68.6 mV','24.2 C (ATC)'),
(1108,1108,1,'5.69 pH','69.6 mV','24.3 C (ATC)'),
(1109,1109,1,'7.71 pH','-48.3 mV','25.0 C (ATC)'),
(1110,1110,1,'7.63 pH','-43.7 mV','24.6 C (ATC)'),
(1111,1111,1,'7.37 pH','-20.3 mV','24.7 C (ATC)'),
(1112,1112,1,'7.27 pH','-15.0 mV','25.5 C (ATC)'),
(1113,1113,1,'7.32 pH','-17.6 mV','25.2 C (ATC)'),
(1114,1114,1,'7.34 pH','-26.5 mV','24.9 C (ATC)'),
(1115,1115,1,'1.68 pH','303.5 mV','24.8 C (ATC)'),
(1116,1116,1,'4.03 pH','167.1 mV','25.3 C (ATC)'),
(1117,1117,1,'7.04 pH','-8.5 mV','25.1 C (ATC)'),
(1118,1118,1,'10.06 pH','-184.8 mV','25.1 C (ATC)'),
(1119,1119,1,'12.50 pH','-326.8 mV','24.9 C (ATC)'),
(1120,1120,1,'3.89 pH','165.9 mV','23.8 C (ATC)'),
(1121,1121,1,'6.53 pH','14.6 mV','23.7 C (ATC)'),
(1122,1122,1,'4.16 pH','151.0 mV','25.4 C (ATC)'),
(1123,1123,1,'3.32 pH','209.8 mV','26.8 C (ATC)'),
(1124,1124,1,'7.49 pH','-35.3 mV','28.7 C (ATC)'),
(1125,1125,1,'4.43 pH','134.8 mV','23.2 C (ATC)'),
(1126,1126,1,'5.50 pH','82.8 mV','31.4 C (ATC)'),
(1127,1127,1,'6.02 pH','51.4 mV','28.2 C (ATC)'),
(1128,1128,1,'1.66 pH','310.7 mV','25.4 C (ATC)'),
(1129,1129,1,'4.03 pH','172.9 mV','25.3 C (ATC)'),
(1130,1130,1,'7.01 pH','-1.0 mV','25.5 C (ATC)'),
(1131,1131,1,'10.03 pH','-176.9 mV','25.6 C (ATC)'),
(1132,1132,1,'12.49 pH','-320.7 mV','25.9 C (ATC)'),
(1133,1133,1,'4.98 pH','113.7 mV','29.9 C (ATC)'),
(1134,1134,1,'9.16 pH','-132.1 mV','24.6 C (ATC)'),
(1135,1135,1,'9.14 pH','-130.8 mV','24.7 C (ATC)'),
(1136,1136,1,'6.56 pH','25.4 mV','26.6 C (ATC)'),
(1137,1137,1,'9.21 pH','-134.7 mV','24.7 C (ATC)'),
(1138,1138,1,'6.32 pH','26.5 mV','23.3 C (ATC)'),
(1139,1139,1,'6.34 pH','25.7 mV','23.2 C (ATC)'),
(1140,1140,1,'7.35 pH','-20.9 mV','26.6 C (ATC)'),
(1141,1141,1,'7.39 pH','-22.9 mV','25.2 C (ATC)'),
(1142,1142,1,'5.50 pH','80.8 mV','24.0 C (ATC)'),
(1143,1143,1,'5.59 pH','75.5 mV','24.1 C (ATC)'),
(1144,1144,1,'5.78 pH','64.9 mV','24.1 C (ATC)'),
(1145,1145,1,'5.68 pH','70.4 mV','24.3 C (ATC)'),
(1146,1146,1,'5.69 pH','69.7 mV','24.1 C (ATC)'),
(1147,1147,1,'5.73 pH','67.3 mV','23.9 C (ATC)'),
(1148,1148,1,'5.77 pH','65.3 mV','25.5 C (ATC)'),
(1149,1149,1,'4.34 pH','148.9 mV','24.5 C (ATC)'),
(1150,1150,1,'6.21 pH','39.7 mV','24.5 C (ATC)'),
(1151,1151,1,'6.80 pH','11.1 mV','25.7 C (ATC)'),
(1152,1152,1,'6.71 pH','16.5 mV','26.8 C (ATC)'),
(1153,1153,1,'5.51 pH','82.4 mV','30.8 C (ATC)'),
(1154,1154,1,'6.18 pH','41.7 mV','25.4 C (ATC)'),
(1155,1155,1,'6.16 pH','43.1 mV','25.6 C (ATC)'),
(1156,1156,1,'7.47 pH','-27.0 mV','18.9 C (ATC)'),
(1157,1157,1,'7.40 pH','-23.4 mV','23.3 C (ATC)'),
(1158,1158,1,'6.62 pH','21.7 mV','25.4 C (ATC)'),
(1159,1159,1,'1.71 pH','306.5 mV','25.0 C (ATC)'),
(1160,1160,1,'4.03 pH','170.9 mV','25.6 C (ATC)'),
(1161,1161,1,'7.02 pH','-4.9 mV','24.9 C (ATC)'),
(1162,1162,1,'10.04 pH','-181.6 mV','25.0 C (ATC)'),
(1163,1163,1,'12.50 pH','-325.8 mV','25.1 C (ATC)'),
(1164,1164,1,'1.66 pH','311.3 mV','25.1 C (ATC)'),
(1165,1165,1,'4.01 pH','174.0 mV','25.2 C (ATC)'),
(1166,1166,1,'7.00 pH','-0.0 mV','25.3 C (ATC)'),
(1167,1167,1,'9.99 pH','-174.5 mV','25.5 C (ATC)'),
(1168,1168,1,'12.48 pH','-320.3 mV','25.7 C (ATC)'),
(1169,1169,1,'5.49 pH','86.3 mV','30.1 C (ATC)'),
(1170,1170,1,'6.18 pH','44.3 mV','24.3 C (ATC)'),
(1171,1171,1,'5.72 pH','71.0 mV','23.7 C (ATC)'),
(1172,1172,1,'6.01 pH','54.3 mV','24.8 C (ATC)'),
(1173,1173,1,'9.16 pH','-129.3 mV','23.6 C (ATC)'),
(1174,1174,1,'5.98 pH','55.8 mV','24.8 C (ATC)'),
(1175,1175,1,'6.90 pH','5.5 mV','26.1 C (ATC)'),
(1176,1176,1,'7.41 pH','-24.2 mV','25.7 C (ATC)'),
(1177,1177,1,'5.61 pH','81.2 mV','25.0 C (ATC)'),
(1178,1178,1,'7.13 pH','-7.6 mV','23.8 C (ATC)'),
(1179,1179,1,'5.72 pH','71.1 mV','24.6 C (ATC)'),
(1180,1180,1,'5.73 pH','70.6 mV','24.4 C (ATC)'),
(1181,1181,1,'5.72 pH','71.3 mV','23.6 C (ATC)'),
(1182,1182,1,'5.76 pH','68.9 mV','24.7 C (ATC)'),
(1183,1183,1,'5.71 pH','71.9 mV','25.0 C (ATC)'),
(1184,1184,1,'5.78 pH','67.7 mV','24.2 C (ATC)'),
(1185,1185,1,'2.74 pH','246.2 mV','24.9 C (ATC)'),
(1186,1186,1,'2.42 pH','264.7 mV','24.8 C (ATC)'),
(1187,1187,1,'2.47 pH','261.4 mV','24.8 C (ATC)'),
(1188,1188,1,'2.27 pH','273.4 mV','24.7 C (ATC)'),
(1189,1189,1,'2.47 pH','261.8 mV','25.0 C (ATC)'),
(1190,1190,1,'2.62 pH','252.9 mV','24.9 C (ATC)'),
(1191,1191,1,'2.71 pH','247.1 mV','24.3 C (ATC)'),
(1192,1192,1,'2.51 pH','259.0 mV','24.4 C (ATC)'),
(1193,1193,1,'2.60 pH','254.5 mV','24.9 C (ATC)'),
(1194,1194,1,'2.44 pH','263.2 mV','24.7 C (ATC)'),
(1195,1195,1,'2.56 pH','256.2 mV','24.8 C (ATC)'),
(1196,1196,1,'2.63 pH','252.1 mV','24.4 C (ATC)'),
(1197,1197,1,'2.56 pH','256.1 mV','24.1 C (ATC)'),
(1198,1198,1,'2.63 pH','252.5 mV','24.6 C (ATC)'),
(1199,1199,1,'2.48 pH','260.8 mV','24.5 C (ATC)'),
(1200,1200,1,'2.62 pH','252.8 mV','24.8 C (ATC)'),
(1201,1201,1,'6.65 pH','20.6 mV','26.2 C (ATC)'),
(1202,1202,1,'2.46 pH','262.6 mV','24.8 C (ATC)'),
(1203,1203,1,'2.61 pH','253.8 mV','25.0 C (ATC)'),
(1204,1204,1,'2.10 pH','283.0 mV','24.4 C (ATC)'),
(1205,1205,1,'2.19 pH','277.8 mV','24.6 C (ATC)'),
(1206,1206,1,'2.31 pH','271.1 mV','24.5 C (ATC)'),
(1207,1207,1,'3.96 pH','161.9 mV','24.9 C (ATC)'),
(1208,1208,1,'2.38 pH','266.9 mV','24.4 C (ATC)'),
(1209,1209,1,'2.48 pH','261.1 mV','24.4 C (ATC)'),
(1210,1210,1,'2.58 pH','254.9 mV','24.4 C (ATC)'),
(1211,1211,1,'2.63 pH','252.0 mV','24.4 C (ATC)'),
(1212,1212,1,'2.72 pH','246.9 mV','24.2 C (ATC)'),
(1213,1213,1,'2.74 pH','245.4 mV','24.1 C (ATC)'),
(1214,1214,1,'2.40 pH','265.2 mV','24.2 C (ATC)'),
(1215,1215,1,'2.51 pH','258.5 mV','23.9 C (ATC)'),
(1216,1216,1,'2.61 pH','252.9 mV','23.9 C (ATC)'),
(1217,1217,1,'2.65 pH','250.7 mV','23.8 C (ATC)'),
(1218,1218,1,'2.86 pH','238.6 mV','24.5 C (ATC)'),
(1219,1219,1,'4.14 pH','152.1 mV','26.0 C (ATC)'),
(1220,1220,1,'5.61 pH','80.6 mV','26.2 C (ATC)'),
(1221,1221,1,'2.19 pH','277.2 mV','24.0 C (ATC)'),
(1222,1222,1,'2.31 pH','270.5 mV','24.1 C (ATC)'),
(1223,1223,1,'2.46 pH','261.9 mV','24.6 C (ATC)'),
(1224,1224,1,'2.89 pH','237.0 mV','24.8 C (ATC)'),
(1225,1225,1,'2.52 pH','259.0 mV','24.7 C (ATC)'),
(1226,1226,1,'7.03 pH','-1.8 mV','25.2 C (ATC)'),
(1227,1227,1,'2.61 pH','253.2 mV','24.4 C (ATC)'),
(1228,1228,1,'7.46 pH','-26.9 mV','26.1 C (ATC)'),
(1229,1229,1,'2.79 pH','242.4 mV','24.1 C (ATC)'),
(1230,1230,1,'6.37 pH','36.8 mV','25.6 C (ATC)'),
(1231,1231,1,'4.61 pH','124.4 mV','24.9 C (ATC)'),
(1232,1232,1,'2.92 pH','234.9 mV','24.1 C (ATC)'),
(1233,1233,1,'2.55 pH','256.9 mV','24.3 C (ATC)'),
(1234,1234,1,'2.37 pH','267.0 mV','24.2 C (ATC)'),
(1235,1235,1,'2.69 pH','247.9 mV','23.9 C (ATC)'),
(1236,1236,1,'2.46 pH','261.6 mV','24.0 C (ATC)'),
(1237,1237,1,'2.92 pH','234.8 mV','23.9 C (ATC)'),
(1238,1238,1,'2.44 pH','263.2 mV','24.3 C (ATC)'),
(1239,1239,1,'2.45 pH','262.9 mV','24.7 C (ATC)'),
(1240,1240,1,'2.77 pH','243.9 mV','24.2 C (ATC)'),
(1241,1241,1,'6.41 pH','20.6 mV','26.4 C (ATC)'),
(1242,1242,1,'6.81 pH','10.8 mV','25.6 C (ATC)'),
(1243,1243,1,'7.41 pH','-27.6 mV','24.8 C (ATC)'),
(1244,1244,1,'6.27 pH','28.8 mV','25.8 C (ATC)'),
(1245,1245,1,'3.87 pH','167.7 mV','25.5 C (ATC)'),
(1246,1246,1,'4.13 pH','151.3 mV','23.3 C (ATC)'),
(1247,1247,1,'4.30 pH','141.6 mV','23.6 C (ATC)'),
(1248,1248,1,'6.10 pH','38.7 mV','25.3 C (ATC)'),
(1249,1249,1,'6.05 pH','38.6 mV','24.2 C (ATC)'),
(1250,1250,1,'1.66 pH','304.7 mV','24.6 C (ATC)'),
(1251,1251,1,'4.05 pH','164.5 mV','24.6 C (ATC)'),
(1252,1252,1,'7.00 pH','-8.0 mV','24.6 C (ATC)'),
(1253,1253,1,'6.47 pH','17.4 mV','25.4 C (ATC)'),
(1254,1254,1,'9.98 pH','-182.2 mV','24.8 C (ATC)'),
(1255,1255,1,'12.46 pH','-327.5 mV','24.7 C (ATC)'),
(1256,1256,1,'5.49 pH','82.4 mV','31.9 C (ATC)'),
(1257,1257,1,'5.55 pH','77.5 mV','25.5 C (ATC)'),
(1258,1258,1,'5.54 pH','77.8 mV','25.1 C (ATC)'),
(1259,1259,1,'6.31 pH','26.3 mV','25.5 C (ATC)'),
(1260,1260,1,'4.97 pH','110.8 mV','24.3 C (ATC)'),
(1261,1261,1,'5.49 pH','81.1 mV','26.1 C (ATC)'),
(1262,1262,1,'6.04 pH','48.2 mV','25.0 C (ATC)'),
(1263,1263,1,'5.63 pH','72.2 mV','25.1 C (ATC)'),
(1264,1264,1,'4.29 pH','150.6 mV','24.4 C (ATC)'),
(1265,1265,1,'6.13 pH','43.0 mV','25.0 C (ATC)'),
(1266,1266,1,'4.62 pH','131.6 mV','25.3 C (ATC)'),
(1267,1267,1,'6.06 pH','47.4 mV','25.4 C (ATC)'),
(1268,1268,1,'6.05 pH','47.8 mV','25.3 C (ATC)'),
(1269,1269,1,'5.56 pH','76.9 mV','25.8 C (ATC)'),
(1270,1270,1,'5.72 pH','67.1 mV','25.7 C (ATC)'),
(1271,1271,1,'5.69 pH','69.2 mV','25.6 C (ATC)'),
(1272,1272,1,'5.97 pH','52.4 mV','25.0 C (ATC)'),
(1273,1273,1,'6.10 pH','45.1 mV','25.8 C (ATC)'),
(1274,1274,1,'5.70 pH','68.5 mV','24.6 C (ATC)'),
(1275,1275,1,'5.74 pH','65.8 mV','24.6 C (ATC)'),
(1276,1276,1,'5.72 pH','67.2 mV','24.9 C (ATC)'),
(1277,1277,1,'5.75 pH','65.3 mV','24.5 C (ATC)'),
(1278,1278,1,'5.74 pH','65.8 mV','24.6 C (ATC)'),
(1279,1279,1,'5.78 pH','63.3 mV','24.6 C (ATC)'),
(1280,1280,1,'5.68 pH','69.0 mV','24.4 C (ATC)'),
(1281,1281,1,'5.74 pH','65.5 mV','24.3 C (ATC)'),
(1282,1282,1,'5.75 pH','65.2 mV','24.1 C (ATC)'),
(1283,1283,1,'5.77 pH','63.8 mV','24.0 C (ATC)'),
(1284,1284,1,'1.67 pH','303.7 mV','24.5 C (ATC)'),
(1285,1285,1,'4.03 pH','165.9 mV','24.6 C (ATC)'),
(1286,1286,1,'7.01 pH','-8.9 mV','24.5 C (ATC)'),
(1287,1287,1,'9.96 pH','-181.7 mV','24.7 C (ATC)'),
(1288,1288,1,'12.47 pH','-328.8 mV','24.9 C (ATC)'),
(1289,1289,1,'1.65 pH','309.5 mV','24.8 C (ATC)'),
(1290,1290,1,'4.01 pH','172.1 mV','25.3 C (ATC)'),
(1291,1291,1,'6.98 pH','-1.1 mV','25.3 C (ATC)'),
(1292,1292,1,'10.01 pH','-176.2 mV','23.4 C (ATC)'),
(1293,1293,1,'12.46 pH','-319.4 mV','24.4 C (ATC)'),
(1294,1294,1,'5.50 pH','81.5 mV','31.9 C (ATC)'),
(1295,1295,1,'5.98 pH','51.3 mV','25.0 C (ATC)'),
(1296,1296,1,'5.94 pH','54.0 mV','25.2 C (ATC)'),
(1297,1297,1,'7.32 pH','-20.6 mV','26.7 C (ATC)'),
(1298,1298,1,'7.30 pH','-19.7 mV','24.7 C (ATC)'),
(1299,1299,1,'7.29 pH','-18.8 mV','25.7 C (ATC)'),
(1300,1300,1,'7.66 pH','-47.0 mV','24.8 C (ATC)'),
(1301,1301,1,'5.63 pH','71.9 mV','24.1 C (ATC)'),
(1302,1302,1,'5.60 pH','73.5 mV','23.8 C (ATC)'),
(1303,1303,1,'5.54 pH','76.8 mV','23.9 C (ATC)'),
(1304,1304,1,'5.51 pH','79.2 mV','24.8 C (ATC)'),
(1305,1305,1,'5.43 pH','83.9 mV','24.3 C (ATC)'),
(1306,1306,1,'5.32 pH','90.1 mV','24.8 C (ATC)'),
(1307,1307,1,'5.36 pH','87.7 mV','24.7 C (ATC)'),
(1308,1308,1,'5.83 pH','60.4 mV','25.2 C (ATC)'),
(1309,1309,1,'9.20 pH','-137.1 mV','25.0 C (ATC)'),
(1310,1310,1,'7.38 pH','-24.3 mV','26.6 C (ATC)'),
(1311,1311,1,'7.35 pH','-22.8 mV','26.5 C (ATC)'),
(1312,1312,1,'7.39 pH','-24.7 mV','26.3 C (ATC)'),
(1313,1313,1,'7.22 pH','-14.9 mV','23.2 C (ATC)'),
(1314,1314,1,'7.20 pH','-13.9 mV','24.3 C (ATC)'),
(1315,1315,1,'7.21 pH','-14.5 mV','25.0 C (ATC)'),
(1316,1316,1,'6.03 pH','49.0 mV','26.6 C (ATC)'),
(1317,1317,1,'4.41 pH','134.9 mV','25.3 C (ATC)'),
(1318,1318,1,'5.96 pH','45.9 mV','24.6 C (ATC)'),
(1319,1319,1,'5.99 pH','44.0 mV','24.2 C (ATC)'),
(1320,1320,1,'6.34 pH','24.8 mV','26.2 C (ATC)'),
(1321,1321,1,'6.44 pH','18.6 mV','23.4 C (ATC)'),
(1322,1322,1,'6.25 pH','29.2 mV','24.6 C (ATC)'),
(1323,1323,1,'6.27 pH','27.8 mV','23.5 C (ATC)'),
(1324,1324,1,'1.70 pH','308.3 mV','25.7 C (ATC)'),
(1325,1325,1,'4.05 pH','170.6 mV','25.1 C (ATC)'),
(1326,1326,1,'7.02 pH','-3.2 mV','23.6 C (ATC)'),
(1327,1327,1,'10.01 pH','-177.7 mV','24.3 C (ATC)'),
(1328,1328,1,'12.47 pH','-321.3 mV','24.4 C (ATC)'),
(1329,1329,1,'5.53 pH','83.7 mV','23.3 C (ATC)'),
(1330,1330,1,'5.52 pH','84.4 mV','25.0 C (ATC)'),
(1331,1331,1,'5.52 pH','84.4 mV','25.0 C (ATC)'),
(1332,1332,1,'6.02 pH','55.6 mV','28.1 C (ATC)'),
(1333,1333,1,'5.54 pH','85.7 mV','33.4 C (ATC)'),
(1334,1334,1,'5.53 pH','86.0 mV','33.0 C (ATC)'),
(1335,1335,1,'6.19 pH','45.2 mV','26.0 C (ATC)'),
(1336,1336,1,'5.53 pH','83.7 mV','24.7 C (ATC)'),
(1337,1337,1,'5.54 pH','83.7 mV','24.0 C (ATC)'),
(1338,1338,1,'6.75 pH','12.6 mV','25.7 C (ATC)'),
(1339,1339,1,'6.96 pH','0.0 mV','26.2 C (ATC)'),
(1340,1340,1,'5.91 pH','61.8 mV','24.9 C (ATC)'),
(1341,1341,1,'5.83 pH','66.3 mV','24.8 C (ATC)'),
(1342,1342,1,'5.77 pH','69.5 mV','24.7 C (ATC)'),
(1343,1343,1,'5.74 pH','71.5 mV','24.7 C (ATC)'),
(1344,1344,1,'5.71 pH','73.4 mV','24.9 C (ATC)'),
(1345,1345,1,'5.70 pH','73.9 mV','25.0 C (ATC)'),
(1346,1346,1,'5.69 pH','74.4 mV','25.0 C (ATC)'),
(1347,1347,1,'5.66 pH','75.8 mV','24.9 C (ATC)'),
(1348,1348,1,'1.66 pH','303.4 mV','24.7 C (ATC)'),
(1349,1349,1,'4.03 pH','165.3 mV','24.8 C (ATC)'),
(1350,1350,1,'7.04 pH','-9.7 mV','24.8 C (ATC)'),
(1351,1351,1,'10.05 pH','-185.2 mV','24.8 C (ATC)'),
(1352,1352,1,'12.49 pH','-328.0 mV','25.0 C (ATC)'),
(1353,1353,1,'4.31 pH','140.9 mV','26.1 C (ATC)'),
(1354,1354,1,'5.50 pH','87.9 mV','32.1 C (ATC)'),
(1355,1355,1,'5.02 pH','115.2 mV','28.3 C (ATC)'),
(1356,1356,1,'6.92 pH','2.5 mV','25.6 C (ATC)'),
(1357,1357,1,'6.83 pH','8.0 mV','25.2 C (ATC)'),
(1358,1358,1,'5.87 pH','51.2 mV','25.0 C (ATC)'),
(1359,1359,1,'5.91 pH','48.7 mV','25.5 C (ATC)'),
(1360,1360,1,'6.28 pH','27.2 mV','24.8 C (ATC)'),
(1361,1361,1,'6.37 pH','22.5 mV','25.0 C (ATC)'),
(1362,1362,1,'1.67 pH','303.0 mV','24.5 C (ATC)'),
(1363,1363,1,'4.02 pH','165.2 mV','24.6 C (ATC)'),
(1364,1364,1,'7.02 pH','-10.3 mV','24.7 C (ATC)'),
(1365,1365,1,'10.03 pH','-186.5 mV','24.9 C (ATC)'),
(1366,1366,1,'12.42 pH','-327.2 mV','25.2 C (ATC)'),
(1367,1367,1,'5.49 pH','79.3 mV','25.1 C (ATC)'),
(1368,1368,1,'1.64 pH','311.5 mV','26.0 C (ATC)'),
(1369,1369,1,'4.01 pH','172.8 mV','25.4 C (ATC)'),
(1370,1370,1,'7.01 pH','-1.4 mV','25.6 C (ATC)'),
(1371,1371,1,'5.00 pH','110.1 mV','29.9 C (ATC)'),
(1372,1372,1,'10.03 pH','-177.2 mV','24.8 C (ATC)'),
(1373,1373,1,'12.48 pH','-319.4 mV','24.6 C (ATC)'),
(1374,1374,1,'6.02 pH','48.6 mV','25.2 C (ATC)'),
(1375,1375,1,'6.00 pH','50.6 mV','29.6 C (ATC)'),
(1376,1376,1,'5.61 pH','72.5 mV','24.3 C (ATC)'),
(1377,1377,1,'5.79 pH','62.1 mV','25.5 C (ATC)'),
(1378,1378,1,'5.55 pH','75.7 mV','24.8 C (ATC)'),
(1379,1379,1,'5.48 pH','80.2 mV','25.4 C (ATC)'),
(1380,1380,1,'5.64 pH','70.0 mV','23.2 C (ATC)'),
(1381,1381,1,'5.52 pH','77.5 mV','24.6 C (ATC)'),
(1382,1382,1,'5.33 pH','88.9 mV','25.2 C (ATC)'),
(1383,1383,1,'5.70 pH','67.3 mV','25.5 C (ATC)'),
(1384,1384,1,'8.90 pH','-120.5 mV','25.2 C (ATC)'),
(1385,1385,1,'8.94 pH','-123.0 mV','25.4 C (ATC)'),
(1386,1386,1,'5.56 pH','75.1 mV','24.8 C (ATC)'),
(1387,1387,1,'5.65 pH','69.9 mV','24.5 C (ATC)'),
(1388,1388,1,'5.57 pH','74.5 mV','24.8 C (ATC)'),
(1389,1389,1,'5.53 pH','77.2 mV','25.0 C (ATC)'),
(1390,1390,1,'5.59 pH','73.6 mV','24.2 C (ATC)'),
(1391,1391,1,'5.51 pH','78.2 mV','24.4 C (ATC)'),
(1392,1392,1,'5.56 pH','75.2 mV','24.8 C (ATC)'),
(1393,1393,1,'5.70 pH','67.0 mV','24.4 C (ATC)'),
(1394,1394,1,'5.60 pH','72.8 mV','24.2 C (ATC)'),
(1395,1395,1,'7.05 pH','-3.8 mV','23.4 C (ATC)'),
(1396,1396,1,'7.07 pH','-5.1 mV','23.3 C (ATC)'),
(1397,1397,1,'9.14 pH','-134.7 mV','25.0 C (ATC)'),
(1398,1398,1,'9.10 pH','-132.0 mV','25.5 C (ATC)'),
(1399,1399,1,'9.15 pH','-134.8 mV','25.1 C (ATC)'),
(1400,1400,1,'7.07 pH','-13.3 mV','25.7 C (ATC)'),
(1401,1401,1,'6.75 pH','5.6 mV','26.0 C (ATC)'),
(1402,1402,1,'7.44 pH','-34.5 mV','25.4 C (ATC)'),
(1403,1403,1,'7.34 pH','-28.7 mV','25.3 C (ATC)'),
(1404,1404,1,'8.96 pH','-124.4 mV','25.8 C (ATC)'),
(1405,1405,1,'7.34 pH','-29.1 mV','24.6 C (ATC)'),
(1406,1406,1,'6.98 pH','-0.1 mV','24.7 C (ATC)'),
(1407,1407,1,'7.08 pH','-5.6 mV','25.7 C (ATC)'),
(1408,1408,1,'4.63 pH','123.1 mV','24.6 C (ATC)'),
(1409,1409,1,'5.95 pH','47.6 mV','25.7 C (ATC)'),
(1410,1410,1,'5.97 pH','46.3 mV','24.9 C (ATC)'),
(1411,1411,1,'6.28 pH','28.6 mV','26.7 C (ATC)'),
(1412,1412,1,'1.66 pH','302.8 mV','24.5 C (ATC)'),
(1413,1413,1,'4.02 pH','164.9 mV','24.5 C (ATC)'),
(1414,1414,1,'7.03 pH','-10.9 mV','24.7 C (ATC)'),
(1415,1415,1,'10.04 pH','-186.4 mV','24.7 C (ATC)'),
(1416,1416,1,'12.45 pH','-327.6 mV','24.7 C (ATC)'),
(1417,1417,1,'6.47 pH','17.6 mV','26.6 C (ATC)'),
(1418,1418,1,'5.52 pH','79.4 mV','31.0 C (ATC)'),
(1419,1419,1,'5.53 pH','78.9 mV','30.2 C (ATC)'),
(1420,1420,1,'6.18 pH','34.3 mV','24.6 C (ATC)'),
(1421,1421,1,'6.04 pH','47.4 mV','28.1 C (ATC)'),
(1422,1422,1,'6.28 pH','28.3 mV','23.6 C (ATC)'),
(1423,1423,1,'1.69 pH','309.5 mV','24.9 C (ATC)'),
(1424,1424,1,'4.01 pH','172.4 mV','24.1 C (ATC)'),
(1425,1425,1,'6.98 pH','-2.3 mV','25.3 C (ATC)'),
(1426,1426,1,'9.98 pH','-179.8 mV','25.5 C (ATC)'),
(1427,1427,1,'12.47 pH','-326.7 mV','25.5 C (ATC)'),
(1428,1428,1,'7.32 pH','-22.4 mV','26.8 C (ATC)'),
(1429,1429,1,'6.99 pH','-2.9 mV','24.7 C (ATC)'),
(1430,1430,1,'6.04 pH','47.6 mV','27.6 C (ATC)'),
(1431,1431,1,'7.28 pH','-20.1 mV','25.8 C (ATC)'),
(1432,1432,1,'6.11 pH','43.1 mV','25.6 C (ATC)'),
(1433,1433,1,'9.09 pH','-131.1 mV','24.8 C (ATC)'),
(1434,1434,1,'5.78 pH','62.4 mV','25.5 C (ATC)'),
(1435,1435,1,'6.01 pH','48.8 mV','24.3 C (ATC)'),
(1436,1436,1,'5.67 pH','68.6 mV','24.0 C (ATC)'),
(1437,1437,1,'5.62 pH','71.2 mV','23.8 C (ATC)'),
(1438,1438,1,'5.61 pH','71.9 mV','24.8 C (ATC)'),
(1439,1439,1,'5.62 pH','71.4 mV','24.8 C (ATC)'),
(1440,1440,1,'5.69 pH','67.4 mV','23.7 C (ATC)'),
(1441,1441,1,'5.58 pH','73.7 mV','24.8 C (ATC)'),
(1442,1442,1,'5.64 pH','70.1 mV','24.2 C (ATC)'),
(1443,1443,1,'5.56 pH','75.0 mV','24.7 C (ATC)'),
(1444,1444,1,'7.31 pH','-21.8 mV','24.4 C (ATC)'),
(1445,1445,1,'7.26 pH','-19.0 mV','24.0 C (ATC)'),
(1446,1446,1,'7.03 pH','-5.0 mV','24.9 C (ATC)'),
(1447,1447,1,'6.52 pH','24.8 mV','25.4 C (ATC)'),
(1448,1448,1,'6.57 pH','24.5 mV','25.4 C (ATC)'),
(1449,1449,1,'6.50 pH','25.9 mV','25.4 C (ATC)'),
(1450,1450,1,'1.69 pH','303.0 mV','24.7 C (ATC)'),
(1451,1451,1,'4.03 pH','164.8 mV','24.9 C (ATC)'),
(1452,1452,1,'7.00 pH','-10.5 mV','24.7 C (ATC)'),
(1453,1453,1,'9.98 pH','-186.1 mV','24.6 C (ATC)'),
(1454,1454,1,'12.48 pH','-333.6 mV','24.5 C (ATC)'),
(1455,1455,1,'5.00 pH','110.4 mV','31.6 C (ATC)'),
(1456,1456,1,'5.53 pH','76.3 mV','25.2 C (ATC)'),
(1457,1457,1,'5.53 pH','76.6 mV','26.2 C (ATC)'),
(1458,1458,1,'5.53 pH','76.3 mV','24.9 C (ATC)'),
(1459,1459,1,'1.68 pH','308.2 mV','25.6 C (ATC)'),
(1460,1460,1,'4.04 pH','169.4 mV','24.0 C (ATC)'),
(1461,1461,1,'7.02 pH','-3.2 mV','24.5 C (ATC)'),
(1462,1462,1,'10.04 pH','-179.1 mV','25.4 C (ATC)'),
(1463,1463,1,'12.49 pH','-322.9 mV','25.7 C (ATC)'),
(1464,1464,1,'6.03 pH','46.8 mV','26.4 C (ATC)'),
(1465,1465,1,'5.02 pH','108.4 mV','29.6 C (ATC)'),
(1466,1466,1,'5.00 pH','110.4 mV','32.4 C (ATC)'),
(1467,1467,1,'5.02 pH','106.5 mV','24.9 C (ATC)'),
(1468,1468,1,'6.90 pH','3.7 mV','24.1 C (ATC)'),
(1469,1469,1,'5.50 pH','80.0 mV','31.4 C (ATC)'),
(1470,1470,1,'5.54 pH','77.9 mV','33.1 C (ATC)'),
(1471,1471,1,'5.53 pH','78.4 mV','32.4 C (ATC)'),
(1472,1472,1,'5.86 pH','56.8 mV','26.5 C (ATC)'),
(1473,1473,1,'6.10 pH','42.8 mV','25.1 C (ATC)'),
(1474,1474,1,'5.74 pH','54.4 mV','25.0 C (ATC)'),
(1475,1475,1,'6.31 pH','30.1 mV','26.1 C (ATC)'),
(1476,1476,1,'6.50 pH','19.0 mV','26.1 C (ATC)'),
(1477,1477,1,'6.10 pH','42.7 mV','25.2 C (ATC)'),
(1478,1478,1,'5.31 pH','89.3 mV','25.3 C (ATC)'),
(1479,1479,1,'5.57 pH','73.8 mV','24.6 C (ATC)'),
(1480,1480,1,'5.83 pH','58.7 mV','25.3 C (ATC)'),
(1481,1481,1,'6.27 pH','32.9 mV','25.5 C (ATC)'),
(1482,1482,1,'5.58 pH','73.7 mV','25.6 C (ATC)'),
(1483,1483,1,'5.58 pH','73.0 mV','24.7 C (ATC)'),
(1484,1484,1,'5.59 pH','72.6 mV','24.9 C (ATC)'),
(1485,1485,1,'5.55 pH','75.4 mV','25.0 C (ATC)'),
(1486,1486,1,'5.52 pH','76.7 mV','25.0 C (ATC)'),
(1487,1487,1,'5.51 pH','77.4 mV','25.0 C (ATC)'),
(1488,1488,1,'5.50 pH','77.9 mV','25.0 C (ATC)'),
(1489,1489,1,'5.54 pH','75.9 mV','25.0 C (ATC)'),
(1490,1490,1,'5.53 pH','76.1 mV','25.0 C (ATC)'),
(1491,1491,1,'6.99 pH','-1.4 mV','25.4 C (ATC)'),
(1492,1492,1,'4.52 pH','125.8 mV','24.6 C (ATC)'),
(1493,1493,1,'5.86 pH','47.6 mV','25.1 C (ATC)'),
(1494,1494,1,'5.89 pH','45.7 mV','25.6 C (ATC)'),
(1495,1495,1,'6.24 pH','25.2 mV','25.4 C (ATC)'),
(1496,1496,1,'6.43 pH','13.8 mV','24.2 C (ATC)'),
(1497,1497,1,'1.66 pH','301.3 mV','24.6 C (ATC)'),
(1498,1498,1,'4.03 pH','162.5 mV','25.0 C (ATC)'),
(1499,1499,1,'7.00 pH','-12.5 mV','24.9 C (ATC)'),
(1500,1500,1,'9.96 pH','-186.8 mV','25.1 C (ATC)'),
(1501,1501,1,'12.45 pH','-333.4 mV','25.1 C (ATC)'),
(1502,1502,1,'5.71 pH','63.2 mV','25.4 C (ATC)'),
(1503,1503,1,'5.61 pH','69.1 mV','25.4 C (ATC)'),
(1504,1504,1,'6.29 pH','22.2 mV','24.6 C (ATC)'),
(1505,1505,1,'7.33 pH','-32.0 mV','24.8 C (ATC)'),
(1506,1506,1,'6.26 pH','23.1 mV','24.2 C (ATC)'),
(1507,1507,1,'1.70 pH','308.0 mV','24.9 C (ATC)'),
(1508,1508,1,'4.04 pH','171.5 mV','25.2 C (ATC)'),
(1509,1509,1,'7.04 pH','-2.9 mV','25.2 C (ATC)'),
(1510,1510,1,'10.00 pH','-175.2 mV','25.0 C (ATC)'),
(1511,1511,1,'12.47 pH','-319.0 mV','25.1 C (ATC)'),
(1512,1512,1,'5.51 pH','77.4 mV','31.6 C (ATC)'),
(1513,1513,1,'5.51 pH','77.4 mV','31.8 C (ATC)'),
(1514,1514,1,'7.21 pH','-25.0 mV','24.6 C (ATC)'),
(1515,1515,1,'7.13 pH','-19.9 mV','25.6 C (ATC)'),
(1516,1516,1,'5.90 pH','52.1 mV','23.8 C (ATC)'),
(1517,1517,1,'7.44 pH','-26.4 mV','24.0 C (ATC)'),
(1518,1518,1,'7.38 pH','-23.0 mV','22.8 C (ATC)'),
(1519,1519,1,'5.69 pH','64.5 mV','23.8 C (ATC)'),
(1520,1520,1,'5.61 pH','68.8 mV','23.8 C (ATC)'),
(1521,1521,1,'5.53 pH','73.4 mV','23.7 C (ATC)'),
(1522,1522,1,'5.54 pH','73.1 mV','23.5 C (ATC)'),
(1523,1523,1,'5.57 pH','71.1 mV','23.6 C (ATC)'),
(1524,1524,1,'5.20 pH','93.1 mV','23.6 C (ATC)'),
(1525,1525,1,'5.29 pH','87.5 mV','23.6 C (ATC)'),
(1526,1526,1,'5.34 pH','84.8 mV','23.7 C (ATC)'),
(1527,1527,1,'3.82 pH','177.7 mV','29.1 C (ATC)'),
(1528,1528,1,'7.48 pH','-28.9 mV','25.5 C (ATC)'),
(1529,1529,1,'7.34 pH','-20.6 mV','26.3 C (ATC)'),
(1530,1530,1,'1.66 pH','297.6 mV','24.6 C (ATC)'),
(1531,1531,1,'4.03 pH','159.2 mV','24.6 C (ATC)'),
(1532,1532,1,'7.03 pH','-16.5 mV','24.8 C (ATC)'),
(1533,1533,1,'9.98 pH','-189.5 mV','24.9 C (ATC)'),
(1534,1534,1,'12.47 pH','-335.9 mV','24.9 C (ATC)'),
(1535,1535,1,'5.51 pH','74.0 mV','31.7 C (ATC)'),
(1536,1536,1,'5.03 pH','101.7 mV','28.9 C (ATC)'),
(1537,1537,1,'4.99 pH','105.5 mV','32.2 C (ATC)'),
(1538,1538,1,'6.02 pH','43.3 mV','28.0 C (ATC)'),
(1539,1539,1,'5.02 pH','103.5 mV','31.4 C (ATC)'),
(1540,1540,1,'4.69 pH','117.6 mV','24.1 C (ATC)'),
(1541,1541,1,'9.11 pH','-139.3 mV','26.3 C (ATC)'),
(1542,1542,1,'9.07 pH','-136.6 mV','25.9 C (ATC)'),
(1543,1543,1,'9.04 pH','-134.7 mV','25.8 C (ATC)'),
(1544,1544,1,'5.98 pH','44.9 mV','24.7 C (ATC)'),
(1545,1545,1,'5.97 pH','45.5 mV','24.8 C (ATC)'),
(1546,1546,1,'5.86 pH','52.0 mV','25.1 C (ATC)'),
(1547,1547,1,'7.37 pH','-36.8 mV','24.7 C (ATC)'),
(1548,1548,1,'5.63 pH','65.3 mV','24.8 C (ATC)'),
(1549,1549,1,'5.66 pH','63.3 mV','25.0 C (ATC)'),
(1550,1550,1,'5.61 pH','66.2 mV','24.3 C (ATC)'),
(1551,1551,1,'5.55 pH','69.7 mV','24.2 C (ATC)'),
(1552,1552,1,'5.50 pH','72.8 mV','24.2 C (ATC)'),
(1553,1553,1,'5.51 pH','72.3 mV','24.6 C (ATC)'),
(1554,1554,1,'5.44 pH','76.2 mV','24.5 C (ATC)'),
(1555,1555,1,'5.57 pH','68.8 mV','24.3 C (ATC)'),
(1556,1556,1,'5.50 pH','73.0 mV','24.3 C (ATC)'),
(1557,1557,1,'5.01 pH','104.9 mV','32.4 C (ATC)'),
(1558,1558,1,'5.91 pH','46.1 mV','24.7 C (ATC)'),
(1559,1559,1,'5.92 pH','45.0 mV','24.3 C (ATC)'),
(1560,1560,1,'6.20 pH','29.0 mV','26.3 C (ATC)'),
(1561,1561,1,'6.37 pH','19.2 mV','26.2 C (ATC)'),
(1562,1562,1,'6.28 pH','24.0 mV','24.9 C (ATC)'),
(1563,1563,1,'6.31 pH','22.7 mV','26.4 C (ATC)'),
(1564,1564,1,'1.66 pH','301.8 mV','24.6 C (ATC)'),
(1565,1565,1,'4.02 pH','163.6 mV','24.6 C (ATC)'),
(1566,1566,1,'7.03 pH','-12.5 mV','24.7 C (ATC)'),
(1567,1567,1,'10.04 pH','-188.4 mV','24.7 C (ATC)'),
(1568,1568,1,'12.49 pH','-332.4 mV','24.9 C (ATC)'),
(1569,1569,1,'6.05 pH','44.9 mV','24.3 C (ATC)'),
(1570,1570,1,'4.40 pH','132.8 mV','24.7 C (ATC)'),
(1571,1571,1,'5.90 pH','45.8 mV','26.9 C (ATC)'),
(1572,1572,1,'5.89 pH','44.0 mV','23.1 C (ATC)'),
(1573,1573,1,'5.91 pH','44.2 mV','25.6 C (ATC)'),
(1574,1574,1,'6.22 pH','26.5 mV','23.9 C (ATC)'),
(1575,1575,1,'6.42 pH','14.8 mV','26.1 C (ATC)'),
(1576,1576,1,'6.22 pH','26.2 mV','25.8 C (ATC)'),
(1577,1577,1,'6.23 pH','25.8 mV','23.9 C (ATC)'),
(1578,1578,1,'1.66 pH','302.3 mV','24.9 C (ATC)'),
(1579,1579,1,'4.03 pH','164.1 mV','24.9 C (ATC)'),
(1580,1580,1,'7.04 pH','-11.8 mV','25.0 C (ATC)'),
(1581,1581,1,'10.06 pH','-188.1 mV','24.8 C (ATC)'),
(1582,1582,1,'12.50 pH','-331.4 mV','25.0 C (ATC)'),
(1583,1583,1,'1.69 pH','309.1 mV','25.8 C (ATC)'),
(1584,1584,1,'5.51 pH','79.6 mV','32.8 C (ATC)'),
(1585,1585,1,'4.03 pH','171.0 mV','26.0 C (ATC)'),
(1586,1586,1,'7.04 pH','-6.3 mV','24.2 C (ATC)'),
(1587,1587,1,'9.99 pH','-179.8 mV','25.9 C (ATC)'),
(1588,1588,1,'12.49 pH','-327.0 mV','25.7 C (ATC)'),
(1589,1589,1,'5.99 pH','50.1 mV','27.8 C (ATC)'),
(1590,1590,1,'6.48 pH','20.8 mV','24.6 C (ATC)'),
(1591,1591,1,'5.62 pH','70.8 mV','24.7 C (ATC)'),
(1592,1592,1,'5.60 pH','72.3 mV','24.6 C (ATC)'),
(1593,1593,1,'5.51 pH','80.9 mV','36.4 C (ATC)'),
(1594,1594,1,'5.64 pH','69.9 mV','24.5 C (ATC)'),
(1595,1595,1,'5.69 pH','66.6 mV','24.5 C (ATC)'),
(1596,1596,1,'5.68 pH','67.1 mV','24.7 C (ATC)'),
(1597,1597,1,'5.69 pH','67.0 mV','25.0 C (ATC)'),
(1598,1598,1,'5.63 pH','70.2 mV','24.6 C (ATC)'),
(1599,1599,1,'5.66 pH','68.5 mV','24.3 C (ATC)'),
(1600,1600,1,'5.60 pH','72.0 mV','24.9 C (ATC)'),
(1601,1601,1,'5.50 pH','80.3 mV','32.8 C (ATC)'),
(1602,1602,1,'5.52 pH','79.1 mV','32.3 C (ATC)'),
(1603,1603,1,'6.00 pH','48.7 mV','25.6 C (ATC)'),
(1604,1604,1,'5.97 pH','50.9 mV','25.5 C (ATC)'),
(1605,1605,1,'5.94 pH','52.3 mV','25.6 C (ATC)'),
(1606,1606,1,'5.99 pH','49.3 mV','25.5 C (ATC)'),
(1607,1607,1,'6.95 pH','-5.3 mV','26.3 C (ATC)'),
(1608,1608,1,'6.04 pH','46.3 mV','25.6 C (ATC)'),
(1609,1609,1,'5.78 pH','61.6 mV','25.2 C (ATC)'),
(1610,1610,1,'5.64 pH','69.9 mV','25.2 C (ATC)'),
(1611,1611,1,'5.53 pH','76.4 mV','25.1 C (ATC)'),
(1612,1612,1,'7.36 pH','-24.7 mV','24.7 C (ATC)'),
(1613,1613,1,'5.87 pH','56.2 mV','24.4 C (ATC)'),
(1614,1614,1,'5.94 pH','52.1 mV','24.9 C (ATC)'),
(1615,1615,1,'6.98 pH','-2.7 mV','24.5 C (ATC)'),
(1616,1616,1,'7.41 pH','-27.4 mV','23.2 C (ATC)'),
(1617,1617,1,'1.65 pH','302.6 mV','25.0 C (ATC)'),
(1618,1618,1,'4.01 pH','164.3 mV','25.0 C (ATC)'),
(1619,1619,1,'7.03 pH','-12.5 mV','25.3 C (ATC)'),
(1620,1620,1,'10.03 pH','-187.8 mV','24.9 C (ATC)'),
(1621,1621,1,'12.47 pH','-330.4 mV','24.8 C (ATC)'),
(1622,1622,1,'3.50 pH','198.7 mV','30.9 C (ATC)'),
(1623,1623,1,'4.56 pH','122.9 mV','23.9 C (ATC)'),
(1624,1624,1,'7.52 pH','-41.7 mV','32.7 C (ATC)'),
(1625,1625,1,'7.03 pH','-12.0 mV','31.5 C (ATC)'),
(1626,1626,1,'6.05 pH','45.8 mV','28.4 C (ATC)'),
(1627,1627,1,'6.14 pH','39.7 mV','25.7 C (ATC)'),
(1628,1628,1,'5.93 pH','52.4 mV','25.7 C (ATC)'),
(1629,1629,1,'5.93 pH','52.3 mV','25.6 C (ATC)'),
(1630,1630,1,'5.99 pH','49.0 mV','25.5 C (ATC)'),
(1631,1631,1,'5.86 pH','46.8 mV','24.9 C (ATC)'),
(1632,1632,1,'6.07 pH','43.8 mV','25.2 C (ATC)'),
(1633,1633,1,'6.08 pH','43.5 mV','24.9 C (ATC)'),
(1634,1634,1,'6.24 pH','33.7 mV','25.1 C (ATC)'),
(1635,1635,1,'6.12 pH','41.1 mV','25.0 C (ATC)'),
(1636,1636,1,'5.88 pH','45.8 mV','24.4 C (ATC)'),
(1637,1637,1,'1.67 pH','309.3 mV','26.6 C (ATC)'),
(1638,1638,1,'4.04 pH','170.6 mV','26.6 C (ATC)'),
(1639,1639,1,'5.50 pH','80.7 mV','36.9 C (ATC)'),
(1640,1640,1,'7.04 pH','-5.6 mV','24.0 C (ATC)'),
(1641,1641,1,'5.65 pH','68.1 mV','24.7 C (ATC)'),
(1642,1642,1,'10.02 pH','-179.6 mV','25.4 C (ATC)'),
(1643,1643,1,'12.48 pH','-325.5 mV','26.9 C (ATC)'),
(1644,1644,1,'5.65 pH','68.6 mV','24.6 C (ATC)'),
(1645,1645,1,'5.53 pH','75.5 mV','24.8 C (ATC)'),
(1646,1646,1,'5.49 pH','77.6 mV','24.8 C (ATC)'),
(1647,1647,1,'6.19 pH','27.5 mV','24.9 C (ATC)'),
(1648,1648,1,'5.62 pH','70.2 mV','24.8 C (ATC)'),
(1649,1649,1,'5.52 pH','75.9 mV','24.2 C (ATC)'),
(1650,1650,1,'5.57 pH','73.3 mV','24.6 C (ATC)'),
(1651,1651,1,'5.58 pH','72.4 mV','24.2 C (ATC)'),
(1652,1652,1,'5.52 pH','76.1 mV','24.4 C (ATC)'),
(1653,1653,1,'5.50 pH','77.2 mV','24.4 C (ATC)'),
(1654,1654,1,'6.42 pH','13.4 mV','24.6 C (ATC)'),
(1655,1655,1,'6.00 pH','48.3 mV','25.9 C (ATC)'),
(1656,1656,1,'5.95 pH','50.9 mV','25.5 C (ATC)'),
(1657,1657,1,'6.25 pH','33.4 mV','25.1 C (ATC)'),
(1658,1658,1,'7.38 pH','-25.5 mV','26.3 C (ATC)'),
(1659,1659,1,'6.24 pH','24.1 mV','24.4 C (ATC)'),
(1660,1660,1,'6.22 pH','25.4 mV','23.6 C (ATC)'),
(1661,1661,1,'4.45 pH','149.9 mV','25.4 C (ATC)'),
(1662,1662,1,'7.37 pH','-25.0 mV','24.3 C (ATC)'),
(1663,1663,1,'6.48 pH','34.7 mV','26.4 C (ATC)'),
(1664,1664,1,'6.33 pH','43.0 mV','26.5 C (ATC)'),
(1665,1665,1,'6.35 pH','41.5 mV','24.8 C (ATC)'),
(1666,1666,1,'1.68 pH','302.7 mV','24.8 C (ATC)'),
(1667,1667,1,'4.02 pH','164.7 mV','24.9 C (ATC)'),
(1668,1668,1,'7.01 pH','-11.2 mV','24.8 C (ATC)'),
(1669,1669,1,'9.99 pH','-187.1 mV','24.9 C (ATC)'),
(1670,1670,1,'12.48 pH','-334.4 mV','24.9 C (ATC)'),
(1671,1671,1,'4.99 pH','110.2 mV','32.0 C (ATC)'),
(1672,1672,1,'1.64 pH','303.9 mV','26.4 C (ATC)'),
(1673,1673,1,'4.00 pH','166.6 mV','25.8 C (ATC)'),
(1674,1674,1,'6.97 pH','-5.8 mV','24.4 C (ATC)'),
(1675,1675,1,'10.02 pH','-182.5 mV','24.7 C (ATC)'),
(1676,1676,1,'12.47 pH','-325.7 mV','26.1 C (ATC)'),
(1677,1677,1,'6.07 pH','43.8 mV','24.7 C (ATC)'),
(1678,1678,1,'5.50 pH','79.3 mV','31.1 C (ATC)'),
(1679,1679,1,'5.59 pH','71.9 mV','24.8 C (ATC)'),
(1680,1680,1,'4.44 pH','139.8 mV','24.4 C (ATC)'),
(1681,1681,1,'6.18 pH','37.3 mV','24.0 C (ATC)'),
(1682,1682,1,'5.01 pH','109.1 mV','31.9 C (ATC)'),
(1683,1683,1,'5.54 pH','77.5 mV','33.1 C (ATC)'),
(1684,1684,1,'5.51 pH','79.3 mV','32.4 C (ATC)'),
(1685,1685,1,'5.52 pH','78.4 mV','31.5 C (ATC)'),
(1686,1686,1,'5.00 pH','106.9 mV','25.0 C (ATC)'),
(1687,1687,1,'5.75 pH','62.6 mV','23.0 C (ATC)'),
(1688,1688,1,'5.59 pH','72.4 mV','25.1 C (ATC)'),
(1689,1689,1,'5.57 pH','73.4 mV','25.2 C (ATC)'),
(1690,1690,1,'5.58 pH','73.0 mV','25.1 C (ATC)'),
(1691,1691,1,'9.21 pH','-141.6 mV','25.6 C (ATC)'),
(1692,1692,1,'9.19 pH','-140.1 mV','25.1 C (ATC)'),
(1693,1693,1,'2.68 pH','242.7 mV','23.5 C (ATC)'),
(1694,1694,1,'2.74 pH','239.3 mV','23.6 C (ATC)'),
(1695,1695,1,'5.49 pH','78.3 mV','24.7 C (ATC)'),
(1696,1696,1,'5.57 pH','73.4 mV','24.7 C (ATC)'),
(1697,1697,1,'5.59 pH','72.1 mV','24.6 C (ATC)'),
(1698,1698,1,'5.67 pH','67.6 mV','24.8 C (ATC)'),
(1699,1699,1,'5.69 pH','66.2 mV','23.6 C (ATC)'),
(1700,1700,1,'5.71 pH','65.0 mV','23.7 C (ATC)'),
(1701,1701,1,'5.67 pH','67.4 mV','23.6 C (ATC)'),
(1702,1702,1,'5.64 pH','69.1 mV','23.6 C (ATC)'),
(1703,1703,1,'5.66 pH','68.0 mV','24.4 C (ATC)'),
(1704,1704,1,'5.61 pH','71.0 mV','24.2 C (ATC)'),
(1705,1705,1,'5.60 pH','71.6 mV','24.2 C (ATC)'),
(1706,1706,1,'5.56 pH','73.6 mV','24.1 C (ATC)'),
(1707,1707,1,'5.69 pH','66.1 mV','23.8 C (ATC)'),
(1708,1708,1,'6.03 pH','47.9 mV','23.6 C (ATC)'),
(1709,1709,1,'5.77 pH','63.6 mV','24.9 C (ATC)'),
(1710,1710,1,'7.42 pH','-32.3 mV','26.7 C (ATC)'),
(1711,1711,1,'5.86 pH','56.2 mV','25.0 C (ATC)'),
(1712,1712,1,'4.57 pH','122.7 mV','24.8 C (ATC)'),
(1713,1713,1,'7.23 pH','-20.7 mV','24.2 C (ATC)'),
(1714,1714,1,'5.76 pH','63.5 mV','23.3 C (ATC)'),
(1715,1715,1,'6.40 pH','27.3 mV','25.7 C (ATC)'),
(1716,1716,1,'5.85 pH','47.5 mV','24.5 C (ATC)'),
(1717,1717,1,'6.21 pH','26.8 mV','24.6 C (ATC)'),
(1718,1718,1,'6.49 pH','10.0 mV','24.1 C (ATC)'),
(1719,1719,1,'6.25 pH','24.5 mV','26.0 C (ATC)'),
(1720,1720,1,'6.22 pH','26.5 mV','25.0 C (ATC)'),
(1721,1721,1,'1.68 pH','302.0 mV','24.4 C (ATC)'),
(1722,1722,1,'4.02 pH','164.1 mV','24.4 C (ATC)'),
(1723,1723,1,'7.01 pH','-12.2 mV','24.5 C (ATC)'),
(1724,1724,1,'9.99 pH','-187.5 mV','24.7 C (ATC)'),
(1725,1725,1,'12.50 pH','-335.6 mV','24.8 C (ATC)'),
(1726,1726,1,'5.52 pH','77.1 mV','29.4 C (ATC)'),
(1727,1727,1,'5.52 pH','75.1 mV','23.0 C (ATC)'),
(1728,1728,1,'5.54 pH','74.5 mV','23.7 C (ATC)'),
(1729,1729,1,'5.54 pH','74.8 mV','26.0 C (ATC)'),
(1730,1730,1,'5.54 pH','74.7 mV','25.6 C (ATC)'),
(1731,1731,1,'5.56 pH','73.8 mV','25.1 C (ATC)'),
(1732,1732,1,'5.50 pH','77.1 mV','25.1 C (ATC)'),
(1733,1733,1,'6.02 pH','47.3 mV','29.7 C (ATC)'),
(1734,1734,1,'4.40 pH','153.8 mV','25.9 C (ATC)'),
(1735,1735,1,'6.18 pH','36.8 mV','25.3 C (ATC)'),
(1736,1736,1,'1.66 pH','300.5 mV','26.7 C (ATC)'),
(1737,1737,1,'4.03 pH','163.9 mV','26.6 C (ATC)'),
(1738,1738,1,'7.00 pH','-7.6 mV','26.6 C (ATC)'),
(1739,1739,1,'10.02 pH','-181.4 mV','25.7 C (ATC)'),
(1740,1740,1,'12.47 pH','-321.6 mV','25.4 C (ATC)'),
(1741,1741,1,'5.74 pH','63.2 mV','26.6 C (ATC)'),
(1742,1742,1,'5.73 pH','63.4 mV','24.5 C (ATC)'),
(1743,1743,1,'5.67 pH','67.1 mV','24.6 C (ATC)'),
(1744,1744,1,'5.62 pH','70.0 mV','24.6 C (ATC)'),
(1745,1745,1,'5.60 pH','71.4 mV','24.9 C (ATC)'),
(1746,1746,1,'5.53 pH','75.3 mV','24.9 C (ATC)'),
(1747,1747,1,'5.64 pH','69.0 mV','25.0 C (ATC)'),
(1748,1748,1,'5.70 pH','65.1 mV','24.9 C (ATC)'),
(1749,1749,1,'5.61 pH','70.3 mV','24.2 C (ATC)'),
(1750,1750,1,'4.39 pH','142.5 mV','24.6 C (ATC)'),
(1751,1751,1,'7.22 pH','-20.2 mV','25.5 C (ATC)'),
(1752,1752,1,'6.32 pH','45.2 mV','23.6 C (ATC)'),
(1753,1753,1,'6.92 pH','-3.0 mV','25.6 C (ATC)'),
(1754,1754,1,'6.23 pH','33.8 mV','24.5 C (ATC)'),
(1755,1755,1,'6.25 pH','49.2 mV','24.6 C (ATC)'),
(1756,1756,1,'6.28 pH','47.1 mV','20.3 C (ATC)'),
(1757,1757,1,'6.92 pH','-2.7 mV','25.3 C (ATC)'),
(1758,1758,1,'6.29 pH','46.7 mV','25.1 C (ATC)'),
(1759,1759,1,'1.69 pH','302.3 mV','24.8 C (ATC)'),
(1760,1760,1,'4.03 pH','164.4 mV','24.8 C (ATC)'),
(1761,1761,1,'7.01 pH','-11.4 mV','25.0 C (ATC)'),
(1762,1762,1,'9.96 pH','-185.6 mV','24.7 C (ATC)'),
(1763,1763,1,'12.48 pH','-334.1 mV','24.8 C (ATC)'),
(1764,1764,1,'4.47 pH','127.7 mV','23.8 C (ATC)'),
(1765,1765,1,'5.52 pH','78.4 mV','32.4 C (ATC)'),
(1766,1766,1,'1.65 pH','300.7 mV','25.9 C (ATC)'),
(1767,1767,1,'4.03 pH','162.3 mV','25.8 C (ATC)'),
(1768,1768,1,'5.53 pH','78.1 mV','32.4 C (ATC)'),
(1769,1769,1,'6.99 pH','-9.1 mV','25.7 C (ATC)'),
(1770,1770,1,'9.98 pH','-182.5 mV','25.2 C (ATC)'),
(1771,1771,1,'5.52 pH','78.7 mV','32.2 C (ATC)'),
(1772,1772,1,'12.43 pH','-321.8 mV','23.2 C (ATC)'),
(1773,1773,1,'5.50 pH','79.8 mV','32.9 C (ATC)'),
(1774,1774,1,'5.01 pH','109.1 mV','32.0 C (ATC)'),
(1775,1775,1,'5.67 pH','67.7 mV','25.3 C (ATC)'),
(1776,1776,1,'5.91 pH','53.1 mV','25.3 C (ATC)'),
(1777,1777,1,'5.89 pH','54.7 mV','25.6 C (ATC)'),
(1778,1778,1,'5.88 pH','55.3 mV','25.5 C (ATC)'),
(1779,1779,1,'5.83 pH','48.1 mV','25.0 C (ATC)'),
(1780,1780,1,'7.36 pH','-30.7 mV','25.4 C (ATC)'),
(1781,1781,1,'7.37 pH','-31.3 mV','25.6 C (ATC)'),
(1782,1782,1,'7.36 pH','-30.6 mV','25.9 C (ATC)'),
(1783,1783,1,'5.86 pH','46.6 mV','25.7 C (ATC)'),
(1784,1784,1,'6.20 pH','26.2 mV','25.5 C (ATC)'),
(1785,1785,1,'9.10 pH','-134.7 mV','25.0 C (ATC)'),
(1786,1786,1,'9.08 pH','-133.9 mV','25.4 C (ATC)'),
(1787,1787,1,'6.48 pH','9.9 mV','23.4 C (ATC)'),
(1788,1788,1,'5.52 pH','76.5 mV','25.3 C (ATC)'),
(1789,1789,1,'5.62 pH','70.3 mV','25.6 C (ATC)'),
(1790,1790,1,'5.64 pH','69.1 mV','25.5 C (ATC)'),
(1791,1791,1,'5.79 pH','60.6 mV','25.7 C (ATC)'),
(1792,1792,1,'8.05 pH','-72.7 mV','24.0 C (ATC)'),
(1793,1793,1,'6.13 pH','40.0 mV','25.0 C (ATC)'),
(1794,1794,1,'6.09 pH','42.9 mV','24.8 C (ATC)'),
(1795,1795,1,'5.60 pH','71.6 mV','24.8 C (ATC)'),
(1796,1796,1,'5.50 pH','77.1 mV','24.3 C (ATC)'),
(1797,1797,1,'6.27 pH','21.7 mV','25.7 C (ATC)'),
(1798,1798,1,'5.58 pH','72.5 mV','24.8 C (ATC)'),
(1799,1799,1,'5.54 pH','75.2 mV','24.8 C (ATC)'),
(1800,1800,1,'5.56 pH','74.1 mV','24.9 C (ATC)'),
(1801,1801,1,'5.52 pH','76.2 mV','24.6 C (ATC)'),
(1802,1802,1,'6.23 pH','24.3 mV','23.4 C (ATC)'),
(1803,1803,1,'7.31 pH','-27.7 mV','25.6 C (ATC)'),
(1804,1804,1,'7.25 pH','-24.1 mV','25.9 C (ATC)'),
(1805,1805,1,'7.26 pH','-24.7 mV','25.9 C (ATC)'),
(1806,1806,1,'5.87 pH','55.5 mV','23.5 C (ATC)'),
(1807,1807,1,'7.24 pH','-23.7 mV','26.0 C (ATC)'),
(1808,1808,1,'4.39 pH','151.9 mV','23.6 C (ATC)'),
(1809,1809,1,'6.39 pH','39.3 mV','25.3 C (ATC)'),
(1810,1810,1,'6.35 pH','41.5 mV','24.5 C (ATC)'),
(1811,1811,1,'1.69 pH','302.2 mV','24.7 C (ATC)'),
(1812,1812,1,'4.03 pH','164.1 mV','24.7 C (ATC)'),
(1813,1813,1,'7.00 pH','-11.3 mV','24.8 C (ATC)'),
(1814,1814,1,'9.99 pH','-187.8 mV','25.0 C (ATC)'),
(1815,1815,1,'12.50 pH','-335.7 mV','25.1 C (ATC)'),
(1816,1816,1,'6.36 pH','41.3 mV','25.5 C (ATC)'),
(1817,1817,1,'6.21 pH','35.3 mV','24.0 C (ATC)'),
(1818,1818,1,'5.30 pH','89.5 mV','25.6 C (ATC)'),
(1819,1819,1,'5.95 pH','51.1 mV','25.4 C (ATC)'),
(1820,1820,1,'5.02 pH','108.5 mV','32.9 C (ATC)'),
(1821,1821,1,'1.66 pH','295.9 mV','23.2 C (ATC)'),
(1822,1822,1,'4.03 pH','162.6 mV','26.4 C (ATC)'),
(1823,1823,1,'7.00 pH','-9.0 mV','26.4 C (ATC)'),
(1824,1824,1,'9.99 pH','-182.5 mV','26.9 C (ATC)'),
(1825,1825,1,'5.02 pH','107.0 mV','28.8 C (ATC)'),
(1826,1826,1,'5.48 pH','80.5 mV','32.0 C (ATC)'),
(1827,1827,1,'12.48 pH','-322.6 mV','23.3 C (ATC)'),
(1828,1828,1,'5.00 pH','108.5 mV','28.4 C (ATC)'),
(1829,1829,1,'4.72 pH','122.9 mV','23.9 C (ATC)'),
(1830,1830,1,'6.14 pH','39.4 mV','23.9 C (ATC)'),
(1831,1831,1,'5.95 pH','50.7 mV','24.7 C (ATC)'),
(1832,1832,1,'9.18 pH','-139.8 mV','25.4 C (ATC)'),
(1833,1833,1,'9.19 pH','-140.8 mV','25.3 C (ATC)'),
(1834,1834,1,'5.72 pH','64.3 mV','25.5 C (ATC)'),
(1835,1835,1,'6.28 pH','31.5 mV','25.0 C (ATC)'),
(1836,1836,1,'5.57 pH','73.1 mV','24.2 C (ATC)'),
(1837,1837,1,'7.38 pH','-30.8 mV','25.0 C (ATC)'),
(1838,1838,1,'5.60 pH','71.5 mV','24.6 C (ATC)'),
(1839,1839,1,'7.02 pH','-10.4 mV','26.4 C (ATC)'),
(1840,1840,1,'5.54 pH','74.9 mV','24.0 C (ATC)'),
(1841,1841,1,'5.53 pH','75.6 mV','24.0 C (ATC)'),
(1842,1842,1,'5.50 pH','77.5 mV','24.5 C (ATC)'),
(1843,1843,1,'5.57 pH','72.8 mV','24.2 C (ATC)'),
(1844,1844,1,'5.54 pH','74.8 mV','24.0 C (ATC)'),
(1845,1845,1,'7.32 pH','-28.1 mV','25.2 C (ATC)'),
(1846,1846,1,'7.09 pH','-14.7 mV','25.7 C (ATC)'),
(1847,1847,1,'5.34 pH','86.8 mV','25.7 C (ATC)'),
(1848,1848,1,'5.70 pH','65.9 mV','25.6 C (ATC)'),
(1849,1849,1,'5.32 pH','88.3 mV','25.6 C (ATC)'),
(1850,1850,1,'5.72 pH','64.3 mV','25.4 C (ATC)'),
(1851,1851,1,'5.95 pH','50.4 mV','24.4 C (ATC)'),
(1852,1852,1,'5.96 pH','50.5 mV','25.3 C (ATC)'),
(1853,1853,1,'5.94 pH','51.3 mV','25.2 C (ATC)'),
(1854,1854,1,'2.37 pH','261.7 mV','24.5 C (ATC)'),
(1855,1855,1,'2.38 pH','261.2 mV','24.5 C (ATC)'),
(1856,1856,1,'2.43 pH','258.6 mV','24.6 C (ATC)'),
(1857,1857,1,'2.48 pH','255.0 mV','24.2 C (ATC)'),
(1858,1858,1,'2.60 pH','247.8 mV','24.2 C (ATC)'),
(1859,1859,1,'2.68 pH','243.7 mV','24.7 C (ATC)'),
(1860,1860,1,'2.74 pH','240.3 mV','24.8 C (ATC)'),
(1861,1861,1,'2.30 pH','266.2 mV','24.9 C (ATC)'),
(1862,1862,1,'2.40 pH','260.0 mV','24.4 C (ATC)'),
(1863,1863,1,'2.48 pH','255.3 mV','24.3 C (ATC)'),
(1864,1864,1,'2.58 pH','249.0 mV','24.5 C (ATC)'),
(1865,1865,1,'2.37 pH','261.6 mV','24.4 C (ATC)'),
(1866,1866,1,'2.42 pH','258.6 mV','24.5 C (ATC)'),
(1867,1867,1,'2.51 pH','253.3 mV','24.6 C (ATC)'),
(1868,1868,1,'2.57 pH','249.9 mV','24.5 C (ATC)'),
(1869,1869,1,'2.61 pH','247.7 mV','24.7 C (ATC)'),
(1870,1870,1,'2.61 pH','247.3 mV','24.4 C (ATC)'),
(1871,1871,1,'2.56 pH','251.0 mV','24.8 C (ATC)'),
(1872,1872,1,'2.21 pH','271.1 mV','24.5 C (ATC)'),
(1873,1873,1,'2.37 pH','262.0 mV','24.7 C (ATC)'),
(1874,1874,1,'2.45 pH','257.3 mV','24.7 C (ATC)'),
(1875,1875,1,'2.53 pH','252.1 mV','24.5 C (ATC)'),
(1876,1876,1,'2.57 pH','249.4 mV','23.9 C (ATC)'),
(1877,1877,1,'2.66 pH','244.1 mV','24.0 C (ATC)'),
(1878,1878,1,'2.77 pH','238.2 mV','24.3 C (ATC)'),
(1879,1879,1,'2.39 pH','260.5 mV','24.2 C (ATC)'),
(1880,1880,1,'2.54 pH','251.4 mV','24.4 C (ATC)'),
(1881,1881,1,'2.78 pH','237.0 mV','24.2 C (ATC)'),
(1882,1882,1,'2.55 pH','250.6 mV','24.0 C (ATC)'),
(1883,1883,1,'2.62 pH','246.6 mV','24.2 C (ATC)'),
(1884,1884,1,'2.69 pH','242.7 mV','24.1 C (ATC)'),
(1885,1885,1,'2.27 pH','267.1 mV','24.0 C (ATC)'),
(1886,1886,1,'7.03 pH','-10.8 mV','25.8 C (ATC)'),
(1887,1887,1,'7.23 pH','-22.6 mV','25.2 C (ATC)'),
(1888,1888,1,'7.31 pH','-27.1 mV','24.3 C (ATC)'),
(1889,1889,1,'1.67 pH','302.6 mV','24.3 C (ATC)'),
(1890,1890,1,'4.02 pH','164.6 mV','24.4 C (ATC)'),
(1891,1891,1,'7.02 pH','-11.6 mV','24.4 C (ATC)'),
(1892,1892,1,'10.02 pH','-187.4 mV','24.1 C (ATC)'),
(1893,1893,1,'12.50 pH','-334.2 mV','25.1 C (ATC)'),
(1894,1894,1,'5.50 pH','80.2 mV','31.7 C (ATC)'),
(1895,1895,1,'5.51 pH','79.4 mV','32.2 C (ATC)'),
(1896,1896,1,'5.64 pH','69.7 mV','25.7 C (ATC)'),
(1897,1897,1,'5.78 pH','62.0 mV','26.0 C (ATC)'),
(1898,1898,1,'5.82 pH','59.3 mV','25.8 C (ATC)'),
(1899,1899,1,'5.83 pH','58.6 mV','25.7 C (ATC)'),
(1900,1900,1,'5.85 pH','57.7 mV','25.6 C (ATC)'),
(1901,1901,1,'4.51 pH','144.9 mV','24.9 C (ATC)'),
(1902,1902,1,'6.48 pH','34.2 mV','24.2 C (ATC)'),
(1903,1903,1,'6.34 pH','41.9 mV','24.3 C (ATC)'),
(1904,1904,1,'6.35 pH','41.3 mV','24.2 C (ATC)'),
(1905,1905,1,'1.67 pH','302.6 mV','24.6 C (ATC)'),
(1906,1906,1,'4.02 pH','164.7 mV','24.5 C (ATC)'),
(1907,1907,1,'7.03 pH','-11.4 mV','24.6 C (ATC)'),
(1908,1908,1,'10.03 pH','-187.4 mV','24.5 C (ATC)'),
(1909,1909,1,'12.50 pH','-332.5 mV','24.6 C (ATC)'),
(1910,1910,1,'4.51 pH','126.1 mV','24.0 C (ATC)'),
(1911,1911,1,'5.04 pH','107.2 mV','30.6 C (ATC)'),
(1912,1912,1,'5.51 pH','77.2 mV','24.1 C (ATC)'),
(1913,1913,1,'5.50 pH','78.2 mV','25.5 C (ATC)'),
(1914,1914,1,'1.68 pH','301.4 mV','25.0 C (ATC)'),
(1915,1915,1,'4.05 pH','163.7 mV','25.7 C (ATC)'),
(1916,1916,1,'7.00 pH','-8.1 mV','24.0 C (ATC)'),
(1917,1917,1,'10.02 pH','-183.9 mV','24.4 C (ATC)'),
(1918,1918,1,'12.43 pH','-324.1 mV','24.5 C (ATC)'),
(1919,1919,1,'5.01 pH','108.7 mV','29.6 C (ATC)'),
(1920,1920,1,'6.03 pH','47.3 mV','27.6 C (ATC)'),
(1921,1921,1,'4.99 pH','109.9 mV','29.5 C (ATC)'),
(1922,1922,1,'5.51 pH','80.5 mV','34.6 C (ATC)'),
(1923,1923,1,'5.83 pH','48.1 mV','24.9 C (ATC)'),
(1924,1924,1,'8.97 pH','-126.0 mV','25.8 C (ATC)'),
(1925,1925,1,'8.96 pH','-125.1 mV','25.1 C (ATC)'),
(1926,1926,1,'8.97 pH','-126.0 mV','25.2 C (ATC)'),
(1927,1927,1,'8.95 pH','-124.6 mV','25.3 C (ATC)'),
(1928,1928,1,'8.98 pH','-126.2 mV','25.2 C (ATC)'),
(1929,1929,1,'5.90 pH','43.5 mV','23.9 C (ATC)'),
(1930,1930,1,'5.29 pH','89.7 mV','23.4 C (ATC)'),
(1931,1931,1,'5.69 pH','66.4 mV','23.1 C (ATC)'),
(1932,1932,1,'5.54 pH','75.5 mV','23.1 C (ATC)'),
(1933,1933,1,'5.50 pH','77.4 mV','23.2 C (ATC)'),
(1934,1934,1,'5.63 pH','70.2 mV','23.8 C (ATC)'),
(1935,1935,1,'5.54 pH','75.2 mV','23.3 C (ATC)'),
(1936,1936,1,'5.52 pH','76.7 mV','23.2 C (ATC)'),
(1937,1937,1,'5.56 pH','74.2 mV','23.2 C (ATC)'),
(1938,1938,1,'6.16 pH','28.3 mV','25.0 C (ATC)'),
(1939,1939,1,'7.02 pH','-13.5 mV','25.2 C (ATC)'),
(1940,1940,1,'6.46 pH','11.1 mV','26.7 C (ATC)'),
(1941,1941,1,'5.01 pH','109.3 mV','30.2 C (ATC)'),
(1942,1942,1,'5.01 pH','107.4 mV','25.3 C (ATC)'),
(1943,1943,1,'6.25 pH','23.6 mV','23.9 C (ATC)'),
(1944,1944,1,'6.28 pH','21.4 mV','23.2 C (ATC)'),
(1945,1945,1,'7.07 pH','-12.2 mV','25.4 C (ATC)'),
(1946,1946,1,'5.89 pH','55.4 mV','25.4 C (ATC)'),
(1947,1947,1,'4.44 pH','148.4 mV','24.3 C (ATC)'),
(1948,1948,1,'6.40 pH','38.5 mV','25.2 C (ATC)'),
(1949,1949,1,'1.67 pH','301.5 mV','24.4 C (ATC)'),
(1950,1950,1,'4.02 pH','163.8 mV','24.4 C (ATC)'),
(1951,1951,1,'7.01 pH','-11.5 mV','24.5 C (ATC)'),
(1952,1952,1,'10.01 pH','-187.6 mV','24.7 C (ATC)'),
(1953,1953,1,'12.47 pH','-332.6 mV','24.7 C (ATC)'),
(1954,1954,1,'5.51 pH','78.4 mV','31.3 C (ATC)'),
(1955,1955,1,'1.68 pH','298.9 mV','24.5 C (ATC)'),
(1956,1956,1,'4.00 pH','165.6 mV','24.9 C (ATC)'),
(1957,1957,1,'7.00 pH','-8.6 mV','25.2 C (ATC)'),
(1958,1958,1,'9.94 pH','-181.5 mV','24.3 C (ATC)'),
(1959,1959,1,'5.50 pH','79.7 mV','32.6 C (ATC)'),
(1960,1960,1,'6.36 pH','41.0 mV','25.1 C (ATC)'),
(1961,1961,1,'5.01 pH','108.6 mV','31.9 C (ATC)'),
(1962,1962,1,'5.51 pH','77.8 mV','29.0 C (ATC)'),
(1963,1963,1,'6.24 pH','33.5 mV','25.4 C (ATC)'),
(1964,1964,1,'12.42 pH','-321.8 mV','24.0 C (ATC)'),
(1965,1965,1,'6.34 pH','42.0 mV','25.3 C (ATC)'),
(1966,1966,1,'9.19 pH','-139.5 mV','24.7 C (ATC)'),
(1967,1967,1,'6.44 pH','22.2 mV','25.8 C (ATC)'),
(1968,1968,1,'6.28 pH','31.6 mV','25.4 C (ATC)'),
(1969,1969,1,'6.19 pH','36.8 mV','25.6 C (ATC)'),
(1970,1970,1,'3.80 pH','178.7 mV','27.8 C (ATC)'),
(1971,1971,1,'5.48 pH','78.2 mV','24.2 C (ATC)'),
(1972,1972,1,'5.55 pH','74.0 mV','23.9 C (ATC)'),
(1973,1973,1,'5.52 pH','76.0 mV','24.2 C (ATC)'),
(1974,1974,1,'5.53 pH','74.9 mV','24.1 C (ATC)'),
(1975,1975,1,'5.48 pH','78.1 mV','23.8 C (ATC)'),
(1976,1976,1,'5.48 pH','78.2 mV','23.7 C (ATC)'),
(1977,1977,1,'5.53 pH','75.2 mV','23.9 C (ATC)'),
(1978,1978,1,'5.52 pH','76.0 mV','23.9 C (ATC)'),
(1979,1979,1,'5.52 pH','75.6 mV','24.2 C (ATC)'),
(1980,1980,1,'1.72 pH','297.1 mV','23.4 C (ATC)'),
(1981,1981,1,'4.01 pH','169.5 mV','26.4 C (ATC)'),
(1982,1982,1,'5.49 pH','81.0 mV','36.7 C (ATC)'),
(1983,1983,1,'7.04 pH','-4.2 mV','25.3 C (ATC)'),
(1984,1984,1,'5.81 pH','59.1 mV','24.9 C (ATC)'),
(1985,1985,1,'10.03 pH','-175.5 mV','26.6 C (ATC)'),
(1986,1986,1,'5.50 pH','77.6 mV','25.4 C (ATC)'),
(1987,1987,1,'5.55 pH','74.6 mV','25.4 C (ATC)'),
(1988,1988,1,'12.44 pH','-312.8 mV','26.0 C (ATC)'),
(1989,1989,1,'5.48 pH','78.3 mV','25.3 C (ATC)'),
(1990,1990,1,'5.81 pH','59.1 mV','25.4 C (ATC)'),
(1991,1991,1,'5.83 pH','58.0 mV','25.4 C (ATC)'),
(1992,1992,1,'6.19 pH','36.4 mV','25.0 C (ATC)'),
(1993,1993,1,'7.02 pH','-3.0 mV','23.4 C (ATC)'),
(1994,1994,1,'7.06 pH','-5.3 mV','25.2 C (ATC)'),
(1995,1995,1,'7.05 pH','-4.8 mV','24.4 C (ATC)'),
(1996,1996,1,'7.08 pH','-6.6 mV','25.6 C (ATC)'),
(1997,1997,1,'4.60 pH','121.7 mV','24.1 C (ATC)'),
(1998,1998,1,'1.67 pH','302.0 mV','24.5 C (ATC)'),
(1999,1999,1,'4.02 pH','164.3 mV','24.5 C (ATC)'),
(2000,2000,1,'7.01 pH','-11.1 mV','24.5 C (ATC)'),
(2001,2001,1,'10.01 pH','-186.7 mV','24.6 C (ATC)'),
(2002,2002,1,'12.48 pH','-331.6 mV','24.3 C (ATC)'),
(2003,2003,1,'5.88 pH','46.8 mV','24.2 C (ATC)'),
(2004,2004,1,'5.90 pH','45.8 mV','23.8 C (ATC)'),
(2005,2005,1,'5.50 pH','79.1 mV','29.9 C (ATC)'),
(2006,2006,1,'5.52 pH','76.2 mV','23.7 C (ATC)'),
(2007,2007,1,'5.00 pH','109.5 mV','32.0 C (ATC)'),
(2008,2008,1,'6.24 pH','26.2 mV','26.3 C (ATC)'),
(2009,2009,1,'5.52 pH','76.9 mV','25.5 C (ATC)'),
(2010,2010,1,'5.99 pH','50.5 mV','31.9 C (ATC)'),
(2011,2011,1,'6.45 pH','13.6 mV','24.3 C (ATC)'),
(2012,2012,1,'6.00 pH','57.9 mV','29.4 C (ATC)'),
(2013,2013,1,'5.71 pH','73.4 mV','23.6 C (ATC)'),
(2014,2014,1,'5.69 pH','74.6 mV','23.5 C (ATC)'),
(2015,2015,1,'4.49 pH','147.9 mV','24.2 C (ATC)'),
(2016,2016,1,'5.66 pH','76.0 mV','23.6 C (ATC)'),
(2017,2017,1,'5.73 pH','71.9 mV','23.6 C (ATC)'),
(2018,2018,1,'5.74 pH','71.8 mV','23.7 C (ATC)'),
(2019,2019,1,'5.95 pH','59.5 mV','23.3 C (ATC)'),
(2020,2020,1,'5.94 pH','59.8 mV','23.3 C (ATC)'),
(2021,2021,1,'5.69 pH','74.3 mV','23.6 C (ATC)'),
(2022,2022,1,'5.69 pH','74.5 mV','23.4 C (ATC)'),
(2023,2023,1,'6.20 pH','28.6 mV','26.6 C (ATC)'),
(2024,2024,1,'1.65 pH','307.9 mV','25.5 C (ATC)'),
(2025,2025,1,'4.02 pH','169.3 mV','26.0 C (ATC)'),
(2026,2026,1,'7.02 pH','-6.2 mV','25.7 C (ATC)'),
(2027,2027,1,'10.01 pH','-180.8 mV','25.6 C (ATC)'),
(2028,2028,1,'12.44 pH','-323.4 mV','26.1 C (ATC)');

/*Table structure for table `tbl_weighment_detail_ph_ise` */


CREATE TABLE `tbl_weighment_detail_ph_ise` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `RepSerNo` int(11) DEFAULT NULL,
  `RecSeqNo` int(11) DEFAULT NULL,
  `ElectrodeType` varchar(50) DEFAULT 'NULL',
  `Concentration` varchar(100) DEFAULT '0',
  `Temp` varchar(100) DEFAULT '0',
  `mV` varchar(100) DEFAULT '0',
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_weighment_detail_ph_ise` */

/*Table structure for table `tbl_weighment_detail_ph_ise_archived` */


CREATE TABLE `tbl_weighment_detail_ph_ise_archived` (
  `RecNo` int(11) DEFAULT NULL,
  `RepSerNo` int(11) DEFAULT NULL,
  `RecSeqNo` int(11) DEFAULT NULL,
  `ElectrodeType` varchar(50) DEFAULT 'NULL',
  `Concentration` varchar(100) DEFAULT '0',
  `Temp` varchar(100) DEFAULT '0',
  `mV` varchar(100) DEFAULT '0'
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_weighment_detail_ph_ise_archived` */

/*Table structure for table `tbl_weighment_master_conductivity` */


CREATE TABLE `tbl_weighment_master_conductivity` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `dt` date DEFAULT NULL,
  `tm` time DEFAULT NULL,
  `NegTol` varchar(50) DEFAULT '0',
  `PosTol` varchar(50) DEFAULT '0',
  `ProductName` varchar(500) DEFAULT 'NULL',
  `SampleID` varchar(500) DEFAULT 'NULL',
  `BatchNo` varchar(500) DEFAULT 'NULL',
  `ArNo` varchar(500) DEFAULT 'NULL',
  `ProductType` varchar(500) DEFAULT 'NULL',
  `MeterID` varchar(500) DEFAULT 'NULL',
  `ProbID` varchar(500) DEFAULT 'NULL',
  `ChannelNo` varchar(500) DEFAULT 'NULL',
  `PrinterName` varchar(500) DEFAULT 'NULL',
  `UserID` varchar(500) DEFAULT 'NULL',
  `UserName` varchar(500) DEFAULT 'NULL',
  `DP` int(5) DEFAULT '0',
  `PrintNo` int(11) DEFAULT NULL,
  `IDSNo` int(11) DEFAULT '0',
  `ProbType` varchar(50) DEFAULT 'Conductivity',
  `ProdID` varchar(500) DEFAULT 'NULL',
  `archived` int(1) DEFAULT '0',
  `CheckedByID` varchar(25) DEFAULT 'NULL',
  `CheckedByName` varchar(200) DEFAULT 'NULL',
  `CheckedByDate` datetime DEFAULT '1992-12-08 00:00:00',
  `VerifyRemark` varchar(500) DEFAULT 'NULL',
  `AllData` text,
  `IsUPW` int(1) DEFAULT '0',
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB AUTO_INCREMENT=3 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_weighment_master_conductivity` */

/*Table structure for table `tbl_weighment_master_conductivity_archived` */


CREATE TABLE `tbl_weighment_master_conductivity_archived` (
  `RecNo` int(11) DEFAULT NULL,
  `dt` date DEFAULT NULL,
  `tm` time DEFAULT NULL,
  `NegTol` varchar(50) DEFAULT '0',
  `PosTol` varchar(50) DEFAULT '0',
  `ProductName` varchar(500) DEFAULT 'NULL',
  `SampleID` varchar(500) DEFAULT 'NULL',
  `BatchNo` varchar(500) DEFAULT 'NULL',
  `ArNo` varchar(500) DEFAULT 'NULL',
  `ProductType` varchar(500) DEFAULT 'NULL',
  `MeterID` varchar(500) DEFAULT 'NULL',
  `ProbID` varchar(500) DEFAULT 'NULL',
  `ChannelNo` varchar(500) DEFAULT 'NULL',
  `PrinterName` varchar(500) DEFAULT 'NULL',
  `UserID` varchar(500) DEFAULT 'NULL',
  `UserName` varchar(500) DEFAULT 'NULL',
  `DP` int(5) DEFAULT '0',
  `PrintNo` int(11) DEFAULT NULL,
  `IDSNo` int(11) DEFAULT '0',
  `ProbType` varchar(50) DEFAULT 'Conductivity',
  `ProdID` varchar(500) DEFAULT 'NULL',
  `archived` int(1) DEFAULT '0',
  `CheckedByID` varchar(25) DEFAULT 'NULL',
  `CheckedByName` varchar(200) DEFAULT 'NULL',
  `CheckedByDate` datetime DEFAULT '1992-12-08 00:00:00',
  `VerifyRemark` varchar(500) DEFAULT 'NULL',
  `AllData` text,
  `IsUPW` int(1) DEFAULT '0'
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_weighment_master_conductivity_archived` */

insert  into `tbl_weighment_master_conductivity_archived`(`RecNo`,`dt`,`tm`,`NegTol`,`PosTol`,`ProductName`,`SampleID`,`BatchNo`,`ArNo`,`ProductType`,`MeterID`,`ProbID`,`ChannelNo`,`PrinterName`,`UserID`,`UserName`,`DP`,`PrintNo`,`IDSNo`,`ProbType`,`ProdID`,`archived`,`CheckedByID`,`CheckedByName`,`CheckedByDate`,`VerifyRemark`,`AllData`,`IsUPW`) values 
(1,'2024-11-26','12:56:37','0','0','100 microsemens','NA','NA','NA','Calibration Buffer','QC-CONMZ-002','ZO1-16440','1','NA','V-trident','Trident',2,3,106,'Conductivity','Conductivity standard',1,'V-trident','Trident','2024-09-13 13:01:46','As part of OQ','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V18806\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA30043\r\nMode                 Conductivity\r\nConductivity         98.25 uS/cm\r\nConductance          1056.489 uS\r\nTemperature          24.7 C (ATC)\r\nTemp. Coefficient    Off\r\nCell Constant        0.0930 /cm\r\nMethod               100\r\nCalLog #             0\r\nLast Cal             -----\r\nLog #                27\r\n\r\n\r\n',0),
(2,'2024-11-26','16:28:08','0','0','100 microsemens','NA','NA','NA','Calibration Buffer','QC-CONMZ-002','ZO1-16440','1','NA','V-trident','Trident',1,2,106,'Conductivity','Conductivity standard',1,'V-trident','Trident','2024-09-17 16:41:50','As part of PQ','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V18806\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA30043\r\nMode                 Conductivity\r\nConductivity         100.5 uS/cm\r\nConductance          1080.608 uS\r\nTemperature          26.1 C (ATC)\r\nTemp. Coefficient    Off\r\nCell Constant        0.0930 /cm\r\nMethod               100\r\nCalLog #             0\r\nLast Cal             -----\r\n\r\n\r\n',1);

/*Table structure for table `tbl_weighment_master_do` */


CREATE TABLE `tbl_weighment_master_do` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `dt` date DEFAULT NULL,
  `tm` time DEFAULT NULL,
  `NegTol` varchar(50) DEFAULT '0',
  `PosTol` varchar(50) DEFAULT '0',
  `ProductName` varchar(500) DEFAULT 'NULL',
  `SampleID` varchar(500) DEFAULT 'NULL',
  `BatchNo` varchar(500) DEFAULT 'NULL',
  `ArNo` varchar(500) DEFAULT 'NULL',
  `ProductType` varchar(500) DEFAULT 'NULL',
  `MeterID` varchar(500) DEFAULT 'NULL',
  `ProbID` varchar(500) DEFAULT 'NULL',
  `ChannelNo` varchar(500) DEFAULT 'NULL',
  `PrinterName` varchar(500) DEFAULT 'NULL',
  `UserID` varchar(500) DEFAULT 'NULL',
  `UserName` varchar(500) DEFAULT 'NULL',
  `DP` int(5) DEFAULT '0',
  `PrintNo` int(11) DEFAULT NULL,
  `IDSNo` int(11) DEFAULT '0',
  `ProbType` varchar(50) DEFAULT 'DO',
  `ProdID` varchar(500) DEFAULT 'NULL',
  `archived` int(1) DEFAULT '0',
  `Slope` varchar(50) DEFAULT '0',
  `Saturation` varchar(50) DEFAULT '0',
  `Concentration` varchar(50) DEFAULT '0',
  `Current` varchar(50) DEFAULT '0',
  `TempSolution` varchar(50) DEFAULT '0',
  `TempMembrane` varchar(50) DEFAULT '0',
  `BioPressure` varchar(50) DEFAULT '0',
  `SalinityCorrection` varchar(50) DEFAULT '0',
  `CheckedByID` varchar(25) DEFAULT 'NULL',
  `CheckedByName` varchar(200) DEFAULT 'NULL',
  `CheckedByDate` datetime DEFAULT '1992-12-08 00:00:00',
  `VerifyRemark` varchar(500) DEFAULT 'NULL',
  `AllData` text,
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_weighment_master_do` */

/*Table structure for table `tbl_weighment_master_do_archived` */


CREATE TABLE `tbl_weighment_master_do_archived` (
  `RecNo` int(11) DEFAULT NULL,
  `dt` date DEFAULT NULL,
  `tm` time DEFAULT NULL,
  `NegTol` varchar(50) DEFAULT '0',
  `PosTol` varchar(50) DEFAULT '0',
  `ProductName` varchar(500) DEFAULT 'NULL',
  `SampleID` varchar(500) DEFAULT 'NULL',
  `BatchNo` varchar(500) DEFAULT 'NULL',
  `ArNo` varchar(500) DEFAULT 'NULL',
  `ProductType` varchar(500) DEFAULT 'NULL',
  `MeterID` varchar(500) DEFAULT 'NULL',
  `ProbID` varchar(500) DEFAULT 'NULL',
  `ChannelNo` varchar(500) DEFAULT 'NULL',
  `PrinterName` varchar(500) DEFAULT 'NULL',
  `UserID` varchar(500) DEFAULT 'NULL',
  `UserName` varchar(500) DEFAULT 'NULL',
  `DP` int(5) DEFAULT '0',
  `PrintNo` int(11) DEFAULT NULL,
  `IDSNo` int(11) DEFAULT '0',
  `ProbType` varchar(50) DEFAULT 'DO',
  `ProdID` varchar(500) DEFAULT 'NULL',
  `archived` int(1) DEFAULT '0',
  `Slope` varchar(50) DEFAULT '0',
  `Saturation` varchar(50) DEFAULT '0',
  `Concentration` varchar(50) DEFAULT '0',
  `Current` varchar(50) DEFAULT '0',
  `TempSolution` varchar(50) DEFAULT '0',
  `TempMembrane` varchar(50) DEFAULT '0',
  `BioPressure` varchar(50) DEFAULT '0',
  `SalinityCorrection` varchar(50) DEFAULT '0',
  `CheckedByID` varchar(25) DEFAULT 'NULL',
  `CheckedByName` varchar(200) DEFAULT 'NULL',
  `CheckedByDate` datetime DEFAULT '1992-12-08 00:00:00',
  `VerifyRemark` varchar(500) DEFAULT 'NULL',
  `AllData` text
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_weighment_master_do_archived` */

/*Table structure for table `tbl_weighment_master_ph` */


CREATE TABLE `tbl_weighment_master_ph` (
  `RecNo` int(11) NOT NULL AUTO_INCREMENT,
  `dt` date DEFAULT NULL,
  `tm` time DEFAULT NULL,
  `NegTol` varchar(50) DEFAULT '0',
  `PosTol` varchar(50) DEFAULT '0',
  `ProductName` varchar(500) DEFAULT 'NULL',
  `SampleID` varchar(500) DEFAULT 'NULL',
  `BatchNo` varchar(500) DEFAULT 'NULL',
  `ArNo` varchar(500) DEFAULT 'NULL',
  `ProductType` varchar(500) DEFAULT 'NULL',
  `MeterID` varchar(500) DEFAULT 'NULL',
  `ProbID` varchar(500) DEFAULT 'NULL',
  `ChannelNo` varchar(500) DEFAULT 'NULL',
  `PrinterName` varchar(500) DEFAULT 'NULL',
  `UserID` varchar(500) DEFAULT 'NULL',
  `UserName` varchar(500) DEFAULT 'NULL',
  `DP` int(5) DEFAULT '0',
  `PrintNo` int(11) DEFAULT NULL,
  `IDSNo` int(11) DEFAULT '0',
  `ProbType` varchar(50) DEFAULT 'pH',
  `ProdID` varchar(500) DEFAULT 'NULL',
  `archived` int(1) DEFAULT '0',
  `CheckedByID` varchar(25) DEFAULT 'NULL',
  `CheckedByName` varchar(200) DEFAULT 'NULL',
  `CheckedByDate` datetime DEFAULT '1992-12-08 00:00:00',
  `VerifyRemark` varchar(500) DEFAULT 'NULL',
  `AllData` text,
  PRIMARY KEY (`RecNo`)
) ENGINE=InnoDB AUTO_INCREMENT=2029 DEFAULT CHARSET=latin1;

/*Data for the table `tbl_weighment_master_ph` */

/*Table structure for table `tbl_weighment_master_ph_archived` */


CREATE TABLE `tbl_weighment_master_ph_archived` (
  `RecNo` int(11) DEFAULT NULL,
  `dt` date DEFAULT NULL,
  `tm` time DEFAULT NULL,
  `NegTol` varchar(50) DEFAULT '0',
  `PosTol` varchar(50) DEFAULT '0',
  `ProductName` varchar(500) DEFAULT 'NULL',
  `SampleID` varchar(500) DEFAULT 'NULL',
  `BatchNo` varchar(500) DEFAULT 'NULL',
  `ArNo` varchar(500) DEFAULT 'NULL',
  `ProductType` varchar(500) DEFAULT 'NULL',
  `MeterID` varchar(500) DEFAULT 'NULL',
  `ProbID` varchar(500) DEFAULT 'NULL',
  `ChannelNo` varchar(500) DEFAULT 'NULL',
  `PrinterName` varchar(500) DEFAULT 'NULL',
  `UserID` varchar(500) DEFAULT 'NULL',
  `UserName` varchar(500) DEFAULT 'NULL',
  `DP` int(5) DEFAULT '0',
  `PrintNo` int(11) DEFAULT NULL,
  `IDSNo` int(11) DEFAULT '0',
  `ProbType` varchar(50) DEFAULT 'pH',
  `ProdID` varchar(500) DEFAULT 'NULL',
  `archived` int(1) DEFAULT '0',
  `CheckedByID` varchar(25) DEFAULT 'NULL',
  `CheckedByName` varchar(200) DEFAULT 'NULL',
  `CheckedByDate` datetime DEFAULT '1992-12-08 00:00:00',
  `VerifyRemark` varchar(500) DEFAULT 'NULL',
  `AllData` text
) ENGINE=InnoDB DEFAULT CHARSET=latin1;

/*Data for the table `tbl_weighment_master_ph_archived` */

insert  into `tbl_weighment_master_ph_archived`(`RecNo`,`dt`,`tm`,`NegTol`,`PosTol`,`ProductName`,`SampleID`,`BatchNo`,`ArNo`,`ProductType`,`MeterID`,`ProbID`,`ChannelNo`,`PrinterName`,`UserID`,`UserName`,`DP`,`PrintNo`,`IDSNo`,`ProbType`,`ProdID`,`archived`,`CheckedByID`,`CheckedByName`,`CheckedByDate`,`VerifyRemark`,`AllData`) values 
(1,'2024-09-13','10:29:53','90.00000','105.00000','pH Buffer','NA','NA','NA','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','V-trident','Trident',2,0,101,'pH','4.01pH',1,'85395','Ravi Kumar','2024-09-24 14:59:06','For E-lims interface','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   0.0 mV\r\nTemperature          25.0 C (MAN)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             13-09-24,09:35\r\n\r\n\r\n'),
(2,'2024-09-17','15:02:34','90.00000','105.00000','pH Buffer','NA','NA','NA','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','V-trident','Trident',2,1,101,'pH','7.00pH',1,'V-trident','Trident','2024-09-17 15:34:39','As part of PQ','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   0.0 mV\r\nTemperature          25.0 C (MAN)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             17-09-24,15:01\r\n\r\n\r\n'),
(3,'2024-09-17','15:16:31','90.00000','105.00000','pH Buffer','NA','NA','NA','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','V-trident','Trident',2,1,102,'pH','7.00pH',1,'V-trident','Trident','2024-09-17 15:35:11','As part of PQ','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   0.0 mV\r\nTemperature          25.0 C (MAN)\r\nAverage Slope        100.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             17-09-24,15:07\r\n\r\n\r\n'),
(4,'2024-09-17','16:35:54','90.00000','105.00000','pH Buffer','NA','NA','NA','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','V-trident','Trident',2,1,103,'pH','7.00pH',1,'V-trident','Trident','2024-09-17 16:42:16','As part of PQ','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   0.0 mV\r\nTemperature          25.0 C (MAN)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             17-09-24,16:34\r\n\r\n\r\n'),
(5,'2024-09-18','14:30:04','90.00000','105.00000','pH Buffer','NA','NA','NA','pH Buffer','PR-PHMTZ-003','CW1-12031','1','NULL','V-trident','Trident',2,1,105,'pH','7.00pH',1,'V-trident','Trident','2024-09-18 15:50:43','As part of PQ','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122332\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36314\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   0.0 mV\r\nTemperature          25.0 C (MAN)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             18-09-24,14:16\r\n\r\n\r\n'),
(6,'2024-09-18','15:42:22','90.00000','105.00000','pH Buffer','NA','NA','NA','pH Buffer','PR-PHMTZ-004','CX1-15689','1','NULL','V-trident','Trident',2,1,104,'pH','7.00pH',1,'V-trident','Trident','2024-09-18 15:51:21','As part of PQ','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   0.0 mV\r\nTemperature          25.0 C (MAN)\r\nAverage Slope        100.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_103\r\nLast Cal             18-09-24,15:40\r\n\r\n\r\n'),
(7,'2024-09-19','10:40:01','90.00000','105.00000','pH Buffer','NA','NA','NA','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','V-trident','Trident',2,0,102,'pH','7.00pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.96 pH\r\nmV                   13.9 mV\r\nTemperature          22.9 C (ATC)\r\nAverage Slope        100.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             19-09-24,10:37\r\n\r\n\r\n'),
(8,'2024-09-19','12:02:41','90.00000','105.00000','pH Buffer','NA','NA','NA','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','V-trident','Trident',2,0,101,'pH','7.00pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.97 pH\r\nmV                   -1.5 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             19-09-24,12:00\r\n\r\n\r\n'),
(9,'2024-09-19','12:30:02','90.00000','105.00000','pH Buffer','NA','NA','NA','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','V-trident','Trident',2,0,101,'pH','7.00pH',1,'85395','Ravi Kumar','2024-09-24 15:42:48','For Lims Interface','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -0.3 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             19-09-24,12:28\r\n\r\n\r\n'),
(10,'2024-09-19','12:45:25','90.00000','105.00000','pH Buffer','NA','NA','NA','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','V-trident','Trident',2,0,101,'pH','7.00pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -0.3 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             19-09-24,12:28\r\n\r\n\r\n'),
(11,'2024-09-19','14:18:08','90.00000','105.00000','pH Buffer','NA','NA','NA','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','V-trident','Trident',2,0,101,'pH','4.01pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -0.2 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             19-09-24,12:28\r\n\r\n\r\n'),
(12,'2024-09-30','17:02:04','90.00000','105.00000','pH Buffer','verification','na','E4INS24005157','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-09-30 17:54:32','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   309.7 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-09-24,16:49\r\nLog #                23\r\n\r\n\r\n'),
(13,'2024-09-30','17:11:33','90.00000','105.00000','pH Buffer','VERIFICATION','NA','E4INS24005158','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205066','Sreeram Jakkaraiah',2,1,102,'pH','1.68 pH',1,'86253','D. Jayapal Reddy','2024-09-30 17:46:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   311.5 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             30-09-24,16:49\r\nLog #                21\r\n\r\n\r\n'),
(14,'2024-09-30','17:15:23','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005157','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-09-30 17:54:47','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   173.6 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-09-24,16:49\r\nLog #                24\r\n\r\n\r\n'),
(15,'2024-09-30','17:18:58','90.00000','105.00000','pH Buffer','VERIFICATION','NA','E4INS24005158','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205066','Sreeram Jakkaraiah',2,1,102,'pH','4.01pH',1,'86253','D. Jayapal Reddy','2024-09-30 17:46:57','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   4.00 pH\r\nmV                   176.4 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             30-09-24,16:49\r\nLog #                22\r\n\r\n\r\n'),
(16,'2024-09-30','17:20:29','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005157','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-09-30 17:55:17','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.98 pH\r\nmV                   -2.6 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-09-24,16:49\r\nLog #                25\r\n\r\n\r\n'),
(17,'2024-09-30','17:23:47','90.00000','105.00000','pH Buffer','VERIFICATION','NA','E4INS24005158','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205066','Sreeram Jakkaraiah',2,1,102,'pH','7.00pH',1,'86253','D. Jayapal Reddy','2024-09-30 17:47:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   0.4 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             30-09-24,16:49\r\nLog #                23\r\n\r\n\r\n'),
(18,'2024-09-30','17:27:43','90.00000','105.00000','pH Buffer','VERIFICATION','NA','E4INS24005158','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205066','Sreeram Jakkaraiah',2,1,102,'pH','10.01 pH',1,'86253','D. Jayapal Reddy','2024-09-30 17:47:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   9.99 pH\r\nmV                   -170.9 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             30-09-24,16:49\r\nLog #                24\r\n\r\n\r\n'),
(19,'2024-09-30','17:25:36','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005157','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-09-30 17:55:44','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.03 pH\r\nmV                   -174.5 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-09-24,16:49\r\nLog #                26\r\n\r\n\r\n'),
(20,'2024-09-30','17:30:54','90.00000','105.00000','pH Buffer','VERIFICATION','NA','E4INS24005158','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205066','Sreeram Jakkaraiah',2,1,102,'pH','12.46 pH',1,'86253','D. Jayapal Reddy','2024-09-30 17:47:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -319.6 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             30-09-24,16:49\r\nLog #                25\r\n\r\n\r\n'),
(21,'2024-09-30','17:30:11','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005157','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-09-30 17:56:33','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.49 pH\r\nmV                   -318.6 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.6 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             30-09-24,16:49\r\nLog #                27\r\n\r\n\r\n'),
(22,'2024-09-30','18:00:17','90.00000','105.00000','Piperacillin and Tazobactam for Injection','PIPERTAZO','PA0424001A','E4STFP24000622','Stability Study','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,1,102,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-09-30 18:12:34','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.37 pH\r\nmV                   37.3 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             30-09-24,16:49\r\n\r\n\r\n'),
(23,'2024-10-01','07:21:55','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005160','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,1,102,'pH','1.68 pH',1,'90778','B. Bhaskar','2024-10-01 07:52:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   1.71 pH\r\nmV                   312.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\nLog #                27\r\n\r\n\r\n'),
(24,'2024-10-01','07:23:31','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005160','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,1,102,'pH','4.01pH',1,'90778','B. Bhaskar','2024-10-01 07:52:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   173.7 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\nLog #                28\r\n\r\n\r\n'),
(25,'2024-10-01','07:26:31','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005160','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,1,102,'pH','7.00pH',1,'90778','B. Bhaskar','2024-10-01 07:53:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -0.6 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\nLog #                29\r\n\r\n\r\n'),
(26,'2024-10-01','07:28:17','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005160','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,1,102,'pH','10.01 pH',1,'90778','B. Bhaskar','2024-10-01 07:54:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   10.03 pH\r\nmV                   -173.1 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\nLog #                30\r\n\r\n\r\n'),
(27,'2024-10-01','07:29:59','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005160','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,1,102,'pH','12.46 pH',1,'90778','B. Bhaskar','2024-10-01 07:54:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   12.44 pH\r\nmV                   -317.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\nLog #                31\r\n\r\n\r\n'),
(28,'2024-10-01','07:24:12','90.00000','105.00000','pH Buffer','VERIFICATION','NA','E4INS24005159','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','1.68 pH',1,'90778','B. Bhaskar','2024-10-01 08:01:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.67 pH\r\nmV                   310.1 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             01-10-24,07:10\r\nLog #                28\r\n\r\n\r\n'),
(29,'2024-10-01','07:26:55','90.00000','105.00000','pH Buffer','VERIFICATION','NA','E4INS24005159','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','4.01pH',1,'90778','B. Bhaskar','2024-10-01 08:02:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   170.7 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             01-10-24,07:10\r\nLog #                29\r\n\r\n\r\n'),
(30,'2024-10-01','07:29:04','90.00000','105.00000','pH Buffer','VERIFICATION','NA','E4INS24005159','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','7.00pH',1,'90778','B. Bhaskar','2024-10-01 08:02:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.98 pH\r\nmV                   -2.9 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             01-10-24,07:10\r\nLog #                30\r\n\r\n\r\n'),
(31,'2024-10-01','07:30:27','90.00000','105.00000','pH Buffer','VERIFICATION','NA','E4INS24005159','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','10.01 pH',1,'90778','B. Bhaskar','2024-10-01 08:09:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.02 pH\r\nmV                   -173.2 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             01-10-24,07:10\r\nLog #                31\r\n\r\n\r\n'),
(32,'2024-10-01','07:35:11','90.00000','105.00000','pH Buffer','VERIFICATION','NA','E4INS24005159','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88571','V. Ramu',2,1,101,'pH','12.46 pH',1,'90778','B. Bhaskar','2024-10-01 08:09:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -315.8 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        97.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             01-10-24,07:10\r\nLog #                32\r\n\r\n\r\n'),
(33,'2024-10-01','09:58:04','90.00000','105.00000','pH Buffer','1.68','NA','E4INS24005172','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-01 12:02:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   319.7 mV\r\nTemperature          26.5 C (ATC)\r\nAverage Slope        98.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,09:47\r\nLog #                18\r\n\r\n\r\n'),
(34,'2024-10-01','10:01:44','90.00000','105.00000','pH Buffer','4.0','NA','E4INS24005172','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-01 12:03:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   180.3 mV\r\nTemperature          26.5 C (ATC)\r\nAverage Slope        98.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,09:47\r\nLog #                20\r\n\r\n\r\n'),
(35,'2024-10-01','10:02:51','90.00000','105.00000','pH Buffer','7.0','NA','E4INS24005172','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-01 12:04:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.98 pH\r\nmV                   4.9 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        98.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,09:47\r\nLog #                21\r\n\r\n\r\n'),
(36,'2024-10-01','10:04:30','90.00000','105.00000','pH Buffer','10.01','NA','E4INS24005172','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-01 12:04:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   9.98 pH\r\nmV                   -168.7 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        98.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,09:47\r\nLog #                22\r\n\r\n\r\n'),
(37,'2024-10-01','10:06:32','90.00000','105.00000','pH Buffer','12.46','NA','E4INS24005172','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-01 12:04:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.43 pH\r\nmV                   -307.7 mV\r\nTemperature          26.7 C (ATC)\r\nAverage Slope        98.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,09:47\r\nLog #                23\r\n\r\n\r\n'),
(38,'2024-10-01','10:38:14','90.00000','105.00000','Sensor Calibration','PH METER TEMPERATURE SENSOR CALIBRATION','NA','E4INS24005176','Temperature Sensor','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,1,102,'pH','Purified Water',1,'90778','B. Bhaskar','2024-10-01 11:13:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.64 pH\r\nmV                   21.4 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\nLog #                32\r\n\r\n\r\n'),
(39,'2024-10-01','10:46:46','90.00000','105.00000','PIPERACILLIN','NA','TZ0024090','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,1,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-01 14:36:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.85 pH\r\nmV                   68.9 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             01-10-24,09:12\r\n\r\n\r\n'),
(40,'2024-10-01','11:40:12','90.00000','105.00000','PIPERACILLIN','NA','TZ0024090','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-01 14:36:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.90 pH\r\nmV                   65.8 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             01-10-24,09:12\r\n\r\n\r\n'),
(41,'2024-10-01','11:42:03','90.00000','105.00000','PIPERACILLIN','NA','TZ0024090','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-01 14:37:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.95 pH\r\nmV                   62.6 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             01-10-24,09:12\r\n\r\n\r\n'),
(42,'2024-10-01','12:16:21','90.00000','105.00000','TAZOBACTAM','NA','TZ0024090','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,1,104,'pH','1',1,'95681','T. Sivanarayana','2024-10-01 14:35:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.48 pH\r\nmV                   149.4 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             01-10-24,09:12\r\n\r\n\r\n'),
(43,'2024-10-01','12:53:03','90.00000','105.00000','Rubber Stopper','Heavy metals','31743664','E4PM24000575','Packing Material','QC-PHMTZ-009','CX1-16015','1','NULL','88571','V. Ramu',2,1,102,'pH','3.5 Acetate Buffer',1,'86253','D. Jayapal Reddy','2024-10-01 13:12:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   3.52 pH\r\nmV                   205.3 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\nLog #                33\r\n\r\n\r\n'),
(44,'2024-10-01','13:32:42','90.00000','105.00000','PIPERACILLIN','NA','TZ0024090','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-01 14:40:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.44 pH\r\nmV                   33.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             01-10-24,09:12\r\n\r\n\r\n'),
(45,'2024-10-01','13:34:04','90.00000','105.00000','PIPERACILLIN','na','TZ0024090','na','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-01 14:40:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.43 pH\r\nmV                   34.2 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             01-10-24,09:12\r\n\r\n\r\n'),
(46,'2024-10-01','13:48:14','90.00000','105.00000','Flucloxacillin Sodium','Assay by HPLC','FL0124003','E4WS24000011','Raw Material','QC-PHMTZ-009','CX1-16015','1','NULL','205189','P. Ravi Kumar',2,1,102,'pH','5.0 Phosphate Buffer',1,'82735','M.D. Babar Qhadri','2024-10-01 14:12:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.00 pH\r\nmV                   118.0 mV\r\nTemperature          27.5 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\nLog #                34\r\n\r\n\r\n'),
(47,'2024-10-01','13:58:14','90.00000','105.00000','Flucloxacillin Sodium','Assay by HPLC (ANALYST -2)','FL0124003','E4WS24000011','Raw Material','QC-PHMTZ-009','CX1-16015','1','NULL','205048','P. Nagesh',2,1,102,'pH','5.0 Phosphate Buffer',1,'82735','M.D. Babar Qhadri','2024-10-01 14:11:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   4.99 pH\r\nmV                   118.1 mV\r\nTemperature          27.3 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\nLog #                35\r\n\r\n\r\n'),
(48,'2024-10-01','14:30:04','90.00000','105.00000','Sensor Calibration','Temprerature verification','NA','E4INS24005178','Temperature Sensor','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','Purified Water',1,'205143','Ch. Prasad','2024-10-01 14:36:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.66 pH\r\nmV                   82.5 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,09:47\r\n\r\n\r\n'),
(49,'2024-10-01','15:14:23','90.00000','105.00000','Purified water','pH','PW_SP06',' E4RM24005303','Purified water','QC-PHMTZ-009','CX1-16015','1','NULL','205066','Sreeram Jakkaraiah',2,0,102,'pH','pH Test',1,'84378','Shivanand. W','2024-10-01 16:13:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.77 pH\r\nmV                   72.1 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\nLog #                36\r\n\r\n\r\n'),
(50,'2024-10-01','15:18:55','90.00000','105.00000','Purified water','pH','PW_SP37','E4RM24005304','Purified water','QC-PHMTZ-009','CX1-16015','1','NULL','205066','Sreeram Jakkaraiah',2,0,102,'pH','pH Test',1,'84378','Shivanand. W','2024-10-01 16:13:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.73 pH\r\nmV                   73.8 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\n\r\n\r\n'),
(51,'2024-10-01','15:21:49','90.00000','105.00000','Water For Injection','pH','WFI_SP14','E4RM24005305','Water For Injection','QC-PHMTZ-009','CX1-16015','1','NULL','205066','Sreeram Jakkaraiah',2,0,102,'pH','pH',1,'84378','Shivanand. W','2024-10-01 16:14:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.76 pH\r\nmV                   72.4 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\n\r\n\r\n'),
(52,'2024-10-01','15:24:54','90.00000','105.00000','Water For Injection','pH','WFI_SP35','E4RM24005306','Water For Injection','QC-PHMTZ-009','CX1-16015','1','NULL','205066','Sreeram Jakkaraiah',2,0,102,'pH','pH',1,'84378','Shivanand. W','2024-10-01 16:15:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   73.4 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\n\r\n\r\n'),
(53,'2024-10-01','15:27:44','90.00000','105.00000','Water For Injection','pH','WFI_SP13','E4RM24005307','Water For Injection','QC-PHMTZ-009','CX1-16015','1','NULL','205066','Sreeram Jakkaraiah',2,0,102,'pH','pH',1,'84378','Shivanand. W','2024-10-01 16:15:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.67 pH\r\nmV                   77.7 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\n\r\n\r\n'),
(54,'2024-10-01','15:30:44','90.00000','105.00000','Pure steam','pH','PSC_SP10','E4RM24005308','Pure steam Condensate','QC-PHMTZ-009','CX1-16015','1','NULL','205066','Sreeram Jakkaraiah',2,0,102,'pH','pH_Pure Steam',1,'84378','Shivanand. W','2024-10-01 16:15:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   75.9 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\n\r\n\r\n'),
(55,'2024-10-01','15:34:03','90.00000','105.00000','Purified water','pH','PW_SP13','E4RM24005309','Purified water','QC-PHMTZ-009','CX1-16015','1','NULL','205066','Sreeram Jakkaraiah',2,0,102,'pH','pH Test',1,'84378','Shivanand. W','2024-10-01 16:16:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   72.7 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\n\r\n\r\n'),
(56,'2024-10-01','15:39:24','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024090','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','3',1,'88629','Raghuram Reddy','2024-10-01 15:41:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.25 pH\r\nmV                   45.3 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             01-10-24,09:12\r\n\r\n\r\n'),
(57,'2024-10-01','15:37:50','90.00000','105.00000','Purified water','pH','PW_SP48','E4RM24005310','Purified water','QC-PHMTZ-009','CX1-16015','1','NULL','205066','Sreeram Jakkaraiah',2,0,102,'pH','pH Test',1,'84378','Shivanand. W','2024-10-01 16:16:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.77 pH\r\nmV                   72.0 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\nLog #                43\r\n\r\n\r\n'),
(58,'2024-10-01','15:44:56','90.00000','105.00000','Water For Injection','pH','WFI_SP16','E4RM24005311','Water For Injection','QC-PHMTZ-009','CX1-16015','1','NULL','205066','Sreeram Jakkaraiah',2,0,102,'pH','pH',1,'84378','Shivanand. W','2024-10-01 16:16:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   75.9 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\n\r\n\r\n'),
(59,'2024-10-01','15:52:55','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024090','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','3',1,'88629','Raghuram Reddy','2024-10-01 15:58:56','Reveiwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.27 pH\r\nmV                   43.5 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             01-10-24,09:12\r\n\r\n\r\n'),
(60,'2024-10-01','15:59:03','90.00000','105.00000','Piperacillin and Tazobactam for Injection','ASSAY','PN0324060A','E4STFP24000603','Stability Study','QC-PHMTZ-009','CX1-16015','1','NULL','89853','J.Ravi',2,1,102,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-01 16:17:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   87.3 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,07:18\r\nLog #                45\r\n\r\n\r\n'),
(61,'2024-10-01','16:21:14','90.00000','105.00000','Macckankey broth','RS013/24006','RS013/24006','E4SO24000593','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','NA',1,'205143','Ch. Prasad','2024-10-01 16:48:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.31 pH\r\nmV                   -14.1 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,09:47\r\nLog #                27\r\n\r\n\r\n'),
(62,'2024-10-01','18:41:00','90.00000','105.00000','Macckankey broth','RS013/24006','RS013/24006','	E4SO24000593','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','NA',1,'205143','Ch. Prasad','2024-10-03 12:25:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.17 pH\r\nmV                   -6.1 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             01-10-24,09:47\r\n\r\n\r\n'),
(63,'2024-10-02','17:50:54','90.00000','105.00000','TAZOBACTAM','NA','TR0024002','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','1',1,'88629','Raghuram Reddy','2024-10-02 17:54:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.31 pH\r\nmV                   156.5 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             02-10-24,08:21\r\n\r\n\r\n'),
(64,'2024-10-02','22:44:07','90.00000','105.00000','PIPERACILLIN','NA','TR0024002','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','Siva Shankar Reddy',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-02 22:52:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.32 pH\r\nmV                   42.9 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        98.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_105\r\nLast Cal             02-10-24,08:21\r\n\r\n\r\n'),
(65,'2024-10-03','03:34:00','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TR0024002','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','Siva Shankar Reddy',2,0,104,'pH','3',1,'84472','G. Ashok','2024-10-03 05:43:01','Reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.16 pH\r\nmV                   50.6 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             03-10-24,00:53\r\n\r\n\r\n'),
(66,'2024-10-03','03:56:55','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TR0024002','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','Siva Shankar Reddy',2,0,104,'pH','3',1,'84472','G. Ashok','2024-10-03 05:44:27','Reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.18 pH\r\nmV                   49.2 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_106\r\nLast Cal             03-10-24,00:53\r\n\r\n\r\n'),
(67,'2024-10-03','07:47:24','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005190','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205211','S. Sailatha',2,0,102,'pH','1.68 pH',1,'84378','Shivanand. W','2024-10-03 08:16:17','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   311.9 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                46\r\n\r\n\r\n'),
(68,'2024-10-03','07:49:18','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005190','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205211','S. Sailatha',2,0,102,'pH','4.01pH',1,'84378','Shivanand. W','2024-10-03 08:16:46','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   4.00 pH\r\nmV                   173.2 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                47\r\n\r\n\r\n'),
(69,'2024-10-03','07:51:24','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005190','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205211','S. Sailatha',2,0,102,'pH','7.00pH',1,'84378','Shivanand. W','2024-10-03 08:17:07','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -3.1 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                48\r\n\r\n\r\n'),
(70,'2024-10-03','07:53:17','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005190','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205211','S. Sailatha',2,0,102,'pH','10.01 pH',1,'84378','Shivanand. W','2024-10-03 08:17:37','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   10.03 pH\r\nmV                   -174.9 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                49\r\n\r\n\r\n'),
(71,'2024-10-03','07:54:29','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005190','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205211','S. Sailatha',2,0,102,'pH','12.46 pH',1,'84378','Shivanand. W','2024-10-03 08:18:08','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   12.45 pH\r\nmV                   -321.2 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                50\r\n\r\n\r\n'),
(72,'2024-10-03','08:52:34','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS By HPLC ','PN0324060A',' E4STFP24000603','Stability Study','QC-PHMTZ-009','CX1-16015','1','NULL','87697','M. Raghavendr Reddy',2,1,102,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-03 09:04:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.04 pH\r\nmV                   53.4 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                51\r\n\r\n\r\n'),
(73,'2024-10-03','09:37:43','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','ASSAY BY HPLC','TR0024002','E4PP24001567','BULK','QC-PHMTZ-009','CX1-16015','1','NULL','88655','V. Prashanth Kumar',2,1,102,'pH','PIPERTAZ',1,'84378','Shivanand. W','2024-10-03 10:10:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   87.6 mV\r\nTemperature          35.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                52\r\n\r\n\r\n'),
(74,'2024-10-03','11:10:48','90.00000','105.00000','Penicillin G Benzathine','DISSOLUTION','PG0124003A','E4STFP24000566','Stability Study','QC-PHMTZ-009','CX1-16015','1','NULL','89180','R. Harish Reddy',2,1,102,'pH','Penicillin G',1,'84378','Shivanand. W','2024-10-03 11:43:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   3.31 pH\r\nmV                   217.5 mV\r\nTemperature          30.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(75,'2024-10-03','11:18:39','90.00000','105.00000','Penicillin G Benzathine','DISSOLUTION','PG0124003A','E4STFP24000566','Stability Study','QC-PHMTZ-009','CX1-16015','1','NULL','89180','R. Harish Reddy',2,1,102,'pH','Penicillin G',1,'84378','Shivanand. W','2024-10-03 13:31:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   7.54 pH\r\nmV                   -33.7 mV\r\nTemperature          31.2 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(76,'2024-10-03','11:56:40','90.00000','105.00000','pH Buffer','NA','pH Buffer','E4INS24005204','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-03 12:17:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   318.5 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             03-10-24,11:46\r\nLog #                30\r\n\r\n\r\n'),
(77,'2024-10-03','11:57:14','90.00000','105.00000','Piperacillin and Tazobactam For injection','Assay','TZ0424086','E4FP24000668','Finished Product','QC-PHMTZ-009','CX1-16015','1','NULL','205119','S.M. Subhani',2,0,102,'pH','Pipertazoinj',1,'84378','Shivanand. W','2024-10-03 12:41:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   85.8 mV\r\nTemperature          31.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                55\r\n\r\n\r\n'),
(78,'2024-10-03','12:04:57','90.00000','105.00000','pH Buffer','NA','pH Buffer','E4INS24005204','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-03 12:18:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   180.1 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             03-10-24,11:46\r\nLog #                31\r\n\r\n\r\n'),
(79,'2024-10-03','12:09:20','90.00000','105.00000','pH Buffer','NA','pH Buffer','E4INS24005204','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-03 12:18:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   4.8 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             03-10-24,11:46\r\n\r\n\r\n'),
(80,'2024-10-03','12:11:07','90.00000','105.00000','pH Buffer','NA','pH Buffer','E4INS24005204','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-03 12:18:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -168.6 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             03-10-24,11:46\r\n\r\n\r\n'),
(81,'2024-10-03','12:12:45','90.00000','105.00000','pH Buffer','NA','pH Buffer','E4INS24005204','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-03 12:19:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.44 pH\r\nmV                   -310.7 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             03-10-24,11:46\r\n\r\n\r\n'),
(82,'2024-10-03','12:16:57','90.00000','105.00000','Penicillin G Benzathine','RS','PG0124001A','E4STFP24000548','Stability Study','QC-PHMTZ-009','CX1-16015','1','NULL','205043','T.V. Subbaiah',2,1,102,'pH','Penicillin G',1,'84378','Shivanand. W','2024-10-03 13:31:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   3.31 pH\r\nmV                   216.4 mV\r\nTemperature          28.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(83,'2024-10-03','12:25:42','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC','TZ0424084','E4FP24000663','Finished Product','QC-PHMTZ-009','CX1-16015','1','NULL','80903','D. naga Prasad',2,1,102,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-04 12:10:24','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   82.9 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(84,'2024-10-03','12:32:30','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC','TZ0424084','E4FP24000663','Finished Product','QC-PHMTZ-009','CX1-16015','1','NULL','80903','D. naga Prasad',2,1,102,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-04 12:11:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   82.9 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(85,'2024-10-03','12:57:19','90.00000','105.00000','Peptone','NA','RS007/24029','E4SO24000598','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-03 16:58:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.07 pH\r\nmV                   0.7 mV\r\nTemperature          28.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             03-10-24,11:46\r\n\r\n\r\n'),
(86,'2024-10-03','13:50:24','90.00000','105.00000','Floxapen powder for solution for injection for infusion','pH','FP0124001A','E4STFP24000626','Stability Study','QC-PHMTZ-009','CX1-16015','1','NULL','205211','S. Sailatha',2,0,102,'pH','Floxapen powder',1,'82735','M.D. Babar Qhadri','2024-10-03 14:11:25','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.71 pH\r\nmV                   72.7 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(87,'2024-10-03','13:54:13','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TI0323006A','E4STFP24000627','Stability Study','QC-PHMTZ-009','CX1-16015','1','NULL','205211','S. Sailatha',2,0,102,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-03 14:12:22','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.22 pH\r\nmV                   42.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(88,'2024-10-03','14:13:30','90.00000','105.00000','Purified water','pH','PW_SP06','E4RM24005312','Purified water','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-03 18:54:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.78 pH\r\nmV                   68.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(89,'2024-10-03','14:19:05','90.00000','105.00000','Purified water','pH','PW_SP37','E4RM24005313','Purified water','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-03 18:55:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   70.7 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(90,'2024-10-03','14:25:23','90.00000','105.00000','Water For Injection','pH','WFI_SP14','E4RM24005314','Water For Injection','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','pH',1,'90018','Ch. Satyam','2024-10-03 18:56:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.81 pH\r\nmV                   67.0 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                63\r\n\r\n\r\n'),
(91,'2024-10-03','14:41:39','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','pH','TZ0024090','E4IP24001214','BULK','QC-PHMTZ-009','CX1-16015','1','NULL','205210','B. Anitha',2,0,102,'pH','3',1,'90018','Ch. Satyam','2024-10-03 16:01:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   70.2 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(92,'2024-10-03','14:56:22','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024002','E4PP24001563','BULK','QC-PHMTZ-009','CX1-16015','1','NULL','205210','B. Anitha',2,0,102,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-10-03 16:00:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   4.23 pH\r\nmV                   158.8 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(93,'2024-10-03','15:05:27','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024002','E4PP24001565','BULK','QC-PHMTZ-009','CX1-16015','1','NULL','205210','B. Anitha',2,0,102,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-10-03 15:59:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.25 pH\r\nmV                   41.4 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                66\r\n\r\n\r\n'),
(94,'2024-10-03','15:09:14','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024002','E4PP24001566','BULK','QC-PHMTZ-009','CX1-16015','1','NULL','205210','B. Anitha',2,0,102,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-10-03 16:02:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.13 pH\r\nmV                   48.5 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(95,'2024-10-03','15:12:28','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024002','E4PP24001568','BULK','QC-PHMTZ-009','CX1-16015','1','NULL','205210','B. Anitha',2,0,102,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-10-03 16:04:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.33 pH\r\nmV                   36.2 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(96,'2024-10-03','15:15:27','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024002','E4PP24001570','BULK','QC-PHMTZ-009','CX1-16015','1','NULL','205210','B. Anitha',2,0,102,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-10-03 16:04:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.33 pH\r\nmV                   36.5 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                69\r\n\r\n\r\n'),
(97,'2024-10-03','15:19:18','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024002','E4PP24001572','BULK','QC-PHMTZ-009','CX1-16015','1','NULL','205210','B. Anitha',2,0,102,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-10-03 16:04:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.33 pH\r\nmV                   36.3 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(98,'2024-10-03','15:23:24','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0424002','E4PP24001575','BULK','QC-PHMTZ-009','CX1-16015','1','NULL','205210','B. Anitha',2,0,102,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-10-03 16:05:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.46 pH\r\nmV                   28.8 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(99,'2024-10-03','15:29:24','90.00000','105.00000','Water For Injection','pH','WFI_SP35','E4RM24005315','Water For Injection','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','pH',1,'90018','Ch. Satyam','2024-10-03 18:57:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   70.8 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(100,'2024-10-03','15:35:01','90.00000','105.00000','Purified water','pH','PW_SP05','E4RM24005317','Purified water','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-03 18:58:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.79 pH\r\nmV                   67.9 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                73\r\n\r\n\r\n'),
(101,'2024-10-03','15:40:48','90.00000','105.00000','Pure steam','pH','PSC_SP20','E4RM24005318','Pure steam Condensate','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','pH_Pure Steam',1,'90018','Ch. Satyam','2024-10-03 18:58:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.80 pH\r\nmV                   67.1 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(102,'2024-10-03','15:43:53','90.00000','105.00000','Purified water','pH','PW_SP23','E4RM24005319','Purified water','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-03 18:59:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.79 pH\r\nmV                   67.6 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(103,'2024-10-03','15:48:08','90.00000','105.00000','Water For Injection','pH','WFI_SP24','E4RM24005320','Water For Injection','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','pH',1,'90018','Ch. Satyam','2024-10-03 19:00:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.76 pH\r\nmV                   69.7 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(104,'2024-10-03','15:51:07','90.00000','105.00000','Water For Injection','pH','WFI_SP41','E4RM24005321','Water For Injection','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','pH',1,'90018','Ch. Satyam','2024-10-03 19:01:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.81 pH\r\nmV                   66.8 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(105,'2024-10-03','15:54:21','90.00000','105.00000','Water For Injection','pH','WFI_SP33','E4RM24005322','Water For Injection','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','pH',1,'90018','Ch. Satyam','2024-10-03 19:02:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.79 pH\r\nmV                   67.8 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                78\r\n\r\n\r\n'),
(106,'2024-10-03','15:58:45','90.00000','105.00000','Pure steam','pH','PSC_SP21','E4RM24005323','Pure steam Condensate','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','pH_Pure Steam',1,'90018','Ch. Satyam','2024-10-03 19:03:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.72 pH\r\nmV                   71.9 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(107,'2024-10-03','16:01:34','90.00000','105.00000','Purified water','pH','PW_SP17','E4RM24005324','Purified water','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-03 19:03:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.79 pH\r\nmV                   67.7 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\n\r\n\r\n'),
(108,'2024-10-03','16:04:23','90.00000','105.00000','Water For Injection','pH','WFI_SP18','E4RM24005325','Water For Injection','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','pH',1,'90018','Ch. Satyam','2024-10-03 19:04:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.79 pH\r\nmV                   68.1 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                81\r\n\r\n\r\n'),
(109,'2024-10-03','15:56:07','90.00000','105.00000','Soyabean casein digest broth for sterilty','NA','RS033/24007','E4SO24000603','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDBSIH',1,'205143','Ch. Prasad','2024-10-03 16:59:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.23 pH\r\nmV                   -13.6 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             30-09-24,16:12\r\nLog #                40\r\n\r\n\r\n'),
(110,'2024-10-03','16:31:26','90.00000','105.00000','Fluid thioglycollate meddiuam',' NA','RS026/24008','E4SO24000604','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','FTGMIH',1,'205143','Ch. Prasad','2024-10-03 16:59:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.23 pH\r\nmV                   -13.4 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             30-09-24,16:12\r\n\r\n\r\n'),
(111,'2024-10-03','16:43:03','90.00000','105.00000','Peptone','NA','RS007/24030','ESO24000602','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-03 16:59:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -1.6 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             30-09-24,16:12\r\nLog #                42\r\n\r\n\r\n'),
(112,'2024-10-03','16:46:07','90.00000','105.00000','Aseptic Media Fill Validation','pH','DP24007','E4AMV24000131','Finished Product','QC-PHMTZ-009','CX1-16015','1','NULL','205210','B. Anitha',2,0,102,'pH','pH  Test',1,'90018','Ch. Satyam','2024-10-03 19:13:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   7.33 pH\r\nmV                   -21.5 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                82\r\n\r\n\r\n'),
(113,'2024-10-03','16:53:28','90.00000','105.00000','Peptone','NA','RS007/24029','E4SO24000598','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-03 16:59:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.04 pH\r\nmV                   -2.2 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             30-09-24,16:12\r\n\r\n\r\n'),
(114,'2024-10-03','17:10:33','90.00000','105.00000','Tryptone soya agar TSA with 0.05%soya lecithin&0.5% tween 80 and beta lactamase 55mm plates(RODAC)','NA','SPU342','E4SO24000606','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','205207','P. Hanmanthu',2,0,103,'pH','RUTSA',1,'205143','Ch. Prasad','2024-10-05 14:05:16','For details refer PNC-EP4-MB-24-0015','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -1.2 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             30-09-24,16:12\r\n\r\n\r\n'),
(115,'2024-10-03','17:29:39','90.00000','105.00000','Tryptone soya agar TSA with 0.05%soya lecithin&0.5% tween 80 and beta lactamase 55mm plates(RODAC)','NA','SPU377','E4SO24000606','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','205207','P. Hanmanthu',2,0,103,'pH','RUTSA',1,'205143','Ch. Prasad','2024-10-05 14:06:00','For details refer PNC-EP4-MB-24-0015','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.12 pH\r\nmV                   -7.1 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             30-09-24,16:12\r\nLog #                45\r\n\r\n\r\n'),
(116,'2024-10-03','17:34:26','90.00000','105.00000','Tryptone soya agar TSA with 0.05%soya lecithin&0.5% tween 80 and beta lactamase 55mm plates(RODAC)','NA','SPU377','E4SO24000607','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','205207','P. Hanmanthu',2,0,103,'pH','RUTSA',1,'205143','Ch. Prasad','2024-10-05 14:06:33','For details refer PNC-EP4-MB-24-0015','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.12 pH\r\nmV                   -7.5 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             30-09-24,16:12\r\n\r\n\r\n'),
(117,'2024-10-03','17:36:30','90.00000','105.00000','Tryptone soya agar TSA with 0.05%soya lecithin&0.5% tween 80 and beta lactamase 55mm plates(RODAC)','NA','SPU342','E4SO24000608','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','205207','P. Hanmanthu',2,0,103,'pH','RUTSA',1,'205143','Ch. Prasad','2024-10-05 14:07:17','For details refer PNC-EP4-MB-24-0015','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.18 pH\r\nmV                   -10.9 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             30-09-24,16:12\r\n\r\n\r\n'),
(118,'2024-10-03','17:39:13','90.00000','105.00000','Soyabean casean digest agar SCDA With 0.05%soya lecithin 05% tween 80 and beta lactamase (Finger dab)','NA','MPT0494','E4SO24000609','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','205207','P. Hanmanthu',2,0,103,'pH','RUSCDAF',1,'205143','Ch. Prasad','2024-10-05 14:08:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.21 pH\r\nmV                   -12.6 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             30-09-24,16:12\r\n\r\n\r\n'),
(119,'2024-10-03','18:10:23','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','PH','TR0424002','E4PP24001576','BULK','QC-PHMTZ-009','CX1-16015','1','NULL','205075','G. Shyam Kumar',2,0,102,'pH','PIPERTAZ',1,'90018','Ch. Satyam','2024-10-03 18:51:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.80 pH\r\nmV                   9.2 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             03-10-24,07:44\r\nLog #                83\r\n\r\n\r\n'),
(120,'2024-10-03','19:37:38','90.00000','105.00000','Soyabean casein digest broth for sterilty','NA','RS033/24007','ESO24000603','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDBSIH',1,'205143','Ch. Prasad','2024-10-05 14:08:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.26 pH\r\nmV                   -15.4 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             30-09-24,16:12\r\n\r\n\r\n'),
(121,'2024-10-03','19:40:12','90.00000','105.00000','Fluid thioglycollate meddiuam','NA','RS026/24008','E4SO24000604','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','FTGMIH',1,'205143','Ch. Prasad','2024-10-05 14:09:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -0.8 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             30-09-24,16:12\r\n\r\n\r\n'),
(122,'2024-10-03','19:42:39','90.00000','105.00000','Peptone','NA','RS007/24030','E4SO24000602','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-05 14:10:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.10 pH\r\nmV                   -5.7 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        100.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_104\r\nLast Cal             30-09-24,16:12\r\n\r\n\r\n'),
(123,'2024-10-04','06:32:41','90.00000','105.00000','TAZOBACTAM','NA','TR0024003','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','1',1,'88629','Raghuram Reddy','2024-10-04 21:36:12','Reveiwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.63 pH\r\nmV                   137.3 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             04-10-24,03:34\r\n\r\n\r\n'),
(124,'2024-10-04','07:58:23','90.00000','105.00000','pH Buffer','1.68','na','E4INS24005219','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,1,102,'pH','1.68 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-04 08:31:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   1.71 pH\r\nmV                   311.5 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\nLog #                84\r\n\r\n\r\n'),
(125,'2024-10-04','08:09:33','90.00000','105.00000','pH Buffer','4.01','na','E4INS24005219','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,0,102,'pH','4.01pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-04 08:32:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   173.3 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\n\r\n\r\n'),
(126,'2024-10-04','08:13:20','90.00000','105.00000','pH Buffer','7.00','na','E4INS24005219','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,0,102,'pH','7.00pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-04 08:33:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -1.8 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\n\r\n\r\n'),
(127,'2024-10-04','08:15:45','90.00000','105.00000','pH Buffer','10.01','na','E4INS24005219','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,0,102,'pH','10.01 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-04 08:33:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   10.02 pH\r\nmV                   -168.9 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\n\r\n\r\n'),
(128,'2024-10-04','08:19:52','90.00000','105.00000','pH Buffer','12.46','na','E4INS24005219','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,0,102,'pH','12.46 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-04 08:34:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   12.48 pH\r\nmV                   -321.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\n\r\n\r\n'),
(129,'2024-10-04','09:41:22','90.00000','105.00000','Piperacillin and Tazobactam For injection','UOD BY HPLC','PT0424106','E4FP24000667','Finished Product','QC-PHMTZ-009','CX1-16015','1','NULL','205232','K. Narender Reddy',2,1,102,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-04 11:15:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   87.9 mV\r\nTemperature          27.2 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\n\r\n\r\n'),
(130,'2024-10-04','10:33:53','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005222','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-04 11:00:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   316.1 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\n\r\n\r\n'),
(131,'2024-10-04','10:36:22','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005222','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-04 11:00:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   179.5 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\nLog #                54\r\n\r\n\r\n'),
(132,'2024-10-04','10:40:28','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005222','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-04 11:01:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   4.4 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\n\r\n\r\n'),
(133,'2024-10-04','10:42:25','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005222','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-04 11:01:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -167.9 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\n\r\n\r\n'),
(134,'2024-10-04','10:44:22','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005222','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-04 11:01:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.46 pH\r\nmV                   -313.0 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\n\r\n\r\n'),
(135,'2024-10-04','10:44:10','90.00000','105.00000','Piperacillin and Tazobactam For injection','RS BY HPLC','PT0424106','E4FP24000667','Finished Product','QC-PHMTZ-009','CX1-16015','1','NULL','205058','C. Srinivasulu',2,0,102,'pH','Pipertazoinj',1,'205208','S. Siddhartha Kumar Reddy','2024-10-04 11:18:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.96 pH\r\nmV                   60.4 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\nLog #                90\r\n\r\n\r\n'),
(136,'2024-10-04','12:09:00','90.00000','105.00000','Soyabean casein digest agar','NA','RS022/24005','E4SO24000612','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDAIH',1,'205143','Ch. Prasad','2024-10-15 12:27:09','PNC NO-PNC-EP4-MB-24-0015','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.72 pH\r\nmV                   21.1 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\n\r\n\r\n'),
(137,'2024-10-04','12:07:18','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024003','E4PP24001580','BULK','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,0,102,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-10-04 12:23:45','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   4.70 pH\r\nmV                   133.8 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\nLog #                91\r\n\r\n\r\n'),
(138,'2024-10-04','12:27:05','90.00000','105.00000','PIPERACILLIN','NA','TR0024003','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-04 21:36:03','Reveiwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.38 pH\r\nmV                   38.0 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             04-10-24,03:34\r\n\r\n\r\n'),
(139,'2024-10-04','12:50:43','90.00000','105.00000','Soyabean casein digest agar','NA','RS022/24005','E4SO24000612','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SCDAIH',1,'205143','Ch. Prasad','2024-10-05 14:14:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.48 pH\r\nmV                   -22.5 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\n\r\n\r\n'),
(140,'2024-10-04','13:00:58','90.00000','105.00000','0.9% SALINE','NA','RS008/24010','E4SO24000611','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SALIH',1,'205143','Ch. Prasad','2024-10-15 12:45:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.36 pH\r\nmV                   42.2 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\n\r\n\r\n'),
(141,'2024-10-04','13:05:37','90.00000','105.00000','Sabourarud dextrose agar','NA','RS021/24005','E4SO24000613','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SDAIH',1,'205143','Ch. Prasad','2024-10-15 12:45:45','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   78.0 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\n\r\n\r\n'),
(142,'2024-10-04','13:12:12','90.00000','105.00000','Phosphate buffered saline','NA','RS014/24003','E4SO24000614','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','PBSIH',1,'205143','Ch. Prasad','2024-10-15 12:48:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.40 pH\r\nmV                   -18.2 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\n\r\n\r\n'),
(143,'2024-10-04','13:18:03','90.00000','105.00000','R2AIH AGAR','NA','RS020/24004','E4SO24000618','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','R2AIH',1,'205143','Ch. Prasad','2024-10-15 12:48:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.24 pH\r\nmV                   -8.8 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\n\r\n\r\n'),
(144,'2024-10-04','14:20:53','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024003','E4PP24001582','BULK','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,0,102,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-10-04 14:37:12','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.43 pH\r\nmV                   33.3 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\n\r\n\r\n'),
(145,'2024-10-04','14:45:28','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0424002','E4PP24001578','BULK','QC-PHMTZ-009','CX1-16015','1','NULL','88655','V. Prashanth Kumar',2,0,102,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-10-04 14:53:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.49 pH\r\nmV                   30.0 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\n\r\n\r\n'),
(146,'2024-10-04','16:03:42','90.00000','105.00000','Purified water','pH','PW_SP06','E4RM24005326','Purified water','QC-PHMTZ-009','CX1-16015','1','NULL','205090','M. Srikrishnudu',2,0,102,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-04 16:52:27','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.84 pH\r\nmV                   67.6 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\n\r\n\r\n'),
(147,'2024-10-04','16:08:56','90.00000','105.00000','Purified water','pH','PW_SP37','E4RM24005327','Purified water','QC-PHMTZ-009','CX1-16015','1','NULL','205090','M. Srikrishnudu',2,0,102,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-04 16:52:58','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   68.4 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\nLog #                95\r\n\r\n\r\n'),
(148,'2024-10-04','16:14:33','90.00000','105.00000','Water For Injection','pH','WFI_SP14','E4RM24005328','Water For Injection','QC-PHMTZ-009','CX1-16015','1','NULL','205090','M. Srikrishnudu',2,0,102,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-10-04 16:53:09','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.62 pH\r\nmV                   80.6 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\n\r\n\r\n'),
(149,'2024-10-04','16:18:13','90.00000','105.00000','Water For Injection','pH','WFI_SP35','E4RM24005329','Water For Injection','QC-PHMTZ-009','CX1-16015','1','NULL','205090','M. Srikrishnudu',2,0,102,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-10-04 16:53:36','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.61 pH\r\nmV                   81.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\nLog #                97\r\n\r\n\r\n'),
(150,'2024-10-04','16:24:04','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TR0024003','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','Siva Shankar Reddy',2,0,104,'pH','3',1,'88629','Raghuram Reddy','2024-10-04 16:25:21','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.42 pH\r\nmV                   35.8 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             04-10-24,03:34\r\n\r\n\r\n'),
(151,'2024-10-04','16:26:25','90.00000','105.00000','Purified water','pH','PW_SP34','E4RM24005330','Purified water','QC-PHMTZ-009','CX1-16015','1','NULL','205090','M. Srikrishnudu',2,0,102,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-04 16:53:49','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.78 pH\r\nmV                   71.6 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\n\r\n\r\n'),
(152,'2024-10-04','16:29:10','90.00000','105.00000','Purified water','pH','PW_SP36','E4RM24005331','Purified water','QC-PHMTZ-009','CX1-16015','1','NULL','205090','M. Srikrishnudu',2,0,102,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-04 16:54:02','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.76 pH\r\nmV                   72.4 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\nLog #                99\r\n\r\n\r\n'),
(153,'2024-10-04','16:34:17','90.00000','105.00000','Purified water','pH','PW_SP33','E4RM24005332','Purified water','QC-PHMTZ-009','CX1-16015','1','NULL','205090','M. Srikrishnudu',2,0,102,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-04 16:54:18','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.81 pH\r\nmV                   69.1 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\n\r\n\r\n'),
(154,'2024-10-04','16:36:49','90.00000','105.00000','Water For Injection','pH','WFI_SP40','E4RM24005333','Water For Injection','QC-PHMTZ-009','CX1-16015','1','NULL','205090','M. Srikrishnudu',2,0,102,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-10-04 16:54:10','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   86.3 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\nLog #                101\r\n\r\n\r\n'),
(155,'2024-10-04','16:45:24','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TR0024003','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','Siva Shankar Reddy',2,0,104,'pH','3',1,'88629','Raghuram Reddy','2024-10-04 21:35:47','Reveiwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.49 pH\r\nmV                   33.5 mV\r\nTemperature          23.1 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_107\r\nLast Cal             04-10-24,03:34\r\n\r\n\r\n'),
(156,'2024-10-04','16:46:00','90.00000','105.00000','Phosphate buffered saline','NA','RS014/24003','E4SO24000614','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','PBSIH',1,'205143','Ch. Prasad','2024-10-15 12:49:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.34 pH\r\nmV                   -14.8 mV\r\nTemperature          26.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\nLog #                67\r\n\r\n\r\n'),
(157,'2024-10-04','16:51:42','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024003','E4PP24001583','BULK','QC-PHMTZ-009','CX1-16015','1','NULL','205211','S. Sailatha',2,0,102,'pH','PIPERTAZ',1,'82735','M.D. Babar Qhadri','2024-10-04 17:03:31','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.77 pH\r\nmV                   13.3 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\nLog #                102\r\n\r\n\r\n'),
(158,'2024-10-04','16:56:44','90.00000','105.00000','0.9% SALINE','NA','RS008/24010','E4SO24000611','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SALIH',1,'205143','Ch. Prasad','2024-10-15 12:49:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.91 pH\r\nmV                   9.9 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\nLog #                68\r\n\r\n\r\n'),
(159,'2024-10-04','17:03:31','90.00000','105.00000','Soyabean casein digest agar','NA','RS022/24005','E4SO24000612','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SCDAIH',1,'205143','Ch. Prasad','2024-10-15 12:49:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.13 pH\r\nmV                   -4.5 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\n\r\n\r\n'),
(160,'2024-10-04','17:10:57','90.00000','105.00000','Sabourarud dextrose agar','NA','RS021/24005','E4SO24000613','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SDAIH',1,'205143','Ch. Prasad','2024-10-15 12:50:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   5.63 pH\r\nmV                   85.0 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\n\r\n\r\n'),
(161,'2024-10-04','17:21:13','90.00000','105.00000','R2AIH AGAR','NA','RS020/24004','	E4SO24000618','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','R2AIH',1,'205143','Ch. Prasad','2024-10-15 12:52:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.21 pH\r\nmV                   -7.2 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             04-10-24,10:27\r\nLog #                71\r\n\r\n\r\n'),
(162,'2024-10-04','18:03:25','90.00000','105.00000','Aseptic Media Fill Validation','PH','DP24007','E4AMV24000137','Finished Product','QC-PHMTZ-009','CX1-16015','1','NULL','89853','J.Ravi',2,0,102,'pH','pH  Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 09:36:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   7.40 pH\r\nmV                   -22.7 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             04-10-24,07:44\r\nLog #                103\r\n\r\n\r\n'),
(163,'2024-10-05','08:53:44','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005239','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,0,101,'pH','1.68 pH',1,'86253','D. Jayapal Reddy','2024-10-05 09:19:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.67 pH\r\nmV                   309.9 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(164,'2024-10-05','08:55:41','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005239','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,0,101,'pH','4.01pH',1,'86253','D. Jayapal Reddy','2024-10-05 09:21:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   171.6 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(165,'2024-10-05','08:57:11','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005239','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,0,101,'pH','7.00pH',1,'86253','D. Jayapal Reddy','2024-10-05 09:22:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   -2.8 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(166,'2024-10-05','08:58:39','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005239','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,0,101,'pH','10.01 pH',1,'86253','D. Jayapal Reddy','2024-10-05 09:23:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.02 pH\r\nmV                   -174.6 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(167,'2024-10-05','09:00:31','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005239','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,0,101,'pH','12.46 pH',1,'86253','D. Jayapal Reddy','2024-10-05 09:23:59','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.45 pH\r\nmV                   -320.1 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(168,'2024-10-05','09:42:55','90.00000','105.00000','Penicillin G Benzathine','DISSOLUTION','PG0124003A','E4STFP24000570','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','89180','R. Harish Reddy',2,0,101,'pH','Penicillin G',1,'86253','D. Jayapal Reddy','2024-10-05 09:58:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.53 pH\r\nmV                   -32.8 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(169,'2024-10-05','09:54:50','90.00000','105.00000','Piperacillin and Tazobactam For injection','UOD','PT0424108','E4FP24000671','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,0,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-10-05 10:08:13','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   84.3 mV\r\nTemperature          29.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\nLog #                39\r\n\r\n\r\n'),
(170,'2024-10-05','10:07:09','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','ASSAY','TR0024003','E4PP24001584','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205119','S.M. Subhani',2,0,101,'pH','PIPERTAZ',1,'86253','D. Jayapal Reddy','2024-10-05 11:47:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   84.1 mV\r\nTemperature          28.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(171,'2024-10-05','10:22:33','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS BY HPLC MP A','TZ0424036A','E4STFP24000616','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,1,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-10-05 10:59:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   82.8 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\nLog #                41\r\n\r\n\r\n'),
(172,'2024-10-05','10:36:39','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS BH HPLB MP B','TZ0424036A','E4STFP24000616','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,1,101,'pH','PIPERTAZO',1,'86253','D. Jayapal Reddy','2024-10-05 11:00:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   83.6 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\nLog #                42\r\n\r\n\r\n'),
(173,'2024-10-05','10:43:40','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005244','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-05 10:54:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.70 pH\r\nmV                   318.5 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             05-10-24,10:39\r\n\r\n\r\n'),
(174,'2024-10-05','10:45:26','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005244','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-05 10:55:10','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   179.8 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             05-10-24,10:39\r\n\r\n\r\n'),
(175,'2024-10-05','10:47:22','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005244','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-05 10:55:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   4.6 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             05-10-24,10:39\r\n\r\n\r\n'),
(176,'2024-10-05','10:49:12','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005244','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-05 10:55:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -168.1 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             05-10-24,10:39\r\n\r\n\r\n'),
(177,'2024-10-05','10:51:05','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005244','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-05 10:56:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.46 pH\r\nmV                   -313.7 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             05-10-24,10:39\r\n\r\n\r\n'),
(178,'2024-10-05','11:36:31','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024003','E4PP24001585','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,0,101,'pH','PIPERTAZ',1,'86253','D. Jayapal Reddy','2024-10-05 12:20:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.42 pH\r\nmV                   31.3 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(179,'2024-10-05','11:39:29','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024003','E4PP24001586','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,0,101,'pH','PIPERTAZ',1,'86253','D. Jayapal Reddy','2024-10-05 12:20:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.45 pH\r\nmV                   29.0 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(180,'2024-10-05','11:41:42','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0024003','E4PP24001588','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,0,101,'pH','PIPERTAZ',1,'86253','D. Jayapal Reddy','2024-10-05 12:20:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.39 pH\r\nmV                   31.8 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(181,'2024-10-05','11:51:40','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0424003','E4PP24001592','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,0,101,'pH','PIPERTAZ',1,'86253','D. Jayapal Reddy','2024-10-05 12:20:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.34 pH\r\nmV                   35.8 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(182,'2024-10-05','11:54:45','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0424003','E4PP24001593','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,0,101,'pH','PIPERTAZ',1,'86253','D. Jayapal Reddy','2024-10-05 12:21:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.50 pH\r\nmV                   26.3 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(183,'2024-10-05','14:14:20','90.00000','105.00000','Purified water','pH','PW_SP06','E4RM24005334','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 16:03:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   72.9 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(184,'2024-10-05','14:17:15','90.00000','105.00000','Purified water','pH','PW_SP37','E4RM24005335','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 16:04:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.64 pH\r\nmV                   75.9 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(185,'2024-10-05','14:20:31','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24013','E4SO24000623','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','SCDBIH',1,'205143','Ch. Prasad','2024-10-05 17:24:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.31 pH\r\nmV                   -13.4 mV\r\nTemperature          26.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             05-10-24,10:39\r\nLog #                77\r\n\r\n\r\n'),
(186,'2024-10-05','14:20:12','90.00000','105.00000','Water For Injection','pH','WFI_SP14','E4RM24005336','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 16:05:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.73 pH\r\nmV                   71.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\nLog #                50\r\n\r\n\r\n'),
(187,'2024-10-05','14:25:07','90.00000','105.00000','Water For Injection','pH','WFI_SP35','E4RM24005337','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 16:05:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   70.2 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(188,'2024-10-05','14:30:35','90.00000','105.00000','Dey-engley neutralizer broth','NA','RS024/24003','E4SO24000624','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','DEBIH',1,'205143','Ch. Prasad','2024-10-05 17:25:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.59 pH\r\nmV                   -29.7 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             05-10-24,10:39\r\nLog #                78\r\n\r\n\r\n'),
(189,'2024-10-05','14:30:31','90.00000','105.00000','Water For Injection','pH','WFI_SP34','E4RM24005338','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 16:06:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.72 pH\r\nmV                   71.6 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\nLog #                52\r\n\r\n\r\n'),
(190,'2024-10-05','14:35:41','90.00000','105.00000','Purified water','pH','PW_SP20','E4RM24005339','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH Test',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 16:07:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.65 pH\r\nmV                   76.4 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(191,'2024-10-05','14:38:40','90.00000','105.00000','Water For Injection','pH','WFI_SP21','E4RM24005340','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 16:07:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.63 pH\r\nmV                   72.0 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(192,'2024-10-05','14:46:29','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005236','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205210','B. Anitha',2,0,102,'pH','1.68 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 16:17:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   311.6 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             05-10-24,14:43\r\nLog #                104\r\n\r\n\r\n'),
(193,'2024-10-05','14:53:51','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005236','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205210','B. Anitha',2,0,102,'pH','4.01pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 16:17:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   172.6 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             05-10-24,14:43\r\n\r\n\r\n'),
(194,'2024-10-05','14:56:01','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005236','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205210','B. Anitha',2,0,102,'pH','7.00pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 16:18:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.99 pH\r\nmV                   -1.6 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             05-10-24,14:43\r\nLog #                106\r\n\r\n\r\n'),
(195,'2024-10-05','14:59:01','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005236','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205210','B. Anitha',2,0,102,'pH','10.01 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 16:18:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   10.00 pH\r\nmV                   -172.3 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             05-10-24,14:43\r\n\r\n\r\n'),
(196,'2024-10-05','15:02:22','90.00000','105.00000','pH Buffer','verification','NA','E4INS24005236','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205210','B. Anitha',2,0,102,'pH','12.46 pH',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 16:19:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   12.43 pH\r\nmV                   -320.7 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             05-10-24,14:43\r\n\r\n\r\n'),
(197,'2024-10-05','15:01:57','90.00000','105.00000','Floxapen powder for Solution for Injection','ASSAY BY HPLC','FP0224002A','E4FP24000675','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,1,101,'pH','Floxapen',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 16:10:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.99 pH\r\nmV                   115.0 mV\r\nTemperature          27.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\nLog #                55\r\n\r\n\r\n'),
(198,'2024-10-05','15:17:50','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM FOR INJECTION (PPQ)','pH','TR0424003','E4PP24001595','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,0,101,'pH','PIPERTAZ',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 15:38:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.41 pH\r\nmV                   31.5 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(199,'2024-10-05','15:52:11','90.00000','105.00000','potable water pH','pH','PR_SP11','E4RM24005341','potable water ','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,0,101,'pH','potable water',1,'205208','S. Siddhartha Kumar Reddy','2024-10-05 16:09:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -3.6 mV\r\nTemperature          23.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             05-10-24,08:47\r\n\r\n\r\n'),
(200,'2024-10-05','17:05:07','90.00000','105.00000','Soyabean casein digest agar (SCDA)with beta-lactamase','NA','RS040/24004','E4SO24000626','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','RUSCDA',1,'205143','Ch. Prasad','2024-10-05 17:25:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.40 pH\r\nmV                   -18.4 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             05-10-24,10:39\r\n\r\n\r\n'),
(201,'2024-10-05','17:10:08','90.00000','105.00000','Soyabean casean digest agar SCDA With 0.05%soya lecithin 05% tween 80 and beta lactamase (Finger dab)','NA','RS041/24005','E4SO24000625','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','RUSCDAF',1,'205143','Ch. Prasad','2024-10-05 17:26:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.18 pH\r\nmV                   -5.6 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             05-10-24,10:39\r\n\r\n\r\n'),
(202,'2024-10-05','20:41:06','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24013','E4SO24000623','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','SCDBIH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.29 pH\r\nmV                   -12.0 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             05-10-24,10:39\r\n\r\n\r\n'),
(203,'2024-10-05','20:47:38','90.00000','105.00000','Dey-engley neutralizer broth','NA','RS024/24003','E4SO24000624','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','DEBIH',1,'205143','Ch. Prasad','2024-10-15 12:54:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.56 pH\r\nmV                   -27.6 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             05-10-24,10:39\r\n\r\n\r\n'),
(204,'2024-10-05','23:12:35','90.00000','105.00000','TAZOBACTAM','NA','TZ0024091','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','1',1,'95681','T. Sivanarayana','2024-10-05 23:44:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.70 pH\r\nmV                   111.4 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        99.5 %\r\nMethod               100\r\nCalLog #             PH_CAL_108\r\nLast Cal             05-10-24,12:24\r\n\r\n\r\n'),
(205,'2024-10-06','04:49:39','90.00000','105.00000','PIPERACILLIN','NA','TZ0024091','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-06 06:22:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.95 pH\r\nmV                   43.9 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        100.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             06-10-24,00:39\r\n\r\n\r\n'),
(206,'2024-10-06','05:43:09','90.00000','105.00000','PIPERACILLIN','NA','TZ0024091','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-06 06:24:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.09 pH\r\nmV                   35.4 mV\r\nTemperature          25.2 C (ATC)\r\nAverage Slope        100.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             06-10-24,00:39\r\n\r\n\r\n'),
(207,'2024-10-06','06:31:21','90.00000','105.00000','PIPERACILLIN','NA','TZ0024091','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205190','B. Prabhudas',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-06 07:28:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.18 pH\r\nmV                   30.3 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        100.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             06-10-24,00:39\r\n\r\n\r\n'),
(208,'2024-10-06','07:20:02','90.00000','105.00000','PIPERACILLIN','NA','TZ0024091','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','2',1,'88629','Raghuram Reddy','2024-10-06 07:28:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.47 pH\r\nmV                   13.4 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        100.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             06-10-24,00:39\r\n\r\n\r\n'),
(209,'2024-10-06','08:53:12','90.00000','105.00000','pH Buffer','NA','167723H1','E4INS24005250','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,1,102,'pH','1.68 pH',1,'90018','Ch. Satyam','2024-10-06 09:53:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   311.0 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-10-24,08:43\r\n\r\n\r\n'),
(210,'2024-10-06','08:57:24','90.00000','105.00000','pH Buffer','NA','423L1','E4INS24005250','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,1,102,'pH','4.01pH',1,'90018','Ch. Satyam','2024-10-06 09:53:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   3.99 pH\r\nmV                   174.6 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-10-24,08:43\r\n\r\n\r\n'),
(211,'2024-10-06','09:00:12','90.00000','105.00000','pH Buffer','NA','72523G1','E4INS24005250','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,1,102,'pH','7.00pH',1,'90018','Ch. Satyam','2024-10-06 09:53:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.98 pH\r\nmV                   -1.3 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-10-24,08:43\r\nLog #                111\r\n\r\n\r\n'),
(212,'2024-10-06','09:06:44','90.00000','105.00000','pH Buffer','NA','116223D1','E4INS24005250','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,1,102,'pH','10.01 pH',1,'90018','Ch. Satyam','2024-10-06 09:53:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -173.5 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-10-24,08:43\r\n\r\n\r\n'),
(213,'2024-10-06','09:09:35','90.00000','105.00000','pH Buffer','NA','112454R23G1','E4INS24005250','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,1,102,'pH','12.46 pH',1,'90018','Ch. Satyam','2024-10-06 09:53:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -322.8 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-10-24,08:43\r\n\r\n\r\n'),
(214,'2024-10-06','09:42:20','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024091','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','3',1,'88629','Raghuram Reddy','2024-10-06 09:48:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.25 pH\r\nmV                   25.9 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        100.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             06-10-24,00:39\r\n\r\n\r\n'),
(215,'2024-10-06','09:55:19','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024091','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','3',1,'88629','Raghuram Reddy','2024-10-06 10:12:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   24.4 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        100.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_109\r\nLast Cal             06-10-24,00:39\r\n\r\n\r\n'),
(216,'2024-10-06','12:34:54','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH test','TZ0424086','E4FP24000668','Finished Product','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,1,102,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-10-06 14:18:40','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.92 pH\r\nmV                   61.0 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-10-24,08:43\r\n\r\n\r\n'),
(217,'2024-10-06','12:40:18','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH Test','TZ0424087','E4FP24000670','Finished Product','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,1,102,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-10-06 14:18:59','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   5.87 pH\r\nmV                   63.7 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-10-24,08:43\r\n\r\n\r\n'),
(218,'2024-10-06','14:02:55','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','pH Test','TZ0024091','E4IP24001225','BULK','QC-PHMTZ-009','CX1-16015','1','NULL','205187','B. Ramvikas',2,1,102,'pH','3',1,'82735','M.D. Babar Qhadri','2024-10-06 14:19:20','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   6.11 pH\r\nmV                   49.5 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        99.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             06-10-24,08:43\r\n\r\n\r\n'),
(219,'2024-10-07','07:11:22','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005252','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','1.68 pH',1,'90018','Ch. Satyam','2024-10-07 07:52:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   1.66 pH\r\nmV                   310.4 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-10-24,07:08\r\n\r\n\r\n'),
(220,'2024-10-07','07:13:53','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005252','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','4.01pH',1,'90018','Ch. Satyam','2024-10-07 07:52:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   174.3 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-10-24,07:08\r\n\r\n\r\n'),
(221,'2024-10-07','07:16:12','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005252','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','7.00pH',1,'90018','Ch. Satyam','2024-10-07 07:52:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -1.5 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-10-24,07:08\r\n\r\n\r\n'),
(222,'2024-10-07','07:18:19','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005252','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','10.01 pH',1,'90018','Ch. Satyam','2024-10-07 07:52:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -173.2 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-10-24,07:08\r\n\r\n\r\n'),
(223,'2024-10-07','07:26:35','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005252','pH Buffer','QC-PHMTZ-009','CX1-16015','1','NULL','205089','M. Venkata Ramana',2,0,102,'pH','12.46 pH',1,'90018','Ch. Satyam','2024-10-07 07:52:26','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   12.49 pH\r\nmV                   -321.5 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-10-24,07:08\r\nLog #                121\r\n\r\n\r\n'),
(224,'2024-10-07','08:28:27','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005255','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','1.68 pH',1,'90018','Ch. Satyam','2024-10-07 09:08:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.70 pH\r\nmV                   308.6 mV\r\nTemperature          23.2 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,08:25\r\nLog #                58\r\n\r\n\r\n'),
(225,'2024-10-07','08:30:38','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005255','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','4.01pH',1,'90018','Ch. Satyam','2024-10-07 09:08:51','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   174.2 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,08:25\r\nLog #                59\r\n\r\n\r\n'),
(226,'2024-10-07','08:32:45','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005255','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','7.00pH',1,'90018','Ch. Satyam','2024-10-07 09:09:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -2.0 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,08:25\r\n\r\n\r\n'),
(227,'2024-10-07','08:34:59','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005255','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','10.01 pH',1,'90018','Ch. Satyam','2024-10-07 09:09:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.03 pH\r\nmV                   -172.0 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,08:25\r\n\r\n\r\n'),
(228,'2024-10-07','08:37:05','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005255','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,1,101,'pH','12.46 pH',1,'90018','Ch. Satyam','2024-10-07 09:09:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.46 pH\r\nmV                   -319.8 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,08:25\r\n\r\n\r\n'),
(229,'2024-10-07','08:55:01','90.00000','105.00000','Penicillin G Benzathine','dissolution','PG0124001A','E4STFP24000552','Stability Study','QC-PHMTZ-009','CX1-16015','1','NULL','89180','R. Harish Reddy',2,0,102,'pH','Penicillin G',1,'90018','Ch. Satyam','2024-10-07 09:58:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   3.26 pH\r\nmV                   217.8 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-10-24,07:08\r\n\r\n\r\n'),
(230,'2024-10-07','09:00:07','90.00000','105.00000','Penicillin G Benzathine','dissolution','PG0124001A','E4STFP24000552','Stability Study','QC-PHMTZ-009','CX1-16015','1','NULL','89180','R. Harish Reddy',2,0,102,'pH','Penicillin G',1,'90018','Ch. Satyam','2024-10-07 09:59:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   7.48 pH\r\nmV                   -27.7 mV\r\nTemperature          26.8 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-10-24,07:08\r\nLog #                123\r\n\r\n\r\n'),
(231,'2024-10-07','11:31:35','90.00000','105.00000','Nafcillin For Injection','NA','NF1022001A','E4STFP24000606','Stability Study','QC-PHMTZ-009','CX1-16015','1','NULL','90017','G. Srishailam',2,1,102,'pH','Assay Test',1,'90018','Ch. Satyam','2024-10-07 12:06:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   7.51 pH\r\nmV                   -29.4 mV\r\nTemperature          28.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-10-24,07:08\r\nLog #                124\r\n\r\n\r\n'),
(232,'2024-10-07','11:39:28','90.00000','105.00000','Nafcillin For Injection','NA','NF1022001A','E4STFP24000606','Stability Study','QC-PHMTZ-009','CX1-16015','1','NULL','90017','G. Srishailam',2,1,102,'pH','Assay Test',1,'90018','Ch. Satyam','2024-10-07 12:07:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122318\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA32174\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -0.4 mV\r\nTemperature          28.6 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-10-24,07:08\r\nLog #                125\r\n\r\n\r\n'),
(233,'2024-10-07','11:50:50','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005266','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-07 15:27:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   316.9 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-10-24,11:47\r\n\r\n\r\n'),
(234,'2024-10-07','11:53:00','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005266','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-07 15:32:53','For Details refer PNC No- PNC-EP4-MB-24-0015','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.61 pH\r\nmV                   143.2 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-10-24,11:47\r\n\r\n\r\n'),
(235,'2024-10-07','12:11:23','90.00000','105.00000','Purified water','pH','PW_SP06','E4RM24005355','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-07 13:59:42','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.81 pH\r\nmV                   68.8 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,08:25\r\nLog #                63\r\n\r\n\r\n'),
(236,'2024-10-07','12:16:06','90.00000','105.00000','Purified water','pH','PW_SP37','E4RM24005356','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-07 14:00:00','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.86 pH\r\nmV                   66.4 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,08:25\r\n\r\n\r\n'),
(237,'2024-10-07','12:14:00','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005266','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-07 15:33:48','PNC No- PNC-EP4-MB-24-0015','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   178.5 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-10-24,11:47\r\nLog #                85\r\n\r\n\r\n'),
(238,'2024-10-07','12:21:30','90.00000','105.00000','Water For Injection','pH','WFI_SP14','E4RM24005357','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-10-07 14:00:25','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.89 pH\r\nmV                   63.7 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,08:25\r\nLog #                66\r\n\r\n\r\n'),
(239,'2024-10-07','12:23:12','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005266','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-07 15:34:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.98 pH\r\nmV                   3.3 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-10-24,11:47\r\nLog #                86\r\n\r\n\r\n'),
(240,'2024-10-07','12:25:30','90.00000','105.00000','Water For Injection','pH','WFI_SP35','E4RM24005358','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'82735','M.D. Babar Qhadri','2024-10-07 14:00:45','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.85 pH\r\nmV                   64.9 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,08:25\r\n\r\n\r\n'),
(241,'2024-10-07','12:28:17','90.00000','105.00000','Purified water','pH','PW_SP03','E4RM24005359','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'82735','M.D. Babar Qhadri','2024-10-07 14:01:09','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.82 pH\r\nmV                   59.4 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,08:25\r\n\r\n\r\n'),
(242,'2024-10-07','12:30:36','90.00000','105.00000','Pure steam','pH','PSC_SP09','E4RM24005360','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'90018','Ch. Satyam','2024-10-07 14:53:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.91 pH\r\nmV                   62.3 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,08:25\r\n\r\n\r\n'),
(243,'2024-10-07','12:34:46','90.00000','105.00000','TAZOBACTAM','NA','TZ0024092','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205223','Jithendra',2,0,104,'pH','1',1,'88629','Raghuram Reddy','2024-10-07 12:39:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.51 pH\r\nmV                   124.1 mV\r\nTemperature          23.0 C (ATC)\r\nAverage Slope        100.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,07:41\r\n\r\n\r\n'),
(244,'2024-10-07','12:32:43','90.00000','105.00000','Purified water','pH','PW_SP38','E4RM24005361','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'90018','Ch. Satyam','2024-10-07 14:53:55','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.86 pH\r\nmV                   65.4 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,08:25\r\nLog #                70\r\n\r\n\r\n'),
(245,'2024-10-07','12:37:55','90.00000','105.00000','Pure steam','pH','PSC_SP07','E4RM24005362','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'90018','Ch. Satyam','2024-10-07 14:54:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.90 pH\r\nmV                   63.7 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,08:25\r\n\r\n\r\n'),
(246,'2024-10-07','12:32:27','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005266','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-07 15:35:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.04 pH\r\nmV                   -173.7 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-10-24,11:47\r\nLog #                87\r\n\r\n\r\n'),
(247,'2024-10-07','12:38:29','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005266','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-07 15:35:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.48 pH\r\nmV                   -317.1 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             07-10-24,11:47\r\nLog #                88\r\n\r\n\r\n'),
(248,'2024-10-07','15:47:23','90.00000','105.00000','PIPERACILLIN','NA','TZ0024092','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','Siva Shankar Reddy',2,0,104,'pH','2',1,'84472','G. Ashok','2024-10-07 18:52:42','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.79 pH\r\nmV                   51.9 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        100.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,07:41\r\n\r\n\r\n'),
(249,'2024-10-07','16:15:29','90.00000','105.00000','Sensor Calibration','NA','NA','E4INS24005268','Temperature Sensor','QC-PHMTZ-007','CX1-15733','1','NULL','87624','N. Nani babu',2,1,101,'pH','Purified Water',1,'82735','M.D. Babar Qhadri','2024-10-07 16:39:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.43 pH\r\nmV                   31.7 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        99.4 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,08:25\r\nLog #                72\r\n\r\n\r\n'),
(250,'2024-10-07','16:43:37','90.00000','105.00000','PIPERACILLIN','NA','TZ0024092','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','Siva Shankar Reddy',2,0,104,'pH','2',1,'84472','G. Ashok','2024-10-07 18:53:14','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   49.7 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        100.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,07:41\r\n\r\n\r\n'),
(251,'2024-10-07','17:38:30','90.00000','105.00000','PIPERACILLIN','NA','TZ0024092','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','Siva Shankar Reddy',2,0,104,'pH','2',1,'84472','G. Ashok','2024-10-07 18:53:36','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.14 pH\r\nmV                   31.7 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        100.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,07:41\r\n\r\n\r\n'),
(252,'2024-10-07','18:04:05','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005266','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-07 18:11:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.64 pH\r\nmV                   314.4 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             07-10-24,17:59\r\n\r\n\r\n'),
(253,'2024-10-07','18:05:36','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005266','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-07 18:11:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   177.2 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             07-10-24,17:59\r\n\r\n\r\n'),
(254,'2024-10-07','18:07:09','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005266','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-07 18:11:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.00 pH\r\nmV                   2.0 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             07-10-24,17:59\r\n\r\n\r\n'),
(255,'2024-10-07','18:09:15','90.00000','105.00000','pH Buffer','NA','pH  Verification','E4INS24005266','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-07 18:12:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   9.99 pH\r\nmV                   -173.7 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             07-10-24,17:59\r\n\r\n\r\n'),
(256,'2024-10-07','18:12:59','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005266','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-07 18:14:56','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.43 pH\r\nmV                   -316.5 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             07-10-24,17:59\r\n\r\n\r\n'),
(257,'2024-10-07','18:18:15','90.00000','105.00000','PIPERACILLIN','NA','TZ0024092','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','Siva Shankar Reddy',2,0,104,'pH','2',1,'84472','G. Ashok','2024-10-07 18:53:56','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.42 pH\r\nmV                   16.0 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        100.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,07:41\r\n\r\n\r\n'),
(258,'2024-10-07','19:28:26','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24015','E4SO24000635','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','SCDBIH',1,'205143','Ch. Prasad','2024-10-08 11:45:27','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.26 pH\r\nmV                   -13.0 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             07-10-24,17:59\r\n\r\n\r\n'),
(259,'2024-10-07','19:32:06','90.00000','105.00000','Nacl with newttralizers','NA','RS023/24002','E4SO24000636','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','NSALIH',1,'205143','Ch. Prasad','2024-10-08 11:46:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   1.5 mV\r\nTemperature          26.9 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             07-10-24,17:59\r\n\r\n\r\n'),
(260,'2024-10-07','19:34:44','90.00000','105.00000','SCDM WITH 4%Tween80 &0.5 Soya lecthin','NA','RS011/24007','E4SO2400637','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','NSCDBIH',1,'205143','Ch. Prasad','2024-10-08 11:47:23','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.25 pH\r\nmV                   -12.5 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             07-10-24,17:59\r\n\r\n\r\n'),
(261,'2024-10-07','19:37:15','90.00000','105.00000','SCDM with 1% Tween 80','NA','RS034/24006','E4SO24000638','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','NSSCDBSIH',1,'205143','Ch. Prasad','2024-10-08 12:07:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.30 pH\r\nmV                   -15.7 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             07-10-24,17:59\r\n\r\n\r\n'),
(262,'2024-10-07','20:10:16','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024092','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','Siva Shankar Reddy',2,0,104,'pH','3',1,'84472','G. Ashok','2024-10-07 22:13:05','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.32 pH\r\nmV                   21.7 mV\r\nTemperature          26.1 C (ATC)\r\nAverage Slope        100.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,07:41\r\n\r\n\r\n'),
(263,'2024-10-07','20:21:44','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024092','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205104','Siva Shankar Reddy',2,0,104,'pH','3',1,'84472','G. Ashok','2024-10-07 22:13:36','reviwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.34 pH\r\nmV                   20.8 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        100.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_110\r\nLast Cal             07-10-24,07:41\r\n\r\n\r\n'),
(264,'2024-10-07','21:50:39','90.00000','105.00000','Soyabean casein digest broth','NA','RS009/24015','E4SO24000635','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','SCDBIH',1,'205143','Ch. Prasad','2024-10-08 11:48:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.26 pH\r\nmV                   -13.1 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             07-10-24,17:59\r\n\r\n\r\n'),
(265,'2024-10-07','21:53:30','90.00000','105.00000','Nacl with newttralizers','NA','RS023/24002','E4SO24000636','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','NSALIH',1,'205143','Ch. Prasad','2024-10-08 12:08:09','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   0.6 mV\r\nTemperature          26.7 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             07-10-24,17:59\r\n\r\n\r\n'),
(266,'2024-10-07','21:58:16','90.00000','105.00000','SCDM WITH 4%Tween80 &0.5 Soya lecthin','NA','RS011/24007','E4SO24000637','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','NSCDBIH',1,'205143','Ch. Prasad','2024-10-08 12:08:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.29 pH\r\nmV                   -14.7 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             07-10-24,17:59\r\n\r\n\r\n'),
(267,'2024-10-07','22:03:48','90.00000','105.00000','SCDM with 1% Tween 80','NA','RS034/24006','E4SO24000638','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','NSSCDBSIH',1,'205143','Ch. Prasad','2024-10-08 11:50:11','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.30 pH\r\nmV                   -15.6 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        99.0 %\r\nMethod               100\r\nCalLog #             PH_CAL_115\r\nLast Cal             07-10-24,17:59\r\n\r\n\r\n'),
(268,'2024-10-08','07:19:22','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005273','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,0,101,'pH','1.68 pH',1,'90018','Ch. Satyam','2024-10-08 08:11:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   310.3 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(269,'2024-10-08','07:22:08','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005273','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,0,101,'pH','4.01pH',1,'90018','Ch. Satyam','2024-10-08 08:11:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   173.9 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(270,'2024-10-08','07:25:34','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005273','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,0,101,'pH','7.00pH',1,'90018','Ch. Satyam','2024-10-08 08:11:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -2.0 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(271,'2024-10-08','07:39:00','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005273','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,0,101,'pH','10.01 pH',1,'90018','Ch. Satyam','2024-10-08 08:11:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   10.00 pH\r\nmV                   -175.6 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(272,'2024-10-08','07:56:41','90.00000','105.00000','pH Buffer','Verification','NA','E4INS24005273','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205089','M. Venkata Ramana',2,0,101,'pH','12.46 pH',1,'90018','Ch. Satyam','2024-10-08 08:12:03','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -321.8 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(273,'2024-10-08','08:57:17','90.00000','105.00000','Piperacillin and Tazobactam for Injection','PIPERATAZO','TI0323002A','E4STFP24000596','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','80903','D. naga Prasad',2,1,101,'pH','PIPERTAZO',1,'90018','Ch. Satyam','2024-10-08 09:11:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.03 pH\r\nmV                   56.6 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(274,'2024-10-08','10:11:46','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005288','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-08 10:33:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   312.5 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             08-10-24,10:09\r\nLog #                102\r\n\r\n\r\n'),
(275,'2024-10-08','10:15:30','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005288','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-08 10:33:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   175.5 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             08-10-24,10:09\r\n\r\n\r\n'),
(276,'2024-10-08','10:17:13','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005288','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-08 10:33:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -0.9 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             08-10-24,10:09\r\nLog #                104\r\n\r\n\r\n'),
(277,'2024-10-08','10:20:19','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005288','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-08 10:34:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.00 pH\r\nmV                   -176.1 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             08-10-24,10:09\r\n\r\n\r\n'),
(278,'2024-10-08','10:26:39','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005288','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-08 10:34:28','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.43 pH\r\nmV                   -319.7 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             08-10-24,10:09\r\n\r\n\r\n'),
(279,'2024-10-08','10:27:07','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS BY HPLC MP A','PA0423014A','E4STFP24000555','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,1,101,'pH','PIPERTAZO',1,'90018','Ch. Satyam','2024-10-08 11:22:12','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   85.9 mV\r\nTemperature          25.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\nLog #                80\r\n\r\n\r\n'),
(280,'2024-10-08','10:37:05','90.00000','105.00000','Piperacillin and Tazobactam for Injection','RS BY HPLC MP B','PA0423014A','E4STFP24000555','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','87697','M. Raghavendr Reddy',2,1,101,'pH','PIPERTAZO',1,'90018','Ch. Satyam','2024-10-08 11:23:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   84.9 mV\r\nTemperature          23.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\nLog #                81\r\n\r\n\r\n'),
(281,'2024-10-08','10:49:10','90.00000','105.00000','Floxapen powder for Solution for Injection','RS','FP0124013','E4FP24000664','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205043','T.V. Subbaiah',2,1,101,'pH','Floxapen',1,'90018','Ch. Satyam','2024-10-08 11:25:18','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.99 pH\r\nmV                   119.0 mV\r\nTemperature          29.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\nLog #                82\r\n\r\n\r\n'),
(282,'2024-10-08','11:08:40','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0024092','EIP24001230','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','pH (Bulk)',1,'90018','Ch. Satyam','2024-10-08 11:27:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.05 pH\r\nmV                   55.0 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(283,'2024-10-08','11:30:37','90.00000','105.00000','Tazobactam USP','ASSAY BY HPLC','A2403A020','E4AQR24000013','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205189','P. Ravi Kumar',2,1,101,'pH','Assay',1,'90018','Ch. Satyam','2024-10-08 11:40:49','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.53 pH\r\nmV                   88.2 mV\r\nTemperature          35.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(284,'2024-10-08','12:17:42','90.00000','105.00000','Peptone','NA','RS007/24031','E4SO24000641','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-09 10:06:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -0.3 mV\r\nTemperature          23.5 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             08-10-24,10:09\r\nLog #                107\r\n\r\n\r\n'),
(285,'2024-10-08','12:22:40','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0223002A','E4STFP24000623','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-08 15:20:44','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.79 pH\r\nmV                   70.4 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(286,'2024-10-08','12:26:56','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0223003A','E4STFP24000624','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-08 15:22:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.54 pH\r\nmV                   85.1 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(287,'2024-10-08','12:30:57','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0324059A','E4STFP24000628','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-08 15:25:01','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.68 pH\r\nmV                   76.8 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(288,'2024-10-08','12:34:08','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0324059A','E4STFP24000629','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-08 15:25:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.86 pH\r\nmV                   66.6 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(289,'2024-10-08','13:19:50','90.00000','105.00000','Purified water','pH','PW_SP06','E4RM24005365','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-10-08 15:00:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.82 pH\r\nmV                   68.0 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(290,'2024-10-08','13:22:08','90.00000','105.00000','Purified water','pH','PW_SP37','E4RM24005366','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-10-08 15:02:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.79 pH\r\nmV                   69.5 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(291,'2024-10-08','13:25:09','90.00000','105.00000','Water For Injection','pH','WFI_SP14','E4RM24005367','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-10-08 15:03:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   76.4 mV\r\nTemperature          24.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(292,'2024-10-08','13:27:19','90.00000','105.00000','Water For Injection','pH','WFI_SP35','E4RM24005368','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-10-08 15:10:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   75.8 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(293,'2024-10-08','13:29:10','90.00000','105.00000','Water For Injection','pH','WFI_SP13','E4RM24005369','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-10-08 15:11:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   75.4 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(294,'2024-10-08','13:31:02','90.00000','105.00000','Pure steam','pH','PSC_SP10','E4RM24005370','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'84378','Shivanand. W','2024-10-08 15:12:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.69 pH\r\nmV                   74.9 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(295,'2024-10-08','13:32:33','90.00000','105.00000','Purified water','pH','PW_SP22','E4RM24005371','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-10-08 15:13:38','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.70 pH\r\nmV                   75.3 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(296,'2024-10-08','13:35:13','90.00000','105.00000','Purified water','pH','PW_SP49','E4RM24005372','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'84378','Shivanand. W','2024-10-08 15:14:30','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.77 pH\r\nmV                   71.3 mV\r\nTemperature          24.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(297,'2024-10-08','13:36:31','90.00000','105.00000','Water For Injection','pH','WFI_SP23','E4RM24005373','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'84378','Shivanand. W','2024-10-08 15:15:34','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   72.8 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\nLog #                97\r\n\r\n\r\n'),
(298,'2024-10-08','13:48:49','90.00000','105.00000','Ampicillin and sulbactam for Injection','pH','AS0123001A','E4STFP24000625','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','Ampisulbactam',1,'84378','Shivanand. W','2024-10-08 15:23:53','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.20 pH\r\nmV                   -129.0 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(299,'2024-10-08','14:08:40','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424107','E4FP24000669','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Pipertazoinj',1,'84378','Shivanand. W','2024-10-08 14:56:39','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.88 pH\r\nmV                   64.9 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(300,'2024-10-08','14:14:20','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','PT0424108','E4FP24000671','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205247','R. Puneetha',2,1,101,'pH','Pipertazoinj',1,'84378','Shivanand. W','2024-10-08 14:57:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.82 pH\r\nmV                   68.5 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(301,'2024-10-08','14:34:45','90.00000','105.00000','Peptone','NA','RS007/24032','E4SO24000643','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-09 10:07:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.61 pH\r\nmV                   23.2 mV\r\nTemperature          25.6 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             08-10-24,10:09\r\n\r\n\r\n'),
(302,'2024-10-08','15:11:01','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0324059A','E4STFP24000630','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-08 15:27:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.94 pH\r\nmV                   61.4 mV\r\nTemperature          25.5 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(303,'2024-10-08','15:20:51','90.00000','105.00000','Floxapen powder for Solution for Injection','NA','FP0124013','E4FP24000664','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','90017','G. Srishailam',2,1,101,'pH','Floxapen',1,'84378','Shivanand. W','2024-10-08 15:34:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.00 pH\r\nmV                   119.8 mV\r\nTemperature          33.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\nLog #                103\r\n\r\n\r\n'),
(304,'2024-10-08','15:36:04','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0324057A','E4STFP24000631','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-08 16:07:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.92 pH\r\nmV                   62.6 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(305,'2024-10-08','15:38:33','90.00000','105.00000','Peptone','NA','RS007/24031','E4SO24000641','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-09 10:07:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -1.6 mV\r\nTemperature          26.4 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             08-10-24,10:09\r\nLog #                109\r\n\r\n\r\n'),
(306,'2024-10-08','15:39:36','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0324057A','E4STFP24000632','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-08 16:08:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.25 pH\r\nmV                   43.1 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\nLog #                105\r\n\r\n\r\n'),
(307,'2024-10-08','15:44:53','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','PN0324057A','E4STFP24000633','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','PIPERTAZO',1,'84378','Shivanand. W','2024-10-08 16:09:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.12 pH\r\nmV                   51.8 mV\r\nTemperature          25.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_111\r\nLast Cal             08-10-24,07:15\r\n\r\n\r\n'),
(308,'2024-10-08','16:59:32','90.00000','105.00000','pH Buffer','1.68','na','na','pH Buffer','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','1.68 pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   293.7 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             08-10-24,16:55\r\n\r\n\r\n'),
(309,'2024-10-08','17:04:46','90.00000','105.00000','pH Buffer','4.01','na','na','pH Buffer','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','4.01pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.01 pH\r\nmV                   159.0 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             08-10-24,16:55\r\n\r\n\r\n'),
(310,'2024-10-08','17:07:01','90.00000','105.00000','pH Buffer','7.00','na','na','pH Buffer','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','7.00pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -16.8 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             08-10-24,16:55\r\n\r\n\r\n'),
(311,'2024-10-08','17:21:10','90.00000','105.00000','pH Buffer','10.01','na','na','pH Buffer','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','10.01 pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   9.98 pH\r\nmV                   -190.4 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             08-10-24,16:55\r\n\r\n\r\n'),
(312,'2024-10-08','17:23:08','90.00000','105.00000','pH Buffer','12.46','na','na','pH Buffer','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','12.46 pH',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   12.42 pH\r\nmV                   -334.1 mV\r\nTemperature          26.9 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             08-10-24,16:55\r\n\r\n\r\n'),
(313,'2024-10-08','17:41:24','90.00000','105.00000','Peptone','NA','RS007/24032','E4SO24000643','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-09 10:08:50','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.88 pH\r\nmV                   7.0 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        99.3 %\r\nMethod               100\r\nCalLog #             PH_CAL_116\r\nLast Cal             08-10-24,10:09\r\n\r\n\r\n'),
(314,'2024-10-08','21:44:49','90.00000','105.00000','TAZOBACTAM','na','TZ0024093','na','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','205104','N. Siva Sekhar Reddy',2,0,104,'pH','1',1,'95681','T. Sivanarayana','2024-10-08 22:56:42','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.54 pH\r\nmV                   127.2 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             08-10-24,16:55\r\n\r\n\r\n'),
(315,'2024-10-09','00:36:05','90.00000','105.00000','pH Buffer','1.68','NA','NA','pH Buffer','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','1.68 pH',1,'95681','T. Sivanarayana','2024-10-09 05:56:55','Reveiwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   1.68 pH\r\nmV                   294.7 mV\r\nTemperature          23.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             09-10-24,00:31\r\n\r\n\r\n'),
(316,'2024-10-09','00:41:03','90.00000','105.00000','pH Buffer','4.01','NA','NA','pH Buffer','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','4.01pH',1,'95681','T. Sivanarayana','2024-10-09 05:57:14','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   159.2 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             09-10-24,00:31\r\n\r\n\r\n'),
(317,'2024-10-09','00:48:36','90.00000','105.00000','pH Buffer','7.00','NA','NA','pH Buffer','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','7.00pH',1,'95681','T. Sivanarayana','2024-10-09 05:57:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -16.8 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             09-10-24,00:31\r\n\r\n\r\n'),
(318,'2024-10-09','01:18:44','90.00000','105.00000','pH Buffer','10.01','NA','NA','pH Buffer','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','10.01 pH',1,'95681','T. Sivanarayana','2024-10-09 05:58:02','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   9.97 pH\r\nmV                   -189.0 mV\r\nTemperature          26.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             09-10-24,00:31\r\n\r\n\r\n'),
(319,'2024-10-09','01:23:27','90.00000','105.00000','pH Buffer','12.46','NA','NA','pH Buffer','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','12.46 pH',1,'95681','T. Sivanarayana','2024-10-09 05:58:17','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   12.47 pH\r\nmV                   -335.0 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             09-10-24,00:31\r\n\r\n\r\n'),
(320,'2024-10-09','01:25:45','90.00000','105.00000','PIPERACILLIN','NA','TZ0024093','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-09 05:56:28','Reveiwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.85 pH\r\nmV                   52.0 mV\r\nTemperature          23.7 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             09-10-24,00:31\r\n\r\n\r\n'),
(321,'2024-10-09','02:13:04','90.00000','105.00000','PIPERACILLIN','NA','TZ0024093','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'NULL','NULL','1992-12-08 00:00:00','NULL','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   5.87 pH\r\nmV                   51.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             09-10-24,00:31\r\n\r\n\r\n'),
(322,'2024-10-09','03:26:39','90.00000','105.00000','PIPERACILLIN','NA','TZ0024093','NA','IINPROCESS BULK(PIPERACILLIN)','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','2',1,'95681','T. Sivanarayana','2024-10-09 05:55:26','Reveiwed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.47 pH\r\nmV                   16.0 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             09-10-24,00:31\r\n\r\n\r\n'),
(323,'2024-10-09','04:59:43','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024093','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','3',1,'95681','T. Sivanarayana','2024-10-09 05:58:32','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   27.1 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             09-10-24,00:31\r\n\r\n\r\n'),
(324,'2024-10-09','05:21:47','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024093','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','3',1,'95681','T. Sivanarayana','2024-10-09 05:58:46','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.28 pH\r\nmV                   27.3 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             09-10-24,00:31\r\n\r\n\r\n'),
(325,'2024-10-09','05:23:18','90.00000','105.00000','PIPERACILLIN AND TAZOBACTAM','NA','TZ0024093','NA','BULK','PR-PHMTZ-004','CX1-15689','1','NULL','205044','Prem kumar',2,0,104,'pH','3',1,'95681','T. Sivanarayana','2024-10-09 05:59:07','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   6.27 pH\r\nmV                   28.0 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             09-10-24,00:31\r\n\r\n\r\n'),
(326,'2024-10-09','07:25:48','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005297','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','1.68 pH',1,'82735','M.D. Babar Qhadri','2024-10-09 08:00:52','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   309.3 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\nLog #                107\r\n\r\n\r\n'),
(327,'2024-10-09','07:28:57','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005297','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','4.01pH',1,'82735','M.D. Babar Qhadri','2024-10-09 08:01:00','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   172.9 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(328,'2024-10-09','07:31:22','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005297','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','7.00pH',1,'82735','M.D. Babar Qhadri','2024-10-09 08:01:08','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.02 pH\r\nmV                   -1.8 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(329,'2024-10-09','07:40:04','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005297','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','10.01 pH',1,'82735','M.D. Babar Qhadri','2024-10-09 08:01:17','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.97 pH\r\nmV                   -174.4 mV\r\nTemperature          25.7 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(330,'2024-10-09','07:44:42','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005297','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','88599','MD Mahamad',2,1,101,'pH','12.46 pH',1,'82735','M.D. Babar Qhadri','2024-10-09 08:01:24','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   12.50 pH\r\nmV                   -321.8 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(331,'2024-10-09','08:32:37','90.00000','105.00000','Piperacillin and Tazobactam For injection','Assay','TR0424002','E4FP24000680','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','89853','J.Ravi',2,1,101,'pH','Pipertazoinj',1,'82735','M.D. Babar Qhadri','2024-10-09 09:23:00','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.52 pH\r\nmV                   87.7 mV\r\nTemperature          31.7 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\nLog #                112\r\n\r\n\r\n'),
(332,'2024-10-09','08:48:22','90.00000','105.00000','pH Buffer','NA','pH Verifiction','E4INS24005300','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','1.68 pH',1,'205143','Ch. Prasad','2024-10-09 10:03:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   1.65 pH\r\nmV                   310.8 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             09-10-24,08:45\r\n\r\n\r\n'),
(333,'2024-10-09','08:52:14','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005300','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','4.01pH',1,'205143','Ch. Prasad','2024-10-09 10:03:54','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   4.02 pH\r\nmV                   173.5 mV\r\nTemperature          26.2 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             09-10-24,08:45\r\n\r\n\r\n'),
(334,'2024-10-09','08:54:04','90.00000','105.00000','pH Buffer','NA','pH  verification','E4INS24005300','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','7.00pH',1,'205143','Ch. Prasad','2024-10-09 10:04:36','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -2.2 mV\r\nTemperature          25.8 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             09-10-24,08:45\r\n\r\n\r\n'),
(335,'2024-10-09','08:56:10','90.00000','105.00000','pH Buffer','NA','pH verification','E4INS24005300','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','10.01 pH',1,'205143','Ch. Prasad','2024-10-09 10:04:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   10.01 pH\r\nmV                   -177.6 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             09-10-24,08:45\r\n\r\n\r\n'),
(336,'2024-10-09','08:58:25','90.00000','105.00000','pH Buffer','NA','pH Verification','E4INS24005300','pH Buffer','QC-PHMTZ-008','CW1-12012','1','NULL','92513','Ch. Chinna Rao',2,0,103,'pH','12.46 pH',1,'205143','Ch. Prasad','2024-10-09 10:05:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   12.43 pH\r\nmV                   -321.0 mV\r\nTemperature          26.7 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             09-10-24,08:45\r\n\r\n\r\n'),
(337,'2024-10-09','09:27:57','90.00000','105.00000','Piperacillin and Tazobactam for Injection','pH','TZ0024093','E4IP24001235','BULK','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','pH (Bulk)',1,'82735','M.D. Babar Qhadri','2024-10-09 13:38:57','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.87 pH\r\nmV                   64.7 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(338,'2024-10-09','10:25:37','90.00000','105.00000','Piperacillin and Tazobactam for Injection','Assay','TZ0424035A','E4STFP24000611','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','90017','G. Srishailam',2,1,101,'pH','PIPERTAZO',1,'82735','M.D. Babar Qhadri','2024-10-09 10:56:32','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.51 pH\r\nmV                   88.3 mV\r\nTemperature          32.5 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\nLog #                114\r\n\r\n\r\n'),
(339,'2024-10-09','10:49:24','90.00000','105.00000','Nafcillin For Injection','pH','NF0122001A','E4STFP24000621','Stability Study','QC-PHMTZ-007','CX1-15733','1','NULL','205210','B. Anitha',2,1,101,'pH','pH_Test',1,'82735','M.D. Babar Qhadri','2024-10-09 13:40:21','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.97 pH\r\nmV                   1.7 mV\r\nTemperature          25.0 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(340,'2024-10-09','11:58:51','90.00000','105.00000','Sodium Hydrogen Carbonate','pH','SL/214/170','E4RM24005374','Raw Material','QC-PHMTZ-007','CX1-15733','1','NULL','205090','M. Srikrishnudu',2,1,101,'pH','pH   Test',1,'82735','M.D. Babar Qhadri','2024-10-09 12:38:17','reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   8.16 pH\r\nmV                   -68.2 mV\r\nTemperature          24.3 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(341,'2024-10-09','12:20:20','90.00000','105.00000','Peptone','NA','RS007/24033','E4SO24000649','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-15 12:58:33','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.61 pH\r\nmV                   21.0 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             09-10-24,08:45\r\n\r\n\r\n'),
(342,'2024-10-09','14:28:10','90.00000','105.00000','Purified water','pH','PW/SP06','E4RM24005419','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-09 15:29:25','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.78 pH\r\nmV                   70.4 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(343,'2024-10-09','14:30:16','90.00000','105.00000','Purified water','pH','PW/SP37','E4RM24005420','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-09 15:29:48','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.74 pH\r\nmV                   72.3 mV\r\nTemperature          24.0 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(344,'2024-10-09','14:32:10','90.00000','105.00000','Water For Injection','pH','WFI/SP14','E4RM24005421','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-10-09 15:30:00','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.81 pH\r\nmV                   68.4 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(345,'2024-10-09','14:33:49','90.00000','105.00000','Water For Injection','pH','WFI/SP35','E4RM24005422','Water For Injection','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH',1,'86253','D. Jayapal Reddy','2024-10-09 15:30:15','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.83 pH\r\nmV                   67.3 mV\r\nTemperature          23.8 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(346,'2024-10-09','14:35:27','90.00000','105.00000','Purified water','pH','PW/SP05','E4RM24005423','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-09 15:30:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.80 pH\r\nmV                   69.3 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(347,'2024-10-09','14:37:00','90.00000','105.00000','Pure steam','pH','PSC/SP20','E4RM24005424','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'86253','D. Jayapal Reddy','2024-10-09 15:31:05','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.84 pH\r\nmV                   66.9 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(348,'2024-10-09','14:38:41','90.00000','105.00000','Purified water','pH','PW/SP39','E4RM24005425','Purified water','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH Test',1,'86253','D. Jayapal Reddy','2024-10-09 15:31:16','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.75 pH\r\nmV                   71.9 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(349,'2024-10-09','14:40:25','90.00000','105.00000','Pure steam','pH','PSC/SP17','E4RM24005426','Pure steam Condensate','QC-PHMTZ-007','CX1-15733','1','NULL','205066','Sreeram Jakkaraiah',2,1,101,'pH','pH_Pure Steam',1,'86253','D. Jayapal Reddy','2024-10-09 15:31:41','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.78 pH\r\nmV                   70.0 mV\r\nTemperature          23.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(350,'2024-10-09','15:13:07','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0424088','E4FP24000672','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205043','T.V. Subbaiah',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-10-09 15:24:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   5.94 pH\r\nmV                   60.4 mV\r\nTemperature          22.0 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(351,'2024-10-09','15:15:38','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0424089','E4FP24000673','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205043','T.V. Subbaiah',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-10-09 15:24:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.19 pH\r\nmV                   46.7 mV\r\nTemperature          24.4 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(352,'2024-10-09','15:18:29','90.00000','105.00000','Piperacillin and Tazobactam For injection','pH','TZ0424090','E4FP24000674','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','205043','T.V. Subbaiah',2,1,101,'pH','Pipertazoinj',1,'86253','D. Jayapal Reddy','2024-10-09 15:25:19','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   6.10 pH\r\nmV                   51.9 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(353,'2024-10-09','15:50:56','90.00000','105.00000','Peptone','NA','RS007/24034','E4SO24000651','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','88819','M. Ramya',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-15 12:58:06','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.66 pH\r\nmV                   18.7 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             09-10-24,08:45\r\n\r\n\r\n'),
(354,'2024-10-09','16:43:26','90.00000','105.00000','Peptone','NA','RS007/24033','E4SO24000649','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-15 12:59:29','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.69 pH\r\nmV                   16.2 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             09-10-24,08:45\r\n\r\n\r\n'),
(355,'2024-10-09','17:27:32','90.00000','105.00000','AMPICILLIN FOR INJECTION','pH','AP0224021','E4FP24000679','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','Ampicillin pH',1,'86253','D. Jayapal Reddy','2024-10-09 17:47:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.23 pH\r\nmV                   -131.2 mV\r\nTemperature          25.1 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(356,'2024-10-09','17:33:15','90.00000','105.00000','AMPICILLIN FOR INJECTION','pH','AP0224022(LOT-A)','E4FP24000682','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','Ampicillin pH',1,'86253','D. Jayapal Reddy','2024-10-09 17:47:31','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.15 pH\r\nmV                   -126.1 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\n\r\n\r\n'),
(357,'2024-10-09','17:37:26','90.00000','105.00000','AMPICILLIN FOR INJECTION','pH','AP0224022(LOT-B)','E4FP24000683','Finished Product','QC-PHMTZ-007','CX1-15733','1','NULL','88655','V. Prashanth Kumar',2,1,101,'pH','Ampicillin pH',1,'86253','D. Jayapal Reddy','2024-10-09 17:47:40','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.22 pH\r\nmV                   -130.3 mV\r\nTemperature          24.5 C (ATC)\r\nAverage Slope        98.7 %\r\nMethod               100\r\nCalLog #             PH_CAL_112\r\nLast Cal             09-10-24,07:12\r\nLog #                130\r\n\r\n\r\n'),
(358,'2024-10-09','17:49:08','90.00000','105.00000','XLDA','NA','MPZZ1003','E4SO24000646','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','XLDA',1,'205143','Ch. Prasad','2024-10-15 12:59:08','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.53 pH\r\nmV                   -32.6 mV\r\nTemperature          25.4 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             09-10-24,08:45\r\n\r\n\r\n'),
(359,'2024-10-09','17:51:37','90.00000','105.00000','Cetramide agar','NA','MPZZ1047','E4SO24000647','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','CA',1,'205143','Ch. Prasad','2024-10-15 13:00:04','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.14 pH\r\nmV                   -9.3 mV\r\nTemperature          24.9 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             09-10-24,08:45\r\n\r\n\r\n'),
(360,'2024-10-09','19:17:09','90.00000','105.00000','Peptone','NA','RS007/24034','E4SO24000651','Media Preparation','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','PEPIH',1,'205143','Ch. Prasad','2024-10-15 13:00:22','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   7.01 pH\r\nmV                   -1.8 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             09-10-24,08:45\r\n\r\n\r\n'),
(361,'2024-10-09','19:26:50','90.00000','105.00000','Penicillin G','NA','PG0124001A','E4STFP240000551','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-15 13:00:35','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.79 pH\r\nmV                   11.0 mV\r\nTemperature          26.3 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             09-10-24,08:45\r\n\r\n\r\n'),
(362,'2024-10-09','19:30:03','90.00000','105.00000','Penicillin G','NA','PG0124002A','E4STFP24000558','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-15 13:00:43','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.44 pH\r\nmV                   32.7 mV\r\nTemperature          26.5 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             09-10-24,08:45\r\n\r\n\r\n'),
(363,'2024-10-09','19:32:13','90.00000','105.00000','Penicillin G','NA','PG0124003A','E4STFP24000566','Stability Study','QC-PHMTZ-008','CW1-12012','1','NULL','95422','M. Shankar',2,0,103,'pH','Penicillin G Micro',1,'205143','Ch. Prasad','2024-10-15 13:00:58','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122322\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36473\r\nMode                 pH\r\npH                   6.38 pH\r\nmV                   35.0 mV\r\nTemperature          26.6 C (ATC)\r\nAverage Slope        98.9 %\r\nMethod               100\r\nCalLog #             PH_CAL_117\r\nLast Cal             09-10-24,08:45\r\n\r\n\r\n'),
(364,'2024-10-10','06:59:01','90.00000','105.00000','TAZOBACTAM','NA','TZ0024094','NA','INPROCESS BULK(TAZOBACTAM)','PR-PHMTZ-004','CX1-15689','1','NULL','84342','G.S.A. Naidu',2,0,104,'pH','1',1,'88629','Raghuram Reddy','2024-10-10 07:21:09','Verified','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122329\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36273\r\nMode                 pH\r\npH                   4.39 pH\r\nmV                   136.0 mV\r\nTemperature          23.3 C (ATC)\r\nAverage Slope        99.1 %\r\nMethod               100\r\nCalLog #             PH_CAL_114\r\nLast Cal             10-10-24,05:17\r\nLog #                112\r\n\r\n\r\n'),
(365,'2024-10-10','07:39:58','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005316','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,1,101,'pH','1.68 pH',1,'84378','Shivanand. W','2024-10-10 08:03:47','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   1.69 pH\r\nmV                   308.4 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(366,'2024-10-10','07:43:20','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005316','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,1,101,'pH','4.01pH',1,'84378','Shivanand. W','2024-10-10 08:04:20','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   4.03 pH\r\nmV                   172.2 mV\r\nTemperature          24.2 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(367,'2024-10-10','07:46:46','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005316','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,1,101,'pH','7.00pH',1,'84378','Shivanand. W','2024-10-10 08:04:52','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   7.03 pH\r\nmV                   -2.8 mV\r\nTemperature          24.1 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
(368,'2024-10-10','07:49:17','90.00000','105.00000','pH Buffer','NA','NA','E4INS24005316','pH Buffer','QC-PHMTZ-007','CX1-15733','1','NULL','205075','G. Shyam Kumar',2,1,101,'pH','10.01 pH',1,'84378','Shivanand. W','2024-10-10 08:05:37','Reviewed','Thermo Scientific (c) 2011\r\n\r\n          Versa Star Pro    \r\n\r\nMeter S/N               V122320\r\nSW Rev                  11.50\r\n\r\n\r\nChannel - 1 \r\n\r\nModule S/n #         VA36002\r\nMode                 pH\r\npH                   9.99 pH\r\nmV                   -175.6 mV\r\nTemperature          24.6 C (ATC)\r\nAverage Slope        98.8 %\r\nMethod               100\r\nCalLog #             PH_CAL_113\r\nLast Cal             10-10-24,07:29\r\n\r\n\r\n'),
